Quantitative accuracy of 177Lu SPECT reconstruction using different compensation methods: phantom and patient studies
BackgroundIn targeted radionuclide therapy (TRT) accurate quantification using SPECT/CT images is important for optimizing radiation dose delivered to both the tumour and healthy tissue. Quantitative SPECT images are regularly reconstructed using the ordered subset expectation maximization (OSEM) algorithm with various compensation methods such as attenuation (A) scatter (S) and detector and collimator response (R) In this study, different combinations of the compensation methods are applied during OSEM reconstruction and the effect on the 177Lu quantification accuracy is studied in an anthropomorphic torso phantom. In addition, the phantom results are reflected to (177)Lu DOTA Tyr3 octreotate (177Lu DOTATATE) treated patient data and kidney absorbed dose estimates.MethodsThe torso phantom was imaged with nine various sized (0.4 104.4 cm3) spherical inserts, filled with known 177Lu activity ranging from 0.5 to 105.5 MBq. Images were reconstructed using OSEM algorithm using A, AR and ARS compensation method combinations. The compensation method combinations were compared by calculating the concentration recovery coefficient (cRC) for each insert. In addition, ten 177Lu DOTATATE treated patient's post therapy dosimetry acquisitions were reconstructed, and the absorbed dose to kidneys was estimated.ResultscRC values depend on the insert size for all compensation methods. AR and ARS produced significantly higher cRC values than attenuation correction alone. There were no cRC value differences between the methods for the smallest 1 cm diameter insert, cRC being 0.18. However, the collimator and detector response compensation method (R) made the 1.3 cm diameter insert clearly visible and improved cRC estimate from 0.19 to 0.43. ARS produced slightly higher cRC values for small and medium sized inserts than AR. On the patient data, a similar trend could be seen. AR and ARS produced higher kidney activities than using attenuation correction alone; the total absorbed doses to the right and left kidneys were on average 15 and 20 higher for AR and 19 and 25 higher for ARS, respectively. The effective half life decay estimated from time activity curves however showed no notable difference between the compensation methods.ConclusionsThe highest cRC values were achieved by applying ARS compensation during reconstruction. The results were notably higher than those using attenuation correction alone. Similarly, higher activity estimates and thus higher absorbed dose estimates were found in patient data when all compensation methods were applied. ARS improved cRC especially in small sized sources, and it thus might aid tumour dosimetry for 177Lu PRRT treatments.
EJNMMI Research