Apparatus and method for measuring the phase and magnitude of microwave signals
This is a continuation of U.S. Ser. No. 07/764,975, filed Sep. 23, 1993, now abandoned, U.S. Ser. No. 07/644,684, filed Jan. 22, 1991, now abandoned, U.S. Ser. No. 507,109, filed Apr. 9, 1990, now abandoned, and U.S. Ser. No. 176,202, filed Mar. 31, 1988, now abandoned, assigned to the assignee of the present application, and incorporated by reference herein. 
    
    
     LIMITED COPYRIGHT WAIVER 
     A portion of this patent document, contains material to which a claim of copyright protection is made. The copyright owner has no objection to the facsimile reproduction by anyone of the patent document, or the patent disclosure as it appears in the U.S. Patent and Trademark Office patent file records, but reserves all other rights whatsoever. 
     BACKGROUND OF THE INVENTION 
     1. Field of the Invention 
     The invention relates generally to systems for measuring microwave frequency signals and more particularly to systems used to extract phase and magnitude information from microwave frequency signals. 
     2. Description of the Related Art 
     Earlier vector network analyzers with integral processors exist which can perform both phase and magnitude measurements of microwave frequency signals. Unfortunately, there have been shortcomings with these earlier analyzers. In particular, such earlier analyzers often employed relatively expensive techniques to achieve frequency accuracy from a microwave frequency signal source. Furthermore, such earlier analyzers often performed measurements of signals relatively slowly, were relatively clumsy to operate and failed to conveniently provide to an analyzer user desired microwave frequency measurement information. 
     Thus, there has been a need for a more economical microwave frequency signal measurement system which takes more rapid measurements of microwave frequency signals, which is relatively easy to use and which can, at any moment, provide a user with a wider range of microwave frequency measurement information. Furthermore, there has been a need for such a system which provides relatively low noise signals for phase and magnitude measurement. The present invention meets these needs. 
     SUMMARY OF THE INVENTION 
     The present invention provides a measurement system. The system includes a source of signals at discrete microwave frequencies in a prescribed microwave frequency range. The signal source receives feedback signals. The discrete microwave frequency signals are substantially phase-locked to at least one downconverted signal in response to the feedback signals. Downconverting circuitry linearly downconverts the signals at the discrete frequencies and provides the at least one downconverted signal. A phase detector receives the at least one downconverted signal and provides the feedback signals. 
    
    
     BRIEF DESCRIPTION OF THE DRAWINGS 
     Further details of the present invention are explained with the help of the attached drawings in which: 
     FIG. 1 shows a block diagram illustrating the measurement system of the present invention; 
     FIG. 2A illustrates the system of FIG. 1 operating in the harmonic mode; 
     FIG. 2B illustrates the system of FIG. 1 operating in the direct mode; 
     FIG. 3A illustrates further details of the test set 106 of FIG. 1; 
     FIG. 3B illustrates details of the the analyzer 108 of FIG. 1; 
     FIG. 4 illustrates details of the splitter 110 of FIG. 1; 
     FIG. 5 illustrates details of one of the harmonic mixers 174 used in the present invention; 
     FIG. 5A illustrates the nature of the harmonic mixer of FIG. 5 in a harmonic mode; 
     FIG. 5B illustrates the filter behavior of the harmonic mixer of FIG. 5 in a harmonic mode; 
     FIG. 6 illustrates details of the first local oscillator 122 of FIG. 1; 
     FIG. 7 illustrates details of the phase lock circuit 252 used to control the frequency of operation of the first local oscillator of FIG. 6; 
     FIG. 8 illustrates details of the harmonic generator 170 of the present invention; 
     FIG. 9 illustrates details of the second local oscillator 124 of FIG. 1; 
     FIG. 10 illustrates details of the phase lock circuit 314 used to control the frequency of operation of the second local oscillator 124 of FIG. 9; 
     FIG. 11 illustrates details of image reject mixer 183 and its associated variable driver assembly 177 for the present invention; 
     FIG. 12 illustrates details of the calibration oscillator 134 and the third local oscillator 128 of FIG. 1; 
     FIG. 13 illustrates details of a respective fourth local oscillator 146 and the details of one of the three synchronous detectors 130 and two of the six sample/hold circuits 148 from FIG. 1; 
     FIG. 14 illustrates details of the source lock circuitry 154 of FIG. 1; 
     FIG. 15 illustrates the ability of the system to select unevenly spaced discrete frequency points; 
     FIG. 16 illustrates the ability of the system to select evenly spaced frequencies; 
     FIG. 17 illustrates conditions necessary to create a spurious If signal; 
     FIG. 18 shows an algorithm for automatically checking for spurious harmonics; 
     FIGS. 19-20 provide schematic diagrams illustrating details of the test set I/O circuit 534 of FIG. 3B; 
     FIG. 21 provides a timing diagram illustrating the processor pipeline timing of the system of FIG. 1 for three sequential stimulus signal frequencies: F N-1 , F N  and F N+1  ; 
     FIG. 22 shows exemplary components of a source locking circuit 154 of FIG. 1; 
     FIGS. 23-32 illustrate details of the test port connectors of the present invention; 
     FIG. 33 illustrates the types of measurements that the model 360 is capable of making; 
     FIG. 34 shows the network analyzer is a tuned receiver; 
     FIG. 35 shows a phase reference can be obtained by splitting the microwave source signal; 
     FIG. 36 illustrates phase measurement using a split microwave source signal and replacing a DUT with a length of transmission line; 
     FIG. 37 illustrates a test signal lagging the reference signal by 360 degrees; 
     FIG. 38 illustrates a test signal path length 0.1 wavelength longer than that of the reference signal; 
     FIG. 39 shows a plot of measured phase vs. frequency due to length differences between a reference and test signal connection; 
     FIG. 40 illustrates insertion of a length of line to make reference and test signal paths equal; 
     FIG. 41 displays the phase vs. frequency response of a device with a path lengh difference between reference and test signals; 
     FIG. 42 shows the resultant measurement path length difference between reference and test signals with software compensation; 
     FIG. 43 shows definition of a reference enabling definition of reflection characteristics from forward and reference directions; 
     FIG. 44 illustrates the four scattering parameters; 
     FIG. 45 illustrates display of linear phase vs. frequency; 
     FIG. 46 illustrates polar display; 
     FIG. 47 illustrates a Smith Chart display; 
     FIG. 48 illustrates vector error correction; 
     FIGS. 49-50 show WILTRON Test Sets; 
     FIG. 51 illustrates smoothing; 
     FIG. 52 illustrates the control panel layout; 
     FIG. 53 illustrates the calibration keys; 
     FIG. 54 provides a flowchart describing the calibration sequence; 
     FIGS. 55-57 show calibration menus; 
     FIG. 58 is a utility menus key flow chart; 
     FIG. 59 shows available display formats; 
     FIG. 60 illustrates marker annotation for different graph-types; 
     FIG. 61 shows the data, menu and sweep indicator areas of the display; 
     FIG. 62 is a flow chart of a display control algorithm for selection of an active channel; 
     FIG. 63 shows a Tabular Printout Format; 
     FIG. 64 shows an Alternate Data Format; 
     FIG. 65 shows a header which prints before the screen data prints; 
     FIGS. 66-68 illustrate measurement of group delay; 
     FIG. 69 illustrates a set up for testing active devices; 
     FIG. 70 illustrates that measurements of active devices are made with a reference plane outside the device; 
     FIG. 71 illustrates testing an ideal device; 
     FIG. 72 illustrates the nonideal transition from a coax structure which affects freely moving the reference plane; 
     FIG. 73 shows some of the special test-fixture calibration standards that are available; 
     FIG. 74 shows open, short and termination measurement configurations with transition from a coax to permit solving for unknowns; 
     FIG. 75 illustrates a calibration device including a varactor with three known impedances; 
     FIG. 76 illustrates an impulse response for different circuit elements; 
     FIG. 77 illustrates the impulse response of a stepped attenuator; 
     FIG. 78 illustrates the impuilse response for a series inductance; 
     FIG. 79 illustrates use of the impulse response to locate a discontinuity in a cable; 
     FIG. 80 shows the bandpass-impulse response for various impedance discontinuities; 
     FIG. 81 illustrates a phasor-impulse response for various devices; 
     FIG. 82 ilustrates utilization of lowpass impulse, lowpass step and bandpass responses for a complex impedance device; 
     FIG. 83 shows phasor-impulse response displays one discontinuity at a time; 
     FIG. 84 illustrates windowing; 
     FIG. 85 illustrates gating; 
     FIG. 86 illustrates gating of an antenna transmission measurement to remove unwanted ground reflections; 
     FIG. 87 illustrates bus interface connections; 
     FIG. 88 illustrates typical handshake operation; and 
     FIG. 89 illustrates a binary data transfer message format. 
    
