113702675
Characterization of Thin Pixel Sensor Modules Interconnected with SLID Technology Irradiated to a Fluence of 2$\cdot 10^{15}$\u2009n$_{\mathrm{eq}}$/cm$^2$
73653251
Imaging performance of a DEPFET pixel Bioscope system in Tritium autoradiography
Abstract A Bioscope system based on a 64 × 64 -matrix of DEPFET pixels has been developed for real-time digital autoradiography in biomedical applications. It provides very good energy and spatial resolution at the same time. The DEPFET pixel cells consist of p -channel JFETs integrated into the high-ohmic n -silicon substrate. This leads to a very low noise performance at room temperature. Two dedicated ASIC chips have been developed for steering and readout of the pixel matrix. The measured system noise of ( 69 ± 4 ) e - at 35 ∘ C is mostly due to sensor leakage current and can be largely reduced by cooling or a faster readout. The spatial resolution is examined by measurements with a laser as well as with a dedicated test chart and is determined to be ( 6.7 ± 0.7 ) μ m for 6 keV-X-rays and 4.3 ± 0.8 μ m for 22 keV-X-rays. The measurements are compared to simulation results with different reconstruction algorithms. Due to the thin entrance window at the detector backside and the low noise performance, a detection of low-energy β -emitters such as 3H is possible. Measurements with 3H-labeled biological samples were taken. The ability of the DEPFET pixel Bioscope to separate different radiolabels in the same sample is demonstrated by a measurement with 3H and 14C.
34283775
Tracking and level-1 triggering in the forward region of the ATLAS muon spectrometer at sLHC
In the endcap region of the ATLAS Muon Spectrometer (η  >  1) precision tracking and Level-1 triggering are performed by different types of chambers. Monitored Drift Tube chambers (MDT) and Cathode Strip Chambers (CSC) are used for precision tracking, while Thin Gap Chambers (TGC) form the Level-1 muon trigger, selecting muons with high transverse momentum (pT). When by 2018 the LHC peak luminosity of 1034 cm−2s−1 will be increased by a factor of ~ 2 and by another factor of ~ 2–2.5 in about a decade from now (``SLHC''), an improvement of both systems, precision tracking and Level-1 triggering, will become mandatory in order to cope with the high rate of uncorrelated background hits (``cavern background'') and to stay below the maximum trigger rate for the muon system, which is in the range of 10–20 % of the 100 kHz rate, allowed for ATLAS. For the Level-1 trigger of the ATLAS Muon Spectrometer this means a stronger suppression of sub-threshold muons in the high-pT trigger as well as a better rejection of tracks not coming from the primary interaction point. Both requirements, however, can only be fulfilled if spatial resolution and angular pointing accuracy of the trigger chambers, in particular of those in the Inner Station of the endcap, are improved by a large factor. This calls for a complete replacement of the currrently used TGC chambers by a new type of trigger chambers with better performance. In parallel, the precision tracking chambers must be replaced by chambers with higher rate capability to be able to cope with the intense cavern background. In this article we present concepts to decisively improve the Level-1 trigger with newly developed trigger chambers, being characterized by excellent spatial resolution, good time resolution and sufficiently short latency. We also present new types of precision chambers, designed to maintain excellent tracking efficiency and spatial resolution in the presence of high levels of uncorrelated background hits, as generated by γ and neutron conversions.
108645383
The ATLAS Silicon Pixel Sensors
Abstract Prototype sensors for the ATLAS silicon pixel detector have been developed. The design of the sensors is guided by the need to operate them in the severe LHC radiation environment at up to several hundred volts while maintaining a good signal-to-noise ratio, small cell size, and minimal multiple scattering. The ability to be operated under full bias for electrical characterization prior to attachment of the readout integrated circuit electronics is also desired.
195868697
Irradiation tests of double-sided silicon strip detectors with a special guard ring structure
The results of the first irradiation tests of newly designed silicon microstrip detectors performed with 21 MeV protons at the Max-Planck-Institut in Heidelberg are presented. The detectors were developed and produced by the semiconductor laboratory of the Max-Planck-Institut in Munich. Novel guard ring structures allow operation of the detectors at voltages exceeding 300 V.
