Production follow-up of the LHC main dipoles through magnetic measurements at room temperature
In this paper, we review the tools used for controlling the production of the LHC main dipoles through warm magnetic measurements. For the collared coil measurements, control limits are based on the statistics relative to the pre-series production. For the cold mass, the difference between collared coil and cold mass is considered, allowing a very stringent test. In both cases, measurements are split in straight part average, variations and coil ends contributions. Two different alarm levels exist in case the measured field is out of limits. The analysis can be carried out at the manufacturer and allows detection of anomalies in the measured magnetic field. These can be either due to wrong measurements or caused by assembly defects. Techniques used to work out information on the magnet assembly from the field harmonics are outlined. We summarize the experience gathered on about 180 collared coils and 120 cold masses, pointing out the bad cases and investigating the reliability of the measurements.
Modeling of random geometric errors in superconducting magnets with applications to the CERN Large Hadron Collider
Estimates of random field-shape errors induced by cable mispositioning in superconducting magnets are presented and specific applications to the Large Hadron Collider (LHC) main dipoles and quadrupoles are extensively discussed. Numerical simulations obtained with Monte Carlo methods are compared to analytic estimates and are used to interpret the experimental data for the LHC dipole and quadrupole prototypes. The proposed approach can predict the effect of magnet tolerances on geometric components of random field-shape errors, and it is a useful tool to monitor the obtained tolerances during magnet production.
rPET detectors design and data processing
Small animal PET systems based on rotating planar detectors posses some interesting advantages for high sensitivity, high resolution imaging. We have designed the rPET detectors based on MLS crystals assembled on a 30times30 matrix optically coupled to a flat-panel PS-PMT. Weighted position readout circuits pre-process the 64 signals from the 8times8 anodes matrix, which are digitized using a charge-integrating converter. The amplification electronics, including the trigger output for coincidence detection, and the high voltage supply are integrated in a three PCBs stack that forms the base attached to the back of the PMT. The whole assembly is enclosed in a light tight, lead (Pb) shielded aluminum box. The detectors are mounted on a rotating gantry with more than 180 degrees rotation span. The digitized events are screened and histogramed, and a modified center of gravity algorithm removes from the position calculation those signals with poor signal to noise ratio. Apparent mean crystal size on the 511 keV field-flood images is 0.6 mm, mean peak-to-valley ratio is better than 8, and intrinsic resolution is 1.5 mm at the central row, with the energy window wide open. Sensitivity (CPS) for a pair of these detectors set in coincidence at 160 mm distance is 1%