Gravitational eccentric correction optics (GECO): an optical-gravitational device to compensate for flexures in astronomical spectrographs.
Mechanical flexure is a source of major failures in astronomical spectrographs, for which the reimaging of a focal-plane pinhole has to be maintained in position within a fraction of a CCD pixel that has dimensions of the order of 15 microm. The d.o.lo.res. (an acronym for device optimized for low resolution) spectrograph for the Italian national telescope, Galileo, showed displacements of the image of the pinhole more than 10 times greater than expected. The mechanical failure was overcome by the insertion of a passive optical wedge that can add an out-of-phase circle to the flexure ellipse. The results encourage the use of the gravitational eccentric correction optics (GECO) optical-gravitational device in all astronomical observations made with the d.o.lo.res. spectrograph.
PAOLO: A Polarimeter Add-On for the LRS Optics at a Nasmyth focus of the TNG†
We describe a new polarimetric facility available at the Istituto Nazionale di AstroFisica / Telescopio Nazionale Galileo at La Palma, Canary islands. This facility, PAOLO (Polarimetric Add-On for the LRS Optics), is located at a Nasmyth focus of an alt-az telescope and requires a specific modeling in order to remove the time- and pointing position-dependent instrumental polarization. We also describe the opto-mechanical structure of the instrument and its calibration and present early examples of applications. (© 2014 WILEY-VCH Verlag GmbH & Co. KGaA, Weinheim)
Integral field spectroscopy with the GEMINI multiobject spectrographs
We describe the integral field unit (IFU) which converts the Gemini Multiobject Spectrograph (GMOS) installed on the Gemini-North telescope to an integral field spectrograph,which produces spectra over a contiguous field of view of 7 × 5 arcsec with spatial sampling of 0.2 arcsecover the wavelength range 0.4-1.0 μm.GMOS is converted to this mode by the remote insertion of the IFU into thebeam in place of the masks used for the multiobject mode. A separate fieldof half the area of the main field, but otherwise identical, is alsoprovided to improve background subtraction. The IFU contains 1500lenslet-coupled fibres and was the first facility of any type for integralfield spectroscopy employed on an 8/10 m telescope.We describe the design, construction and testing of the GMOS IFU and present measurements of the throughput both in the laboratory and at the telescope. We compare these with a theoretical prediction made before construction started. All are in good agreement with each other, with the on-telescope throughput exceeding 60% (averaged over wavelength). Finallywe show an example of data obtained during commissioning to illustrate the power of the device.