Monte Carlo calculations and experimental measurements of the TG 43U1 recommended dosimetric parameters of 125I (Model IR Seed2) brachytherapy source
A new design of I125 (Model IR Seed2) brachytherapy source has been manufactured recently at the Applied Radiation Research School, Nuclear Science and Technology Research Institute in Iran. The source consists of six resin beads (0.5 mm diameter) that are sealed in a cylindrical titanium capsule of 0.7 mm internal and 0.8 mm external diameters. This work aims to evaluate the dosimetric parameters of the newly designed I125 source using experimental measurements and Monte Carlo (MC) simulations. Dosimetric characteristics (dose rate constant, radial dose function, and 2D and 1D anisotropy functions) of the IR Seed2 were determined using experimental measurements and MC simulations following the recommendations by the Task Group 43 (TG 43U1) report of the American Association of Physicists in Medicine (AAPM) MC simulations were performed using the MCNP5 code in water and Plexiglas, and experimental measurements were carried out using thermoluminescent dosimeters (TLD GR207A) in Plexiglas phantoms. The measured dose to water in Plexiglas data were used for verification of the accuracy of the source and phantom geometry in the Monte Carlo simulations. The final MC simulated data to water in water were recommended for clinical applications. The MC calculated dose rate constant (L) of the IR Seed2 I125 seed in water was found to be 0.992+ 0.025 cGy h 1U 1. Additionally, its radial dose function by line and point source approximations, gL(r) and gp(r) calculated for distances from 0.1 cm to 7 cm. The values of gL(r) at radial distances from 0.5 cm to 5 cm were measured in a Plexiglas phantom to be between 1.212 and 0.413. The calculated and measured of values for 2D anisotropy function, F(r,th) were obtained for the radial distances ranging from 1.5 cm to 5 cm and angular range of 0deg 90deg in a Plexiglas phantom. Also, the 2D anisotropy function was calculated in water for the clinical application. The results of these investigations show that the uncertainty of the experimental data is within 7% between the measured and simulated data in Plexiglas. Based on these results, the MC simulated dosimetric parameters of the new I125 source model in water are presented for its clinical applications in brachytherapy treatments. PACS number(s) 87.56.bg
Journal of applied clinical medical physics