system
Medical AI Inquiry: You was designed to assist in medical decision-making based on the provided user questions. It is crucial to provide accurate and comprehensive information to receive a well-informed response.  Labels: OBJECTIVE, METHODS. Meshes: Aged, Aged, 80 and over, Aortic Aneurysm, Abdominal, Aortic Rupture, Blood Vessel Prosthesis, Blood Vessel Prosthesis Implantation, Device Removal, Endoleak, Endovascular Procedures, Fatal Outcome, Humans, Male, Prosthesis Design, Prosthesis Failure, Reoperation, Stents, Treatment Failure.
user
Is type IIIb endoleak an important cause of failure following endovascular aneurysm repair? Context: To present confirmed cases of type IIIb endoleak in second and third-generation stent-grafts used for endovascular aneurysm repair (EVAR). Four patients developed type IIIb endoleak caused by fabric tears between 4 and 13 years following their initial EVAR. Three patients presented with rupture and one with aneurysm expansion of unknown cause. In each case, the type IIIb endoleak was confirmed at open surgery after imaging proved non-diagnostic. Only one patient survived. Had the cause for the expansion or ruptures been found prior to open reintervention, relining of the stent-graft may have been possible.
assistant
Type IIIb endoleak remains difficult to diagnose. Avoidance of the high mortality associated with open secondary intervention requires a high degree of suspicion and it should be considered in any post-EVAR aneurysm expansion without an obvious cause. Final Decision: yes