Technical pitfalls in middle ear gas studies: Errors introduced by the gas permeability of tubing and additional dead space
Conclusions We demonstrated errors introduced by the gas permeability of the tubing and additional dead space. Materials with practically no permeability, such as glass, must be used to overcome the loss or gain of gas through the walls of tubes used for studying ME gas variations. Experiments conducted at a constant volume (variable pressure) require the smallest possible tubing volume in order to obtain good sensitivity and improve the accuracy of the results. Objectives Experimental studies that investigate middle ear (ME) gas exchanges, using either pressure measurements or volume changes, are conducted using various tubing connections between the ME and a measuring device. The aims of this study were to highlight experimental problems due to the errors introduced by (i) the gas permeability of the tubing used and (ii) additional dead space in experiments conducted at constant volume. Material and methods The problem of the gas permeability of the tubing was addressed by comparing three types of tube (silicone, polyethylene, PVC) with a glass tube. Horizontally placed tubes were connected to a syringe filled with pure CO2 via a gas-tight valve. The end of each tube tested was plunged into colored water (5% Coomassie Brilliant Blue R-250). The tube was washed out with CO2 until gas bubbles were seen leaving it. The valve was then closed and the movement of water in the tube was observed. The same experiments were repeated for pure O2. The problem of the error introduced by the additional dead space volume was addressed at a theoretical level using the well-known gas laws. Two conditions were studied: condition A, in which the experiment was conducted at constant pressure and the volume was measured; and condition B, in which the experiment was conducted at constant volume and the pressure was measured. The main outcome measure of each condition was the calculated variation in the final number of moles of gas involved. Results No water movement was observed in glass tubes. In contrast, plastic tubes exhibited significant gas permeability effects for both CO2 and O2. The colored solution advanced at a faster rate with CO2 than O2 but differently for each type of tubing. For condition A, gas transfer was independent of the volume of the external tubing and was accurately measured by the displacement of the droplet in the lumen. In contrast, for condition B, the pressure variations were influenced by the volume of the tubing.
Volume of Mastoid Pneumatization: Three-Dimensional Reconstruction with Ultrahigh-Resolution Computed Tomography
The volume of the mastoid air cell system was measured in 69 patients with normal middle ears. All patients underwent axial ultrahigh-resolution computed tomography. Mastoid pneumatization was marked on each axial slice, and 3-dimensional reconstruction was performed. The volumes were measured with a volumetric algorithm. A polyethylene tubing phantom with a density similar to that of bone on computed tomography was devised. The polyethylene tubing was tied in a particular fashion so as to create interconnecting air spaces with a known volume. The phantom was scanned with the imaging parameters used for scanning the temporal bone. The air in the tubing was marked, and 3-dimensional reconstruction for the marked phantom air was performed. The volume of the interconnecting air spaces was measured and found to be identical to its known volume, thereby verifying the accuracy of the method used. The mean mastoid volume was 6.61 cm3. The smallest volume measured was 1.3 cm3, and the largest was 12.7 cm3. The importance of this technique lies in its high accuracy, ease of use, and ability to directly correlate mastoid size and clinical findings.
Trends in kiln drives
The heart of the cement plant is the rotary kiln, and continuous quality production of cement clinker depends on the continuous accurate performance of the rotary kiln drive.