Enhancement of contrast in living and fixed specimens by the use of fibre optics
A method is described which enhances the contrast of living and fixed specimens examined with the stereomicroscope. It consists of immersing the ends of flexible fibre optic light sources together with the specimen in the fluid used for examination. It is reported that not only does this method increase the contrast of living specimens but that it may also be applied to specimens being prepared as thin sections or freeze fracture surfaces for examination with the transmission electron microscope. A further method of enhancement of contrast is suggested which involves the fitting of light filters of complementary colours, one to each of the fibre optic light sources, before immersion with the specimen.
A fibre optic ring illumination system for use in low‐powered dark field (including Rheinberg method) video, cine and photomicrography
A fibre optic ring illuminator normally used as a reflected light source with a low‐powered microscope is here used as a substage illuminator with a stereo zoom microscope, to give uniform dark field (dark ground) illumination. This development was primarily for use in colour video recordings giving good contrast pictures but can also be used successfully in cine and photomicrography, at magnifications ranging from x 1ṁ5 to x 124. Its application is particularly suitable for the observation of biological specimens.
New Low-Profile Applicators for Local Heating of Tissues
A new type of low-profile applicator is described giving efficient power transfer to tissue with low leakage. It consists of a resonant element spaced from a ground plane by low-loss material having high relative permittivity and some relative permeability. This combination is covered by similar material which can be placed in direct contact with the tissue to be heated, or separated from it by a water bolus as required. Laboratory measurements using phantom materials show that these applicators are well matched and produce well defined and relatively uniform heating profiles. Applicators having maximum dimensions of 3-15 cm covering the frequency range 1000-100 MHz, respectively, are described. Penetration depths of at least 4 cm at 100-200 MHz are achieved. Tests on anesthetised pigs and some clinical assessment have confirmed the usefulness of this compact low-frequency type of applicator and have shown that performance compares favorably with that of other recently reported types. The design can be optimized for treatment of specific sites, and suitable choice of dimensions and electrical characteristics of the layer material allow the low-profile concept to be extended over a wide range of operating frequencies.