Ultrasonic NDE in a reactor core
We have irradiated bulk single crystal piezoelectric AlN operating as an ultrasonic transducer up to a fast and thermal neutron fluence of 1.85×1018 n/cm2 and 5.8×1018 n/cm2 respectively along with a gamma dose of 26.8 MGy. Our primary interest was in the radiation effects on the piezoelectric performance of the AlN crystal. Ideally standard procedures for extracting material properties from the impedance spectra would have been applied. However, this procedure was anticipated to be very error prone due the effects of radiation on passive components, such as electrodes and cabling. As a result the transducer was coupled to an aluminum cylinder and operated in pulse echo mode throughout the irradiation. In addition to the pulse echo testing impedance data were obtained. Although, material parameters such as d33 could not be measured in-situ during irradiation with our procedure, our work is the only publication, to the author's knowledge, showing the operation of an ultrasonic transducer in a reactor core. Further, the piezoelectric coefficient was measured prior to irradiation and found to be 5.5 pC/N which is unchanged from as-grown samples, and in fact higher than the measured d33 for many as-grown samples. Further proof of the viability of utilizing ultrasonic transducers in a reactor core is provided by means of successful time of flight (TOF) measurements throughout the irradiation. The TOF data was used to measure the gamma heating of an aluminum cylinder. The heating due to short lived isotopes in the Al was also detected via TOF.
Irradiation Testing of Ultrasonic Transducers
Ultrasonic technologies offer the potential for high accuracy and resolution in-pile measurement of numerous parameters, including geometry changes, temperature, crack initiation and growth, gas pressure and composition, and microstructural changes. Many Department of Energy-Office of Nuclear Energy (DOE-NE) programs are exploring the use of ultrasonic technologies to provide enhanced sensors for in-pile instrumentation during irradiation testing. For example, the ability of single, small diameter ultrasonic thermometers (UTs) to provide a temperature profile in candidate metallic and oxide fuel would provide much needed data for validating new fuel performance models. Other efforts include an ultrasonic technique to detect morphology changes (such as crack initiation and growth) and acoustic techniques to evaluate fission gas composition and pressure. These efforts are limited by the lack of existing knowledge of ultrasonic transducer material survivability under irradiation conditions. To address this need, the Pennsylvania State University (PSU) was awarded an Advanced Test Reactor National Scientific User Facility (ATR NSUF) project to evaluate promising magnetostrictive and piezoelectric transducer performance in the Massachusetts Institute of Technology Research Reactor (MITR) up to a fast fluence of at least 1021 n/cm2 (E>0.1 MeV). This test will be an instrumented lead test; and real-time transducer performance data will be collected along with temperature and neutron and gamma flux data. By characterizing magnetostrictive and piezoelectric transducer survivability during irradiation, test results will enable the development of novel radiation tolerant ultrasonic sensors for use in Material and Test Reactors (MTRs). The current work bridges the gap between proven out-of-pile ultrasonic techniques and in-pile deployment of ultrasonic sensors by acquiring the data necessary to demonstrate the performance of ultrasonic transducers.
Right coronary artery stenosis associated with tricuspid valve ring annuloplasty
A 70-year-old man with severe symptomatic functional mitral valve regurgitation underwent successful mitral valve repair combined with tricuspid valve ring annuloplasty. Pre-operative coronary angiography (CAG) showed no significant stenosis. One-and-a-half years later, the patient presented with an acute exacerbation of heart failure. Repeat CAG demonstrated tight stenosis in the right coronary artery (RCA) with arterial kinking that corresponded to the same point as the stenosis where the RCA is the closest to the tricuspid valve ring. The new lesion probably occurred as a consequence of the tricuspid valve ring annuloplasty.