page_title: Survey meter
section_title: Measurement Techniques and Interpretation
hierarchical_section_title: Survey meter / Measurement Techniques and Interpretation
caption_attribution_description: A radiation technician conducts a spot survey on debris at Corrective Action Unit 511, Site A located on the Nevada Test Site
context_page_description: Survey meters in radiation protection are hand-held ionising radiation measurement instruments used to check such as personnel, equipment and the environment for radioactive contamination and ambient radiation. The hand-held survey meter is probably the most familiar radiation measuring device owing to its wide and visible use.
context_section_description: The user must have an awareness of the types of radiation to be encountered so that the correct instrument is used. A further complication is the possible presence of "mixed radiation fields" where more than one form of radiation is present. Many instruments are sensitive to more than one type of radiation; alpha and beta, or beta and gamma, for instance, and the operator must know how to discriminate between these. The necessary skills in using a hand-held instrument are not only to manipulate the instrument, but also to interpret results of the rate of radiation exposure and the type of radiation being detected.
  For instance, a Geiger end-window instrument cannot discriminate between alpha and beta, but moving the detector away from the source of radiation will reveal a drop off in alpha as the detector tube must normally be within 10mm of the alpha source to obtain a reasonable counting efficiency. The operator can now deduce that both alpha and beta is present. Likewise for a beta/gamma geiger instrument, the beta may have an effect at a range in the order of metres, depending on the energy of the beta, which may give rise to the false assumption that only gamma is being detected, but if a sliding shield type detector is used, the beta can be shielded out manually, leaving only the gamma reading.
  For this reason, an instrument such as the dual phosphor scintillation probe, which will discriminate between alpha and beta, is used where routine checking will come across alpha and beta emitters simultaneously. This type of counter is known as "dual channel" and can discriminate between radiation types and give separate readouts for each.
  However, scintillation probes can be affected by high gamma background levels, which must therefore be checked by the skilled operator to allow the instrument to compensate. A common technique is to remove the counter from any proximity to alpha and beta emitters and allow a "background" count of gamma. The instrument can then subtract this in subsequent readings.
  In dose survey work Geiger counters are often just used to locate sources of radiation, and an ion chamber instrument is then used to obtain a more accurate measurement owing to their better accuracy and capability of counting higher dose rates.
  In summary, there are a variety of instrument features and techniques to help the operator to work correctly, but the use by a skilled operator is necessary to ensure reliable results. The UK Health and Safety Executive has issued a guidance note on selecting the correct instrument for the application concerned, and the care and use of such instruments.
caption: Scintillation probe being used to measure surface radioactive contamination. The probe is held as close to the object as practicable
exif: {"Image Make": "Canon", "Image Model": "Canon PowerShot A80", "Image Orientation": "Horizontal (normal)", "Image XResolution": "180", "Image YResolution": "180", "Image ResolutionUnit": "Pixels/Inch", "Image Software": "Adobe Photoshop 7.0", "Image DateTime": "2005:06:27 15:39:26", "Image YCbCrPositioning": "Centered", "Image ExifOffset": "216", "Thumbnail Compression": "JPEG (old-style)", "Thumbnail XResolution": "72", "Thumbnail YResolution": "72", "Thumbnail ResolutionUnit": "Pixels/Inch", "Thumbnail JPEGInterchangeFormat": "1058", "Thumbnail JPEGInterchangeFormatLength": "6403", "EXIF ExposureTime": "1/400", "EXIF FNumber": "28/5", "EXIF ExifVersion": "0220", "EXIF DateTimeOriginal": "2005:02:04 10:27:32", "EXIF DateTimeDigitized": "2005:02:04 10:27:32", "EXIF ComponentsConfiguration": "YCbCr", "EXIF CompressedBitsPerPixel": "3", "EXIF ShutterSpeedValue": "277/32", "EXIF ApertureValue": "159/32", "EXIF ExposureBiasValue": "0", "EXIF MaxApertureValue": "95/32", "EXIF MeteringMode": "Pattern", "EXIF Flash": "Flash did not fire, compulsory flash mode", "EXIF FocalLength": "125/16", "EXIF FlashPixVersion": "0100", "EXIF ColorSpace": "Uncalibrated", "EXIF ExifImageWidth": "2272", "EXIF ExifImageLength": "1704", "EXIF FocalPlaneXResolution": "56800/7", "EXIF FocalPlaneYResolution": "56800/7", "EXIF FocalPlaneResolutionUnit": "2", "EXIF SensingMethod": "One-chip color area", "EXIF FileSource": "Digital Camera", "EXIF CustomRendered": "Normal", "EXIF ExposureMode": "Auto Exposure", "EXIF WhiteBalance": "Auto", "EXIF DigitalZoomRatio": "1", "EXIF SceneCaptureType": "Landscape"}