Measurement of the response-ratio of an ionization chamber filled with dry air to that filled with P-10 gas in the calibration of gas monitors.
We developed a new method to determine the response ratio of an ionization chamber (Ohkura ionization chamber: type I-4096 01/06) filled with dry air to that filled with P-10 gas. The response-ratio was obtained by comparing the output currents of the ionization chamber after introduction of radioactive gasses of identical activities contained in two small vessels. The obtained ratio was 0.690 ± 0.003, which agreed with the results obtained by other methods.
Activity measurement of 85Kr diluted by a large volume balloon technique.
The National Metrology Institute of Japan (NMIJ) participated in the CCRI(II) Key-comparison of (85)Kr activity measurement. The absolute internal gas-counting method was used, with a set of proportional counters of different length. The original (85)Kr gas was diluted by a factor of 2x10(5) in order to achieve the suitable activity concentration for the proportional counters. A new dilution method was introduced, based on a large-volume balloon and gravimetric determination of the dilution volume with small uncertainties.
DEM generation from new VIR sensors: IKONOS, ASTER and Landsat-7
DEM generation is a living R&D topic since new-launched satellite images can be used to generate stereoscopic images. Using a coplanarity equation model developed at CCRS and an automatic matching method developed at PCI, DEMs are generated with new VIR stereo data. Accuracy of 4.9 m with high-resolution IKONOS images, and 33 m and 101 m with medium-resolution ASTER and Landsat-7 ETM/sup +/ images respectively are obtained. The same matching accuracy of one-pixel is translated into worse results with Landsat due to a small B/H ratio of 0.15.