Headspace solid phase microextraction – GC/C-IRMS for δ13CVPDB measurements of mono-aromatic hydrocarbons using EA-IRMS calibration
This work aims at comparing the δ13CVPDB of mono-aromatic hydrocarbons benzene, toluene, ethylbenzene and xylene isomers (BTEX) measured by elemental analyser (EA)-isotope ratio mass spectrometer (IRMS) with the δ13CVPDB measured on the same compounds by headspace solid phase microextraction – GC/C-IRMS (hSPME – GC/C-IRMS) with the final goal of using these compounds as internal standards on the latter system. The EA-IRMS measurements were done using calcium and lithium carbonate isotopic reference materials: NBS19 and L-SVEC for establishing the δ13CVPDB scale. The EA-IRMS measurements with helium dilution of a set of five reference materials (USGS40, USGS41, IAEA-CH-6, IAEA-CH-3 and IAEA-601) show systematic bias of 1‰ relative to their assigned values. This bias due to the dilution mechanism in the used ConfloII interface device could not be avoided. As the selected hydrocarbons: BTEX could not be analysed by EA-IRMS without helium dilution, their δ13CVPDB must be corrected from this observed bias using an external calibration. The CO2 gas calibrated using EA-IRMS without helium dilution, was used as an in-house reference for the δ13CVPDB measurements of the BTEX by the hSPME – GC/C-IRMS system. The comparison made between the δ13CVPDB measured on the same BTEX compounds by EA-IRMS (with external calibration) and by hSPME – GC/C-IRMS techniques showed good agreement.
ConFlo III – an interface for high precision δ13C and δ15N analysis with an extended dynamic range
A newly developed interface coupling a CHN combustion device (elemental analyser 'EA') to an isotope ratio mass spectrometer is described and evaluated. The purpose of the device is to extend the dynamic range of delta(13)C and delta(15)N analysis from less than 2 orders of magnitude to more than 3 orders of magnitude. Carbon isotope ratio measurements of atropine as a model compound have been performed analysing between 1 µg to 5 mg C with acceptable to excellent precision (0.6 to 0.06 per thousand, delta-notation). The correction due to the blank signal is critical for sample amounts smaller than 4 µg C. The maximum sample weight is determined by the combustion capacity of the EA. Larger sample amounts are measured using dilution of a small part of the EA effluent with helium. The dilution mechanism works virtually free of isotope fractionation. Copyright 1999 John Wiley & Sons, Ltd.
Complex thoracic vascular injury repair using deep hypothermia and circulatory arrest.
A 61-year-old man with a penetrating injury to the innominate artery, left common carotid artery, and left subclavian artery at their origins from the aortic arch with associated injuries to both innominate veins and an innominate artery to vein fistula after a single stab wound is described. The patient was managed successfully using cardiopulmonary bypass together with deep hypothermia and circulatory arrest. Presentation and management are discussed.