Displacement Determination of the Guard Electrode for the New Calculable Capacitor at NIM
We present our continuing work on the displacement determination of the guard electrode for the new vertical calculable capacitor. We describe a method for determining the displacement by actively locking the Fabry-Perot interferometer to a stabilized laser at two cavity lengths of interest. In order to lock the interferometer tightly, the modulation width of the laser is optimized for the full-width at half-maximum at the two different cavity lengths. For each of the cavity lengths, the Fabry-Perot interferometer is maintained at its resonance, and the cavity length is quantized with a space of λ/2. Then, its corresponding ratio between the fixed standard capacitor and the calculable capacitor is accurately measured by a capacitance bridge. The two main parameters for determining the displacement between the two cavity lengths-i.e., the difference in order number (AN) of the interferometer and the laser wavelength (λ)-are obtained from the capacitance bridge measurement based on a 1-pF standard capacitor with a priori value and the accurate wavelength measurement of an iodine laser, respectively. Evaluation of the displacement sensing system confirms a relative uncertainty of 5.2 × 10-9 over a range of 205 mm.
Improvements on Laser Interferometer in Calculable Capacitor at NIM
This paper illustrates the improvements on the laser interferometer in the calculable capacitor at nim, including the fiber coupling optical layout, mechanical drift and aging monitor, and parasitic interference reduction. Using the new fiber coupling, we can more flexible to change the laser source without break the beam layout. And the reduction of parasitic interference is more than one order in amplitude with the benefit of antireflection coatings.
Contribution of the Gouy phase to two-pathway coherent control of the photoionization and photodissociation of vinyl chloride.
The electric field of a light wave accumulates a pi phase shift as it passes through a focus. We show here how this effect, known as the Gouy phase, may be used to control the branching ratio of a unimolecular reaction when the products are formed with different numbers of photons. We demonstrate this control method for the ionization and dissociation of vinyl chloride, using absorption of 177 and 532 nm photons to induce a pair of interfering paths. Excellent agreement between the observed and calculated phase shift as a function of the axial coordinate of the laser focus indicates that fragmentation occurs via a ladder switching mechanism. The axial dependence of the modulation depth is evidence of loss of coherence at higher internal temperatures of the molecule.