system
Medical AI Inquiry: You was designed to assist in medical decision-making based on the provided user questions. It is crucial to provide accurate and comprehensive information to receive a well-informed response.  Labels: BACKGROUND, METHODS, RESULTS. Meshes: Animals, Catheter Ablation, Cattle, Kidney, Liver, Microwaves, Muscle, Skeletal, Surgery, Computer-Assisted, Swine, Time Factors.
user
Are optimal ablation volumes achieved at submaximal power settings in a 2.45-GHz microwave ablation system? Context: Local ablative therapies, including microwave ablation (MWA), are common treatment modalities for in situ tumor destruction. Currently, 2.45-GHz ablation systems are gaining prominence because of the shorter application times required. The aims of this study were to determine optimal power and time to ablation volume (AbV) ratios for a new 1.8-mm-2.45-GHz antenna using ex vivo tissue models. The 1.8-mm-2.45-GHz Accu2i MWA system was employed to perform ablations in bovine liver, porcine muscle, and porcine kidney ex vivo. Whole tissues were prewarmed (35°C) and multiple ablations performed at power settings of 60 to 180 W for 2- to 6-minute time intervals. Postablation, tissues were dissected, AbVs calculated, and correlations to power and time settings made. Significant increases in AbV were measured between each of the time points for a constant power setting in all 3 tissues. Increasing power settings led to significant increases in AbV at power settings ≤140 W. However, no significant increase in AbV was obtained at power settings >140 W.
assistant
Optimal efficiency for MWA using a new 1.8-mm-2.45-GHz system is achieved at settings of ≤140 W for 6 minutes in a range of ex vivo tissue and no additional benefit occurs by increasing the power setting to 180 W in these tissues. Final Decision: yes