    
     DETAILED DESCRIPTION OF THE PREFERRED EMBODIMENT 
     Referring to FIG. 1, there is shown a first block diagram illustrating the measurement system 100 of the preferred embodiment, The system generates stimulus signals over a range of individual frequencies. For each frequency, a stimulus signal is applied to a Device Under Test (DUT) 102. For each frequency, the system produces test signals which contain phase and magnitude information which can be used to characterize the DUT 102 at that frequency. The test signals comprise bidirectional signals transmitted through or reflected by the DUT 102 at the stimulus signal frequency. The system also produces reference signals at the stimulus signal frequency. The system downconverts the test and reference signals from the stimulus signal frequency, to a DC in phase and quadrature component. For each stimulus signal frequency, the system produces digital logic level signals carrying phase and magnitude information which characterize the DUT 102 at the particular stimulus signal frequency. The digital logic level signals are provided to digital processing circuitry 190 which, in addition to processing the digital logic level signals and producing a display, controls the signal source 104 via signals on GPIB 265. 
     In the first block diagram, details of the system 100 relating to the processing of the digital logic level signals and to the display of the phase and magnitude information have been omitted in order to simplify the diagram. These components are described in relation to other figures. 
     The measurement system 100 includes a signal source 104 which provides individual stimulus signals over a range of frequencies. The system includes a test set shown within dashed lines 106. The test set 106 splits stimulus signals into two signals at the stimulus signal frequency. One of the two split signals is used as a reference signal, and the other split signal is provided to the DUT 102 to generate test signals at the stimulus signal frequency. The test set 106 can provide split signals to the DUT 102 in either a forward or a reverse direction, as described below, so as to generate forward and reverse scattering parameters for the DUT 102. 
     The test signals comprise split signals transmitted through the DUT 102 or reflected by it. The phase and magnitude of the test signals relative to the reference signal depends upon the characteristics of the DUT 102. Thus, for each individual stimulus signal frequency, the phase and magnitude of the test signals relative to those of the reference signal provides information about the characteristics of the DUT 102 at the individual stimulus signal frequency. 
     In order to facilitate the extraction of the phase and magnitude information from the reference and test signals, the test set 106 downconverts the reference and test signals from the stimulus signal frequency to an intermediate frequency of 83.33 Khz. Since the frequency downconversion is a linear process, the test signals downconverted to 83.33 Khz retain the same phase and magnitude relative to the reference signals downconverted to 83.33 Khz. 
     For each stimulus signal frequency, the test set 106 provides the downconverted 83.33 Khz test and reference signals to an analyzer, indicated by the reference numeral 108. The analyzer 108 analyzes the downconverted test and reference signals to extract the phase and magnitude information. It then provides to the digital processing circuitry 190, digital logic level signals containing phase and magnitude information which can be used to characterize the DUT 102 at each discrete stimulus signal frequency. 
     In the presently preferred embodiment, the signal source 104 can provide stimulus signals over a range of frequencies from 0.01 to 40 Ghz. The WILTRON 360SS69 available from WILTRON Company, Morgan Hill, Calif. is the signal source in the preferred embodiment. The signal source 104 is a sweeper which provides a series of unsynthesized stimulus signals over a range of frequencies for which the characteristics of the DUT 102 are to be tested. 
     The test set 106 includes a signal splitter 110 which receives discrete stimulus signals in the series over a desired frequency range from the signal source 104. In the preferred embodiment, the test set 106 is the 3611A available from WILTRON company. The signal splitter 110 splits each received signal into two signals having the same phase and magnitude. One of the split signals is provided on either line 112 or 118 as a reference signal, and the other split signal is applied to the DUT 102. Constituents of the split signal applied to the DUT 102 which are transmitted through it or reflected by it constitute test signals provided on lines 114 and 116. For each stimulus signal frequency, the test signals typically differ in phase or magnitude from the reference signal. This phase and magnitude difference represents information which can be used to characterize the DUT 102. 
     As will be more fully explained below, the splitter 110 is capable of applying stimulus signals to the DUT 102 in either a forward or a reverse direction. Thus, information can be obtained about both the forward and reverse transmission and reflection of stimulus signals by the DUT 102. When the splitter 110 applies signals to the DUT 102 in the forward direction: a reference signal r 1  is provided on line 112; a reflected test signal is provided on line 114 as signal t 1  ; and a transmitted test signal is provided on line 116 as signal t 2 . When, on the other hand, the splitter 110 provides a signal to the DUT 102 in the reverse direction: a reference signal r 2  is provided to line 118; a transmitted signal is provided to line 114 as signal t 1  ; and a reflected signal is provided on line 116 as signal t 2 . 
     The test set 106 includes a microwave downconverter 120 which receives mixing signals from respective first and second local oscillators 122 and 124. The downconverter 120 downconverts the respective reference and test signals from the stimulus signal frequency generated by the signal source 104 to a second intermediate frequency (IF 2 ) of 2.25 Mhz. 
     The first local oscillator 122 has a frequency range of approximately 357-536.5 Mhz. The second local oscillator 124 has a frequency range of approximately of 91.65±1.875 Mhz when the system is operating in a &#34;harmonic mode&#34;, and has a frequency range of approximately 12.25-272.25 Mhz when the system is operating in a &#34;direct mode&#34;. As explained below, the exact frequency of operation of the respective first and second local oscillators 122 and 124 depends upon the frequency of the stimulus signal provided by the source 104. 
     In operation, the signal source 104 typically provides a series of stimulus signals at discrete frequencies over a prescribed range of frequencies in order to characterize the DUT 102 over that prescribed frequency range. Thus, the respective frequencies of the first and second local oscillators 122 and 124 are varied such that, for each different stimulus signal frequency, the test and reference signals are downconverted to an IF 2  of 2.25 Mhz. 
     A multiplexer/downconverter 126 receives the 2.25 Mhz reference and test signals as well as a 2.33 Mhz signal generated by a third local oscillator 128. It downconverts the 2.25 Mhz reference and test signals to a third intermediate frequency (IF 3 ) of 83.33 Khz. 
     The multiplexer/downconverter 126 also selects either the reference signal r 1  on line 136 or reference signal r 2  on line 138 for provision to synchronous detectors 130. As explained above, at any given moment during the testing of a DUT 102, only one reference signal is provided by the splitter 110. When the splitter 110 applies signals to the DUT 102 in the forward direction, then multiplexer/downconverter 126 selects reference signal r 1 . When the splitter 110 applies signals to the DUT 102 in the reverse direction, then multiplexer/downconverter 126 selects reference signal r 2 . For each discrete stimulus signal frequency, the test set 106 can automatically switch between applying signals in the forward direction to the DUT 102 and applying signals in the reverse direction to the DUT 102. 
     Thus, at any moment during the testing of a DUT 102, the measurement system provides three channels: test channel t 1 , test channel t 2  and reference channel r 1 , r 2 . In operation, as explained below, the splitter 110 can be caused to alternate between applying the stimulus signal to the DUT 102 in the forward direction and in the reverse direction. Consequently, multiplexer/downconverter 126 must be capable of alternatively coupling reference signal r 1  on reference line 136 or reference signal r 2  on line 138 to line 140 such that: when the splitter 110 applies the stimulus signal in the forward direction, signal r 1  is provided on line 140, and when the splitter 110 provides the stimulus signal in the reverse direction, signal r 2  is provided on line 140. 
     Gain ranging/calibration circuitry 132 receives the 83.33 Khz test and reference signals from the multiplexer/downconverter 126. During normal testing operations, this circuitry 132 amplifies these signals and provides them to the synchronous detectors 130 of the analyzer 108. The amplification of the reference and test signals ensures that they are at a high enough signal level to be accurately converted by the synchronous detectors 130 and the ADC 150. 
     Periodically (approximately once every two and one-half minutes in the presently preferred embodiment), in the course of the normal testing operation of the measurement system, the gain ranging/calibration circuitry 136 decouples the test and reference signals on lines 140, 142 and 144 from gain ranging amplifier assemblies (discussed below) and the amplifiers and the synchronous detectors 130, and couples the gain ranging amplifier assemblies and the synchronous detectors 130 to an 83.33 Khz calibration signal provided by a calibration oscillator 134. The calibration signal is used to calibrate a gain ranging amplifier assembly, discussed below, and to calibrate the synchronous detectors 130 to compensate for phase and magnitude drift in the test and reference signals. The manner in which this calibration is accomplished is discussed in more detail below. 
     The analyzer 108 incorporates the synchronous detectors 130. In the preferred embodiment, the analyzer 108 is the WILTRON Model 360 Network Analyzer available from WILTRON Company. 
     The synchronous detectors 130 receive the downconverted 83.33 Khz test and reference signals and an 83.33 Khz signal provided by respective fourth local oscillators 146. The synchronous detectors provide DC Real and DC Imaginary signals containing the phase and magnitude information carried by the respective test and reference signals. 
     A Sample/Hold (S/H) and multiplexer circuit 148 receives the DC Real and DC Imaginary signals. It parallel samples all of the signals and holds the sampled signals until they can be individually applied to an analog-to-digital converter (ADC) 150 which converts the sampled signals to digital logic levels. The digital logic level signals are provided to the processing circuitry 190 which processes the signals and produces a display illustrating the phase and magnitude information characterizing the DUT 102 over the frequency range of interest or upon further processing, displays the time domain representation of the signals. 
     The analyzer 108 includes a 2.25 Mhz oscillator 152 and a source lock circuit 154. The source lock circuit 154 receives on line 156 the 2.25 Mhz downconverted version of the reference signal, r 1  or r 2  (depending upon whether the test set 106 applies the stimulus signal to the DUT 102 in the forward or the reverse direction). At the same time, the source lock circuit 154 receives on line 158 a 2.25 Mhz signal provided by oscillator 152. In response to phase differences between these two received signals, the source lock 154 provides to the signal source 104, via line 160, a correction signal used to remove any phase drift in the stimulus signal frequency. The signal source 104, therefore, is phase-locked to a desired stimulus frequency using a downconverted 2.25 Mhz version of a reference signal, either r 1  or r 2 . 
     An overview of the operation of a presently preferred embodiment of the measurement system will now be provided with reference to FIGS. 2A and 2B. The drawings of FIGS. 2A and 2B are stripped-down representations of the system. These two figures are merely intended to aid in the explanation of the reference and test signal downconversion and the source lock of the signal source using the IF 2  reference signals. The drawings of these two figures are not intended to provide a detailed representation of the system. 
     In order to reduce the noise level of the downconverted versions of the reference and test signals, the system can operate in either a &#34;harmonic mode&#34; or a &#34;direct mode&#34;. The system 100 operating in the harmonic mode is illustrated in FIG. 2A, and the system 100 operating in the direct mode is illustrated in FIG. 2B. The system 100 operates in the harmonic mode when the stimulus signal frequency is in the range of 0.2701-40 Ghz, and the system 100 operates in a direct mode when the stimulus signal frequency is in the range of 10-270 Mhz. 
     In FIG. 2A, the signal source 104 provides a stimulus signal in the range of 0.2701-40 Ghz, and the system 100 operates in the harmonic mode. The signal source 104 is under control of digital processing circuitry 190 via GPIB 265. The splitter 110 provides reference signals on lines 112 and 118, and it provides test signals on lines 114 and 116. The first local oscillator 122 provides a signal at a prescribed frequency in the range of 357-536.5 Mhz, via driver 168, to a harmonic generator 170. The harmonic generator, in turn, provides sampling pulses at a desired sampling rate to harmonic mixers 174. The harmonic mixers 174 sample the reference signals and the test signals, at the sampling rate. The sampled reference and test signals are thereby downconverted from the stimulus signal frequency to a first intermediate frequency (IF 1 ) of 89.4±1.875 Mhz. 
     The precise frequency at which the first local oscillator 122 operates is selected according to an algorithm described below. The selected frequency is chosen such that the harmonic mixers 174 sample the reference and test signals at a prescribed sampling rate so as to downconvert them to the desired first intermediate frequency (IF 1 ). 
     The downconverted reference and test signals at the IF 1  frequency are passed through respective drivers 176. The signals are provided to respective double balanced mixers 178 where they are mixed with a signal at a frequency in the range of 91.65±1.875 Mhz provided by the second local oscillator 124. The precise frequency of the signal provided by the second local oscillator 124 also is chosen in accordance with the algorithm described below. The respective double balanced mixers 178 output, onto lines 135 and 137 the test signals downconverted to the second intermediate frequency (IF 2 ) of 2.25 Mhz and onto lines 136 and 138 the reference signals downconverted to 2.25 Mhz. 
     During normal test measurement operation, test signal t 2  on line 137 is provided to image reject mixer 179. Switch 180 couples the test signal t 1  on 135 to image reject mixer 183. Switches 182 and 184 operate in unison to couple either reference signal r 1  on line 136 or r 2  on line 138 to image reject mixer 185 and to the source lock 154. When the splitter 110 applies signals to the DUT 102 in the forward direction, reference signal r 1 , is coupled to image reject mixer 185 and to the source lock 154, and when the splitter 110 applies signals to the DUT 102 in the reverse direction, reference signal r 2  is so coupled. Thus, the switch positions of switches 182 and 184 depend upon the operation of the splitter 110. 
     The respective 2.25 Mhz test signals t 1  and t 2  are provided, via respective drivers 187, to image reject mixers 179 and 183 which mix them with a 2.33 Mhz signal provided by the third local oscillator 128 and which output versions of the test signals downconverted to 83.33 Khz. 
     As explained above, switch 182 couples either reference signal r 1 , or r 2  to image reject mixer 185. The coupled reference signal is provided to image reject mixer 185 via a respective driver 187. The image reject mixer 185 mixes the selected 2.25 Mhz reference signal with the 2.33 Mhz signal provided by the third local oscillator 128 and outputs the coupled reference signal downconverted to 83.33 Khz. 
     The 83.33 Khz reference and test signals are respectively provided, via variable driver assemblies 177, to the synchronous detectors 130 wherein they are mixed with an 83.33 Khz signal produced by respective fourth local oscillators 146. The resultant DC Real and DC Imaginary reference and test signals are provided to sample and hold (S/H) circuitry 148-1. 
     DC Real and DC Imaginary signals stored by the S/H circuitry 148-1 are serially provided, via multiplexer 148-2, to ADC 150. The ADC 150 provides digital logic level signals to digital processing circuitry 190. 
     In the course of the downconversion of the stimulus signal, the reference signal downconverted to the IF 2  frequency, is used to generate a signal on line 156 which is provided to the source lock 154. At the same time, oscillator 152 provides a 2.25 Mhz signal to the source lock 154. The source lock 154, in turn; feeds back the correction signal, via line 160, to the signal source 104 to correct any drift in the signal source 104. Thus, the signal source 104 is phase-locked to a desired stimulus frequency using the IF 2  reference signal and the 2.25 Mhz signal provided by oscillator 152. 
     In FIG. 2B, the signal source 104 provides a stimulus signal in the range of 10-270 Mhz, and the system operates in the direct mode. One difference in the operation of the system in the direct mode from its operation in the harmonic mode is that, in the direct mode, the respective harmonic mixers 174 are bypassed such that the reference and test signals pass through substantially unaffected. Thus, the reference and test signals are conducted at the stimulus signal frequency directly to the respective double balanced mixers 178. Another difference is that the second local oscillator 124, in the direct mode, operates in a frequency range of 12.25-272.25 Mhz. The exact frequency at which the second local oscillator 124 operates is selected according to an algorithm discussed below. In all other significant respects, the operation of the system in the direct mode is substantially the same as its operation in the harmonic mode. 
     The use of both a harmonic mode and a direct mode improves the noise performance of the system 100 over the broad stimulus signal frequency range of 0.01-40 Ghz. More specifically, it is well known that system noise increases with harmonic number. By incorporating a first local oscillator 122 which, in the harmonic mode, operates in the range of 357-536.5 Mhz, the largest number harmonic of the system is 75 (536.5 Mhz×75=40 Ghz). Relatively speaking, 75 is a low harmonic number, and therefore, system noise is reduced. Thus, by using a relatively large high-end frequency (536.5 Mhz) for the first local oscillator 122, the largest harmonic number of the system, as well as system noise, are reduced. 
     In order to further reduce system noise, the first local oscillator 122 not only uses a relatively large high-end frequency (536.5 Mhz), but also itself operates with relatively low noise. Low noise operation of the first local oscillator 122 is important because any noise carried by its signal is multiplied by the harmonic generator 170. In order to reduce the noise level of signals produced by the first local oscillator 122, a relatively narrow frequency range (357-536.5 Mhz) is used so as to make the oscillator more resistant to noise in the tuning voltage used to tune the oscillator. Since the stimulus signal frequency range extends below the low end (357 Mhz) of the first local oscillator&#39;s frequency range, the system bypasses the harmonic mixers 174 and operates in the direct mode when the stimulus signal is in a relatively lower frequency range (10-270 Mhz). 
     Referring to FIGS. 3A and 3B, there are shown additional details of the system 100. FIG. 3A illustrates further details of the test set 106 which includes the first and second local oscillators 122 and 124. It also includes switches 180, 182 and 183 as well as limiter 522 shown within dashed lines 524. Furthermore, the test set 106 includes a digital interface 526 to analyzer 108. 
     The first local oscillator 122 includes a 357-536.5 Mhz voltage tuned oscillator (VTO) 254, tuning voltage summation circuit (TVS-1) 260 and digital-to-analog converter 266. Additional features of the first local oscillator 122 are described below. 
     The second local oscillator 124 includes a 98-272.5 Mhz VTO 316, tuning voltage summation circuit (TVS-2) 334, digital-to-analog converter 338, and &#34;÷L&#34; components. Additional features of the second local oscillator 124 are described below. 
     FIG. 3B illustrates details of the the analyzer 108 which includes a plurality of microprocessors in its digital processing circuitry 190. I/O processor 190-1 comprises an 8088 microprocessor produced by Intel Corp. which has a place of business in Sunnyvale, Calif. A first main processor 190-2A comprises an 8088 microprocessor, and a first math co-processor 190-2B comprises an Intel 8087 math co-processor. A second main processor 190-3A comprises an 8088 microprocessor, and a second math co-processor 190-3B comprises an Intel 8087 math co-processor. 
     The I/O processor 190-1 and the first main processor 190-2A are coupled to engage in two-way digital communication via FIFO circuitry 528. The first main processor 190-2A and the second main processor 190-3A are coupled to engage in two-way digital communication via FIFO circuitry 530. 
     The I/O processor 190-1 is coupled to the signal source 104 via the GPIB 265. Optionally, the I/O processor 190-1 also can be coupled to a plotter 532 via the GPIB 265. 
     The I/O processor 190-1 is coupled to a test set I/O circuit 534 via bus 269. The test set I/O circuit 534, which is part of the analyzer 108, is coupled to the digital interface 526, which is part of the test set 106. 
     The digital interface 526 is coupled to provide control signals, via I/O bus 267, to components of the test set 106 such as the splitter 110, digital-to-analog converters 266 and 338, the gain ranging amplifier assemblies 177 and the harmonic mixers 174. Thus, the I/O processor 190-1, in effect, controls the operation of the test set 106. 
     The analyzer 108 includes a 10 Mhz crystal controlled reference oscillator 536 which provides 10 Mhz reference signals to the third local oscillator 128 and the calibration oscillator 134, to the source lock 154 and to phase lock circuits 252 and 314. The reference oscillator also provides 10 Mhz signals to the synchronous detectors 130. 
     The system 100 can operate in either a sweeper mode in which the signal source 104 is an unsynthesized sweeper or in a synthesizer mode in which a synthesizer is substituted for the sweeper. A phase lock mode control 539 receives control signals from the source lock 154 and phase lock circuit 314. When the system 100 is in a sweeper mode, the mode control 539 provides the source lock control signal to the signal source 104 and provides the control signal from phase lock circuit 314 to the second local oscillator 124. When the system 100 is in the synthesizer mode, the mode control 539 provides the source lock control signal to the second local oscillator 124. Thus, in the sweeper mode, the signal source 104 is phase locked to the IF 2  reference, and in the synthesizer mode the second local oscillator 124 is phased locked to the IF 2  reference. 
     IF synchronization control circuitry 520 provides an absolute phase reference for the system 100. Circuitry 520 receives 10 Mhz signals generated by the crystal controlled reference signal generator 536. Circuitry 520 provides IF synchronization signals on line 541. It also provides to switching power supply 540, via line 538, signals which place the side bands of the switching power supply signals at approximately 55 Khz and 111 Khz, equally spaced on either side of the 83.33 Khz reference and test signals. The switching power supply 540, therefore, is injection locked by signals provided on line 538 to a frequency which does not interfere with measurements by the system 100. 
     The first main processor 190-2A is interfaced to a disk drive 542 and to a programmable module 544. A software operating system can be booted-up either from the disk drive 542 or the module 544. 
     The second main processor 190-3A is interfaced to a parallel printer 542, to a front panel 552, to the GPIB 265 and to a graphics control processor 544. The graphics control processor 546 in turn, is interfaced to an RGB (Red-Green-Blue) monitor 548. 
     The system 100 can buffer data to be printed. Therefore, the parallel printer 542 can print while the second main processor 190-3A is engaged in other processing. 
     A user can control the system 100 through the front panel 552. The user can monitor system measurements through the RGB monitor 548. 
     Referring to the illustrative drawings of FIG. 4, there are shown details of the splitter 110. The splitter 110 includes first and second splitter circuits 198 and 200 which can be individually coupled to receive the stimulus signal from the signal source 104. PIN diode switches 202 couple either splitter circuit 198 or 200 to the signal source. The test set also includes respective first and second couplers 204 and 206. Coupler 206 is asymmetrically coupled, as described below, in order to increase the dynamic range of scattering parameter S 21  in the forward direction Details of the first and second couplers are provided in co-pending commonly assigned Patent Application entitled DIRECTIONAL COUPLER AND TERMINATION FOR STRIPLINE AND COAXIAL CONDUCTORS, invented by William W. Oldfield, et al., Ser. No. 176,100, now U.S. Pat. No. 5,047,737, which was filed on the same date as this Application and which is hereby incorporated herein in its entirety by this reference. 
     The first splitter circuit 198, when coupled to receive the stimulus signal, splits that signal using two 50 ohm loads 208. A first one of the two split signals is provided on line 112 as reference signal r 1 . A second one of the two split signals is provided to the through-arm of the first coupler 204. 
     The second splitter circuit 200, when coupled to receive the stimulus signal, splits that stimulus signal using two 50 ohm loads 212. A first one of those two split signals is provided on line 118 as reference signal r 2 . A second one of those two split signals is provided to the coupling-arm of the second coupler 206. 
     Respective through-arms of couplers 204 and 206 are connected to the DUT 102 through Ports I and II, respectively. The coupling-arm of the first coupler 204 provides test signals on line 114 as described below, and the through-arm of the second coupler 206 provides test signals on line 116 as described below. 
     In operation, the test set 106 is used to generate reference and test signals used to calculate the &#34;Scattering Parameters&#34; for the DUT 102. The Forward Scattering Parameters are generated by connecting the first splitter circuit 198 to receive the stimulus signals. The first split signals are provided on line 112 as reference signal r 1 . The second split signals provided to the through-arm of the first coupler 204 are conducted through Port I to the DUT 102. Forward Transmission test signals are transmitted through the DUT 102 and through Port II to the through-arm of the second coupler. The Forward Transmission test signals are provided, via the through-arm of the second coupler 206, as test signal t 2  on line 116. Forward Reflection test signals are reflected by the DUT 102 back through Port I. The Forward Reflection test signals are conducted, via the coupling-arm of the first coupler 204, to line 114 as test signal t 1 . 
     The Forward Scattering Parameters are represented by the following equations. In these equations, S 21  represents the Forward Transmission Scattering Parameter, and S 11  represents the Forward Reflection Scattering Parameter. 
     
