Some observations on the breakage of ultrasonic files driven piezoelectrically.
The incidence of breakage of Piezon-Master ultrasonic K files were evaluated. Three groups of unused files were subjected to three treatments, namely; free vibration in air without irrigation, free vibration in root canal while minimizing contact with the wall of canal in the presence of irrigation and light filing in root canal with free flow of irrigation. Cavitation produced by files in contact and free of contact with a glass surface was examined in order to observe the relationship between cavitation defects and breakage. In addition, the fractured and unfractured files were examined under a scanning electron microscope for the presence of cavitation pits. The results indicated that more files broke in air. In water, a higher incidence of breakage occurred when files were allowed to freely vibrate while no breakage occurred when the files were used in filing. All files generated cavitation which resulted in pitting of their surfaces. However, it was considered unlikely that the pits contributed to fracture. Fatigue cracks which could be the result of the manufacturing process were observed at some of the corners of the cross sections of the fractured files and could be the main contributory factor to fracture.
Efficacy of a Nickel-Titanium Ultrasonic Instrument for Biofilm Removal in a Simulated Complex Root Canal
This study evaluated the effectiveness of NiTi ultrasonic tips for Enterococcus faecalis (E. faecalis) biofilm removal in simulated complex root canals. Sixty root canal models consisting of a 30-degree curved main canal and two lateral canals were constructed from polydimethylsiloxane and incubated with E. faecalis. Irrigants in root canals were activated using a manual syringe (SI), a stainless steel (SS) instrument, a nickel-titanium (Ni-Ti) ultrasonic instrument, or a sonic instrument (EA). Instruments of SI, SS, and NiTi-9 groups were placed 9 mm from the apex, whereas those in NiTi-2 and EA groups were placed 2 mm from the apex. The efficacy of each method was determined as the ratio of fluorescence concentration before and after activation. In the apical curved canal, the highest efficacy was found in the NiTi-2 group (99.40%), followed by SI (84.25%), EA (80.38%), SS (76.93%), and NiTi-9 (67.29%) groups. In lateral canals 1 and 2, the efficacy was the highest in the NiTi-2 group and the lowest in the SI group. The NiTi ultrasonic instrument could effectively remove biofilms in the curved canal and lateral canals. This instrument should be introduced close to the working length. An up-and-down motion of the activation instrument is recommended.
Numerical and experimental investigation on air distributor design of fluidized bed reactor of sawdust pyrolysis
Abstract An improved biomass conversion in a lab-scale fluidized bed reactor (FBR) is achieved by optimizing the air distributor orifice size and number. Two approaches for introducing air were numerically and experimentally examined. These included perforated plate and downward orifice sparger. 3 mm and 5 mm orifice diameters, and 7, 9 and 13 holes were simulated at flow velocities of 2, 4, 6, and 8 times the minimum fluidization velocity, Umf. The simulation results revealed that increasing the orifice number has a positive effect on gas holdup, particularly for the perforated plate, which showed a uniform distribution of bubbles. The sparger configuration produced larger bubbles sizes, which could lead to better recirculation of solid flux, while stagnant particle zones were observed at the lower section of both distributors. A pyrolysis unit is also inserted at the top of freeboard of the FBR. For perforated plate with 6Umf, the pyrolysis process achieved thermal and cold gas efficiencies of 82.91% and 52.03%, respectively, while combustible gases with 41.6%mol CO, 8.7%mol CH 4 and 2.17%mol HC, were obtained. For sparger pipe and 6U mf _sp, 35.33 ppm NOx and 111 ppm SO 2 were measured in flue gas from combustion process at bed region.