Radiation skyshine from a 6 MeV medical accelerator
This study assesses the dose level from skyshine produced by a 6 MeV medical accelerator. The analysis of data collected on skyshine yields professional guidance for future investigators as they attempt to quantify and qualify radiation protection concerns in shielding therapy vaults. Survey measurements using various field sizes and at varying distances from a primary barrier have enabled us to identify unique skyshine behavior in comparison to other energies already seen in literature. In order to correctly quantify such measurements outside a shielded barrier, one must take into consideration the fact that a skyshine maximum may not be observed at the same distance for all field sizes. A physical attribute of the skyshine scatter component was shown to increase to a maximum value at 4.6 m from the barrier for the largest field size used. We recommend that the largest field sizes be used in the field for the determination of skyshine effect and that the peak value be further analyzed specifically when considering shielding designs. PACS numbers: 87.52.‐g, 87.52.Df, 87.52.Tr, 87.53.‐j, 87.53.Bn, 87.53.Dq, 87.66.‐a, 89., 89.60.+x
Determination of the shielding for an unoccupied roof of a radiation therapy facility to account for adjacent buildings--modification of a computer program.
An addition has been made to an existing software program for calculating shielding thicknesses of barriers for megavoltage radiation therapy beams. This addition calculates the shielding thicknesses required for the roof of a single-story building when adjacent, multi-level buildings are occupied.
A new look at CT dose measurement: beyond CTDI.
Equations are derived for generating accumulated dose distributions and the dose line integral in a cylindrical dosimetry phantom for a helical CT scan series from the single slice dose profiles using convolution methods. This exposition will better clarify the nature of the dose distribution in helical CT, as well as providing the medical physicist with a better understanding of the physics involved in dose delivery and the measurement process. Also addressed is the concern that as radiation beam widths for multi-slice scanners get wider, the current methodology based on the measurement of the integral of the single slice profile using a 10 cm long ion chamber (CTDI100) may no longer be adequate. It is shown that this measurement would underestimate the equilibrium dose and dose line integral by about 20% in the center of the body phantom, and by about 10% in the center of the head phantom for a 20 mm nominal beam width in a multi-slice scanner. Rather than making the ion chamber even longer to collect the broad scatter tails of the single slice profile, an alternative to the CTDI method is suggested which involves using a small volume ion chamber, and scanning a length of phantom long enough to establish dose equilibrium at the location of the chamber. With a modern CT scanner, such a scan length can be covered in 15 s or less with a helical or axial series, so this method is not significantly more time-consuming than the long chamber method. The method is demonstrated experimentally herein.