Ultrasonic NDE in a reactor core
We have irradiated bulk single crystal piezoelectric AlN operating as an ultrasonic transducer up to a fast and thermal neutron fluence of 1.85×1018 n/cm2 and 5.8×1018 n/cm2 respectively along with a gamma dose of 26.8 MGy. Our primary interest was in the radiation effects on the piezoelectric performance of the AlN crystal. Ideally standard procedures for extracting material properties from the impedance spectra would have been applied. However, this procedure was anticipated to be very error prone due the effects of radiation on passive components, such as electrodes and cabling. As a result the transducer was coupled to an aluminum cylinder and operated in pulse echo mode throughout the irradiation. In addition to the pulse echo testing impedance data were obtained. Although, material parameters such as d33 could not be measured in-situ during irradiation with our procedure, our work is the only publication, to the author's knowledge, showing the operation of an ultrasonic transducer in a reactor core. Further, the piezoelectric coefficient was measured prior to irradiation and found to be 5.5 pC/N which is unchanged from as-grown samples, and in fact higher than the measured d33 for many as-grown samples. Further proof of the viability of utilizing ultrasonic transducers in a reactor core is provided by means of successful time of flight (TOF) measurements throughout the irradiation. The TOF data was used to measure the gamma heating of an aluminum cylinder. The heating due to short lived isotopes in the Al was also detected via TOF.
High-temperature ultrasonic sensor for in-situ monitoring of hot isostatic processing
A sensor has been developed and tested that is capable of emitting and receiving ultrasonic energy at temperatures exceeding 900 degrees C and pressures above 150 MPa. The sensor is based on a unique form of aluminum nitride that retains tits piezoelectric properties at high temperatures. The sensor works with standard ultrasonic pulse-receivers and has demonstrated the capability of measuring workpiece deformation during hot isostatic pressing (HIP). Details of the sensor design, performance, and coupling of the ultrasound to the workpiece are described. Ultrasonic data acquired by the sensor, in situ, during HIP runs and at elevated temperatures in air are presented.
Numerical analysis of surface crack propagation in flexible pavements using XFEM and cohesive zone model
Surface initiated (top-down) cracking in the longitudinal direction of pavements has become a predominant mode of failure in thick asphalt pavements. Most previous researchers studied crack propagation using the traditional Linear Elastic Fracture Mechanics (LEFM) method by calculating the stress intensity factor around the crack tip. In this paper, Extended Finite Element Method (XFEM) was used to study the key factors causing surface initiated crack propagation under surface tension and/or shear. A two-dimensional (2-D) finite element (FE) model for a two-lane flexible pavement was built to predict pavement responses and crack propagation potential under various scenarios. It was found that surface tension is the major driving force for crack propagation when the crack is at some distance away from tire loading. The results show that the distance from the location of crack to tire loading, the depth and direction of the crack, and the modulus ratio between the hot-mix asphalt (HMA) layer and the base layer have significant influences on crack propagation under surface tension. On the other hand, this study shows that shear stress can induce surface initiated crack propagation near the loading area as the shear resistance of material becomes low at the high temperature.