Single shot measurements of the E-field vector with pigtailed optical probes
Summary form only given. Over the variety of available sensors dedicated to electric (E)-field characterization, the use of antennas constitutes the most widespread technique. While such probes are convenient and provide a good sensitivity, they remain invasive and bandwidth limited. At the opposite, fibered electro-optic transducers1 are fully dielectric, millimeter sized and allow to perform measurements of the E-field vector from DC to several gigahertz and even up to terahertz frequencies using equivalent-time sampling. Furthermore, recent developments lead to a simultaneous characterization of 2 transverse E-field vector components with a single EO probe2. Based on polarization state modulation, the EO transducer is linked to a remote (up to 30 meters) optoelectronic set-up including a ultra low noise laser feeding the probe and a real time optical set-up to manage the modulation treatment. The automated and servo controlled measurement bench is temperature dependent free. The available measurement dynamics exceeds 100 dB, ranging from less than 1 V. m-1. Hz-1/2 up to the breakdown electric field in air.An exhaustive comparison between BO sensors and other technologies will be firstly given during the conference. This analysis will be based on intrinsic sensor properties, such as sensitivity, frequency bandwidth, vectorial selectivity, spatial resolution and induced perturbation on the field to be measured. After recalling the principles of the BO effect, the optical arrangement of the optical probes will be described. The characterization of the BO system will be presented together with experimental results illustrating the potentialities of BO sensors. Among these examples, measurements of pulsed B-field in air (pulsed power), water (specific absorption rate evaluation in pulsed regime) or in plasma (real time evolution of the electrical discharge associated B-field) will be shown.
Two electric-field components measurement using a 2-port pigtailed electro-optic sensor
Based on an isotropic electro-optic crystal, a two-port pigtailed electro-optic sensor has been built. The probe allows to measure two orthogonal components of the ambient electric field. The sensor intrinsically presents a temperature-dependent free response. The two measured electric field components are orthogonal to the sensor revolution axis, leading to a transverse electro-optic probe. Magnitude and orientation of the electric field are measured simultaneously with an accuracy of 0.5 dB and 2°, respectively. The rejection of orthogonal components to the electric field vector reaches 30 dB.
Long term response stability of a well-type ionization chamber used in calibration of high dose rate brachytherapy sources
Well-type ionization chamber is often used to measure strength of brachytherapy sources. This study aims to check long term response stability of High Dose Rate (HDR)-1000 Plus well-type ionization chamber in terms of reference air kerma rate (RAKR) of a reference 137Cs brachytherapy source and recommend an optimum frequency of recalibration. An HDR-1000 Plus well-type ionization chamber, a reference 137Cs brachytherapy source (CDCSJ5), and a MAX-4000 electrometer were used in this study. The HDR-1000 Plus well-type chamber was calibrated in terms of reference air kerma rate by the Standards Laboratory of the International Atomic Energy Agency (IAEA), Vienna. The response of the chamber was verified at regular intervals over a period of eight years using the reference 137Cs source. All required correction factors were applied in the calculation of the RAKR of the 137Cs source. This study reveals that the response of the HDR-1000 Plus well-type chamber was well within ±0.5% for about three years after calibration/recalibration. However, it shows deviations larger than ±0.5% after three years of calibration/recalibration and the maximum variation in response of the chamber during an eight year period was 1.71%. The optimum frequency of recalibration of a high dose rate well-type chamber should be three years.