         S.sub.21 t.sub.2 /r.sub.1                                  (1) 
    
     
         S.sub.11 t.sub.1 /r.sub.1                                  (2) 
    
     Reverse Scattering Parameters are generated by connecting the second splitter circuit 200 to receive the stimulus signals. The first split signal is provided on line 118 as reference signal r 2 . The second split signals provided to the coupling-arm of the second coupler 206 are conducted, via Port II, to the DUT 102. The Reverse Transmission test signals are coupled to the coupling-arm of the first coupler 204 and are provided on line 114 as test signal t.sub.. The Reverse Reflection test signals are reflected by the DUT 102 back through Port II and travel, via through-arm of the second coupler 206, to line 116 where they are provided as test signal t 2 . 
     The Reverse Scattering Parameters are represented by the following equations. In these equations, S 12  represents the Reverse Transmission Scattering Parameter, and S 22  represents the Reverse Reflection Scattering Parameter. 
     
         S.sub.12 t.sub.1 /r.sub.2                                  (3) 
    
     
         S.sub.22 t.sub.2 /r.sub.2                                  (4) 
    
     It will be appreciated that the first and second couplers 204 and 206 are not identically coupled to the respective first and second splitter circuits 198 and 200. The first coupler 204 is coupled by its through-arm to the first splitter 198; whereas the second coupler 206 is coupled by its coupling-arm to the second splitter 200. The result is that the Forward Transmission test signals provided on line 116 as test signals t 2  suffer significantly less loss than do the Reverse Transmission test signals provided on line 114 as test signals t 1 . In the preferred embodiment, the Forward Transmission test signals suffer only approximately a 7 dB loss; whereas the Reverse Transmission test signals suffer approximately a 26 dB loss. Since the Forward Transmission test signals generally are of greater interest to users than are the Reverse Transmission test signals, this asymmetrical coupling of couplers 204 and 206 advantageously ensures that the more desirable Forward Transmission test signals are transmitted with less loss. 
     Details of the splitter 110 are described further in co-pending commonly assigned patent application entitled, ASYMMETRICAL COUPLING CIRCUIT, invented by Martin I. Grace, Ser. No. 175,956, now U.S. Pat. No. 4,808,913 filed on the same date as this Application and which is hereby incorporated herein in its entirety by this reference. 
     Referring to the illustrative drawings of FIG. 5, there are shown details of one of the harmonic mixers 174 of the preferred embodiment. Each of the harmonic mixers of the embodiment are substantially identical. The harmonic mixer 174 receives from the splitter 110 signals at the stimulus signal frequency. Sampling signal pulses generated at the sampling rate are provided to the dual slotline 220. The harmonic mixer 174 includes two 0.5 pf capacitors 224, two 100 ohm resistors 226, two 1 nF capacitors 242 as well as a 50 ohm resistor 228 and a 47k ohm resistor 229. The harmonic mixer also includes a tuneable one microhenry inductor 230 as well as two 2.7 nf capacitors 232 and respective 16k ohm resistors 234 and 236, a 100k ohm resistor 238 and a 1k ohm resistor 239. 
     The sampling signal pulses generated by the harmonic generator 170 are provided to a power splitter (not shown). Each of the two split sampling signal pulses then is provided to the dual slotline 220. Details of the dual slotline 220 and the manner in which stimulus frequency signals are provided to the dual slotline 220 are set forth in co-pending commonly assigned Patent Application, entitled DUAL SLOTLINE FEED, invented by W. Moberg, Ser. No. 176,098, now abandoned, filed on the same date as this Application and which is hereby incorporated herein in its entirety by this reference. 
     In operation, each respective sampling signal pulse causes respective diodes 240 to momentarily turn ON resulting in sampling of the signal provided by the splitter 110 at the stimulus signal frequency. The sampled signal is provided, via bypass capacitors 242 to the gate of FET 244. The output signal at the IF 1   frequency of 89.4±1.875 Mhz is provided on the drain terminal of the FET 244 to a driver 176 and then to a 2.25 Mhz notch filter 248. The notch filter 248 blocks spurious signals at the IF 2  so that such signals are not propagated to double balance mixer 178. Mixer 178 provides signals at IF 2  to bandpass filter 249. 
     During operation in the harmonic mode, voltages V 1  and V 3  are allowed to float so that diodes 240 are self-biasing. Voltage V 2  is +15 volts which turns ON PIN diode 250. The PIN diode 250, when turned ON, couples inductor 230 to the harmonic mixer 174. The inductor 230 ensures that the harmonic mixer has a frequency passband at the IF 1  of 89.4±1.875 Mhz. It should be noted that the IF 1  frequency is selected so as to provide continuous frequency coverage in the harmonic mode for the entire range of stimulus signal frequencies in that mode. 
     FIG. 5A illustrates the nature of the harmonic mixer 174 in the harmonic mode. The 3 pF capacitance is due to capacitors 224, 242 and to stray capacitance. The 5.4 nF capacitance is due to capacitors 232. The resistance R D  of 8k ohms is due to diodes 240. 
     FIG. 5B illustrates the filter behavior of the harmonic mixer 174 in the harmonic mode. The frequency of 89.4 Mhz is in the passband, and the frequency of 2.25 Mhz is rejected. 
     During operation in the harmonic mode, when one of the respective harmonic mixers 174 is receiving low level signals, it is sometimes necessary to shut OFF another one of the harmonic mixers 174. The shutting OFF is referred to as &#34;blanking&#34;. Blanking is accomplished by providing V 1  =+2.5 v. and V 3  =-2.5 v. Details of the blanking operation are provided in co-pending commonly assigned Patent Application entitled REFLECTED SIGNAL MEASUREMENT BLANKING CIRCUIT, invented by Glenn Ewart, Ser. No. 176,099, now U.S. Pat. No. 4,896,096 filed on the same day as this Application and which is hereby incorporated herein in its entirety by this reference. 
     During operation in the direct mode, voltage V 1  is -15 volts, and voltage V 3  is +15 volts. Thus, diodes 240 are permanently turned ON in the direct mode allowing signals provided by the splitter 110 at the stimulus signal frequency to be transmitted through the harmonic mixer 174 without being downconverted through sampling. Voltage V 2  is set to 0 volts so as to turn OFF PIN diode 250. Thus, in the direct mode, the inductor 230 is effectively decoupled from the harmonic mixer 174. As a result, the passband of the harmonic mixer in the direct mode is wide enough to accommodate all signals in the stimulus signal frequency range (10-270 Mhz) in that mode. 
     FIG. 6 illustrates details of the first local oscillator 122. FIG. 7 illustrates details of the phase lock circuit 252 used to control the frequency of operation of the first local oscillator 122. 
     The first local oscillator 122 includes a voltage tuned oscillator 254 which provides on line 256 signals in the frequency range 357-536.5 Mhz, and which receives on line 258 tuning voltage signals which tune the oscillator to a desired precise frequency of operation within that range. 
     The first local oscillator 122 includes tuning voltage summation circuit (TVS-1) 260 which receives on line 262 fine frequency steering signals from the phase lock control circuit 252. The TVS-1 260 receives on line 264 coarse frequency steering signals from eight bit digital-to-analog converter 266. The coarse steering signals provided by the digital-to-analog converter 266 are controlled by signals provided by the I/O processor 190-1, via I/O bus 269. 
     The tuning voltage signals generated by TVS-1 260 are provided to a 100 Khz, 150 Khz notch filter 268 used to remove sidebands caused by fractional M division explained below. The filtered signals are provided to a linearizer circuit 270 to obtain a linear frequency versus voltage tuning signal. The linearizer circuit 270, in turn, provides the linearized tuning voltage signals, via line 258, to the voltage tuned oscillator 254. 
     Window compare circuitry 271 receives the fine frequency steering signals on line 262 and provides signals to the I/O processor 190-1 used for diagnostics. The coarse frequency steering signals on line 264 are provided as signal V.sub.φDET to the phase lock circuit 252 as explained below. 
     Signal V.sub.φDET varies within the voltage range of 0 to 12 volts. When V.sub.φDET is at 0 volts then frequency F LO1  (the frequency of operation of the first local oscillator 122) is at its lowest, and when V.sub.φDET is 12 volts then frequency F LO1  is at its highest. 
     The phase lock control circuit 252 includes loop gain compensation control circuitry 272 which receives the V.sub.φDET Signal. The circuitry 272 also receives a biasing voltage to provide nominal gain even when V.sub.φDET is at 0 volts. The loop gain compensation control circuitry 272 provides on line 276 signals which maintain a constant loop gain in the phase lock circuit 252 despite variations in the divisor M discussed below. 
     A phase detector 278 receives the signals provided on line 276 by the loop gain compensation control circuitry 272. It also receives 0.5 Mhz signals provided on line 282 by divide-by-twenty circuit 284. The divide-by-twenty circuit 284 receives the 10 Mhz crystal controlled reference signal on line 285. Finally, the phase detector 278 receives on line 286 signals provided by divide-by-M circuitry 288. 
     The divide-by-M circuitry 288 receives on line 290 signals at the precise frequency of operation of the first local oscillator 122. These signals are provided by the first local oscillator 122, via driver 306. The value of M ranges from 714.0 to 1073.0. The resolution of the divide-by-M circuitry 288 is 50 Khz. In order to achieve this resolution with minimal lock-up time, the circuitry 288 includes a fractional frequency divider described in co-pending commonly assigned Patent Application, entitled DUAL MODULUS FRACTIONAL DIVIDER, invented by Donald A. Bradley, Ser. No. 175,759, filed on the same date as this Application which is hereby incorporated herein in its entirety by this reference. Lock-up time, of course, is the time necessary to achieve phase lock. It will be appreciated that (714.0)×(0.5 Mhz)=357 Mhz, and that (1073.0)×(0.5 Mhz)=536.5 Mhz. 
     The phase detector 278 provides phase correction signals to loop amplifier 292. The loop amplifier, in turn, provides these signals, to a 50 Khz notch filter 294 used to remove sidebands caused by fractional M division. The filtered signals are provided to line 262. 
     The phase detector 278 provides to lock detector 296 signals which indicate when phase lock has been achieved. The lock detector 296 provides signals to the I/O processor 190-1 to indicate when phase lock has been achieved and data measurements can be taken. The signals provided on line 290 to the divide-by-M circuitry 288 also are provided to a level detector 298. The level detector indicates to the I/O processor whether there is sufficient signal amplitude provided on line 290 to accurately achieve locking. 
     The first local oscillator 122 provides, via driver 168, signals on line 302 at a prescribed frequency in the range 357-536.5 Mhz. These signals are provided to the harmonic generator 170. Switch 304 is closed in the harmonic mode and is opened in the direct mode. Thus, in the direct mode, the first local oscillator 122 provides no signals to the harmonic generator 170. 
     FIG. 8 illustrates details of the harmonic generator 170. Signals are provided by the first local oscillator 122 on line 302. These signals are amplified by drivers 306 and are provided, via inductor 308, to the step recovery diode (SRD) 310. The output of the harmonic generator 170 comprises sampling signals provided at the sampling rate to the harmonic mixers 174 as explained above with reference FIG. 5. The operation of a harmonic generator such as that illustrated in FIG. 8 is well-known to those skilled in the art and need not be described in detail herein. 
     FIG. 9 illustrates details of the second local oscillator 124. FIG. 10 illustrates details of the phase lock circuit 314 used to control the frequency of operation of the second local oscillator 124. 
     The second local oscillator 124 includes a voltage tuned oscillator 316 having a range of operation of 98-272.5 Mhz. The output frequency of oscillator 316 is provided both to line 318 and to a first divide-by-two circuit 319. The output frequency of the first divide-by-two circuit 319 is provided both to line 320 and to a second divide-by-two circuit 322. The output frequency of the second divide-by-two circuit 322 is provided both to line 324 and to a third divide-by-two circuit 326. The output frequency of the third divide-by-two circuit is provided to line 328. Switch 330 selects among the signals provided on lines 318, 320, 324, and 328 and provides the selected signal, via respective drivers 332 to the respective double balance mixers 178. 
     The second local oscillator 124 includes a tuning voltage summation circuit (TVS-2) 334 which receives on line 336 coarse frequency steering signals from eight bit digital-to-analog converter 338. The coarse steering signals provided by the digital-to-analog converter 338 are controlled by signals provided by the I/O processor 190-1, via I/O bus 267. 
     TVS-2 334 receives on line 340 fine frequency steering signals from loop gain control circuitry 342. These signals compensate for variations in loop gain caused by the &#34;÷L&#34; components when the source lock 154 provides control signals to the second local oscillator 124 instead of to the signal source 104. The loop gain control circuitry 342 receives on line 344 phase correction signals provided by the phase lock circuit 314 as explained below. 
     The TVS-2 provides voltage tuning signals on line 346 to a linearizer 348 which provides a linear frequency versus voltage tuning signal to the voltage tuned oscillator 316. Furthermore, window compare circuitry 350 receives the fine frequency steering signal on line 344 and provides diagnostic signals to the I/O processor 190-1. Finally, the coarse frequency steering signals on line 336 are provided as signal v.sub.φDET to the phase lock circuit 314 as explained below. 
     The phase lock circuit 314 includes loop gain compensation control circuitry 352 which receives the v.sub.φDET signal. The circuitry 352 also receives a biasing voltage used to provide nominal gain even when V.sub.φDET is at 0 volts. The loop gain compensation control circuitry 352 provides on line 356 signals which maintain a constant loop gain in the phase lock control circuit 314 despite variations in the divisor K discussed below. 
