Kinetics of the aeration of ethylene-oxide sterilized plastics.
The aeration of ethylene-oxide (EO) sterilized medical plastic materials is dealth with mathematically, and subsequently the model found is used for calculation of the EO diffusion coefficient in various plastics with the composition and temperature as variables. Absorption and dissolution of EO in various plastic materials are measured at varying concentrations of EO, temperatures and compositions of plastics. The mathematical model is used to evaluate the EO desorption from medical devices under different physico-chemical conditions as well as in vivo. EO analysis methods are evaluated on the basis of the model and are compared with experimental results.
Aeration of medical plastics.
Because the metabolic reaction that gives ethylene oxide is sterilizing qualities also takes place with human protein, it is imperative that any residual gas be removed before the sterilized material is used on the patient. This paper presents a method of estimating aeration time required following sterilization of polyvinyl choride (PVC) to reduced toxic residues to a desired level. The method proposed involves calculation of the coefficient of diffusivity, and the use of a simplified diffusion equation to estimate aeration time.
Failure analysis for poor solderability of Au surface of PCB
Nonwetting is one of the common problems and is always rejected in electronic assemblies. ENIG pads of a PCBA exhibited nonwetting after lead-free reflow soldering. With the aid of stereomicroscope, SEM&EDS and XRF, a series of analysis was performed. And after visual inspection, surface and cross section analysis, thickness measurement and solderability test, it can be inferred that oxidation of plating surface was the main reason for nonwetting, and corrosion in Ni plating caused an adverse impact on solderalibity.