The NAOS visible wave front sensor
This paper describes the Visible Wave Front Sensor (visible WFS) for the VLT Nasmith Adaptive Optics system (NAOS). This Shack-Hartman-based wave front sensor instrument includes within a continuous flow liquid nitrogen cryostat: - a low noise fast readout CCD camera controlled by the ESO new generation CCD controller FIERA. The readout noise of this system is 3 e- at 50 kilopixel/sec/port, and is only limited by the CCD intrinsic noise. FIERA proposes remotely controlled readout modes with optional binning, windowing and flexible integration time. - two remotely exchangeable micro-lens arrays focusing the analyzed wave front directly on the CCD sensitive surface. The wave front sensor includes also its own atmospheric dispersion compensator. Due to the continuous rotation of the NAOS adapter, the mechanical stiffness of the visible wave front sensor must be very high not to disturb the loop operation (no more than 0.1 μm of lenslet array displacement compared to the CCD location over a 30° rotation angle of the instrument). The following simulations and tests are described: - simulation results providing an estimation of the NAOS maximum operating magnitude - camera optimization - mechanical stiffness measurements.
CONICA: the high-resolution near-infrared camera for the ESO VLT
A high resolution near IR camera (CONICA) for the firs VLT unit is under development, which will provide diffraction limited spatial resolution being combined with the adaptive optics system NAOS. CONICA serves as a multi-mode instrument for the wavelength region between 1.0 and 5.0 micrometers , offering broad band, narrow band or Fabry Perot direct imaging capabilities, polarimetric modes using Wollaston prism or wire grid analyzers and long slit spectroscopy up to a spectral resolution of about 1000 per two pixel. We presented a first concept of CONICA in 1995. In the mean time, large parts of the instrument have been manufactured, the cryostat and the adapter have been finished and first cryogenic test have been performed. This paper describes the actual design and status of development of CONICA focusing on those aspects which have not been described in detail before or the design of which have been changed in the mean time.
NAOS real-time computer for optimized closed-loop and online performance estimation
The Real Time Computer RTC is a key component of the Nasmyth Adaptive Optics System, controlling the 185 actuators of the deformable mirror from a 144 Shack-Hartmann subapertures wavefront sensor at a maximum frequency of 500 Hz. It also provides additional capabilities such as real time optimization of the control loop which is the warranty for NAOS to achieve a very good Strehl Ratio in a broad magnitude range (Mv equals 8 up to 18), on-line turbulence and performance estimations and finally capability to store and process the data necessary to the off-line PSF reconstruction algorithm. This RTC is also designed to be easily upgraded as for Laser Guide Star. Moreover all softwares can be easily adapted to control a curvature sensor as well as the hardware which can be used with the two types of wave front sensors.