     A phase detector 358 receives the signals provided on line 356 by loop gain compensation control circuitry 352. It also receives 0.5 Mhz signals provided on line 360 by divide-by-twenty circuit 362. The divide-by-twenty circuit 362 receives the 10 Mhz crystal controlled reference signals on line 364. Finally, the phase detector 358 receives on line 366 signals provided by divide-by-K circuitry 368. 
     The divide-by-K circuitry 368 receives on line 370 signals provided on line 370 by the voltage tuned oscillator 316 via driver 372. The value of K ranges from the 196.0-544.5. The resolution of the divide-by-K circuitry 368 is 50 Khz. In order to achieve this resolution and to achieve rapid phase-lock of the second local oscillator 124, a fractional frequency divider described in the above-referenced co-pending application is included in the divide-by-K circuitry 368. 
     The phase detector 358 provides phase correction signals to loop amplifier 374. The loop amplifier, in turn, provides the signals, via 50 Khz notch filter 376, to the loop gain control circuitry 342 of the second local oscillator 124. The 50 Khz notch filter 376 removes sidebands caused by fractional K division. 
     The phase detector 358 provides to lock detector 373 signals which indicate when phase lock has been achieved. The lock detector 373 provides signals to the I/O processor 190-1 indicating when the second local oscillator 124 is in phase lock and test measurements can be taken. The signals provided on line 370 are provided to a level detector 371. The level detector 371 provides diagnostic signals to the I/O processor 190-1. 
     Switch 377 alternatively couples line 344 to the phase lock circuit 314 or to TVS-4 502 discussed below in relation to FIG. 14. Switch 377 couples line 344 to TVS-4 when the source lock 154 is to be coupled to the second local oscillator 124 instead of to the signal source 104. 
     This alternative coupling, for example, is used in the synthesizer mode when a synthesized signal source (not shown) is incorporated in the system 100. The phase lock of the IF 2  to the 2.25 Mhz synthesized reference frequency reduces measurement noise regardless of whether the signal source 104 or the second local oscillator 124 is controlled. 
     It will be appreciated that an important advantage of the system 100 is that it permits the use of an unsynthesized signal source 104. This particular advantage is gained by using the source lock 154 to phase lock the signal source 104 to an IF 2  reference signal. Typically, an unsynthesized signal source 104 is significantly less expensive than a synthesized source. Furthermore, the system 100 permits the use of an unsynthesized signal source 104 without suffering a degradation in stimulus signal frequency accuracy. 
     FIG. 11 illustrates details of image reject mixer 183 and its associated variable driver assembly 177. It will be appreciated that the image reject mixers indicated by reference numerals 179 and 185 are substantially identical to that illustrated in FIG. 11. 
     The image reject mixer 183 is operable in either a test mode or a calibration mode. In the test mode, downconverted versions of the test signal t 1  at the IF 2  of 2.25 Mhz are received by driver 176 and are provided to 2 Mhz bandpass filter 382 centered at 2.25 Mhz. The filtered IF 2  signal is split in order to allow for quadrature phasing at IF 2  of the split signals as follows. One split signal is provided to a -45° phase shifter 384, and the other split signal is provided to +45° phase shifter 386. The respective phase shifted signals are provided to respective double balanced mixers 388 and 390 where they are mixed with the 2.33 Mhz signal provided on line 392 by the third local oscillator 128. The mixed signal output by mixer 388 is provided to +45° phase shifter 389, and the mixed signal output by mixer 390 is provided to -45° phase shifter 391. The resulting signals are provided to summation circuit 394 which, in turn, provides their sum to 10 Khz bandpass filter 396 centered at 83.33 Khz. 
     Thus, in test mode the image reject mixer 183 downconverts the IF 2  signal to a third intermediate frequency (IF 3 ) of 83.33 Khz. The 83.33 Khz signal is provided to a gain ranging amplifier assembly 177 which comprises a series of five individual drivers. These drivers can be set by control signals provided on lines 398 by the I/O processor 190-1 to individually provide gain of either one or four. Consequently, a variable gain is provided to signals output by the double balance mixer 183 for provision to a synchronous detector 130. Peak detector 400 monitors the peak amplitude of the signals output by the gain ranging amplifier assembly 177. Window detector 402 provides to the I/O processor 190-1 signals indicating when the signals on line 399 are within a desired signal level suitable for provision to the synchronous detector 130. 
     In the calibration mode, switches 404 and 406 couple line 392 directly to bandpass filter 396, and the calibration oscillator 134 provides on line 392 calibration signals at 83.33 Khz as explained below. In the calibration mode, the 83.33 Khz signal provided by the calibration oscillator 134 is used to calibrate the gain ranging amplifiers 177 and the synchronous detectors 130. 
     Calibration occurs approximately once every two and one-half minutes. During calibration, the ADC 150 is calibrated using a precision DC reference (not shown). The gain and phase shifts experienced by a signal propagated through the gain ranging amplifier assembly 177 due to gain settings of one and four are determined and stored in memory. Also, circularity errors in the synchronous detectors 130 are identified and stored in memory. The stored errors are used later to correct measurement data. 
     FIG. 12 illustrates details of the calibration oscillator 134 and the third local oscillator 128. The third local oscillator 128 receives the 10 Mhz crystal control reference signal on line 404. The divide-by-fifteen circuit 406 downconverts the 10 Mhz signal to the 666 Khz. The downconverted 666 Khz signal is provided to phase detector 408. Divide-by-fourteen circuit 410 receives an input from 9.33 Mhz voltage tuned oscillator (VTO) 412 and provides an output signal at approximately 666 Khz to the phase detector 408. The phase detector 408 provides an output to loop amplifier 414 which, in turn, provides an amplified voltage tuning signal to the 9.33 Mhz VTO 412. The phase detector also provides a signal to lock detector 416 which, in turn, indicates to the I/O processor 190-1 when the third local oscillator 128 is phase locked and measurements can be taken. 
     The 9.33 Mhz VTO 412 provides its output signal both to the divide-by-fourteen circuit 410 and to a divide-by-four circuit 418. The output of the divide-by-four circuit 418 is provided to a 3 Mhz low-pass filter 420. Thus, at terminal 422 a 2.33 Mhz signal is provided by the third local oscillator 128. 
     A byte provided on bus 269 to divide-by-one hundred and twenty circuit 271 indicates the phase of the 83.33 Khz calibration signal relative to the IF synchronization signal provided on line 541 by the IF synchronization circuitry 520. A sine look-up table 275 converts the signals provided by the circuit 271 to a series of pulses which, in turn, are converted by digital-to-analog converter 277 to an 83.33 Khz sine wave. Filter 279 provides filtering to the 83.33 Khz calibration signal on terminal 424. 
     Switch 426 can be used to select between terminals 422 and 424. When terminal 422 is selected, the 2.33 Mhz signal provided by the third local oscillator 128 is provided on line 392 and when terminal 424 is selected, the 83.33 Khz signal provided by the calibration oscillator 134 is provided on line 392. 
     FIG. 13 illustrates details of a respective fourth local oscillator 146 and the details of one of the three synchronous detectors 130 and two of the six sample/hold circuits 148-1. In operation, divide-by-one hundred and twenty circuit 428 receives the 10 Mhz crystal controlled reference signals on line 430. The circuit 428 receives on bus 269 a control byte from the I/O processor 190-1 which indicates the phase of the output signal of the oscillator relative to the IF synchronization signal. The circuit 428 also receives the IF synchronization signal. The circuit 428 provides on lines 432 a digital representation of a 83.33 Khz signal. 
     The synchronous detector 130 includes a respective sine look-up table 436 and a respective cosine look-up table 438 which respectively receive the digital signal representation provided by the divide-by-one hundred and twenty circuit 428. The respective sine and cosine look-up tables 436 and 438 respectively provide digital sine and cosine representations of the 83.33 Khz signal to respective digital-to-analog converters and eight bit multiplier circuits 440 and 442. 
     Circuit 440 converts the signals provided by sine look-up table 436 from digital to analog. Circuit 442 converts the signals provided to it by cosine look-up table 438 from digital to analog. Circuit 440 multiplies the analog versions of the signals provided by the sine look-up table 436 with 83.33 Khz signals provided on line 399. Circuit 442 multiplies the analog versions of the signals provided by the cosine look-up table 438 with the 83.33 Khz signals provided on line 399. 
     The output of circuit 440 is provided to the 100 Khz low-pass filter 444. The output of filter 444 represents the &#34;Real&#34; (or in-phase) component of the signal on line 399. 
     The output of circuit 442 is provided to 100 Khz low-pass filter 446. The output of filter 446 represents the &#34;Imaginary&#34; (or quadrature) component of the signal on line 399. 
     The respective outputs of filters 444 and 446 are provided, via respective drivers 448, to sample-and-hold circuits 148-1. The I/O processor 190-1 controls the operation of the sample-and-hold circuits 148-1 using control signals provided on lines 450. Sampling of the Real and Imaginary components by all three channels occurs simultaneously. The simultaneous sampling advantageously preserves the correlated noise on the channels which is normalized out during S-parameter display. 
     The respective outputs of the filters 444 and 446 can undergo additional filtering to remove noise using either respective 1 Khz filters 452 or using respective 100 hz filters 454. Alternatively, the outputs of respective filters 444 and 446 can be provided directly to respective drivers 448 without further filtering. The selection of the respective filters 452 and 454 is provided by the front panel 552. 
     Filters 444, 446 and 452, 454 advantageously filter out noise in the downconverted signals outside the filter bandwidths without the need for involvement of the processors 190-1, 190-2 or 190-3. More specifically, the filters can be used to filter out noise without the need to take a relatively large number of stimulus signal measurements and then average the results, using the processors, in order to reduce noise. Thus, the processor is free to undertake other tasks, and in fact, the settling time of filters 444, 446 and 452, 454 is short enough such that accurate measurements often can be taken more rapidly using such filtering than if the processors were involved in calculating average measurement values in order to remove noise. Optionally, of course, the processors (instead of the filters) can be used to remove such noise or the two can be used together for even greater noise reduction. 
     In the preferred embodiment, the use of the 1 Khz filters 452 results in a sampling time of approximately 2.1 msec for each discrete stimulus signal frequency, and use of filters 454 results in a sampling time of approximately 11.1 msec. Whereas, use of neither filters 452 nor 454 results in a sampling time of approximately 1.1 msec. The additional sampling time incurred by using either filters 452 or 454 is due to an increased settling time for narrower bandwidth filters. 
     It will be appreciated that the discussion with respect to FIG. 13 applies to each and every one of the synchronous detectors 130. 
     In operation, for each stimulus signal frequency, the Real and Imaginary signal levels are adjusted prior to their actual measurement so as to obtain maximum resolution from the ADC 150. For each stimulus signal frequency tested and for each channel, the I/O microprocessor 190-1 first determines that the first and second local oscillators 122 and 124 and the signal source 104 are phase locked. Then it increases or decreases the gain provided by the gain ranging amplifier assemblies 177 as needed to maintain the downconverted signals on each of the three respective channels within the desired limits of the window detectors 402. For each of the three channels, once proper signal magnitude has been achieved by increasing or decreasing the gain, the I/O processor 190-1 measures the Real and Imaginary outputs of the channel&#39;s synchronous detector 130 and increases or decreases the phase byte provided on bus 269 so as to make the Real and the Imaginary signals equal and positive. This assures that the signal magnitude will be maximized for both the Real and the Imaginary signals. The adjustments just described minimize quantization error in Real and Imaginary signals on each of the three channels converted by the ADC 150. 
     FIG. 14 illustrates details of the source lock circuitry 154. Divide-by-ten circuit 456 receives the 10 Mhz reference signals. Circuit 456, in turn, provides 1 Mhz signals to phase detector 458. Divide-by-nine circuit 460 receives signals from 9 Mhz voltage tuned oscillator (VT0) 462, and provides 1 Mhz signals to the phase detector 458. The phase detector 458 provides a correction signal to a loop amplifier 464 which, in turn, provides an amplified tuning voltage signal to the 9 Mhz VTO. The phase detector 458 also provides a signal to lock detector 466 which provides a lock detection signal to the I/O processor 190-1. 
     The 9 Mhz VTO 462 provides to divide-by-four and phase shifting circuit 468 its output at the 9 Mhz frequency. Circuit 468 provides on line 470 a 2.25 Mhz signal with 0° phase shift and provides on line 472 a 2.25 Mhz signal with a 90° phase shift. The signal on line 470 is provided to double balanced mixer 474 which mixes it with an IF 2  reference signal. Similarly, the signal on line 472 is provided to double balanced mixer 476 which mixes it with an IF 2  reference signal. 
     The output of mixer 474 is used to generate the correction signal provided on line 160 to the signal source 104. The output of mixer 476 is used to detect when the IF 2  reference signal is in phase lock with the signal provided on line 472. 
     The output of mixer 476 is provided to 3 Mhz low-pass filter 478. The output of filter 478 is provided to inverter 480 and to line 481. Switch 482 is preset by the I/O processor 190-1 to couple a positive polarity signal to the positive terminal of comparator 484 upon achieving phase lock with IF 2 . 
     Table 1 indicates the position of switch 482 over various frequency ranges. 
     
