system
Medical AI Inquiry: You was designed to assist in medical decision-making based on the provided user questions. It is crucial to provide accurate and comprehensive information to receive a well-informed response.  Labels: BACKGROUND, OBJECTIVE, METHODS, RESULTS. Meshes: Dental Abutments, Dental Implants, Dental Prosthesis Design, Dental Restoration Failure, Dental Stress Analysis, Elasticity, Equipment Failure Analysis, Humans, Pliability, Statistics, Nonparametric.
user
Do in vitro evaluation of the strength of the conical implant-to-abutment joint in two commercially available implant systems? Context: The cone-screw abutment has been shown to diminish micromovement, reducing the burden of component loosening and fracture. However, it is unclear whether the conical taper and joint design influence strength of the interface, with respect to unfavorable bending moments. This comparative study evaluated the resistance to bending for the ITI Straumann and Astra Tech ST implant systems using an 8- and 11-degree internal cone, respectively. Assembled units from each system were mounted in a 3-point bending apparatus. High load tests were performed, 4 mm from the joint, and bending moments necessary to induce first point of plastic deformation and ultimate failure were measured. All units were inspected to determine the critical zone of failure. Bending moments necessary to induce first point of plastic deformation were considered well above that expected in clinical function for both systems. However, the critical zones of failure differed in that the solid Astra abutment deformed before the cone joint with its 11-degree taper and smooth transition into the neck of the screw, preventing screw fracture. By contrast, all ITI screws fractured at the head of the screw where it met the base of the 8-degree cone. It is unclear which aspects of the joint design were responsible for the difference observed in mode of failure or if it was a direct result of the experimental design.
assistant
For clinically relevant levels of bending moment, no problems were anticipated with respect to component failure for either system. Final Decision: yes