Some observations on the breakage of ultrasonic files driven piezoelectrically.
The incidence of breakage of Piezon-Master ultrasonic K files were evaluated. Three groups of unused files were subjected to three treatments, namely; free vibration in air without irrigation, free vibration in root canal while minimizing contact with the wall of canal in the presence of irrigation and light filing in root canal with free flow of irrigation. Cavitation produced by files in contact and free of contact with a glass surface was examined in order to observe the relationship between cavitation defects and breakage. In addition, the fractured and unfractured files were examined under a scanning electron microscope for the presence of cavitation pits. The results indicated that more files broke in air. In water, a higher incidence of breakage occurred when files were allowed to freely vibrate while no breakage occurred when the files were used in filing. All files generated cavitation which resulted in pitting of their surfaces. However, it was considered unlikely that the pits contributed to fracture. Fatigue cracks which could be the result of the manufacturing process were observed at some of the corners of the cross sections of the fractured files and could be the main contributory factor to fracture.
The vibratory pattern of ultrasonic files driven piezoelectrically.
The pattern of oscillation of a Piezon-Master 400 ultrasonic file driven by a piezoelectric transducer was studied in air and on water. In addition, the displacement amplitudes of the files were measured. The findings were compared with those observed with the Cavi-Endo unit reported in another study (Ahmad 1969). It was observed that the file vibrated such that a standing wave was formed on the file and it exhibited points of maximum deflection (antinode) and points of minimum deflection (node) with the largest deflection occurring at the apical end. This pattern of oscillation was similar to that exhibited by the Cavi-Endo file which employed a magnetostrictive transducer. However, the displacement amplitudes were very much higher than those exhibited by the Cavi-Endo. It is considered that the 120 degrees angle of the file holder inherent in the Piezon-Master 400 unit and the more effective power transmission with the piezoelectric transducer may have contributed to the large amplitudes.
A scanning electron microscope study comparing the efficacy of hand instrumentation with ultrasonic instrumentation of the root canal.
Hand instrumentation and ultrasonic instrumentation were used to prepare the root canals of freshly extracted teeth. Assessment of the efficacy of these techniques for debriding root canals was made by scanning electron microscopy. No difference in the appearance of the root canals was observed when comparing the two techniques.