                       TABLE 1                                                     
______________________________________                                    
Terminal          Frequency Range                                         
______________________________________                                    
(+)               110-270 Mhz                                             
(-)               270.1-447 Mhz                                           
(+)               447.1-626 Mhz                                           
(-)               0.6261-40 Ghz                                           
______________________________________                                    
 
    
     Upon the achievement of phase-lock with IF 2  the comparator 484 provides a signal on line 486 causing switches 488 and 490 to open. Thus, upon achieving phase lock, the first and second integrators 492 and 494 are incorporated into the feedback path of the correction signal provided on line 160. The signal source 104 in conjunction with the first and second integrators 492 and 494 form a third order loop. Optionally, when the second local oscillator 124 is in the feedback loop, it together with the first and second integrators 492 and 494 forms a third order loop. The use of a third order loop significantly reduces noise in the system 100. 
     Additionally, upon the achievement of phase lock with IF 2  the signal provided by comparator 484 disables a search oscillator 496. Before phase-lock is achieved, the search oscillator 496 sweeps in frequency from 3 Mhz above to 3 Mhz below the desired stimulus signal frequency to be generated by the signal source 104. The search oscillator provides its output frequency sweep signal to tuning voltage summation circuitry (TVS-3) 498. 
     TVS-3 circuit 498 also receives from the 8-bit digital-to-analog converter (DAC) 500 coarse tuning signals. The DAC 500 is under control of signals provided on bus 269. TVS-3 498 provides correction signals on line 160 to the signal source 104. 
     Additionally, the DAC 500 provides a signal to tuning voltage summation circuitry (TVS-4) 502 which can provide a control signal to the phase lock circuit 314, via line 379, when the second local oscillator is phase-locked to IF 2 . 
     In order to decrease heat dissipation requirements, weight and volume of the power supply of the system 100, a switching power supply 540 is used. It is well known that switching power supplies typically generate electromagnetic interference (EMI) which potentially can interfere with measurement signals (t 1 , t 2 , r 1 , and r 2 ) if the EMI falls within the passband of the measurement signals. 
     In the system 100 of the present invention, the frequency of operation of the switching power supply 540 is injection locked so as to strategically place sidebands of the EMI produced by the switching power supply 540 in frequency bands (approximately 55 Khz and 111 Khz) equally spaced above and below the highly sensitive 83.33 Khz (IF 3 ) measurement signals. Injection locking ensures long term frequency accuracy of the EMI so as to avoid drift into the IF 3  passband. 
     Details of the switching power supply 540 and injection locking as accomplished in the presently preferred embodiment are provided in commonly assigned co-pending Patent Application, entitled SWITCHING POWER SUPPLY WITH AN INJECTION SIGNAL FREQUENCY LOCKING CIRCUIT, invented by Donald A. Bradley Ser. No. 176,095, now U.S. Pat. No. 4,858,097 filed on the same date as this Application and which is hereby incorporated herein in its entirety by this reference. 
     FIGS. 19-20 provide schematic diagrams illustrating details of the test set I/O circuit 534. Referring to FIG. 20, the I/O circuit 534 includes five overcurrent protection circuits 554. Each respective overprotection circuit 554 is coupled to respective indicator circuitry 556 which indicates when an overcurrent condition has occurred. Each respective indicator circuit 556 is coupled to respective circuitry 558 which communicates with the I/O processor 190-1. 
     Details of the overcurrent protection circuits 554 and their operation are provided in commonly assigned co-pending Patent Application, entitled OVER-CURRENT PROTECTION CIRCUIT, invented by Donald A. Bradley, Ser. No. 175,957, now U.S. Pat. No. 4,926,288 filed on the same date as this Application and which is hereby incorporated herein in its entirety by this reference. 
     Frequency Solution 
     In operation, the network analyzer system of the present invention can provide scattering parameters for a DUT 102 over a range of discrete frequencies. The number of stimulus signal frequencies for which phase and magnitude information can be measured and the frequency range of interest can be selected by the user. In the preferred embodiment, the maximum number of stimulus signal frequency points is 512, and the minimum number is 74. The number of frequency data points can be chosen by the user so as to optimize phase and magnitude resolution versus speed of operation. It will be appreciated that the more frequency points measured, the longer it will take for the system to provide to the user phase and magnitude information for all of these points. 
     The measurement system advantageously permits a user to select discrete frequency points within a desired range of frequencies. Alternatively, the user may designate a range of frequencies of interest, and the measurement system will automatically select discrete frequency points within that range for testing of the DUT 102. 
     The ability of the user to select discrete frequency points is particularly advantageous where the DUT 102 is a device such as a filter in which more frequency points are desired in the skirts than in the pass band of the filter. FIG. 15 illustrates the selection of unevenly spaced discrete frequency points within a range of frequencies of interest bounded by a starting frequency (F Start ) and a stopping frequency (F Stop ). FIG. 16 illustrates a range of frequencies of interest which are evenly spaced within the range of frequencies of interest. In the preferred embodiment, the frequency resolution provided by the signal source 104 is 50 Khz. 
     The measurement system uses an algorithm to select the frequency of operation of the respective first and second local oscillators 122 and 124 for each frequency point selected by the user. The system then stores these oscillator frequencies for each frequency point, and during actual measurement of a DUT 102, for each discrete test frequency, the stored frequencies of operation of the respective first and second local oscillators 122 and 124 are recalled and are used to control the frequency of operation of the first and second local oscillators 122 and 124. 
     The algorithm advantageously seeks to use a relatively low harmonic number to generate the 89.4 Mhz first intermediate frequency (IF 1 ). In practice, this means that it is desirable to use a relatively high frequency of operation for the first local oscillator 122 for each discrete stimulus frequency point. Furthermore, as mentioned above, it is desirable to use a first intermediate frequency of approximately 89.4 Mhz in order to provide continuous frequency coverage in the harmonic mode. The relationship between IF 1  and the frequency of operation of the first local oscillator (F LO1 ) , the harmonic number and the stimulus signal frequency (F ST ) is expressed by the following equation: 
     
         IF.sub.1 =/H(F.sub.LO1)-F.sub.ST /                         (5) 
    
     The algorithm is explained as follows: 
     
         F=(F.sub.ST) (20)                                          (6) 
    
     
         H=1+INT((F+1788)/10730)                                    (7) 
    
     
         Z=5365/(3+H/1250)                                          (8) 
    
     
         F.sub.LO1 =(1+INT((F+Z)/H)/20                              (9) 
    
     
         F.sub.LO2 =(H) (F.sub.LO1)-F/20+2.25                       (10) 
    
     It will be noted that the number 20 is used to normalize the 50 Khz resolution to 1 Mhz in order to simplify the algorithm. 
     The following illustrates use of the above algorithm to determine the respective frequencies of operation of the first and second local oscillators 122 and 124 for a stimulus signal frequency of 10,000.1 Mhz. The calculation is as follows: 
     
         F=(20) (10,000.1)=200,002                                  (11) 
    
     
         H=1+INT(18.81)=19                                          (12) 
    
     
         Z=5365/(3+19)/1250)=1779.3                                 (13) 
    
     
         F.sub.LO1 =(1+INT(200,002+1779.3)/19)/20=531.05            (14) 
    
     
         F.sub.LO2 =19,531.05-200,002/20+2.25=92.1                  (15) 
    
     Thus for a stimulus frequency of 10,000.1 Mhz, the first local oscillator 122 operates on an F LO1  of 531.05 Mhz, and the second local oscillator 124 operates at a frequency of F LO2  -92.1 Mhz. The precise frequencies of operation, however, may be varied in accordance with the spur search and avoidance algorithm which is discussed next. 
     After the determination of the individual values for F LO1  and F LO2  for each of the discrete stimulus signal frequency points selected by a user, the measurement system employs an algorithm to detect spurious signals (spurs) likely to be generated during measurements of a DUT 102. The system uses the algorithm for each discrete selected stimulus signal frequency point for which spurious signals are found, to vary the values of F LO1  and F LO2  so as to remove the spurious signals. The result is a significant reduction of spur-related noise in the downconverted reference and test signals. The values for F LO1  and F LO2  as modified to avoid spurs, then are stored by the system for use during measurement of a DUT 102. 
     Spurious signals can result, for example, from the fact that both harmonics of a discrete stimulus signal frequency F ST  and harmonics of the first local oscillator signal frequency F LO1  exist due to nonlinearities in the harmonic mixers 174. These harmonics may have a mix product in the noise bandwidth of the system 100. 
     In addition, small portions of the original F ST  and F LO1  signals can pass through the harmonic mixers 174 with the desired IF 1  signal. These small portions can mix with harmonics of the signals produced by the respective second and third local oscillators 124 and 128 so as to provoke still further spurious signals. 
     In order to allow maximum dynamic range of the measurement system, it is desirable to eliminate these spurs. The elimination of these spurs using a spur detection and avoidance algorithm is first explained for a particular stimulus signal frequency point at F ST  =3140.90 Mhz. Then the algorithm is explained in general using the flow diagram of FIG. 18. 
     Using the algorithm discussed above for selecting F LO1  and F LO2  the values in Table 2 are selected for F ST  =3140.90 Mhz. 
     
                       TABLE 2                                                     
______________________________________                                    
          F.sub.ST = 3140.90 Mhz                                          
          F.sub.LO1 = 461.50 Mhz                                          
          H = 7                                                           
          IF.sub.1 = 89.6 Mhz                                             
          F.sub.LO2 = 91.85 Mhz                                           
          IF.sub.2 = 2.25 Mhz                                             
          F.sub.LO3 = 2.33 Mhz                                            
          IF.sub.3 = 83.33 Khz                                            
______________________________________                                    
 
    
     In this example, the seventh harmonic of F LO1  (461.50 Mhz×7=3230.5 Mhz) is mixed with the F ST  signal to produce IF 1  (89.6 Mhz). IF 1  is mixed with F LO2  to produce the IF 2  of 2.25 Mhz. The equations are as follows: 
     
         IF.sub.1 =H*F.sub.LO1 -F.sub.ST                            (17) 
    
     
         IF.sub.2 =F.sub.LO2 -IF.sub.1                              (18) 
    
     Combining: 
     
         IF.sub.2 =F.sub.LO2 -(H*F.sub.LO1 -F.sub.ST)               (19) 
    
     Putting in actual values from Table 2: 
     
         2.25=91.85-(7*461.5-3140.9)                                (20) 
    
     Note that the fifth harmonic of F LO2  will mix with the frequency of the F LO1  producing a spurious IF 3  which does not result from an R.F. input: 
     
         461.50-(5×91.85)=2.25                                (21) 
    
     This spur will produce a nonzero IF 3  output without any R.F. input. In fact, the possibility exists that several different harmonics of F LO1  and F LO2  may create spurs. In general, a spur is present if the following conditions are satisfied: 
     
         |(M*F.sub.LO1 -N*F.sub.LO2)|&lt;IF.sub.3 -PB.sub.IF.sbsb.3 /2 and                                                    (22) 
    
     
         |(M*F.sub.LO1 -N*F.sub.LO2)|&gt;IF.sub.3 +PB.sub.IF.sbsb.3 /2(23) 
    
     M and N are small integers whose range depends upon the passband of the system, and PB IF .sbsb.3 is the passband of the bandpass filter 396, e.g., 10 Khz (see FIG. 11). 
     This relationship is illustrated in FIG. 17 in which M=1. The absolute value of the first terms of the inequalities is reflected in a range above and below the F LO1 . In this example, only the fundamental (first harmonic, M=1) of F LO1  is capable of causing a spur; M is limited to the value of 1. If the value of F LO2  is approximately 90 Mhz, for example, and the value of F LO1  can range from 357 Mhz to 536.5 Mhz, then the meaningful values for N would be 4 through 6. So all of the combinations of M=1 and N=4 through N=6 must be tested in inequalities 22 and 23. 
     In order to remove this spur, the values of M and F LO2  can be changed so as to maintain the same resulting IF 3  while avoiding creation of a spur. This can be explained as follows. By rearranging equation (19), we have: 
     
         F.sub.st =H*F.sub.LO1 -F.sub.LO2 +IF.sub.2                 (24) 
    
     Thus, the values of F LO1  and F LO2  can be varied to maintain this equation, while avoiding the conditions of inequalities (22) and (23). In the case of the preferred embodiment, the first and second local oscillators 122 and 124 are programmable in 50 Khz steps; so their frequencies must be changed by multiples of this value. If the value of F LO1  is increased by 1 Mhz/H, or 150 Khz (H=7) at 50 Khz resolution, then the value of F LO2  must be increased by H*150 Khz or 1.05 Mhz at 50 Khz resolution. Substituting into equation (19): 
     
         2.25=92.9-(7*461.65-3140.9)                                (25) 
    
     The new values, corrected for spurs, for F ST  =3140.9 Mhz will be as follows: 
     
                       TABLE 3                                                     
______________________________________                                    
          F.sub.ST = 3140.90 Mhz                                          
          F.sub.LO1 = 461.65 Mhz                                          
          IF.sub.1 = 90.65 Mhz                                            
          F.sub.LO2 = 92.9 Mhz                                            
          IF.sub.2 = 2.25 Mhz                                             
          F.sub.LO3 = 2.33 Mhz                                            
          IF.sub.3 = 83.33 Khz                                            
______________________________________                                    
 
    
     As can clearly be seen from Table 3, the new frequencies F LO1  and F LO2  have no effect upon the final results. Note that IF 1  is slightly higher, but the fairly broad band I.F. filter can be used to pass this frequency as well. FIG. 5 and 5B illustrate filtering of IF 1 . 
     The effect on the spur of the new values in Table 3 is that the fifth harmonic of F LO2  and the harmonic of F LO1  results in a frequency of 2.85 Mhz, outside the passband of filter 396. 
     The algorithm illustrated in FIG. 18 interactively performs a check for combinations of harmonics of F LO1  and F LO2  that cause a spur, and adjusts the frequencies of the first and second local oscillators 122 and 124 to remove the spurs. The algorithm is employed to correct for spurs for each discrete stimulus signal frequency point selected by the user. It will be noted that the frequency range of the first local oscillator 122 is approximately 357-536.5 Mhz, and the frequency range of the second local oscillator 124 is approximately 91.65 ±1.875 MHz (harmonic mode). 
     FIG. 21 provides a timing diagram illustrating the processor pipeline timing of the system 100 of the presently preferred embodiment for three sequential stimulus signal frequencies: F N-1 , F N  and F N+1 . The operation of the processor pipeline is explained in relation to stimulus signal frequency F N . 
     During time periods T 1  -T 7 , the I/O processor 190-1 measures the Real and Imaginary signals provided by the synchronous detectors 130. Table 4 explains details of the measurement process for each time period. 
     
