The analysis of research results concerning heat release rates of ship materials with regard to dynamic parameters of the research station
The analysis of research results concerning heat release rates of ship materials with regard to dynamic parameters of the research station Based on the results of author's research it was stated that significant differences between heat release rates measured according to ISO 5660-1 and FTP Code Part 5 (Res. A. 653 (16)) are not caused by measurement uncertainties. A basic factor determining the lack of correlation is treating the heat release rate in the category of static measurements in the normalized measurement methods. Taking into account dynamic parameters of the test station allowed the coincidence rate (reproducibility and correlation) of the results of heat release rates measured using different methods to be improved, without changing test stations and procedures. Author's research performed on 42 materials has revealed that the relative dynamic errors of the heat release rate measurements are significant (2%÷120%). The average value of these changes for the tested group of materials was equal to 39%.
A method to improve repeatability and reproducibility of the results of examination of the rate of heat released by shipbuilding materials
ABSTRACT A specialised integrated research rig has been designed and built to validate the method of heat release dynamic error correction based on the FTP Code, part 5, and to control the rig operation and calibration. The basic rig component is a programmed gas burner with an option to select the function and parameters of burner’s operation. The use of the programmed burner increased the level of consistency (reproducibility and correlation) of the results of heat release rate measurements
Development of the cone calorimeter—A bench-scale heat release rate apparatus based on oxygen consumption†
A new bench-scale rate of heat release calorimeter utilizing the oxygen consumption principle has been developed for use in fire testing and research. Specimens may be of uniform or composite construction and may be tested in a horizontal, face-up orientation, or, for those which do not melt, in a vertical orientation. An external irradiance of zero to over 100 kW m−2 may be imposed by means of a temperature-controlled radiant heater. The rate of heat release is determined by measuring combustion product gas flow and oxygen depletion, while the mass loss is also recorded simultaneously. The instrument has been designed to be capable of higher accuracy than existing instruments and yet to be simple to operate and moderate in construction cost. The instrument is thermed a ‘cone calorimeter’ because of the geometric arrangement of the electric heater.