Multiply diffracted light in the Czerny-Turner spectrometer.
At certain small angles of the grating, any in-plane spectrometer will pass stray light that has been returned by the reflecting optics for a second diffraction. Paths of this source of stray light and its strueture have been calculated and confirmed experimentally.
Reduction of stray light in in-plane grating spectrometers.
The stray light in a commercial Czerny-Turner grating spectrometer has been investigated at wavelength readings between 1800 A and 5000 A. A simple method is described which eliminates that part of the stray light caused by multiple dispersions. This same method should also decrease stray light in other in-plane grating spectrometers with reflecting optics.
Über den Astigmatismus bei Spiegelspektrometern
ZusammenfassungBei Spiegelspektrometern kSnnen je naeh der Anordnung des Strahlenganges die Fehler, die der erste Hohlspiegel in den Strahl~ngang bringt, dureh den zweiten Hohlspiegel gesteigert oder teilweise rtickgi~ngig gemaeht werden. Der Untersehied in der Strahlenvereinigung bei den beiden Fallen wird an photographisehen Aufnahmen gezeigt. Viele in der Praxis benutzte Anordnungen entsprechen dem ungiinstigen Falle.