                       TABLE 4                                                     
______________________________________                                    
Time           Measurement Activity                                       
______________________________________                                    
T.sub.1        Gain and Phase Ranging                                     
T.sub.2        Delay Due to Settling                                      
               Time of Filters 396, 452,                                  
               454                                                        
T.sub.3        Initialization of ADC 150                                  
T.sub.4        Sample and Hold                                            
T.sub.5        Analog-to-Digital                                          
               Conversion and correction                                  
               for errors in ADC 150                                      
T.sub.6        Send Signal Source 104 to                                  
               F.sub.N+1                                                  
T.sub.7        Format Measurement Data                                    
               for F.sub.N in Floating Point                              
               Format for Transfer to                                     
               Main Processor 190-2A on                                   
               FIFO 528                                                   
______________________________________                                    
 
    
     During time periods T 8  -T 10 , the first main processor 190-2A and the first math co-processor 190-2B correct the measured data for stimulus frequency F N  to remove system-induced errors. Table 5 explains details of the correction process for each time period. 
     
                       TABLE 5                                                     
______________________________________                                    
Time          Measurement Activity                                        
______________________________________                                    
T.sub.8       Remove the Effects of                                       
              Gain and Phase Ranging                                      
T.sub.9       12-Term Error Correction                                    
              (or Subset of 12-Term                                       
              Error Correction)                                           
.sub. T.sub.10                                                            
              Send Corrected                                              
              Measurement Data for F.sub.N to                             
              the Second Main Processor                                   
              190-3A                                                      
______________________________________                                    
 
    
     During time periods T 11  -T 12 , the second main processor 190-3A and the second math co-processor 190-3B display the measurement data for F N . Table 6 explains details of the display process for time periods T 11  -T 12 . 
     
                       TABLE 6                                                     
______________________________________                                    
Time          Measurement Activity                                        
______________________________________                                    
T.sub.11      Convert the Measurement                                     
              Data for F.sub.N to a                                       
              Coordinate Graph Type                                       
              (Dependent Upon Graph                                       
              Type Selected by User)                                      
T.sub.12      Display coordinate Graph                                    
              of Measurement Data on RGB                                  
              Monitor 548                                                 
______________________________________                                    
 
    
     The following equations represent the uncorrected S-parameters. The processors 190-1, 190-2A-190-2B and 190-3A-190-3B cooperate to calculate these parameters in accordance with these equations to correct for system errors and to display the S-parameters as described above. ##EQU1## 
     Details of Ports I and II are described as follows. The connector system illustrated in the drawings is particularly adapted for carrying signals above 20 Ghz and even as high as 46 Ghz, but is not, of course, limited in its utility to such high frequency operation. But at the higher frequencies, challenges exist because of the small size of the microwave components with which the connectors must interface. Further, the coaxial connectors themselves are necessarily very small in order to inhibit unwanted higher order modes when carrying such high frequency signals. The smaller connectors are thus generally more fragile and susceptible to damage. Further, their manufacture within the mechanical tolerances required for satisfactory electrical operation places additional constraints on the configuration and the methods of manufacture. 
     The connector system illustrated in the drawings satisfies these requirements. A microwave component 711 carries a microwave device 713 within a housing 715. The device 713 can be, for example, a microstrip, coplanar waveguide, or suspended substrate. A launcher 717 is provided as the first external element to the microwave device 711 for providing a coaxial connector for the device. The launcher 717 is threaded into the housing 15. A third piece of the connector system illustrated in the drawings is a connector 719 which terminates one end of a coaxial cable 721. The illustrated connector system electrically connects a signal from the microwave device 713 to the outside world through a center conductor 723 of the coaxial cable 721. 
     Details of each of the three primary components of the connector system will now be described. The first element shown in FIGS. 23 and 24 in connecting the microwave device 713 to the outside world is the connection of a conductor 725 thereto by a quantity of solder 727. The conductor 725 is held within a cylindrically shaped glass bead 729 having an outer cylindrical metal layer 731. The glass bead assembly 729 is held in the metal housing 715 by a quantity of solder 733 that is allowed to flow around the glass bead, thereby providing a secure hermetic seal and electrical connection. The opening in the housing wall is stepped so that a smaller opening portion 737 is provided for passage of the conductor 725 without contacting it. This stepped structure is designed to maintain a uniform impedance along the signal path. 
     In order to compensate for fringing capacitances that result from the stepped structure, an interface compensation gap 739 is provided between an end of the microwave device 713 and the vertical wall of the housing 715 that carries the bead 729. Additionally, the length of the conductor 725 from that wall to its end in contact with the microwave device 713 is carefully controlled. Further, another compensation gap 741 between the housing wall and the glass bead 729 is provided. The gaps 739 and 741 introduce a series inductance into the signal path that compensates for undesired fringing capacitances resulting from the diameter steps of the system. 
     A miniscus 743 surrounds the center conductor 725 on one side of the glass bead 729, as a result of the manufacturing process. In order to form a clean wall at the end of a cavity 745 in the housing 715, the bead 729 is oriented so that the miniscus 743 is positioned toward the microwave device 713, as best shown in FIG. 24. Prior connectors, because of the equipment used in installing the conductor 725 in the glass bead 729, place the miniscus on the side of the glass bead of the longest extending length of its inner conductor. The geometry of the fixtures holding the conductor in the glass bead while the glass was heated to melt it for insertion of the conductor resulted in this occurring. The present structure, however, is a result of assembling the conductor 725 through the glass bead 729 in a manner that the miniscus is always on the short pin side of the resulting structure. The result is to eliminate the miniscus from possible interference with fingers forming a female connector 747 at an end of a conductor pin 749 within the launcher 717. Damage to these fingers is avoided, and the fingers may be pushed onto the conductor 725 much closer to the bead 729. These are desirable features in a small, high frequency connector system. 
     As best illustrated in FIG. 25, the launcher 717 includes an outer, cylindrically shaped threaded housing 751. Alternative methods of attaching a launcher to the component 711 are certainly possible, such as by forming a flange on the launcher that is attached to a flat outer surface of the component, the threaded attachment illustrated herein being exemplary. At one end of the housing, a cylindrically shaped insert 753 snugly and precisely fits. A ledge 755 is formed internal of the housing 751 by a change of diameter of its inner opening between the portion in which the insert 753 exists and a smaller diameter portion 757. At another end of the housing 751 is yet another enlarged diameter opening 759, forming a ledge 761 between it and the interior, smaller diameter opening 757. 
     The conductor pin 749 is held in the center of the opening 757, thereby to form a very short coaxial transmission path having an air dielectric. The pin 749 is so held by a solid dielectric disc 763 having an outer diameter slightly larger than that of the opening 757 but smaller than that of the opening in the housing 751 in which the insert 753 is placed. An opening is provided at one end of the insert 753 for snugly receiving the dielectric disc 763. Therefore, the insert 753 causes the disc 763 to be urged tightly against the ledge 755 of the housing 751 to hold its attached center conductor pin 749 firmly in place. The insert 753 is so held by a wall of the microwave device 711 when the launcher is fully threaded into the opening 45 of the housing 715, as shown in FIG. 23. 
     The structure of the launcher 717 has a significant advantage of allowing for tightly controlled tolerance of the compensation steps in the inner conductor 749 and inner surface of the outer conductor formed by the housing 751 and insert 753. By positioning the support bead 763 in the insert 753, thereby allowing the insert 753 to directly contact the ledge 755 of the housing 751, the tolerances that must necessarily exist in each part do not accumulate as they do in the prior art structures. In prior devices, only the bead 763 contacts the ledge 755, the insert 753 being on an opposite side of the bead 763. It can be seen that this prior structure, therefore, has a wider possible tolerance range since any errors in dimensions of the support bead and the insert can add together to make an even larger dimensional error. However, the structure of the launcher 717 illustrated in the drawings avoids this potential undesirable accumulation of dimensional errors. 
     The structure of the launcher 717 has another important advantage. Should the connectors at the ends of the pin 749 become damaged, the unitary pin 747 and bead 763 structure can be removed and replaced by first removing the housing 751, and then the insert 753. Easy repair is thus possible without disturbing or damaging the microwave device 713, for it is mechanically protected by the glass bead 729. 
     Referring to FIGS. 26, 27 and 28, a preferred technique for attaching the insulating bead with the center conductor 749 is shown. Its position along the axial length of the conductor 749 is controlled by the position of a reduced diameter portion 765 having a length equal to the thickness of the support bead 763. An enlarged diameter portion 767 exists between the bead holding slot 765 and a smaller diameter female receptacle 747. The receptacle 747 includes four fingers extended and dimensioned to grip the pin 25 when pushed over it, thereby to form an electrical connection between them. On an opposite side of the recess 765 is an enlarged portion 769. 
     The bead 763 is initially formed into the desired shape, including a center aperture having a diameter substantially equal to that of the diameter of the slot 765 in the pin 749. Other holes illustrated in the support bead are provided to adjust its effective dielectric constant. The material for the bead 763 should be a thermoplastic having the characteristic of returning to its initial size when cooled after heating and stretching. An example of such a material is PPO brand polyphenylene oxide available from the General Electric Company. Grade EN265 of this material has been found to be satisfactory. The bead 763 is machined from such material to the dimensions desired for the bead when assembled onto the conductor 749. The bead 763 is then heated until softened, its center aperture 771 expanded by mechanical force, and the enlarged opening slid over the portion 767 of the pin 749. The bead 763 is then held over the slot 765 until it has cooled. In the cooling process, it shrinks back to its original shape and the opening 771 of the bead tightly grips the pin portion 765. 
     This method of attaching a support bead to a center conductor is much simpler and provides a better result than techniques currently used. One such prior technique is to glue the bead on a uniform diameter pin, but this does not provide the same degree of mechanical strength. Another technique is to form the pin in two segments which are threaded together at a reduced diameter slot, the pin being separated into its two components, a bead being slid onto the reduced diameter portion, and the pin reassembled. Not only is this complicated and expensive, but such a screw connection can also cause operational problems. Another technique is to injection mold the bead right onto the center pin, but this is an extremely expensive process and very difficult to accomplish for such small parts as are required for the high frequency connectors being described. 
     A larger end 773 of the pin 749 is shown in the drawings to be of the female type, having four extended spring fingers. Adapted for mating with this connector is a male pin 775 of the coaxial connector 719. The result of the connections on the microwave component 711 and the use of the launcher 717 provides a signal path from the very small microwave device 713 to a relatively large coaxial female center connector 773. It will be recognized, of course, that a male pin could be substituted for the female receptacle 773 and a mating female receptacle substituted for the pin 775 of a connector component to be connected to it. 
     The connector component shown herein for attachment to the launcher 717 is the coaxial connector 719, but it will be understood that other standard connection components could be substituted. Referring to FIG. 29, the pin 775 is mechanically held into a metal outer conductor connector 777 by an insulating support bead 779. The center connector 723 of the coaxial cable 721 is soldered to a backside 781 of the pin 775. An outer conductor 783 of the coaxial cable 721 is mechanically and electrically attached to an inner surface of a cylindrical aperture in an end of the holder 777. A coupling nut 785 is also carried by the connector 777 and has internal threads attached to the threads of the launcher 717. 
     An end 787 of the connector piece 777 at its end of the male pin 775 is cylindrically shaped with a diameter to fit snugly inside the circular recess 759 of the launcher 717. When fully inserted, the pin 775 is inserted into the female connector portion 773, as shown in FIGS. 23 and 31. The attaching parts of the launcher 717 and the coaxial cable connector 719 are cooperatively dimensioned so that the pin 775 does not engage the receptacle 773 until after the connector portions are coaxially aligned. That is, the two inner conductor connector portions do not touch each other until the mating cylindrical portions 759 (launcher 717) and 787 (coaxial, cable connector 719) are aligned. This is accomplished by making the length of the pin 775 that extends beyond the end of the bead 779 be less than the distance from the end of the launcher 717 at the surface 759 to the closest point of the receptacle 773. As can be seen from FIG. 30, if the connector 719 is positioned at an angle with respect to the launcher 717, no damage will be done to the receptacle 773 or the pin 775 since they do not contact until the cable connector piece 777 and the launcher 717 are coaxially aligned. That is, so long as the angular relationship illustrated in FIG. 30 persists, no connection can be made. It is only when the two connector portions are coaxially aligned that they can be pushed together for the inner conductors to engage each other as shown in FIG. 31. 
     The short pin has a further advantage of eliminating a radio frequency resonance from the frequency band of the connector. This resonance can occur in existing connectors and is particularly a problem as the frequency increases since a resonance condition can be created with very small cavities or irregularities. As shown in FIG. 32, the short pin 775 is contacted only at the entry end of the receptacle 773 by the ends of the four fingers that form the receptacle 773. These fingers are mechanically desirable, but the slots required to form them can result in a small portion of the radio frequency field entering the coaxial section and undesirably resonating. However, with the short pin of the present system, the resonant frequency occurs above the frequency range for which the connector is designed. 
     Various improvements and modifications can be made to the system 100 of the preferred embodiment without departing from the spirit and scope of the invention. For example, instead of using a source lock in conjunction with an analyzer 108, a source lock can be used in conjunction with various other devices used to extract phase and/or magnitude information from signals. 
     Referring to FIG. 22 there is shown exemplary components of a source locking circuit. Components which directly correspond to elements described above for the preferred embodiment are labelled with primed reference numerals identical to the reference numerals used to identify the corresponding elements of the system 100. 
     It will be appreciated that the measurement device identified by reference numeral 560 can be an analyzer such as that disclosed in the system 100. Alternatively, for example, it can be an amplitude detector, a log magnitude detector, a synchronous detector, an oscilloscope, a modulation meter or some other measurement device. 
     Furthermore, although the source lock components in FIG. 22 include two local oscillators 122&#39; and 124&#39; and two mixers 174&#39; and 178&#39;, it will be appreciated that a source lock could be achieved using only a single local oscillator and only a single mixer. The advantage gained by using source locking still would be achieved: accurate stimulus signal frequency without the need to employ a costly synthesizer. 
     OPERATION OF THE MODEL 360 VECTOR NETWORK ANALYZER 
     General Description 
     The Model 360 Vector Network Analyzer System measures the magnitude and phase characteristics of networks, such as filters, amplifiers, attenuators, and antennas. It does so by comparing the incident signal leaving the analyzer with the signal transmitted through the test device or reflected from its input circuit. FIG. 33 illustrates the types of measurements that the 360 is capable of making. 
     The 360 is a self-contained, fully integrated measurement system that includes an optional time domain capability. 
     Vector Network Analyzer Basics 
     The network analyzer is a tuned receiver (FIG. 34). The microwave signal is down converted into the passband of the IF. To measure the phase of this signal, we must have a reference to compare it with. If the phase of a signal is 90 degrees, it is 90 degrees different from the reference signal. The network analyzer would read this as -90 degrees, since the test signal is delayed by 90 degrees with respect to the reference signal. 
     This phase reference can be obtained by splitting off some of the microwave signal before the measurement (FIG. 35). The phase of the microwave signal after it has passed through the device under test (DUT) is then compared with the reference signal. A network analyzer test set automatically samples the reference signal, so no external hardware is needed. 
     Let us consider for a moment that you remove the DUT and substitute a length of transmission line (FIG. 36). Note that the path length of the test signal is longer than that of the reference signal. Now let us see how this affects our measurement. 
     Assume that we are making a measurement at 1 Ghz and that the difference in path-length between the two signals is exactly 1 wavelength. This means that test signal is lagging the reference signal by 360 degrees (FIG. 37). We cannot really tell the difference between one sine wave maxima and the next (they are all identical), so the network analyzer would measure a phase difference of 0 degrees. 
     Now consider that we make this same measurement at 1.1 Ghz. The frequency is higher by 10 percent so therefore the wavelength is shorter by 10 percent. The test signal path length is now 0.1 wavelength longer than that of the reference signal (FIG. 38). This test signal is 
     
