Laser ranging error budget for the TOPEX/POSEIDON satellite.
A laser ranging error budget is detailed, and a specific error budget is derived for the TOPEX/POSEIDON satellite. A ranging uncertainty of 0.76 cm is predicted for TOPEX/POSEIDON at 20 degrees elevation using the presently designed laser retroreflector array and only modest improvements in present system operations. Atmospheric refraction and satellite attitude effects cause the predicted range error to vary with satellite elevation angle from 0.71 cm at zenith to 0.76 cm at 20 degrees elevation. This a priori error budget compares well with the ~1.2-cm rms a posteriori polynomial orbital fit using existing data taken for an extant satellite of similar size and orbit.
Pulse spreading and range correction analysis for satellite laser ranging.
The pulse spreading resulting from light detection and ranging measurements of the range to earth-orbiting satellites is described. An analysis quantifying this pulse spreading and the calculation of corrections to be applied to the lidar range determination of satellites is detailed.
Pulsed multiwavelength laser ranging system
A pulsed multiwavelength laser ranging system for measuring atmospheric delay was built and tested, and its theoretical performance limits were calculated. The system uses a dye modelocked ND:YAG laser, which transmits 70 psec wide pulses simultaneously at 1064, 532, and 355 nm. The differential delay of the 1064 and 355 nm pulses is measured by a specially calibrated waveform digitizer to estimate the dry atmospheric delay. The delay time of the 532 nm pulse is used to measure the target distance. Static crossed field photomultipliers are used as detectors for all wavelengths. Theoretical analysis shows that path curvature and atmospheric turbulence are fundamental limits to the ranging accuracy of both single and multicolor systems operating over horizontal paths. For two color systems, an additional error is caused by the uncertainty in the path averaged water vapor. The standard deviation of the multicolor instrument's timing measurements is directly proportional to the laser pulse width plus photomultiplier jitter divided by the square root of the received photoelectron number. The prototype system's maximum range is km, which is limited by atmospheric and system transmission losses at 355 nm. System signal detection and false alarm calculations are also presented.