Series DP-4000 High-Performance Differential Pressure Transmitter
The DP-4000 is a high-performance differential pressure transmitter with a compact and robust Stainless steel electronics housing. The calibration / programming can be done very easy without test pressure by using the Unique One Touch programming button together with the Graphic display with backlight.
- Pulp and Paper
- Food and Beverage
- Clean liquids, gases and vapours
Principle of Operation
The multilingual graphic display can indicate a number of chosen engineering units including the process temperature as well as the mA output in a bargraph.
Damping times can be adjusted and a 4-20 mA current simulation can be performed. For flow measurement a square root function is built-in with adjustable cut off point. The output is 4-20 mA including HART¬ protocol (HART¬ 7.0).
ATEX EX, II1G Ex ia IIC T6 Ga, IECEx, HART¬ protocol
- Stainless steel housing
- Easy adjustable without test pressure by 1 button
- Accuracy: 0.075%, option: 0.065%
- Turn down up to: 100:1
- Graphic Display with bargraph and backlight
- Stilling Well
- Sewage Plant
Accuracy: 0.075% (standard)
Ranges: 0-10 mbar to 0 _ 20 bar
Turndown: up to 100:1
Protection: IP66/67 (option IP68)