         1.1×360=396 degrees. 
    
     This is 36 degrees different from the phase measurement at 1 Ghz. The network analyzer will display this phase difference as -36 degrees. The test signal at 1.1 GHz is delayed by 36 degrees more than the test signal at 1 Ghz. 
     You can see that if the measurement frequency is 1.2 Ghz we will get a reading of -72 degrees, -108 degrees for 1.3 Ghz, etc. (FIG. 39). There is an electrical delay between the reference and test signals. For this delay we will use the common industry term of reference delay. You may also hear it referred to as phase delay. In older network analyzers you had to equalize the length of the reference arm with that of the test arm in order to make an appropriate measurement of phase vs. frequency. 
     To measure phase on a DUT, we want to remove this phase-change-vs-frequency-due-to changes in the electrical length. This will allow us to view the actual phase characteristics. These characteristics may be much smaller than the phase-change-due-to-electrical-length difference. 
     There are two ways of accomplishing this. The most obvious way is to insert a length of line into the reference signal path to make both paths of equal length (FIG. 40). With perfect transmission lines and a perfect splitter, we would then measure a constant phase as we change the frequency. The problem using this approach is that we must change the line length with each measurement setup. 
     Another approach is to handle the path length difference in software. FIG. 41 displays the phase-v-frequency of a device. This device has different effects on the output phase at different frequencies. Because of these differences, we do not have a perfectly linear phase response. We can easily detect this phase deviation by compensating for the linear phase. The size of the phase difference increases linearly with frequency so we can modify the phase display to eliminate this delay. 
     The 360 offers automatic reference delay compensation with the punch of a button. FIG. 42 shows the resultant measurement when we compensate path length. In a system application you can usually correct for length differences; however, the residual phase characteristics are critical. 
     The algorithm used to calculate the reference delay required to flatten a phase response sums the phase deltas between adjacent points for the entire sweep. This phase sum is used to calculate a reference delay value that equalizes the phase of the start and stop frequencies. 
     The algorithm is based on the &#34;average&#34; phase difference between points or the total phase difference between the start and stop frequency: ##EQU2## where dx=extra line length in system. 
     The value of dx depends upon whether coaxial cable, waveguide or microstrip is the medium for which automatic reference delay is calculated. ##EQU3## 
     Network Analyzer Measurements 
     Now let us consider measuring the DUT. Consider a two port device; that is, a device with a connector on each end. What measurements would be of interest? 
     First, we could measure the reflection characteristics at either end with the other end terminated into 50 ohms. If we designate one end as the normal place for the input, that gives a reference. We can then define the reflection characteristics from the reference end as forward reflection, and those from the other end as reverse reflection (FIG. 43). 
     Second, we can measure the forward and reverse transmission characteristics. However, instead of saying &#34;forward&#34;, &#34;reverse&#34;, &#34;reflection&#34;, and &#34;transmission&#34; all of the time, we use a shorthand. That is all that S-Parameters are, a shorthand! That &#34;S&#34; stands for scattering. The first number is the port that the signal is leaving, while the second is the port to which the signal is being injected. S 11 , therefore, is the signal leaving port 1 relative to the signal entering port 1. The four scattering parameters (FIG. 44): 
     S 11  Forward Reflection 
     S 21  Forward Transmission 
     S 22  Reverse Reflection 
     S 12  Reverse Transmission 
     S-Parameters can be displayed in many ways. An S-Parameter consists of a magnitude and a phase. We can display the magnitude in dB, just like a scaler network analyzer. We often call this term log magnitude. 
     We can display phase as &#34;linear phase&#34; (FIG. 45). As discussed earlier, we can&#39;t tell the difference between one cycle and the next. Therefore, after going through 360 degrees we are back to where we began. We can display the measurement from -180 to +180 degrees. The -180 to +180 approach is more common. It keeps the display discontinuity removed from the important 0 degree area used as the phase reference. 
     There are several ways in which all the information can be displayed on one trace. One method is a polar display (FIG. 46). The radial parameter (distance from the center) is magnitude. The rotation around the circle is a phase. We sometimes use polar displays to view transmission measurements, especially on cascaded devices (devices in series). The transmission result is the addition of the phase and log magnitude (dB) information of each device&#39;s polar display. 
     As we have discussed, the signal reflected from a DUT has both magnitude and phase. This is because the impedance of the device has both a resistive and a reactive term of the form r+jx. We refer to the &#34;r&#34; as the real or resistive term, while we call &#34;x&#34; the imaginary or reactive term. The &#34;j&#34;, which we sometimes denote as &#34;i&#34;, is an imaginary number. It is the square root of -1. If x is positive, the impedance is inductive, if x is negative the impedance is capacitive. 
     The size and polarity of the reactive component x is important in impedance matching. The best match to a complex impedance is the complex conjugate. This complex-sounding term simply means an impedance with the same value of r and x, but with x of opposite polarity. This term is best analyzed using a Smith Chart (FIG. 47), which is a plot of r and x. We will discuss Smith Charts in greater detail below. 
     To display all the information on a single S-parameter requires one or two traces, depending upon format we want. A very common requirement is to view forward reflection on a Smith Chart (one trace) while observing forward transmission in Log Magnitude and Phase (two traces). Let us see how to accomplish this on the 360. 
     The 360 has four channels. Each channel can display a complete S-Parameter in any format on either one or two traces. All four S-Parameters can be seen simultaneously in any desired format. A total of eight traces can be viewed at the same time. While this is a lot of information to digest, the 360&#39;s large color display makes recognizing and analyzing the data surprisingly easy. 
     Another important parameter we can measure when phase information is available is group delay. In linear devices, the phase change through the DUT is linear-with-frequency. Thus, doubling the frequency also doubles the phase change. An important measurement, especially for communications system users, is the rate of change of phase-vs-frequency (group delay). If the rate of phase-change-vs-frequency is not constant, the DUT is nonlinear. This nonlinearity can create distortion in communications systems. We will discuss this in greater detail below. 
     Measurement Error Correction 
     Since we can measure microwave signals in both magnitude and phase, it is possible to correct for six major error terms: 
     Source Test Port Match 
     Load Test Port Math 
     Directivity 
     Isolation 
     Transmission Frequency Response 
     Reflection Frequency Response 
     We can correct for each of these six error terms in both the forward and reverse directions, hence the name 12-term error correction. Since 12-term error correction requires both forward and reverse measurement information, the test set must be reversing. &#34;Reversing&#34; means that it must be able to apply the measurement signal in either the forward or reverse direction. 
     To accomplish this error correction, we measure the magnitude and phase of each error signal (FIG. 48). Magnitude and phase information appear as a vector that is mathematically applied to the measurement signal. This process is termed vector error correction. We will discuss this concept in greater detail below. 
     Test Sets 
     We have now learned about reference delay. We have discussed S-Parameters. We know what they are, how to measure them, and how to display them. We have also learned a little about vector error correction. Let us now turn to the 360 Test Sets. We will see how well they meet our measurement needs. 
     The basic WILTRON Test Set is called a &#34;Reversing Test Set&#34; (FIG. 49). It contains an internal switch to select the direction of the microwave signal. Each port has a directional device. Any S-Parameter can be measured and 12-term error correction can be applied automatically. In fact, we can measure all four S-Parameters simultaneously and apply a 12-term error correction to each. Also, with the dedicated synthesized source, we update the display rapidly enough to allow real-time tuning of the DUT. 
     Note that the coupler on Port 2 is aligned different from the coupler on Port 1, The throughline of the Port 2 coupler goes directly into the forward transmission test sampler. There is no coupling-factor loss. This design approach optimizes the dynamic range of forward-transmission measurements. Coupling-factor loss is the most important requirement for large dynamic range measurements. 
     The two Reference Delay Lines (one for each port) are internal connections. Since the Wiltron system always uses a synthesized signal, we can accurately compensate reference delay in software. While the capability for changing the actual line length is still present, most people will rarely need to use it. 
     The Wiltron 4O Ghz Test Set (Model 3611) does not contain internally accessible reference delay loops. The additional line length would significantly affect dynamic range at high frequencies. You can change these internal line lengths. The Models 3620 and 3621 Active Device Test Sets (FIG. 50) are similar to the two reversing models. The exception is a third step attenuator located in the forward transmission line just ahead of the sampler. This step attenuator provides for measuring devices with output powers greater than -10 dBm. This additional step attenuator can be used to reduce the power into the forward transmission measurement sampler. Reverse transmission and output match measurements are unaffected by the step attenuator. The input match and attenuation of the step attenuator is measured and stored with the calibration data during the calibration process. 
     TUTORIAL 
     This tutorial introduces you to the 360 control panel and three basic measurement operations. It guides you through a typical calibration sequence and two basic network analyzer measurements: transmission and reflection. 
     
         ______________________________________                                    
a. Equipment Needed                                                       
______________________________________                                    
360               Network Analyzer                                        
3610              Test Set                                                
360SS47           Source                                                  
3650              Cal Kit                                                 
3670A50-2         Through Cable                                           
______________________________________                                    
 
