1 Recent Radiation Test Results at JPL
This Bruce E. Pritchard, Member, ZEEE, Bemard G. Rax, and Steven S. McClure paper documents recent TID (including ELDRS) and proton damage test results obtained by JPL. Unusual test results, such as abnormally low or high failure levels or unusual failure or response mechanism, are emphasized.
Total ionizing dose (TID) evaluation results of low dose rate testing for NASA programs
Low dose rate radiation testing in the range of 0.01-0.20 rads(Si)/s has been performed at Goddard Space Flight Center on a wide variety of non-RH (rad-hard) part types in the last few years. This paper reports on recent test results of EEPROMs, Operational Amplifiers, Analog-to-Digital Converters (ADCs) and Digital-to-Analog Converters (DACs), and DC-DC converters. These results have shown that non-RH parts are suitable for many low earth orbiting satellites, where predicted mission total dose requirements may be in the range of 0.5-10 krads(Si). However, a careful characterization of non-RH parts should be performed to verify that the parts meet the requirements of their intended application in the projected mission radiation environment.
rPET detectors design and data processing
Small animal PET systems based on rotating planar detectors posses some interesting advantages for high sensitivity, high resolution imaging. We have designed the rPET detectors based on MLS crystals assembled on a 30times30 matrix optically coupled to a flat-panel PS-PMT. Weighted position readout circuits pre-process the 64 signals from the 8times8 anodes matrix, which are digitized using a charge-integrating converter. The amplification electronics, including the trigger output for coincidence detection, and the high voltage supply are integrated in a three PCBs stack that forms the base attached to the back of the PMT. The whole assembly is enclosed in a light tight, lead (Pb) shielded aluminum box. The detectors are mounted on a rotating gantry with more than 180 degrees rotation span. The digitized events are screened and histogramed, and a modified center of gravity algorithm removes from the position calculation those signals with poor signal to noise ratio. Apparent mean crystal size on the 511 keV field-flood images is 0.6 mm, mean peak-to-valley ratio is better than 8, and intrinsic resolution is 1.5 mm at the central row, with the energy window wide open. Sensitivity (CPS) for a pair of these detectors set in coincidence at 160 mm distance is 1%