INTEGRATION OF THE VIMOS CONTROL SYSTEM
The VIRMOS consortium of French and Italian Institutes (PI: O. Le Fevre, co-PI: G. Vettolani) is manufacturing two wide field imaging multi-object spectrographs for the European Southern Observatory Very Large Telescope (VLT), with emphasis on the ability to carry over spectroscopic surveys of large numbers of sources: the VIsible Multi-Object Spectrograph, VIMOS, and the Near InfraRed MultiObject Spectrograph, NIRMOS. VIMOS is being installed at the Nasmyth focus of the third Unit Telescope of the VLT at Mount Paranal in Chile, after a period of pre-integration in Europe at the Observatoire de Haute Provence. There are 52 motors to be controlled in parallel in the spectrograph, making VIMOS a complex machine to be handled. This paper will focus on the description of the control system, designed in the ESO VLT standard control concepts, and on some integration issues and problem solving strategies.
VIRMOS: visible and infrared multiobject spectrographs for the VLT
We present the current design of the VIsible Multi-Object Spectrograph (VIMOS) and the Near InfraRed Multi-Object Spectrograph (NIRMOS) for the European Southern Observatory Very Large Telescope. The basic scientific requirement is to conduct very deep redshift surveys of large quantities of objects, in a minimum number of nights. The technical specifications are to allow for a large multiplex gain over a wide field, and a high efficiency of the optical train, over the 0.37 to 1.8 micrometer domain. The baseline technical concept is built around 4 channels, covering 4 X 7 X 8 arcmin2 for VIMOS and 4 X 7 X 7 arcmin2 for NIRMOS. Each channel is an imaging spectrograph with a large field adaptation lens, a collimator, grisms or filters, and a F/1.8 camera, coupled to a 2048 X 4096 pixels CCD for VIMOS, and a 20482 HgCdTe Rockwell array for NIRMOS. The unique multiplex gain allows to obtain spectra of up to 840 object simultaneously with VIMOS, and up to 170 with NIRMOS (10 arcsec slits). An integral field spectroscopy mode with more than 6400 fibers coupled to micro-lenses will be available for VIMOS, covering a 1 X 1 arcmin2 field. The VLT-VIRMOS survey of more than 150,000 galaxies is planned down to magnitudes IAB equals 24, coupled to an ultra deep probe to IAB equals 26.
Testing FORS: the first focal reducer for the ESO VLT
FORS1 (FOcal Reducer/low-dispersion Spectrograph) is an all dioptric focal reducer designed for direct imaging, low- dispersion multi-object spectroscopy, imaging polarimetry and spectro-polarimetry of faint objects. Two identical copies of the instrument (FORS 1 and 2) are being built by a consortium of three astronomical institutes (Landessternwarte Heidelberg and the University Observatories of Gottingen and Munich) under contract and in cooperation with ESO. FORS 1 and 2 will be installed, respectively, in 1998 and 2000 at the Cassegrain foci of the ESO VLT unit telescopes nos. 1 and 2. For the tests of FORS in Europe, a telescope and star simulator was built, which allows to incline and rotate the whole instrument and to simulate stars in the field of view at various seeing conditions. FORS 1 was integrated at the telescope simulator and saw its 'first light' in the integration facility in November 1996. Since then the electro-mechanical functions, the image motion due to flexure, the calibration units, the optical performance and the instrument software were tested and optimized. This paper presents a summary of the procedure and the results of the tests.