55910251
An RF system for a compact synchrotron
A 17.4 m circumference 200 MeV proton synchrotron, the Cooler Injector Synchrotron (CIS), is being commissioned at the Indiana University Cyclotron Facility. The compact synchrotron will be used as the injector for the electron-cooled IUCF Cooler Synchrotron. The RF system design uses digital signal processing (DSP) to directly modulate a digital RF synthesizer for beam feedback control. A wide band tunable RF cavity uses non-uniform ferrite biasing to achieve a 10:1 frequency coverage with good VSWR performance and low power requirement.
124338422
FIRST OBSERVATION OF A SNAKE DEPOLARIZING RESONANCE
Using a 104MeV stored polarized proton beam and a full Siberian snake, we recently found evidence for a so-called {open_quotes}snake{close_quotes} depolarizing resonance. A full Siberian snake forces the spin tune {nu}{sub s} to be a half integer. Thus, if the vertical betatron tune {nu}{sub y} is set near a quarter integer, then the {nu}{sub s}=n{plus_minus}2{nu}{sub y} second-order snake resonance can depolarize the beam. Indeed, with a full Siberian snake, we found a deep depolarization dip when {nu}{sub y} was equal to 4.756; moreover, when {nu}{sub y} was changed to 4.781, the deep dip disappeared and the polarization was preserved. {copyright} {ital 1997} {ital The American Physical Society}
120417645
Snakes and spin rotators for high energy accelerators
Abstract A modification to the continuous family of snake and spin rotators, discovered by Steffen, is discussed. A compact snake configuration is found to have the advantages of requiring a smaller total integrated dipole magnet strength and a smaller transverse displacement for circulating particles. A split snake configuration has the advantage of allowing an entire experimental insertion region to be located between the two halves of the snake. Such configurations could serve as both a snake and a 90° spin rotator for experiments requiring different helicity states. General properties of these snake configurations are discussed.
120908048
Comment on “Quasiperiodic spin-orbit motion and spin tunes in storage rings”
96454816
Design and fabrication of a multi-purpose Panofsky magnet
A fast, flexible magnet system, consisting of a Panofsky style ferrite picture-frame magnet and pulsed power supply, has been developed. Magnet specifications are: 8 cm /spl times/8 cm aperture, 23 cm length, 500 A pulsed/l60 A DC, and 1 /spl mu/s risetime. Designed for general accelerator physics studies, the magnet may be quickly converted from dipole to quadrupole or sextupole and higher multipole configurations by easily changing winding end terminations. In the quadrupole configuration a field gradient of 0.5 T/m is achieved at 500 A. This magnet has become an important tool for nonlinear beam dynamics experiments at the IUCF Cooler Ring.<<ETX>>
117036088
Comparison of multipole expansion and exact form of the eddy current field of the AGS Booster
Studies are made on magnetic field representation using a multipole expansion as well as the exact form to calculate the magnetic field produced by eddy currents in the vacuum chamber of the AGS Booster as well as the field produced by three turn correction coils attached to the top and bottom of the vacuum chamber. The multiple representation of the chamber field does not converge to the next field when X > 30mm and limits the particle motion. When the exact form of the chamber field is used, initial amplitudes in the horizontal plane (measured at QF) can be nearly as large as the chamber half aperture. Use of three turn correction coils to compensate the eddy current fields seems to reduce rather than increase the acceptance. 6 figs., 3 tabs.
123360724
Some test results of an anti-Cherenkov photomultiplier tube
Abstract We report on the beam test results of a photomultiplier tube (PMT) specially designed to suppress the signal due to Cherenkov light emitted by window glass when impacted by relativistic charged particles. This signal becomes a concern when it is comparable to the true PMT signal, especially in the case of calorimeters, where the response is at the level of a few photoelectrons per GeV. This design allows for placement of these PMTs behind the calorimeter without concern for hot spots from the Cherenkov background.
118615128
Searches for Physics Beyond the Standard Model at CMS
Recent results on searches for physics beyond the Standard Model at Large Hadron Collider are presented, based on early LHC data in proton-proton collisions at $\sqrt{s} = 7$ TeV collected by the CMS experiment. Prospects of early SUSY searches at CMS are also outlined.
9667954
Object Database for Constants: The common CLEO Online and Offline solution
After the successful conclusion of the CLEO III phase, the CLEO experiment at the Cornell Electron Positron Storage Ring CESR is preparing for its transition to CLEO-c. This new program contains a wide array of Physics studies at e+ e- collisions at center of mass energies between 3 GeV and 5 GeV that will provide new insights into QCD. Because the existing Silicon Vertex Detector needed to be replaced within a short time, a 6 layer vertex drift chamber has been installed in spring 2003. At the same time, the existing Ring Imaging Cherenkov Detector, along with a conventional drift chamber, E.M. calorimeter and muon chambers, will continue to be part of the experiment. 
The CLEO Constants Database system must provide efficient access to detector and analysis constants for a wide variety of purposes, such as hardware configuration, Online operation and monitoring, calibration, and the complete CLEO analysis framework. Since the original project, implemented through the Objectivity Object Database Management System, did not meet those requirements, the system was redesigned from the ground up, using the same technology. It is currently being introduced into the production environment without significant interruptions. In this presentation, we will outline the specifications of the constants database system, and then report on the process that led to the redesign. Performance comparisons and insights on relevant design aspects will be shown.
10444896
Active cooling control of the CLEO detector using a hydrocarbon coolant farm
Abstract We describe a novel approach to particle-detector cooling in which a modular farm of active coolant-control platforms provides independent and regulated heat removal from four recently upgraded subsystems of the CLEO detector: the ring-imaging Cherenkov detector, the drift chamber, the silicon vertex detector, and the beryllium beam pipe. We report on several aspects of the system: the suitability of using the aliphatic-hydrocarbon solvent PF TM -200IG as a heat-transfer fluid, the sensor elements and the mechanical design of the farm platforms, a control system that is founded upon a commercial programmable logic controller employed in industrial process-control applications, and a diagnostic system based on virtual instrumentation. We summarize the system's performance and point out the potential application of the design to future high-energy physics apparatus.
118835651
New physics searches with photons in CDF
A brief review of searches for physics beyond the Standard Model with photons using the CDF detector at the Tevatron is given here. These include searches for supersymmetry, extra dimensions, excited electrons and W/Z+{gamma} production, as well as anomalous photon production. Recent results from CDF Run II experiment are presented, but some results from Run I are also reviewed.