195868697
Irradiation tests of double-sided silicon strip detectors with a special guard ring structure
The results of the first irradiation tests of newly designed silicon microstrip detectors performed with 21 MeV protons at the Max-Planck-Institut in Heidelberg are presented. The detectors were developed and produced by the semiconductor laboratory of the Max-Planck-Institut in Munich. Novel guard ring structures allow operation of the detectors at voltages exceeding 300 V.
19809846
Development of Fast High-Resolution Muon Drift-Tube Detectors for High Counting Rates
Abstract Pressurized drift-tube chambers are efficient detectors for high-precision tracking over large areas. The Monitored Drift-Tube (MDT) chambers of the muon spectrometer of the ATLAS detector at the Large Hadron Collider (LHC) reach a spatial resolution of 35 μ m and almost 100% tracking efficiency with 6 layers of 30 mm diameter drift tubes operated with an Ar:CO 2 (93:7) gas mixture at 3 bar and a gas gain of 20 000. The ATLAS MDT chambers are designed to cope with background counting rates due to neutrons and γ rays of up to about 300 kHz per tube which will be exceeded for LHC luminosities larger than the design value of 10 34  cm −1  s −1 . Decreasing the drift-tube diameter to 15 mm while keeping the other parameters, including the gas gain, unchanged reduces the maximum drift time from about 700 to 200 ns and the drift-tube occupancy by a factor of 7. New drift-tube chambers for the endcap regions of the ATLAS muon spectrometer have been designed. A prototype chamber consisting of 12 times 8 layers of 15 mm diameter drift tubes of 1 m length has been constructed with a sense wire positioning accuracy of 20 μ m . The 15 mm diameter drift-tubes have been tested with cosmic rays in the Gamma Irradiation Facility at CERN at γ counting rates of up to 1.85 MHz.
121460348
Irradiation tests of double-sided silicon strip detectors optimized for the ATLAS-inner-detector-region
Abstract Silicon detectors, which will be used as tracking detectors in LHC-experiments, face special requirements; one major challenge is the long term operation of the detectors in very intense radiation fields. This paper describes a design for double-sided strip detectors, which takes these requirements into account. Irradiation tests have been carried out using 24 GeV protons. The change of detector characteristics in particular guardring-behaviour, punchthrough biasing and interstrip isolation has been studied.
111334701
A DEPFET pixel matrix system for the ILC vertex detector
Abstract The DEPFET detector offers radiation detection and amplification jointly by embedding a field effect transistor into fully depleted silicon. Due to the excellent noise performance ( ENC = 2.2 e - for single pixel at room temperature and 6 μ s shaping time) and the high spatial resolution of the device ( 4.3 μ m at 22 keV using 50 × 50 μ m 2 pixels), the DEPFET concept is attractive for X-ray astronomy, for biomedical application and for tracking in particle physics. For the vertex detector of a future TeV-scale linear collider, like the proposed ILC (International Linear Collider), a highly granulated pixel system operated row-wise with line rates of 20 MHz at a noise level below 100 e - is needed. Addressing these requirements, improved DEPFET sensors with 22 × 36 μ m 2 pixel size have been fabricated. A prototype system has been developed that uses dedicated chips for steering and readout of a 64 × 128 DEPFET pixel matrix. Due to the current based readout mode provided by the readout chip, a thousand times faster matrix operation than achieved so far is possible. The performance of the single components of the system will be summarized and first results obtained with the entire system will be presented.
13052670
Prospects of using silicon photomultipliers for the astroparticle physics experiments EUSO and MAGIC
We discuss the silicon photomultiplier as a novel photon detector for the Major Atmospheric Gamma Imaging Cerenkov Telescope and the Extreme Universe Space Observatory. For these astroparticle experiments we pursue two different developments of the SiPM. One development is pursued at MEPhI, where prototypes are available. In the course of this paper we will describe some characteristics of these devices. The second development, which is using the back illumination principle is at present in its design phase.
