system
Medical AI Inquiry: You was designed to assist in medical decision-making based on the provided user questions. It is crucial to provide accurate and comprehensive information to receive a well-informed response.  Labels: OBJECTIVE, METHODS, RESULTS. Meshes: Adult, Aged, Aged, 80 and over, Biopsy, Needle, Equipment Design, Humans, Male, Middle Aged, Organ Size, Prostate, Prostatic Neoplasms.
user
Does a new core-biopsy instrument with an end-cut technique provide prostate biopsies with increased tissue yield? Context: To evaluate the tissue yield of a new core-biopsy needle with an end-cut mechanism when used for transrectal prostate biopsies. The end-cut instrument has an adjustable stroke length (13, 23 or 33 mm) and an inner and an outer cannula, but no trocar. It was compared with the conventional side-notch needle (Biopty, Bard, Covington, GA, USA) with trocar and fixed stroke length (22 mm). In 60 men, octant biopsies were taken including the apex, mid-medial, mid-lateral and basal positions. At random, the left side of the prostate was biopsied with one of the instruments and the right side with the other. The length and weight of the biopsy specimens were measured. In 40 men, the stroke length of the end-cut instrument was set to 23 mm, and in 20 men it was set to 33 mm. The weight and the weight per length of the biopsies provided by the end-cut instrument with a stroke length of 23 mm were 18.4% and 13.7%, respectively, greater than in the biopsies from the side-notch instrument (P<0.001). In 21.9% of the attempts no tissue was obtained by the end-cut needle, compared with 1.9% with the side-notch needle. With a 33 mm stroke length, the length and weight of the end-cut biopsies were 38.4% and 33.0%, respectively, greater than in biopsies from the side-notch instrument.
assistant
The end-cut instrument provided a greater tissue yield than the side-notch needle, but a significant number of biopsies were lost. Final Decision: yes