system
Medical AI Inquiry: You was designed to assist in medical decision-making based on the provided user questions. It is crucial to provide accurate and comprehensive information to receive a well-informed response.  Labels: OBJECTIVE, METHODS, RESULTS. Meshes: Antibodies, Automation, Humans, Immunoassay, Intrinsic Factor, Reference Values, Reproducibility of Results, Vitamin B 12.
user
Is intrinsic factor blocking antibody interference detected in five automated cobalamin immunoassays? Context: To systematically assess five automated cobalamin assays for intrinsic factor binding antibody (IFBA)-induced interference using pooled serum. Six pools created from IFBA-negative and IFBA-positive serum representing low, normal, and high cobalamin concentrations were analyzed before and after polyethylene glycol (PEG) precipitation of immunoglobulins on five cobalamin assays: the Centaur XP (Siemens Healthcare Diagnostics, Tarrytown, NY), IMMULITE 2000 (Siemens Healthcare Diagnostics), ARCHITECT i2000SR (Abbott Diagnostics, Abbott Park, IL), UniCel Dxl 800 (Beckman Coulter, Chaska, MN), and Modular E170 (Roche Diagnostics, Indianapolis, IN). Cobalamin concentrations before and after PEG treatment were similar, almost all within a 30% total allowable error, with no difference in pattern between the IFBA-negative and IFBA-positive pools regardless of the cobalamin concentration.
assistant
Our results suggest that, under optimal conditions, the five automated cobalamin assays assessed are not susceptible to IFBA-mediated interference. Final Decision: no