Multiply diffracted light in the Czerny-Turner spectrometer.
At certain small angles of the grating, any in-plane spectrometer will pass stray light that has been returned by the reflecting optics for a second diffraction. Paths of this source of stray light and its strueture have been calculated and confirmed experimentally.
High-Throughput, High-Resolution Echelle Deep-UV Raman Spectrometer
We constructed an ultrahigh-throughput, high-resolution ultraviolet (UV) Raman spectrograph that utilizes a high-efficiency filter-stage monochromator and a high-dispersion Echelle spectrograph. The spectrograph utilizes a total of six mirrors and two gratings, with an overall efficiency at 229 nm of ∼18%. The limiting resolution of our spectrometer is 0.6 cm−1 full width half-maximum (FWHM), as measured for 229 nm Rayleigh scattering. Use of a 1 mm–wide entrance slit gives rise to an approximately 10 cm−1 FWHM resolution at 229 nm. The ultrahigh spectrograph throughput enables ultrahigh signal-to-noise ratio, deep UV Raman spectra that allow us to monitor <1% changes in peptide bond composition. The throughput is measured to be 35-fold greater than conventional deep UV Raman spectrometers.
UV-B radiometry and dosimetry for solar measurements.
Biologically effective irradiance or dose of solar UV radiation was determined using a spectroradiometer, two broadband radiometers and two types of passive UV-dosimeters. The absolute erythema irradiance and the actinic irradiance were calculated from the solar spectrum measured with the spectroradiometer. It was demonstrated that the erythema irradiance is proportional to the actinic irradiance of solar radiation. The erythema irradiance was also determined using the two broadband radiometers which utilize a filter transmitting erythema spectra. Personal UV-dosimeters such as polysulphone and CR-39 were used to determine the erythema dose for a selected period of time. These results were used to estimate the accuracy of the broadband radiometers and UV-dosimeters. It was found that the results obtained from the broadband radiometers deviate from the absolute erythema irradiance by less than 20% during clear days between the hours of 11:00 and 13:00 Eastern Standard Time (EST) in Australia. The assessment of the erythema dose using passive dosimeters such as polysulphone and CR-39 could introduce an error as high as 40% if the calibration was not performed before undertaking experimental measurements.