Angles and Range: Initial Orbital Determination with the Air Force Space Surveillance Telescope (AFSST)
Abstract : The institution of a robust, comprehensive program of Space Situational Awareness (SSA) necessarily includes observations by ground-based optical and infrared (OIR) telescopes. The Air Force Space Surveillance Telescope (AFSST) has been proposed as a system of telescopes designed to address the ground-based component of comprehensive SSA. A hallmark of our definition of AFSST is that it be composed of inexpensive, replicable telescopes programmed to accomplish multiple surveillance programs, but designed to achieve the difficult design-driving task of accomplishing an uncued search for small objects in Low Earth Orbit (LEO). We discuss the configuration of AFSST and describe one technique by which AFSST simultaneously acquires angle and range data for accurate real-time determination of LEO orbital elements, even upon first detection of the object.
Laser remote sensing : fundamentals and applications
Electromagnetic Theory of Radiation. Quantum Physics and Radiation Processes. Interaction and Propagation of Radiation. Laser Fundamentals. Laser Systems as Remote Sensors. Laser-Remote-Sensor Equations. Analysis and Interpretation of the Lidar Return Signals. Atmospheric Lidar Applications. Hydrographic Lidar Applications. Concluding Remarks. Index.
Subaru next-generation wide-field camera: HyperSuprime
We summarize the design and the specification of a next generation instrument for Subaru Telescope: a very wide-field (2°φ) CCD camera which we name HyperSuprime. The latest design of the corrector ensures 80% encircled energy diameter of 0".3 from 600 nm to 1100 nm over the 2°φ field of view. The size of the focal plane is 612 mm in diameter and covered by about 170 four side buttable 2kx4k CCDs. Fully depleted CCD which is now being developed is the primary candidate for HyperSuprime. The readout electronics is connected behind the CCD and this CCD package is screwed to the cold plate with three positioning pins. The large entrance window of the dewar is supported with additional ribs so that the dewar is evacuated and CCDs are cooled down to about -80°C. HyperSuprime equips with a filter exchanger which can accommodate four large mosaicked filters and a roll-type shutter.