25787680
Spectroscopic performance of the DePMOS detector/amplifier device with respect to different filtering techniques and operating conditions
A DePMOS structure provides detection and amplification jointly, and it is free of interconnection stray capacitance. To fully exploit the intrinsic low noise of the device an electrical model has been developed. The most relevant parameters have been measured in order to choose an adequate readout electronics. DePMOS can operate in continuous mode, i.e., without applying any clear pulse during the signal processing, and can be read out by a time-continuous shaper amplifier. An unequalled noise of 2.2 electrons rms at room temperature has been measured. In this mode DePMOS can be used, for example, as the readout device for silicon drift detectors. Anyway , DePMOS was developed to be the basic element of an active pixel sensor suitable to cope with the requirements of the XEUS Wide Field Imager. In a matrix arrangement, each pixel must be read out by a time-variant filter. A multichannel integrated shaping amplifier, based on multicorrelated double sampling, has been designed. Spectroscopic resolution obtained filtering the pixel matrix with this readout chip is in agreement with measurements in continuous mode and matches the predictions of the model presented. It has also been experimentally proved that the clear procedure does not introduce additional noise contribution, even in the very low noise range achieved. This qualifies DePMOS as a "reset-noise-free" device
120324718
Integration, commissioning and installation of monitored drift tube chambers for the ATLAS barrel muon spectrometer
The ATLAS experiment at the Large Hadron Collider (LHC) at CERN is currently being assembled to be ready to take data in 2007. Its muon spectrometer is designed to achieve a momentum resolution better than 10% at 1 TeV. In the barrel part, a toroidal air-core magnet is instrumented with three layers of Monitored Drift Tube (MDT) chambers for precision tracking. The Max-Planck-Institut fur Physik and the Ludwig-Maximilians-University in Munich have built 88 MDT chambers for the outermost barrel layer. We report on our experience with the tests of the MDT chambers, their integration with the Resistive Plate trigger chambers and the installation of the muon stations in the experiment. First results of their commissioning in the ATLAS detector will be presented.
121450717
Development of Precision Muon Drift Tube Detectors for the High-Luminosity Upgrade of the LHC
For use at the future Super-LHC a new type of muon detector has been developed. It is based on the proven MDT drift tube design, but with tubes of half the diameter, leading to higher rate capabilities by an order of magnitude. We present test results on efficiency and position resolution at high background rates and describe the practical implementation in a real-size prototype.
19280329
Sharpening the ATLAS muon trigger for high luminosity operation at the LHC
The outer shell of the ATLAS experiment at the LHC consists of a system of toroidal air-core magnets in order to allow for the precise measurement of the transverse momentum (pT) of muons, which in many physics channels are a signature of interesting physics processes [1], [2]. For the precise determination of the muon momentum Monitored Drift Tube chambers (MDT) with high position accuracy are used, while for the fast identification of muon tracks chambers with high time resolution are used, able to select muons above a predefined pT threshold for use in the first Level of the ATLAS triggering system (Level-1 trigger). When the LHC peak luminosity will be increased by a factor of 4–5 in about a decade from now (”SLHC”), an improvement of the selectivity of the ATLAS Level-1 triggering system will be mandatory in order to cope with the maximum allowed trigger rate of 100 kHz. For the Level-1 trigger of the ATLAS muon spectrometer this means an increase of the pT threshold for single muons. Due to the limited spatial resolution of the trigger chambers, however, the selectivity for tracks above ∼20 GeV/c is insufficient for an effective reduction of the Level-1 rate. We describe how the track coordinates measured in the MDT precision chambers can be used to decisively improve the selectivity for high momentum tracks. The resulting increase in latency will also be discussed.
119295158
Performance of Drift-Tube Detectors at High Counting Rates for High-Luminosity LHC Upgrades
Abstract The performance of pressurized drift-tube detectors at very high background rates has been studied at the Gamma Irradiation Facility (GIF) at CERN and in an intense 20 MeV proton beam at the Munich Van-der-Graaf tandem accelerator for applications in large-area precision muon tracking at high-luminosity upgrades of the Large Hadron Collider (LHC). The ATLAS muon drift-tube (MDT) chambers with 30 mm tube diameter have been designed to cope with γ and neutron background hit rates of up to 500 Hz/cm 2 . Background rates of up to 14 kHz/cm 2 are expected at LHC upgrades. The test results with standard MDT readout electronics show that the reduction of the drift-tube diameter to 15 mm, while leaving the operating parameters unchanged, vastly increases the rate capability well beyond the requirements. The development of new small-diameter muon drift-tube (sMDT) chambers for LHC upgrades is completed. Further improvements of tracking efficiency and spatial resolution at high counting rates will be achieved with upgraded readout electronics employing improved signal shaping for high counting rates.
5682028
Computation of D-brane instanton induced superpotential couplings: Majorana masses from string theory
We perform a detailed conformal field theory analysis of D2-brane instanton effects in four-dimensional type IIA string vacua with intersecting D6-branes. In particular, we explicitly compute instanton induced fermion two-point couplings which play the role of perturbatively forbidden Majorana mass terms for right-handed neutrinos or MSSM {mu} terms. These results can readily be extended to higher-dimensional operators. In concrete realizations of such nonperturbative effects, the Euclidean D2-brane has to wrap a rigid, supersymmetric cycle with strong constraints on the zero-mode structure. Their implications for type IIA compactifications on the T{sup 6}/(Z{sub 2}xZ{sub 2}) orientifold with discrete torsion are analyzed. We also construct a local supersymmetric GUT-like model allowing for a class of Euclidean D2-branes whose fermionic zero modes meet all the constraints for generating Majorana masses in the phenomenologically allowed regime. Together with perturbatively realized Dirac masses, these nonperturbative couplings give rise to the seesaw mechanism.