118584194
Construction and Test of MDT Chambers for the ATLAS Muon Spectrometer
Abstract The Monitored Drift Tube (MDT) chambers for the muon spectrometer of the ATLAS detector at the Large Hadron Collider (LHC) consist of 3–4 layers of pressurized drift tubes on either side of a space frame carrying an optical monitoring system to correct deformations. The full-scale prototype of a large MDT chamber has been constructed with methods suitable for large-scale production. X-ray measurements at CERN showed a positioning accuracy of the sense wires in the chamber of better than the required 20 μm (rms). The performance of the chamber was studied in a muon beam at CERN. Chamber production for ATLAS now has started.
126243744
Measurement of the reaction π−p→π+π−n on a transversely polarized target at 17.2 GeV
Preliminary results from a high statistics measurement (1 M events) of the reaction π−p→π+π−n on strong polarization effects which must be attributed to amplitudes corresponding to ’’A1’’ exchange.
120421514
GAS SYSTEM FOR ALEPH TPC
Abstract Operation of the time projection chamber of the ALEPH experiment at LEP requires a high-purity argon/methane gas mixture supplied to the detector under stable conditions to ensure minimum signal attenuation throughout the 43 m 3 detector volume (max. 2.2 m drift length) and maintain near-constant electron drift velocity in the gas over extended periods of time. Design considerations for a gas system to fulfil these conditions are presented and details given of the construction, operational aspects and of the achieved performance.
122359864
Large-scale production of Monitored Drift Tube chambers for the ATLAS muon spectrometer
Abstract Precision drift tube chambers with a sense wire positioning accuracy of better than 20 μm are under construction for the ATLAS muon spectrometer. 70% of the 88 large chambers for the outermost layer of the central part of the spectrometer have been assembled. Measurements during chamber construction of the positions of the sense wires and of the sensors for the optical alignment monitoring system demonstrate that the requirements for the mechanical precision of the chambers are fulfilled.
120324718
Integration, commissioning and installation of monitored drift tube chambers for the ATLAS barrel muon spectrometer
The ATLAS experiment at the Large Hadron Collider (LHC) at CERN is currently being assembled to be ready to take data in 2007. Its muon spectrometer is designed to achieve a momentum resolution better than 10% at 1 TeV. In the barrel part, a toroidal air-core magnet is instrumented with three layers of Monitored Drift Tube (MDT) chambers for precision tracking. The Max-Planck-Institut fur Physik and the Ludwig-Maximilians-University in Munich have built 88 MDT chambers for the outermost barrel layer. We report on our experience with the tests of the MDT chambers, their integration with the Resistive Plate trigger chambers and the installation of the muon stations in the experiment. First results of their commissioning in the ATLAS detector will be presented.
108271361
Concepts for the readout of time projection chambers
While the use of CCD’s for the readout of TPC’s offers a reliable and ever more of the digitization concept. The problem of precision of currently available FADC’s is discussed together with two proposals for a solution.
25787680
Spectroscopic performance of the DePMOS detector/amplifier device with respect to different filtering techniques and operating conditions
A DePMOS structure provides detection and amplification jointly, and it is free of interconnection stray capacitance. To fully exploit the intrinsic low noise of the device an electrical model has been developed. The most relevant parameters have been measured in order to choose an adequate readout electronics. DePMOS can operate in continuous mode, i.e., without applying any clear pulse during the signal processing, and can be read out by a time-continuous shaper amplifier. An unequalled noise of 2.2 electrons rms at room temperature has been measured. In this mode DePMOS can be used, for example, as the readout device for silicon drift detectors. Anyway , DePMOS was developed to be the basic element of an active pixel sensor suitable to cope with the requirements of the XEUS Wide Field Imager. In a matrix arrangement, each pixel must be read out by a time-variant filter. A multichannel integrated shaping amplifier, based on multicorrelated double sampling, has been designed. Spectroscopic resolution obtained filtering the pixel matrix with this readout chip is in agreement with measurements in continuous mode and matches the predictions of the model presented. It has also been experimentally proved that the clear procedure does not introduce additional noise contribution, even in the very low noise range achieved. This qualifies DePMOS as a "reset-noise-free" device
110363376
Design and test of radiation hard p'n silicon strip detectors for the ATLAS SCT
Strip detectors covering radiation hardness and large-scale production ability are developed and produced for the ATLAS experiment at the Large Hadron Collider (LHC) at CERN (Switzerland). Capacitively coupled p‘n detectors (p-type strips on n-type substrate) were developed with implanted bias resistors in order to simplify the detector processing addressing the requirements of large-scale production. The detectors were irradiated with 24 GeV protons up to 3]1014 cm~2 in order to simulate a 10 years operation scenario at LHC. The presented static and signal measurements demonstrate the function of the device concept before and after irradiation. ( 2000 Elsevier Science B.V. All rights reserved.
40263937
Heavy Ion Source Development at the Bevatron
The Bevatron 20 MeV duoplasmatron source is currently being modified with the goal of producing 1 mA of 20Ne3+. Initital tests at 420 KeV show a total beam of 20 mA of which 400 ¿A is 20Ne3+. The quantity of beam in various charge states is determined with a pulse field magnetic spectrometer. Titanium sublimation and cryogenic pumping of the PIG source in the High Voltage Terminal and its resultant effects on the acceleration of carbon and nitrogen will be discussed.
34283775
Tracking and level-1 triggering in the forward region of the ATLAS muon spectrometer at sLHC
In the endcap region of the ATLAS Muon Spectrometer (η  >  1) precision tracking and Level-1 triggering are performed by different types of chambers. Monitored Drift Tube chambers (MDT) and Cathode Strip Chambers (CSC) are used for precision tracking, while Thin Gap Chambers (TGC) form the Level-1 muon trigger, selecting muons with high transverse momentum (pT). When by 2018 the LHC peak luminosity of 1034 cm−2s−1 will be increased by a factor of ~ 2 and by another factor of ~ 2–2.5 in about a decade from now (``SLHC''), an improvement of both systems, precision tracking and Level-1 triggering, will become mandatory in order to cope with the high rate of uncorrelated background hits (``cavern background'') and to stay below the maximum trigger rate for the muon system, which is in the range of 10–20 % of the 100 kHz rate, allowed for ATLAS. For the Level-1 trigger of the ATLAS Muon Spectrometer this means a stronger suppression of sub-threshold muons in the high-pT trigger as well as a better rejection of tracks not coming from the primary interaction point. Both requirements, however, can only be fulfilled if spatial resolution and angular pointing accuracy of the trigger chambers, in particular of those in the Inner Station of the endcap, are improved by a large factor. This calls for a complete replacement of the currrently used TGC chambers by a new type of trigger chambers with better performance. In parallel, the precision tracking chambers must be replaced by chambers with higher rate capability to be able to cope with the intense cavern background. In this article we present concepts to decisively improve the Level-1 trigger with newly developed trigger chambers, being characterized by excellent spatial resolution, good time resolution and sufficiently short latency. We also present new types of precision chambers, designed to maintain excellent tracking efficiency and spatial resolution in the presence of high levels of uncorrelated background hits, as generated by γ and neutron conversions.
15397236
Branes and instantons intersecting at angles
We study in detail the system of D6 branes and euclidean D2-brane instantons intersecting at angles in type IIA string theory. We find that in the absence of orientifolds the system does not contribute to the low energy superpotential, in agreement with expectations based on effective field theory arguments. We also comment on the implications of our results for dual string theory pictures.