Adaptive illumination reduces photobleaching in structured illumination microscopy
Photobleaching is a major factor limiting the observation time in fluorescence microscopy. We achieve photobleaching reduction in structured illumination microscopy (SIM) by locally adjusting the illumination intensities according to the sample. Adaptive SIM is enabled by a digital micro mirror device (DMD) which provides a projection of the grayscale illumination patterns. We demonstrate a reduction in photobleaching by a factor of three in adaptive SIM compared to the non adaptive SIM based on a spot grid scanning approach. Our proof of principle experiments show great potential for DMD based microscopes to become a more useful tool in live cell SIM imaging. c (c) 2016 Optical Society of America OCIS codes: (180.2520) Fluorescence microscopy; (100.6640) Superresolution; (110.1085) Adaptive imaging. References and links 1. E. Betzig, G. H. Patterson, R. Sougrat, O. W. Lindwasser, S. Olenych, J. S. Bonifacino, M. W. Davidson, J. LippincottSchwartz, and H. F. Hess, "Imaging intracellular fluorescent proteins at nanometer resolution," Science 313, 1642 1645 (2006) 2. M. J. Rust, M. Bates, and X. Zhuang, "Sub diffraction limit imaging by stochastic optical reconstruction microscopy (STORM)" Nat. Methods 3, 793 795 (2006) 3. S. T. Hess, T. P. K. Girirajan, and M. D. Mason, "Ultra high resolution imaging by fluorescence photoactivation localization microscopy," Biophys. J. 91, 4258 4272 (2006) 4. R. Heintzmann and C. Cremer, "Laterally modulated excitation microscopy: improvement of resolution by using a diffraction grating," Proc. SPIE 3568, 185 196 (1999) 5. M. G. L. Gustafsson, "Surpassing the lateral resolution limit by a factor of two using structured illumination microscopy," J. Microsc. 198, 82 87 (2000) 6. J. T. Frohn, H. F. Knapp, and A. Stemmer, "True optical resolution beyond the Rayleigh limit achieved by standing wave illumination," Proc. Natl. Acad. Sci. USA 93, 7232 7236 (2000) 7. G. E. Cragg and P. T. So, "Lateral resolution enhancement with standing evanescent waves," Opt. Lett. 25, 46 48 (2000) 8. C. B. Muller and J. Enderlein, "Image Scanning Microscopy," Phys. Rev. Lett. 104, 198101 (2010) 9. G. M. R. De Luca, R. M. P. Breedijk, R. Brandt, C. H. C. Zeelenberg, B. E. de Jong, W. Timmermans, L. N. Azar, R. Hoebe, S. Stallinga, and E. M. M. Manders, "Re scan confocal microscopy: scanning twice for better resolution." Biomed. Opt. Express 4, 2644 2656 (2013) 10. S. W. Hell and J. Wichmann, "Breaking the diffraction resolution limit by stimulated emission: stimulated emissiondepletion fluorescence microscopy," Opt. Lett. 19, 780 782 (1994) 11. P. Kner, B. B. Chhun, E. R. Griffis, L. Winoto, and M. G. L. Gustafsson, "Super resolution video microscopy of live cells by structured illumination," Nat. Methods 6, 339 342 (2009) 12. L. Shao, P. Kner, E. H. Rego, and M. G. L. Gustafsson, "Super resolution 3D microscopy of live whole cells using structured illumination," Nat. Methods 8, 1044 1046 (2011) 13. D. Li, L. Shao, B. C. Chen, X. Zhang, M. Zhang, B. Moses, D. E. Milkie, J. R. Beach, J. A. Hammer, M. Pasham, T. Kirchhausen, M. A. Baird, M. W. Davidson, P. Xu, and E. Betzig, "Extended resolution structured illumination imaging of endocytic and cytoskeletal dynamics," Science 349, aab3500 (2015) 14. S. Cox, "Super resolution imaging in live cells," Dev. Biol. 401, 175 181 (2015) 15. E. Mudry, K. Belkebir, J. Girard, J. Savatier, E. Le Moal, C. Nicoletti, M. Allain, and A. Sentenac, "Structured illumination microscopy using unknown speckle patterns," Nature Photon. 