19809846
Development of Fast High-Resolution Muon Drift-Tube Detectors for High Counting Rates
Abstract Pressurized drift-tube chambers are efficient detectors for high-precision tracking over large areas. The Monitored Drift-Tube (MDT) chambers of the muon spectrometer of the ATLAS detector at the Large Hadron Collider (LHC) reach a spatial resolution of 35 μ m and almost 100% tracking efficiency with 6 layers of 30 mm diameter drift tubes operated with an Ar:CO 2 (93:7) gas mixture at 3 bar and a gas gain of 20 000. The ATLAS MDT chambers are designed to cope with background counting rates due to neutrons and γ rays of up to about 300 kHz per tube which will be exceeded for LHC luminosities larger than the design value of 10 34  cm −1  s −1 . Decreasing the drift-tube diameter to 15 mm while keeping the other parameters, including the gas gain, unchanged reduces the maximum drift time from about 700 to 200 ns and the drift-tube occupancy by a factor of 7. New drift-tube chambers for the endcap regions of the ATLAS muon spectrometer have been designed. A prototype chamber consisting of 12 times 8 layers of 15 mm diameter drift tubes of 1 m length has been constructed with a sense wire positioning accuracy of 20 μ m . The 15 mm diameter drift-tubes have been tested with cosmic rays in the Gamma Irradiation Facility at CERN at γ counting rates of up to 1.85 MHz.
55147901
Light stringy states
A bstractWe carefully study the spectrum of open strings localized at the intersections of D6-branes and identify the lowest massive ‘twisted’ states and their vertex operators, paying particular attention to the signs of the intersection angles. We argue that the masses of the lightest states scale as $ M_{\theta }^2 \approx \theta M_s^2 $ and can thus be parametrically smaller than the string scale. Relying on previous analyses, we compute scattering amplitudes of massless ‘twisted’ open strings and study their factorization, confirming the presence of the light massive states as sub-dominant poles in one of the channels.
23063408
Use of Triple Modular Redundancy (TMR) technology in FPGAs for the reduction of faults due to radiation in the readout of the ATLAS monitored drift tube (MDT) chambers
The Triple Modular Redundancy (TMR) technology allows protection of the functionality of FPGAs against single event upsets (SEUs). Each logic block is implemented three times with a 2-out-of-3 voter at the output. Thus, the correct logical value is available even if there is an upset bit in one location. We applied TMR to the configuration code of a Virtex-II-2000 FPGA, which serves as the on-chamber readout processor of the ATLAS monitored drift tubes (MDTs). We describe the code implementation, results of performance measurements and discuss several limitations of the method. Finally, we present a supplementary technology called “scrubbing”. It permanently checks the configuration memory while the FPGA is operating, and corrects upset configuration bits when necessary.
53533600
String constraints on discrete symmetries in MSSM type II quivers
A bstractWe study the presence of discrete gauge symmetries in D-brane semi-realistic compactifications. After establishing the constraints on the transformation behaviour of the chiral matter for the presence of a discrete gauge symmetry we perform a systematic search for discrete gauge symmetries within local semi-realistic D-brane realizations, based on four D-brane stacks, of the MSSM and the MSSM with three right-handed neutrinos. The systematic search reveals that Proton hexality, a discrete symmetry which ensures the absence of R-parity violating terms as well as the absence of dangerous dimension 5 proton decay operators, is only rarely realized. Moreover, none of the semi-realistic local D-brane configurations exhibit any family dependent discrete gauge symmetry.
119295158
Performance of Drift-Tube Detectors at High Counting Rates for High-Luminosity LHC Upgrades
Abstract The performance of pressurized drift-tube detectors at very high background rates has been studied at the Gamma Irradiation Facility (GIF) at CERN and in an intense 20 MeV proton beam at the Munich Van-der-Graaf tandem accelerator for applications in large-area precision muon tracking at high-luminosity upgrades of the Large Hadron Collider (LHC). The ATLAS muon drift-tube (MDT) chambers with 30 mm tube diameter have been designed to cope with γ and neutron background hit rates of up to 500 Hz/cm 2 . Background rates of up to 14 kHz/cm 2 are expected at LHC upgrades. The test results with standard MDT readout electronics show that the reduction of the drift-tube diameter to 15 mm, while leaving the operating parameters unchanged, vastly increases the rate capability well beyond the requirements. The development of new small-diameter muon drift-tube (sMDT) chambers for LHC upgrades is completed. Further improvements of tracking efficiency and spatial resolution at high counting rates will be achieved with upgraded readout electronics employing improved signal shaping for high counting rates.
50340039
Observation of a Narrow Resonance at 10.02-GeV in e+ e- Annihilations
Abstract The ϒ′ state has been observed as a narrow resonance at M ( ϒ ′) = 10.02 ± 0.02 GeV in e + e − annihilations, using a NaI and lead-glass detector in the DORIS storage ring at DESY. The ratio Г ee Г had /Г tot of electronic, hadronic, and total widths has been measured to be 0.32 ± 0.13 keV. The parameters of the Г particle have also been determined to be/ M (Г)