Magnitude and phase of three-dimensional (3D) velocity vector: Application to measurement of cochlear promontory motion during bone conduction sound transmission
&NA; Recent measurements of vibrational motion to assess sound transmission through ear structures and skull contents have included three‐dimensional (3D) behavior. The 3D motion of a point has been described with the three orthogonal components in the 3D space. In this article, a method to represent the 3D velocity with the magnitude and phase of the resultant velocity is introduced. This method was applied to the measurement of cochlear promontory motion as an indication of bone conduction (BC) sound transmission. The promontory motions were measured on the ipsilateral and contralateral sides, and the transcranial attenuation and phase delay of the contralateral side relative to the ipsilateral side were calculated. The transcranial attenuation and phase delay calculated with the maximum magnitudes and corresponding phases of the resultant were a better fit to the interaural threshold difference and transcranial time interval between the ipsilateral and contralateral sides as reported in the literature, than the attenuation and phase delay calculated with any individual Cartesian motion component. HighlightsA method to calculate the magnitude and phase of the 3D velocity vector is described.The 3D velocity of a point can be represented by the magnitude and phase of the resultant instead of the magnitudes and phases of the three orthogonal components.The method was applied to the measurement of cochlear promontory motion as an indication of bone conduction (BC) sound transmission.The maximum velocity vector of the cochlear promontory motion is a better descriptor of BC hearing than individual Cartesian components.
Performance evaluation of a novel piezoelectric subcutaneous bone conduction device
Objectives: Evaluation of the transfer function efficiency of a newly‐developed piezo‐electric actuator for active subcutaneous bone conduction hearing aid. Methods: The experiments were conducted on four Thiel embalmed whole head cadaver specimens. A novel actuator based on piezo‐electric transduction (PZTA), part of a subcutaneous bone conduction hearing aid device, was sequentially implanted on three locations: 1) Immediately posterior to pinna; 2) 50–60 mm posterior to pinna, approximately the same distance as between the BAHA (bone anchored hearing aid) location and the ear canal, but the same horizontal level as location 1; 3) the traditional BAHA location. Using a single point 3‐dimensional laser Doppler vibrometer (LDV) system, three types of motion measurements were performed at the cochlear promontory for each stimulation location: 1) ipsilateral side, 2) contralateral side, 3) measurements 1 and 2 were repeated after mastoidectomy on the ipsilateral side. Results: On average, stimulation at locations 1 and 2 show a trend for higher promontory motion relative to location 3 (BAHA location) above 1 kHz. Stimulation at location 1 had an average improvement of 1–6 dB at 2–4 kHz, and 1–18 dB at 6–8 kHz. The spatial composition of the motion showed significant contributions from both in‐plane and out‐of‐plane (along ear canal) motion components, with in‐plane components being dominant at mid and high frequencies for locations 2 and 3. Stimulation at locations 1 and 3 produced similar transcranial attenuation at mid frequencies (0.6–4 kHz), with a potential trend of higher attenuation (seen in 3 or the 4 samples) for location 1 at higher frequencies (>4 kHz). The mastoidectomy affected negatively mostly the high frequencies (6–8 kHz) for stimulation at location 1, with no significant change for location 3. Conclusion: The sound transfer function efficacy of a novel subcutaneous bone conduction device has been quantified, and the influence of stimulation location and mastoidectomy have been analyzed based on promontory motion in Thiel‐preserved cadaver heads. HighlightsA novel subcutaneous bone conduction device has been tested in cadaver heads.The new device features a piezo‐electric transduction (PZTA) actuator.Stimulation closer to the skull base improves PZTA coupling efficiency.Stimulation closer to the ear canal tends to have stronger transcranial attenuation.Mastoidectomy negatively affects high‐frequency transduction at nearby points.
The distinctiveness of Belfast medicine and its medical school.
ImagesFig 1Fig 2Fig 3Fig 4Fig 5Fig 6Fig 7Fig 8Fig 9