Loop-gap resonator

A loop-gap resonator (LGR) is an electromagnetic resonator that operates in the radio and microwave frequency ranges. The simplest LGRs are made from a conducting tube with a narrow slit cut along its length. The LGR dimensions are typically much smaller than the free-space wavelength of the electromagnetic fields at the resonant frequency. Therefore, relatively compact LGRs can be designed to operate at frequencies that are too low to be accessed using, for example, cavity resonators. These structures can have very sharp resonances (high quality factors) making them useful for electron spin resonance (ESR) experiments, and precision measurements of electromagnetic material properties (permittivity and permeability).
Background.
Loop-gap resonators (LGRs) can be modelled as lumped-element circuits. The slit along the length of the resonator has an effective capacitance formula_1 and the bore of the resonator has effective inductance formula_2. At, or near, the resonance frequency, a circumferential current is established along the inner wall of the resonator. The effective resistance formula_3 that limits this current is, in part, determined by the resistivity formula_4 and electromagnetic skin depth formula_5 of the conductor used to make the LGR. It is, therefore, possible to model the LGR as an formula_6 circuit. Since the LGR current is a maximum at the resonant frequency, the equivalent circuit model is a series formula_6 circuit. This circuit model works well provided the dimensions of the resonator remain small compared to the free-space wavelength of the electromagnetic fields.
One advantage of the LGR is that it produces regions of uniform electric and magnetic fields that are isolated from one another. A uniform electric field exists within the slit of the LGR and a uniform magnetic field exists within the bore of the resonator. The uniform magnetic field makes the LGR a good source of microwave magnetic fields in ESR experiments. Furthermore, because the electric and magnetic fields are isolated from one another, one can use the LGR to independently probe the electric and magnetic properties of materials. For example, if the gap of the LGR is filled with a dielectric material, the effective capacitance of the LGR will be modified which will change the frequency formula_8 and quality factor formula_9 of the resonance. Measurements of the changes in formula_8 and formula_9 can be used to fully determine the complex permittivity of the dielectric material. Likewise, if the bore of the LGR is filled with a magnetic material, the effective inductance of the LGR will be modified and the resulting changes in formula_8 and formula_9 can be used to extract the complex permeability of the magnetic material.
Resonant Frequency and Quality Factor.
Resonance frequency.
The capacitance of the gap of the LGR is given by
where formula_15 is the permittivity of free space, formula_16 is the thickness of the bore wall, formula_17 is the gap width, and formula_18 is the length of the resonator. The resonator bore acts as a single-turn solenoid with inductance given by
where formula_20 is the permeability of free space and formula_21 is the inner radius of the LGR bore. For a high-formula_9 resonator, the resonant frequency is, to an approximation, given by
where formula_24 is the vacuum speed of light. Therefore, the resonant frequency of the LGR is determined from is geometry and is, to first approximation, independent of its length.
Quality factor.
For a highly underdamped series formula_6 circuit, the quality factor, which determines the sharpness of the resonance, is given by
The effective resistance of a LGR can be estimated by considering the length of conductor through which the current travels and the cross-sectional area available to it. The relevant conductor length is the circumference formula_27 of the conductor's inner surface. The depth that the current penetrates into the inner surface of the LGR bore is determined by the electromagnetic skin depth formula_5. Therefore, the cross-sectional area through which charge flows is formula_29. Combining these results gives an effective resistance
where formula_4 is the resistivity of the conductor. The effective capacitance, inductance, and resistance then lead to a simple expression for the expected quality factor of the LGR
where, for a good conductor, the electromagnetic skin depth at the resonance frequency is given by
and formula_34. For an aluminum resonator with formula_35 and formula_36 the above analysis predicts formula_37.
Radiative losses.
In practice, the measured quality factor a cylindrical LGR, without additional electromagnetic shielding, will be much less than the predicted value of formula_38. The suppression of the quality factor is due to radiative power loss from magnetic field lines that extend out of LGR bore and into free space. An order-of-magnitude estimate of the effective radiation resistance can be made by treating the LGR as a conducting loop. In the limit that the wavelength of the radiation is much larger than the loop radius formula_21, the radiation resistance is
and can be much larger than the resistance formula_41 due to the resistivity of the LGR conductor.
The radiative losses can be suppressed by placing the LGR inside a circular waveguide. Provided that the cutoff frequency of the lowest TE11 waveguide mode is well above the resonant frequency of the LGR, the magnetic field lines will be prevented from propagating into free space. The presence of the electromagnetic shield will alter the resonant frequency and quality factor of the LGR, but typically by only a few percent.
Toroidal LGR.
