Engineering the Alcator C-Mod MSE diagnostic: Solutions to reactor-relevant diagnostic challenges
Deploying precise optical diagnostics in harsh reactor environments poses significant engineering challenges including limited access, large forces, time-varying systematic errors, reflections, low signal-to-background levels and a need for in-situ calibration. Fielding a polarization-based Motional Stark Effect (MSE) diagnostic on Alcator C-Mod has proven to be difficult due to similar challenges, which has led to the development of solutions that enable polarization angle measurements with an accuracy of 0.2 degrees.
A motional stark effect instrument to measure q(R) on the C-mod tokamak
A motional Stark effect (MSE) instrument to measure q(r) using the Torus Experiment for Technology (TEXT) neutral beam on the Alcator C-Mod tokamak has been designed and installed. The neutral beam is on the midplane, aimed radially, and designed to operate in hydrogen at 50 keV. The MSE optics view the outer half of the plasma −0.3<(R−R0)/a<1.05. This geometry results in a spatial resolution of 1 (edge)–3 (center) cm from a beam apertured horizontally to 2 cm. Estimates of the signal levels indicate for Hα that photoelectron statistical errors in the measurement of Bθ/Bφ can be less than 0.2% for ne0<2.0×1014 cm−3 similar to other MSE instruments. The collection optics reside within the vacuum chamber reflecting and imaging the neutral beam through a vacuum window in the neutral beam port. The optics are designed so that beam emission spectroscopy can be carried out simultaneously with MSE measurements. Optical signals are transmitted to remote detectors by fiber optics. Low Verdet glass and dielectric m...
Thermal incompatibility analysis of metal-ceramic systems based on flexural displacement data.
The feasibility of simple tests or analytical methods for prediction of residual stress states in metal-ceramic (MC) prostheses has not been demonstrated. Biomaterial metal-ceramic strips have been proposed to provide sensitive measures of transient and residual stress states through the measurement of midpoint deflection after cooling from the ceramic sintering temperature. The objective of this study was to apply the elastic-viscoelastic analogy to calculate transient and residual midpoint deflections in MC biomaterial strips and to compare these values with deflections measured with a beam-bending viscometer (BBV). Calculations and measurements were made for five MC systems that were found from a clinical study to be "thermally compatible" systems. Metal strips, 64 mm in length, 3 mm wide, and either 0.5 mm, 1.0 mm, or 2.0 mm in thickness, were veneered with four 0.25-mm thick layers of opaque porcelain. Midpoint deflection of the MC strips (ceramic oriented in the posterior position) was measured during cooling from an initial temperature of 700 degrees C. In general, the directions of the measured residual deflections did not agree with the "textbook" convention that negative deflections are associated with positive thermal contraction mismatch (alpha(M) - alpha(c) > 0) regardless of metal thickness. For a metal thickness of 0.5 mm, the residual midpoint deflection for all thermal contraction mismatch cases, except one, was positive (upward deflection) whereas the residual midpoint deflections were all negative when the metal thickness was increased to 1 or 2 mm, independent of the thermal contraction mismatch. The best agreement between calculated and measured values of residual midpoint deflection (+16 microns vs. +14 +/- 2.3 microns, respectively was obtained for MC biomaterial strips with a Ni-Cr alloy (0.5 mm thick) while the largest difference (+346 microns vs. +61 +/- 43.8 microns) was obtained for MC bimaterial strips with a Au-Pd allow (0.5 mm thick). In all but one case, changes in deflection direction as a function of metal thickness were correctly predicted by the viscoelastic analysis. The results of this study indicate that a viscoelastic model is useful for estimating thermal compatibility conditions of MC systems.