The design and function of the eleganza 3XC bed is customised to meet the high demands of intensive care. This modern bed with a column construction offers an integrated programme which helps healthcare staff treat and care for critically ill patients.
The safe gaps concept reduces the risk of patient entrapment and associated injuries.
The Ergoframe® positioning system extends the space in the pelvic area. Tissue compression and the development of pressure ulcers in this area is significantly reduced.
The X-ray translucent mattress platform enables integrated X-ray examination or C-arm scanning.
The cardiac chair position is easily adjusted using a single button on the supervisor control panel. In this sitting position, the vital lung capacity increases and gaseous change improves.
The variable side rails can provide complete falls protection for some patients. The eleganza protector at the foot end can help to prevent falls for confused patients in accordance with the individual risk assesssment.
The automatic integrated scales indicate the actual weight of the patient, changes in time and includes a system for the addition or removal of accessories.
Unbraked beds can cause falls so the unique i-brake feature will automatically activate the brakes when the bed is plugged into electricity
The unique Mobi-Lift handles with bed height controls assist with safer mobilisation and active involvement of the patient.
The practical foot controls allow an easy handsfree bed height adjustment allowing healthcare staff to create optimal access to the patient for hygiene needs, wound care and examinations.
A retractable 5th castor helps to manoeuvre the bed during transportation and can be retracted into the chassis when not in use.
Outer dimensions (Side rails up)
|223 x 105 cm|
|Mattress platform extension||0 / 8 / 17 cm|
|Recommended mattress size||208 x 86 cm|
|Max. mattress height||23 cm|
|Bed height||40 - 78 cm|
|Maximum backrest angle||70°|
|Maximum thigh rest angle||30°|
|Trendelenburg/Reverse Trendelenburg position||13°/13°|
|Height of side rails (above mattress platform)||45 cm|
|Weight (basic equipment)||138 kg|
|Safe working load||250 kg|
|Lifting pole bearing capacity||75 kg|