In some applications requiring high quality factors, the electromagnetic shielding provided by a concentric circular waveguide surrounding a cylindrical LGR can be bulky and awkward to work. A toroidal LGR can be used for high-formula_9 measurements without requiring additional electromagnetic shielding. In the toroidal geometry the two ends of a cylindrical LGR are joined to form a completely closed structure. In this case, the magnetic field is completely confined within the bore of the resonator and there is no radiative power loss. The toroidal LGR consists of two halves that are bolted together along the outer diameter of the structure.
Like the cylindrical LGR, the toroidal LGR can be modelled as a series formula_6 circuit. In general, the effective capacitance, inductance, and resistance of the toroidal LGR will differ from the expressions given above for the cylindrical LGR. However, in limit that the radius of the torus is large compared to the bore radius formula_21, the capacitance, inductance, and resistance of the toroidal LGR are approximated by the expressions above if one takes formula_18 to be equal to the circumference of the torus.
The toroidal LGR is particularly convenient when characterizing the electromagnetic properties of liquid samples or particles suspended in a liquid. In these cases, the bore of the toroidal LGR can be partially filled with the liquid sample without requiring a special sample holder. This setup allows one to characterize the magnetic properties of, for example, a ferrofluid. Alternatively, if the liquid sample is nonmagnetic, the entire toroidal LGR can be submerged in the liquid (or gas). In this case, the dielectric properties of the sample only modify the effective capacitance of the resonator and the changes in formula_8 and formula_9 can be used to determine the complex permittivity of the sample.
Coupling to a LGR.
Inductive coupling loops are typically used to couple magnetic flux into and out of the LGR. The coupling loops are made by first removing a length of outer conductor and dielectric from a semi-rigid coaxial cable. The exposed centre conductor is then bent into a loop and short-circuited to the outer conductor. The opposite end of the coaxial cable is connected to either a signal generator or a receiver. In the case of a signal generator, an oscillating current is established in the coupling loop. By Faraday's law of induction, this current creates and oscillating magnetic flux which can be coupled into the bore of the LGR. This magnetic flux, in turn, induces circumferential currents along the inner wall of the LGR. The induced current, once again by Faraday's law, creates an approximately uniform oscillating magnetic field in the bore of the LGR. A second coupling loop, connected to a receiver, can be used to detect the magnetic flux produced by the LGR. Alternatively, using a vector network analyzer (VNA), a single coupling loop can be used to both inject a signal into the LGR and measure its response. The VNA can measure the ratio of the forward and reflected voltages (formula_48, or reflection coefficient) as a function of microwave frequency. Far away from resonance, the magnitude of the reflection coefficient will be close to one since very little power is coupled into the LGR at these frequencies. However, near the resonance frequency formula_8, the magnitude of the reflection coefficient will fall below one as power is transferred into the LGR. The coupling between the external circuits and the LGR can be tuned by adjusting the relative positions and orientations of the coupling loop and LGR. At critical coupling, impedance matching is achieved and the reflection coefficient approaches zero.
It is also possible to capacitively couple electric fields into and out of the gap of the LGR using suitably-fashioned electrodes at the end of a coaxial cable.
Multi-Loop, Multi-Gap LGRs.
Multi-loop, multi-gap LGRs have also been developed. The simplest of these is the two-loop, one-gap LGR. In this case, magnetic field lines form closed loops by passing through each of the bores of the LGR and the currents on the inner walls propagate in opposite directions - clockwise in one bore and counterclockwise in the other. The equivalent circuit, neglecting losses, is a parallel combination of inductors formula_2 and formula_51 in series with capacitance formula_1. If formula_53, then the resonant frequency of the two-loop, one-gap LGR is formula_54 times greater than that of the conventional one-loop, one-gap LGR having the same bore and gap dimensions. It is also worth noting that, since magnetic field lines pass from one bore to the other, radiative power losses are strongly suppressed and the resonator maintains a high quality factor without requiring additional electromagnetic shielding.
The multi-loop, multi-gap LGRs with more than two loops have more than one resonant mode. If the central bore is singled out as having inductance formula_51, then one of the resonant modes is one in which all of the magnetic flux from each of the external loops of inductance formula_2 is shared with the central loop. For this mode, the resonant frequency of an formula_57-loop, formula_58-gap LGR is given by
where it has been assumed that all loops have the same inductance formula_2.
LGRs and superconductivity.
Loop-gap resonators have been used to make precise measurements of the electrodynamic properties of unconventional superconductors.
Most notably, a LGR was used to reveal the linear temperature dependence of the magnetic penetration depth, characteristic of a d-wave superconductor, in a single crystal of YBa2Cu3O6.95.
