107398289
THE FONT5 PROTOTYPE ILC INTRA-TRAIN FEEDBACK SYSTEM AT ATF2
We present the design and beam test results of a prototype beam-based digital feedback system for the Interaction Point of the International Linear Collider. A custom analogue front-end signal processor, FPGA-based digital signal processing boards, and kicker drive amplifier have been designed, built, deployed and tested with beam in the extraction line of the KEK Accelerator Test Facility (ATF2). The system was used to provide orbit correction to the train of bunches extracted from the ATF damping ring. The latency was measured to be approximately 140 ns.
172130610
ATF2 Commissioning
ATF2 is a final-focus test beam line that aims to focus the low-emittance beam from the ATF damping ring to a beam size of about 37 nm, and at the same time to demonstrate nm beam stability, using numerous advanced beam diagnostics and feedback tools. The construction has been finished at the end of 2008 and the beam commissioning of ATF2 has started in December of 2008. ATF2 is constructed and commissioned by ATF international collaborations with strong US, Asian and European participation.
54225329
DEVELOPMENT OF A FAST,SINGLE-PASS,MICRON-RESOLUTION BEAM POSITION MONITOR SIGNAL PROCESSOR:BEAM TEST RESULTS FROM ATF2
We present the design of a stripline beam position monitor (BPM) signal processor with low latency (c. 10ns) and micron-level spatial resolution in single-pass mode. Such a BPM processor has applications in singlepass beamlines such as those at linear colliders and FELs. The processor was deployed and tested at the Accelerator Test Facility (ATF2) extraction line at KEK, Japan. We report the beam test results and processor performance, including response, linearity, spatial resolution and latency.
58892392
LATEST BEAM TEST RESULTS FROM ATF2 WITH THE FONT ILC PROTOTYPE INTRA-TRAIN BEAM FEEDBACK SYSTEMS
We present the design and beam test results of a prototype beam-based digital feedback system for the Interaction Point of the International Linear Collider. A custom analogue front-end signal processor, FPGA-based digital signal processing boards, and kicker drive amplifier have been designed, built, deployed and tested with beam in the extraction line of the KEK Accelerator Test Facility (ATF2). The system was used to provide orbit correction to the train of bunches extracted from the ATF damping ring. The latency was measured to be approximately 140 ns.
14475692
Latest Beam Test Results of the FONT4 ILC Intra-train Feedback System Prototype
We present the design and preliminary results of a prototype beam-based digital feedback system for the Interaction Point of the International Linear Collider. A custom analogue front-end processor, FPGA-based digital signal processing board, and kicker drive amplifier have been designed, built, and tested on the extraction line of the KEK Accelerator Test Facility (ATF). The system was measured to have a latency of approximately 140 ns.
55766811
LATEST PERFORMANCE RESULTS FROM THE FONT5 INTRA-TRAIN POSITION AND ANGLE FEEDBACK SYSTEM AT ATF2
A prototype Interaction Point beam-based feedback system for future electron-positron colliders, such as the International Linear Collider, has been designed and tested on the extraction line of the KEK Accelerator Test Facility (ATF). The FONT5 intra-train feedback system aims to stabilise the beam orbit by correcting both the position and angle jitter in the vertical plane on bunch-to-bunch timescales, providing micron-level stability at the entrance to the ATF2 final-focus system. The system comprises three stripline beam position monitors (BPMs) and two stripline kickers, custom low-latency analogue front-end BPM processors, a custom FPGA-based digital processing board with fast ADCs, and custom kicker-drive amplifiers. An overview of the hardware, and the latest results from beam tests at ATF2, will be presented. A total system latency as low as approximately 140 ns has been demonstrated.
120429047
The design of the optical components and gas control systems of the CERN Omega ring imaging Cherenkov detector
Abstract A large ring imaging Cherenkov detector (RICH) has been commissioned for use at the CERN Omega spectrometer. In the first of a series of reports we begin by discussing the general design of the device and illustrate the dependence of the attainable spatial resolution and range of particle identification on its optical parameters. We then describe in detail the construction and performance of the major optical components and gas systems of the detector.
122893386
Measurement of the spatial resolution of double-sided double-metal AC-coupled silicon microstrips detectors
Abstract The design and first results from double-sided silicon microstrip detectors designed for use in the DELPHI experiment at LEP are presented. The detectors are AC-coupled on both the n- and p-side. A novel readout scheme using a second metal layer has been implemented, allowing the readout of both coordinates on the same edge of the detector. The detectors have been tested in a high energy beam at the CERN SPS. Results on spatial resolution, pulse-height correlation and charge division are presented. The spatial resolution of the n-side has been measured as a function of the beam particle incident angle from 0 to 60°.
120457455
The DELPHI Microvertex detector
The DELPHI Microvertex detector, which has been in operation since the start of the 1990 LEP run, consists of three layers of silicon microstrip detectors at average radii of 6.3, 9.0 and 11.0 cm. The 73728 readout strips, oriented along the beam, have a total active area of 0.42 m2. The strip pitch is 25 μm and every other strip is read out by low power charge amplifiers, giving a signal to noise ratio of 15:1 for minimum ionizing particles. On-line zero suppression results in an average data size of 4 kbyte for Z0 events. After a mechanical survey and an alignment with tracks, the impact parameter uncertainty as determined from hadronic Z0 decays is well described by (69pt)2 + 242 μm, with pt in GeV/c. For the 45 GeV/c tracks from Z0 → μ− decays we find an uncertainty of 21 μm for the impact parameter, which corresponds to a precision of 8 μm per point. The stability during the run is monitored using light spots and capacitive probes. An analysis of tracks through sector overlaps provides an additional check of the stability. The same analysis also results in a value of 6 μm for the intrinsic precision of the detector.
18645695
The recent operational performance of the CERN omega ring imaging cerenkov detector
We discuss the design and construction of the Time Projection chambers (TPCs) of the Omega Ring Imaging Cerenkov Detector (RICH). Details are given of the TPC high voltage system and its monitoring and control. In addition, the operation and monitoring of the readout is described together with results of tests on the performance of the front end amplifiers. The operation of the RICH TPCs and electronics during the first data run of WA69, in 1984, is discussed together with relevant results from laboratory tests. Results from the preliminary analysis of a sample of data from the 1984 run are also presented.
120959186
Radiation tolerance of single sided silicon microstrips
Abstract The RD20 collaboration is investigating the design and operation of an LHC inner tracking detector based on silicon microstrips. Measurements have been made on prototype detectors after irradiation with electrons, neutrons, photons, and protons for doses up to 5 Mrad and fluences up to 10 15 particles/cm 2 . The annealing of effective doping changes caused by high neutron fluences, one of the major limits to detector lifetime at the LHC, is shown to be strongly inhibited by cooling below room temperature. Detailed results are presented on the critical issue of microstrip capacitance. We have also investigated bulk damage caused by high-energy protons, interstrip isolation after neutron irradiation, and MOS capacitors irradiated with electrons and photons.