6, 312 315 (2012) Vol. 7, No. 10 1 Oct 2016 BIOMEDICAL OPTICS EXPRESS 4263 #272429 Journal (c) 2016 http:/dx.doi.org/10.1364/BOE.7.004263 Received 25 Jul 2016; revised 16 Sep 2016; accepted 18 Sep 2016; published 23 Sep 2016 16. A. G. York, S. H. Parekh, D. Dalle Nogare, R. S. Fischer, K. Temprine, M. Mione, A. B. Chitnis, C. a. Combs, and H. Shroff, "Resolution doubling in live, multicellular organisms via multifocal structured illumination microscopy," Nat. Methods 9, 749 754 (2012) 17. P. KAZizek, I. Raska, and G. M. Hagen, "Flexible structured illumination microscope with a programmable illumination array," Opt. Express 20, 24585 24599 (2012) 18. D. Dan, M. Lei, B. Yao, W. Wang, M. Winterhalder, A. Zumbusch, Y. Qi, L. Xia, S. Yan, Y. Yang, P. Gao, T. Ye, and W. Zhao, "DMD based LED illumination Super resolution and optical sectioning microscopy," Sci. Rep. 3, 01116 (2013) 19. R. A. Hoebe, C. H. Van Oven, T. W. J. Gadella, P. B. Dhonukshe, C. J. F. Van Noorden, and E. M. M. Manders, "Controlled light exposure microscopy reduces photobleaching and phototoxicity in fluorescence live cell imaging," Nat. Biotechnol. 25, 249 253 (2007) 20. R. A. Hoebe, H. T. M. Van der Voort, J. Stap, C. J. F. Van Noorden, and E. M. M. Manders, "Quantitative determination of the reduction of phototoxicity and photobleaching by controlled light exposure microscopy," J. Microsc. 231, 9 20 (2008) 21. W. Caarls, B. Rieger, A. H. B. De Vries, D. J. Arndt Jovin, and T. M. Jovin, "Minimizing light exposure with the programmable array microscope," J. Microsc. 241, 101 110 (2011) 22. P. J. Verveer, Q. S. Hanley, P. W. Verbeek, L. J. v. Vliet, and T. M. Jovin, "Theory of confocal fluorescence imaging in the programmable array microscope PAM " J. Microsc. 189, 192 198 (1998) 23. Q. S. Hanley, P. J. Verveer, M. J. Gemkow, and T. M. Jovin, "An optical sectioning programmable array microscope implemented with a digital micromirror device," J. Microsc. 196, 317 331 (1999) 24. K. K. Chu, D. Lim, and J. Mertz, "Enhanced weak signal sensitivity in two photon microscopy by adaptive illumination," Opt. Lett. 32, 2846 2848 (2007) 25. T. Staudt, A. Engler, E. Rittweger, B. Harke, J. Engelhardt, and S. W. Hell, "Far field optical nanoscopy with reduced number of state transition cycles," Opt. Express 19, 5644 5657 (2011) 26. N. Chakrova, R. Heintzmann, B. Rieger, and S. Stallinga, "Studying different illumination patterns for resolution improvement in fluorescence microscopy," Opt. Express 23, 31367 31383 (2015) 27. L. Song, E. J. Hennik, I. T. Young, and H. J. Tanke, "Photobleaching kinetics of fluorescein in quantitative fluorescence microscopy," Biophys. J. 66, 2588 2600 (1995) 28. N. Chakrova, B. Rieger, and S. Stallinga, "Deconvolution methods for structured illumination microscopy," J. Opt. Soc. Am. A 33, 12 20 (2016) 29. N. Chakrova, B. Rieger, and S. Stallinga, "Development of a DMD based fluorescence microscope," Proc. SPIE 9330, 933008 (2015) 30. J. M. Zwier, G. J. Van Rooij, J. W. Hofstraat, and G. J. Brakenhoff, "Image calibration in fluorescence microscopy." J. Microsc. 216, 15 24 (2004) 31. Z. Wang, A. C. Bovik, H. R. Sheikh, and E. P. Simoncelli, "Image quality assessment: From error visibility to structural similarity," IEEE Trans. Sig. Process. 13, 600 612 (2004)