In these experiments, a superconducting sample is placed inside the bore of a LGR. The diamagnetic response of the superconductor alters in the inductance of the LGR and, therefore, its resonant frequency. As described below, tracking the change in the resonant frequency as the temperature of the sample is changed allows one to deduce the temperature dependence of the magnetic penetration depth.
Theory.
The inductance of the LGR can be expressed as formula_61, where formula_62 is the volume of the LGR bore. Since the resonant frequency formula_8 of the LGR is proportional to formula_64, a small change in the effective volume of the resonator bore will result in a change in the resonant frequency given by
Due to the Meissner effect, when a superconducting sample is place in the bore of a LGR, the magnetic flux is expelled from the interior of the sample to within a penetration depth formula_66 of its surface. Therefore, the effective volume of the resonator bore is reduced by an amount equal to the volume from which the magnetic flux has been excluded. This excluded volume is given by
where formula_68, formula_69, and formula_70 are the sample dimensions along the three crystallographic directions and formula_71 is the sample volume formula_72. In the above expression, it has been assumed that the microwave magnetic field is applied parallel to the formula_69-axis of the sample. Since the presence of the superconductor reduces the LGR volume, formula_74 and
Solving this expression for the formula_68-axis penetration depth yields
Generally, it is not possible to use LGR frequency-shift measurements to determine the absolute value of the penetration depth because it would require knowing the sample thickness formula_70 very precisely. For example, in fully doped YBa2Cu3O7, formula_79 at low temperature.
Therefore, to use the LGR measurement to determine formula_80 to within 10%, one would have to know the value of formula_70 with an accuracy of formula_82 which is typically not possible.
Instead, the strategy is to track the changes in frequency as the sample temperature varies (while keeping the LGR at a fixed temperature). The absolute penetration depth can be expressed as
where formula_84 is temperature, formula_85 is the experimental base temperature, and formula_86 is the change in penetration depth as the sample temperature is increased above the base temperature. One can, therefore, express the change in penetration depth as
Finally, defining formula_88, one has
This final expression shows how the LGR shifts in resonant frequency can be used to determine the temperature dependence of the magnetic penetration depth in a superconducting sample.
Experimental details.
In a d-wave superconductor, the penetration depth typically changes by a few ångströms per degree kelvin, which corresponds to formula_90 for a formula_91 platelet sample in a LGR with a bore volume of formula_92. Measuring such small changes in relative frequency requires an extremely high-formula_9 resonator. The ultrahigh quality factors are obtained by coating the LGR surfaces with a superconducting material, such as a lead-tin alloy. The resonator is then cooled below the superconducting transition temperature of the coating using a bath of superfluid liquid helium. Quality factors of formula_94 have been achieved using copper LGRs coated with lead-tin and cooled to formula_95.
Measuring permittivity and permeability.
This section describes how LGRs can be used to determine the electromagnetic properties of materials. When there are no materials filling either the gap or bore of the resonator, the impedance formula_96 of the LGR can be expressed as
where formula_98. Re-expressed in terms of the resonant frequency formula_99 and quality factor formula_9, the impedance is given by
A measurement of the frequency dependence of the impedance of an empty LGR can be used to determine formula_99 and formula_9. The impedance measurement is most easily done using the vector network analyzer (VNA) to measure the reflection coefficient formula_48 from an inductively-coupled LGR. The impedance and reflection coefficient are related by
where formula_106 is the output impedance of the VNA (usually, formula_107).
Complex permittivity.
Now suppose that the gap of resonator has been completely filled with a dielectric material that has complex relative permittivity formula_108. In this case, the effective capacitance becomes formula_109 and the impedance of the LGR is given by
Separating the real and imaginary terms leads to
This expression shows that a nonzero formula_112 enhances the effective resistance of the LGR and, therefore, lowers its quality factor. A nonzero formula_113, on the other hand, alters the imaginary part of the impedance and modifies the resonant frequency. Written in terms of the empty-resonator resonant frequency and quality factor, the above impedance can be expressed as
Provided that formula_99 and formula_9 are known before hand, a measurement of the frequency dependence of formula_117 can be used to determine formula_118 and formula_112 of the material filling the gap of the LGR. This analysis gives the values of formula_118 and formula_112 at the resonant frequency of the filled LGR.
Complex permeability.
Next, suppose that the bore of a LGR is filled with a magnetic material have complex relative permeability formula_122. In this case, the effective inductance becomes formula_123 and the impedance of the LGR is given by
Separating formula_125 into its real and imaginary components and writing the impedance in terms of formula_99 and formula_9 of the empty LGR yields
Once again, formula_129 contributes additional dissipation which lowers the quality factor of the filled resonator and formula_130 shifts the resonant frequency. A measurement of the frequency dependence of formula_125 can be used to extract the values of formula_130 and formula_129 at the resonant frequency of the filled LGR.