A simple, fast, and accurate thermodynamic-based approach for transfer and prediction of gas chromatography retention times between columns and instruments Part III: Retention time prediction on target column.
This is the third part of a three-part series of papers. In Part I, we presented a method for determining the actual effective geometry of a reference column as well as the thermodynamic-based parameters of a set of probe compounds in an in-house mixture. Part II introduced an approach for estimating the actual effective geometry of a target column by collecting retention data of the same mixture of probe compounds on the target column and using their thermodynamic parameters, acquired on the reference column, as a bridge between both systems. Part III, presented here, demonstrates the retention time transfer and prediction from the reference column to the target column using experimental data for a separate mixture of compounds. To predict the retention time of a new compound, we first estimate its thermodynamic-based parameters on the reference column (using geometric parameters determined previously). The compound's retention time on a second column (of previously determined geometry) is then predicted. The models and the associated optimization algorithms were tested using simulated and experimental data. The accuracy of predicted retention times shows that the proposed approach is simple, fast, and accurate for retention time transfer and prediction between gas chromatography columns.
A standardized method for the calibration of thermodynamic data for the prediction of gas chromatographic retention times.
A new method for calibrating thermodynamic data to be used in the prediction of analyte retention times is presented. The method allows thermodynamic data collected on one column to be used in making predictions across columns of the same stationary phase but with varying geometries. This calibration is essential as slight variances in the column inner diameter and stationary phase film thickness between columns or as a column ages will adversely affect the accuracy of predictions. The calibration technique uses a Grob standard mixture along with a Nelder-Mead simplex algorithm and a previously developed model of GC retention times based on a three-parameter thermodynamic model to estimate both inner diameter and stationary phase film thickness. The calibration method is highly successful with the predicted retention times for a set of alkanes, ketones and alcohols having an average error of 1.6s across three columns.
Direct evidence for the single cell origin of mouse liver cell tumours.
ImagesFigure 1Figure 2Figure 3