Bias in the measurement of radon gas using ionization chambers: Application to SIR.
Two main non-destructive techniques can be used to measure standard 222Rn gas ampoules: well-type ionization chambers and gamma-ray spectrometry, the former being used in the Système International de Référence (SIR) for international comparison purposes. The reliability of these techniques requires that the variability of the flame-sealed gas glass ampoules used have a negligible influence on the detector response. This variability is studied in this work by considering three parameters: the volume of the ampoule, the position of the sealing point and the thickness of the glass. Results showed that variability of the gas ampoules induced measurement bias larger than the uncertainty of the standard sources.
Simulation of the response of an ionization chamber to 214Bi emission. Application to the measurement of 222Rn.
PENELOPE simulations of a Vinten ionization chamber (IC) were performed to investigate the influence of the thickness of glass-ampoules used in 222Rn standardization. The simulation reveals a non-negligible variation of the energy deposited in the chamber gas region (which may induce a proportional variation of the measured current) when considering the β transition emissions of the daughters of 222Rn. This reinforces the idea of using a specialist container (made of metal to preserve the integrity of the container) that would circulate between the metrology laboratories in the context of international comparison exercises using the BIPM international reference system (SIR).
l-Phenylalanine dehydrogenase from Brevibacterium sp. for production of l-phenylalanine by reductive amination of phenylpyruvate
SummaryFor production of l-phenylalanine the reductive amination of phenylpyruvate, catalyzed by phenylalanine-dehydrogenase was examined. To reach high levels and a sufficient stability of the inducible intracellular enzyme, growth conditions of Brevibacterium sp. are optimized. For continuous production of l-phenylalanine in an enzyme membrane reactor, the kinetic parameters of the partially purified enzyme are determined.In continuous production a space time yield of 37.4 g l-Phe l-1 d-1 can be reached.By means of the measured kinetic parameters and simultaneous calculation of the mass balances of all reaction components the behaviour of the reactor can be simulated. For certain conditions the multi-enzyme-system shows multiple steady-states.