Time of Flight Sensor with a Flow Parallel Wire
A time of flight sensor has been equipped with a sensing wire parallel to the flow direction (flow parallel wire, FPW). A heat pulse is generated with a coil in the flow channel. The FPW has a center tap allowing its upstream and downstream parts to join in a half bridge. When a heat pulse passes the FPW, a large output peak is generated. The time between heat pulse generation and recording the peak maximum is only marginally affected by the properties of the fluid. With a combination of two FPWs, a measuring range of approximately 0.01–0.5 m/s can be achieved.
A portable MEMS Coriolis mass flow sensor
In this paper, a micromachined Coriolis mass flow sensor is described. Key to portability, this resonant silicon sensor is vacuum packaged at the chip-level. A new MEMS gettering method was developed to accomplish high-vacuum packaging. Fluids flow through machined silicon microtubes. The tubes are driven electrostatically and monitored capacitively. Test data and applications are presented.
UV-B radiometry and dosimetry for solar measurements.
Biologically effective irradiance or dose of solar UV radiation was determined using a spectroradiometer, two broadband radiometers and two types of passive UV-dosimeters. The absolute erythema irradiance and the actinic irradiance were calculated from the solar spectrum measured with the spectroradiometer. It was demonstrated that the erythema irradiance is proportional to the actinic irradiance of solar radiation. The erythema irradiance was also determined using the two broadband radiometers which utilize a filter transmitting erythema spectra. Personal UV-dosimeters such as polysulphone and CR-39 were used to determine the erythema dose for a selected period of time. These results were used to estimate the accuracy of the broadband radiometers and UV-dosimeters. It was found that the results obtained from the broadband radiometers deviate from the absolute erythema irradiance by less than 20% during clear days between the hours of 11:00 and 13:00 Eastern Standard Time (EST) in Australia. The assessment of the erythema dose using passive dosimeters such as polysulphone and CR-39 could introduce an error as high as 40% if the calibration was not performed before undertaking experimental measurements.