    
     b. Initialize the System 
     Install the system disk and turn the power on. The 360 automatically performs a self test and comes on line with the same control panel setting as when exited last. 
     c. Backup the System Diskette 
     1. Remove the system diskette and install a blank diskette in its place. 
     2. Press the UTILITY MENU key (FIG. 52). 
     3. Using the MENU cursor and ENTER keys (FIG. 52), select the menu options. An appendix attached hereto shows menu option displays for the system of the present invention along with error codes provided from the measurement system and command codes acceptable to the measurement system. 
     4. Select GENERAL DISK UTILITIES when Menu U1, Utility Menu appears. 
     5. Select INITIALIZE DISK WITH PROGRAM when Menu U2, Disk Utilities Functions appears. 
     6. Follow the instructions displayed on the screen to complete the backup procedure. 
     d. Load Calibration Kit Data 
     1. Install the data diskette from the Model 3651, 3652, or 3653 Calibration Kit. 
     2. Press the UTILITY MENU key. 
     3. Select CALIBRATION COMPONENT UTILITIES when menu UI, Utilities Menu appears. 
     4. Select INSTALL CALIBRATION COMPONENT INFORMATION FROM DISK when Menu U4, Calibration Component Utilities appears. 
     5. To ensure that correct calibration data has been loaded, select DISPLAY INSTALLED CALIBRATION COMPONENT INFORMATION on Menu U4. 
     6. Select the appropriate component type (SMA[M], SMA[F], K-Conn[M], etc.) when Menu U5, Display Installed Calibration Components Information appears. 
     7. When the Readout Text associated with Menu U5 appears in the display area of the CRT, check that the serial number for your Open or Short device is correct for the component in your kit. 
     Full 12 Term Calibration, Precision 
     Broadband Termination 
     WILTRON&#39;S precision broadband terminations are the ideal impedance reference for calibrating network analyzers. They are easier to use, more repeatable, and less expensive than sliding loads. However, sliding loads, when correctly used and perfectly aligned can be more accurate. 
     a. Connect the Thruline 
     Install the thru cable, WILTRON Co. Part No. PN3670A50, to Port 2 on the Test Set. We will refer to the unterminated end of this cable as Port 2 for all calibration and measurement steps. 
     b. Begin the Calibration 
     1. Press the DEFAULT PROGRAM key (FIG. 52). 
     2. Press the BEGIN CAL key (FIG. 52). 
     NOTE 
     Selecting these menu options automatically calls the next menu in the listed sequence. 
     3. Select FULL 12 TERM when Menu C5, Select Calibration appears. 
     4. Select NORMAL when Menu C1, Select Calibration Data Points appears. 
     5. Set 0.5-18 GHz when Menu C2, Frequency Range of Calibration appears. 
     6. Select CHANGE OF PORT 1 CONN when Menu C3, Confirm Calibration Parameters appears. 
     7. Select GPC-7 when menu C4, Select Connector Type appears. 
     8. Select CHANGE PORT 2 CONN when Menu C3 reappears. 
     9. Select GPC-7 when menu C4 reappears. 
     10. Select CHANGE LOAD TYPE when Menu C3, reappears. 
     11. Select BROADBAND LOAD when Menu C6, Select Load Type appears. 
     12. Select START CAL when Menu C3 reappears. 
     13. Follow the instruction in each of the upcoming Calibration Sequence menus. Each step allows you to view the calibration data being taken and then retake the data if desired. This saves you from having to repeat the complete calibration because of an undetected error--such as, a poorly mated connection. 
     14. When Menu C10, Calibration Sequence Completed appears, you can choose to store the calibration data on a disk. You should always choose to do this; steps 15 thru 19 show you how. 
     15. Press the SAVE/RECALL MENU key (FIG. 52). 
     16. Select SAVE when Menu SR1, Save/Recall Front Panel Information appears. 
     17. Select SAVE CAL DATA AND FRONT PANEL SETUP ON DISK, when Menu SR2, Recall or Save appears. 
     18. Select CREATE NEW FILE, when the GP1-3, Select File menu appears. 
     19. Enter CAL 1 and your initials using the knob, when the Menu GP5, Select Name appears. When finished, move to DONE and press the ENTER key. You can assign an 8 character file name and up to 15 additional spaces for other information. 
     c. Discussion 
     During calibration, the 360 automatically 
     1. Sets the number of points to maximum--501 points. 
     2. Sets averaging to 128 while the loads are being measured. 
     3. Sets the Video IF bandwidth to the REDUCED value (1 kHz). 
     NOTE 
     The above values are default values that can be changed through menu selection. 
     A lower noise floor can be achieved by reducing the Video IF bandwidth and averaging several measurements. However, the default values have been found to be optimum for providing a compromise between a low noise floor and data-taking speed. Reducing the Video IF bandwidth eliminates unwanted noise by more closely tracking the desired frequency. Averaging several measurements removes random variations and effectively improves noise floor performance. However, reducing video IF bandwidth and increasing the number of averages causes an increase in sweep time. 
     Smoothing is not necessary nor desirable during calibration, since it does not affect the actual measurement data and will mask any rapid response variations displayed. This can lead to a sense of false confidence, both when performing the calibration and when monitoring the displayed calibration data for measurement errors. 
     Measuring Transmission 
     Now let us consider the effect of calibration on transmission measurements. Press the APPLY CAL key, to tune off the calibration. Leave the throughline connected between Port 1 and Port 2. 
     Let us look only at S 21  (forward transmission). Set up the 360 for a single display with Log Mag and Phase using channel 3 and the graph menu, as follows: 
     1. Press the CHANNEL MENU key (FIG. 52). 
     2. Select SINGLE DISPLAY when Menu CM, Select Display Mode appears. 
     3. Press the CH3 key. 
     4. Press the S PARAMS key. 
     5. Select S21 FWD TRANS when Menu SP, Select S-Parameter appears. 
     6. Press the GRAPH TYPE key. 
     7. Select LOG MAGNITUDE AND PHRASE when Menu GT1, Select Graph Type appears. 
     8. Press the AUTOSCALE key. 
     9. Notice the sawtooth pattern of the phase display. This represents the phase shift resulting 
     10. Press the APPLY CAL key to apply the calibration. 
     11. Notice that both the magnitude and phase appear to be flat. The amplitude and phase uncertainties have been removed from the magnitude measurement. A software phase correction has been added. It makes the reference arm and the test arm appear to be the same length. 
     12. The test arm now includes the same cable used in the calibration. Therefore, any deviation from 0 dB magnitude and 0 degree phase constitute the uncertainty of the measurement. This uncertainty results from random errors such as connector repeatability. It is dependent on the quality of both the calibration and the calibration components. 
     13. Press the AUTOSCALE key. 
     14. Set the averaging for best performance, as follows: 
     (a) Press the AVG/SMOOTH MENU key. 
     (b) Enter 50 from the keypad for the AVERAGING XXXX MEAS PER POINT option. 
     (c) Press the X1 terminator key (FIG. 52). 
     NOTE 
     Fifty averages typically provide a good balance between a smoother waveform and its tradeoff, increased total measurement time. 
     (d) Press the AVERAGE key to enable the averaging function. 
     Melting Reflection 
     Now let us consider the effect of calibration on reflection measurements. 
     1. Press the APPLY CAL key (FIG. 52) to disable the calibration. 
     2. Press the AVERAGE key (FIG. 52) to disable the Averaging Function. 
     3. Remove the throughline. 
     4. Connect the broadband load to Port 1. 
     5. Set the start frequency to 2 GHz, as follows: 
     (a) Press the SETUP MENU key. 
     (b) Change the START frequency to 2 GHz and press ENTER, when Menu SU1, Sweep Setup appears. 
     6. Set up the 360 to measure S 11  (return loss) as follows: 
     (a) Press the CHANNEL MENU key. 
     (b) Select SINGLE DISPLAY when Menu CM, Select Display Mode appears. 
     (c) Press the S PARAMS key. 
     (d) Select S 11  FWD REFL when Menu SP, Select S-Parameter appears. 
     (e) Press the CH1 key. 
     (f) Press the GRAPH TYPE key. 
     (g) Select LOG MAGNITUDE when Menu GT1, Select Graph Type appears. 
     7. Press the AUTOSCALE key. 
     8. Using Marker 1 and the Readout Marker feature, find the worst case return loss as follows: 
     (a) Press the MARKER MENU key (FIG. 52). 
     (b) Place the cursor on MARKER 1 and press the ENTER key, when Menu M1, Set Markers appears. 
     (c) Press the READOUT MARKER key. 
     (d) Choose MARKER TO MAX and press the ENTER key, when Marker M4, Readout Marker appears. 
     INSTRUMENT OPERATION-AN OVERVIEW 
     Power-Up Characteristics 
     When initially turned on (powered up), the 360 comes on line with the factory-selected default settings. On subsequent power-ups, it generally returns to the exact status and display that it was in when powered down last. (An exception is when the stored control panel setup data has been lost, then it comes on with the default settings.) 
     After coming on line, the 360 executes a self test. It then attempts to load its operating software from the installed disk. If unable to do so, it attempts to load from an internal ROM cartridge. If neither option is possible, it displays the message: &#34;DISK NOT READY--PRESS ENTER TO TRY AGAIN.&#34; It then waits for you to take the appropriate action. 
     During the self test if the program detects a system fault, it shuts the system down and displays an error message. (A system fault is one that occurs in the 360, the test set, or the source.) If the program detects a fault in a peripheral device (such as the printer), it does not shut the system down. It only displays an error message. 
     Measurement Control 
     Measurement control is provided through selections of start, stop, and marker frequencies, as follows: 
     a. Start and Stop Frequencies 
     Start and stop frequencies must meet the following criteria: 
     1. Be within the range of the frequency source and test set. 
     2. Have a span that provides 100 kHz resolution for the 360 SSXX source. 
     3. Have a start frequency lower than the stop frequency. 
     NOTE 
     You may change the start and stop frequencies after calibration. However, your new frequencies must fall within the calibrated range when the calibration is applied. 
     b. Marker Frequencies 
     If there are markers at frequencies other than the equally spaced set of calibration frequencies, they will be readjusted to a calibration frequency. 
     Data Enhancement 
     Vector Error Correction 
     The 360 provides software correction for inherent measurement-setup error terms. Additionally, you can select software correction for any four measurements: Frequency Response, Reflection only, One Path-Two Port, or Full 12-Term. 
     b. Data Averaging 
     You can average measurements over time for a more accurate readout of noisy, rapidly changing amplitude data. In averaging, you select the number of points for which you wish averaging calculated. The sweep then stops at each frequency point and takes that number of readings. The program then averages the amplitude readings at the frequency point and writes the average value on the displayed graph-type. For calibration, the averaging function defaults to 128. You may reset it to any other value, however. 
     c. Smoothing 
     You can smooth (amplitude only) measurement variations over a frequency span of from 0 to 20 percent of the sweep. The smoothing process uses a raise Hamming window to average the data from a span of frequencies. For example, in FIG. 51 if the program averages all data points from A to B to give point X then the average of all points from A+1 to B+1 is X+1. 
     d. Normalization (Trace Memory) 
     Normalization means taking data from a standard component (filter or attenuator, for example), then later comparing it with another like item. To normalize data means to add, subtract, multiply, or divide it by standard data taken earlier and stored in memory. When the measured data taken from the two components are the same, the 360 displays a straight line. If they differ, however, it displays the degree of departure of the new data from the stored data. 
     Human Interface 
     The 360 interfaces with the user through a system of informative menus coordinated with control panel keys. You are always prompted by the menus or complete an action by pressing one or more keys, If the key is one of the menu-allowed choices, then the 360 responds in one of the following ways: 
     It displays a different menu. 
     It enters a numeric value. 
     It allows a choice in the current menu. 
     If the key is not one of the above choices, the 360 beeps. 
     a. Channel Concert 
     The 360 has four measurement channels that you can display simultaneously, individually, or in pairs. You can display a different S-Parameter on each channel. Or, you can display the same S-Parameter on one or more channels. You can control the four channels separately for some functions and parameters, while others must be the same for all channels. The parameters that can be different for the different channels are as follows: 
     Graph-Type (Rectilinear, Polar, Or Smith Chart) 
     Amplitude Scales 
     Reference Delay Setting 
     Normalization Memory 
     S-Parameters 
     The parameters that must be the same for all four channels are as follows: 
     Start and Stop Frequencies 
     Error Correction Type 
     Calibration Type and Range 
     Averaging 
     IF Bandwidth 
     Smoothing 
     Marker Frequencies, Times, Distances 
     b. Display of Messages 
     The 360 displays on-screen text giving error and other messages. This text concisely states the condition causing the message, specifies an action needed, or both. 
     c. Active Parameter 
     The active parameter is the only one that you can change using the Data Entry keys or knob. We define a parameter as a frequency, time, degree, distance, or other numeric value that you can enter. (You enter a parameter using the keypad or knob and end it using one of the Terminator keys (See Appendix page 6 (App. 6)). You open such a parameter for modification by pointing a menu cursor at it. That makes it active. You close a parameter by moving the cursor away from it. Or, alternatively, you erase or replace the menu. 
     Stored Data 
     The 360 can store control panel setups along with normalization, measurement, and calibration data. The following is a list of the items saved. (The 360 saves these parameters for all active channels.) 
     a. Display Parameters 
     Offset 
     Resolution 
     Reference Line 
     Limits (Enabled, On/Off/Values) 
     Minimum and Maximum Values 
     Selected S-Parameter 
     Display Type 
     Active Channel 
     Display Mode 
     S-Parameter For Each Channel 
     Blank Frequency Display Active, Color Plane 
     Reference Delay 
     Dielectric Used 
     Dielectric Constant 
     Normalize Mode 
     b. Measurement Parameters 
     Start of Sweep 
     End of Sweep 
     Source Power and Attenuator Settings 
     Frequency Resolution 
     Device-Under-Test ID 
     c. Enhancement 
     Smoothing Enabled 
     Averaging Enabled 
     Smoothing % of Sweep 
     Number of Points Averaged 
     IF Bandwidth 
     d. Output 
     Type of Output 
     Options Enabled (Model, Device ID, Date, Operator) 
     Resolution 
     e. Calibration 
     Size 
     Frequencies 
     Port Connector Type 
     Calibration Type 
     Correction Type 
     Load Type 
     Capacitance Coefficient for Connector 
     Connector Offset Length for Each Port 
     f. Miscellaneous 
     Markers (Enabled, On/Off, Values) 
     Delta Reference Mode 
     Marker Frequencies, Times, Distances 
     GPIB Addresses and Termination Unit 
     g. Disk Identification 
     Calibration File Name 
     h. Control Panel Setups 
     You can store the instrument state (measurement parameters and operating modes) in internal non-volatile memory or on the installed disk. You select the storage media using the SAVE/RECALL MENU key and its related menus. You can save up to four control panel setups, along with calibration data, in internal memory and more on the disk. Additionally, the 360 saves certain parameters each time you turn it off. It automatically saves (1) the current control panel setup, (2) all measurement, display, calibration, and other parameters and functions. This allows it to return to its exact same state when powered up next. 
     i. Normalization Memory 
     The 360 can store up to four channels of normalization data (S-Parameter measurements) in volatile RAM. To prevent loss of this data when you turn the system off, you may also save it to the disk. 
     j. Measured Data 
     You can also save measured data on the disk. The 360 stores it as ASCII-encoded test. The format is the same as that used for the tabular printout. This feature lets you make the computer analysis of the measured data, provided your computer has a compatible disk drive. 
     k. Calibration Data 
     The 12-term error correction coefficients for each data point covered by the calibration being saved are stored as 12 single precision (32-bit real, 32-bit imaginary) complex numbers. This results in 96 bytes per point, or 48 KBytes for a 51 point calibration. 
     External and Peripheral Interfaces 
     a. Microfloppy Disks 
     The 360 employs an integrally mounted disk drive for 31/2 inch microfloppy disks. 
     b. GPIB Interface 
     The 360 has two GPIB interfaces:Source Control (Master) and System Control (Slave). You can program each of these interfaces for address, delimiting character, etc. using a menu. The 360 provides GPIB status in a menu. 
     GPIB Specifications are as follows: 
     Interface 
     IEEE-488 standard GPIB 
     System Interface: IEEE-488 port used exclusively by the 360 to control and extract information from a Model 360SSXX frequency source. 
     Addressing: Controller is address=0 and source is address=5. Addresses are settable by menu; for the 360SSXX, it is also settable from the rear panel. 
     Speed: 200 μs/bus cycle (device dependent). 
     Interface Function codes: SH1, AH1, T8, TE0, L4, LE0, SR0, RL0, PP0, DT0, DC0, C1, C2, C3, C27. 
     360 GPIB (System Control or Slave) 
     Interface: IEEE-488 standard GPIB 
     System Interface: IEEE-488 port by which an external controller may take control of the 360. The controller can perform all control panel operations. 
     Addressing: Defaults to address 6, settable by menu control. 
     c. Parallel Printer Port 
     The printer port is compatible with a standard &#34;Centronics&#34; interface. The 360 has the capability for an exact pixel-by-pixel dump of the CRT screen, when used with the WILTRON model 2225C inkjet printer. The output can be any of the following. 
     1. A full-screen dump. 
     2. A data-display dump that does not reproduce the menu. 
     3. A tabulated listing of the data. 
     d. Test Set Control Interface 
     The Power and Data interfaces use 37-pin &#34;D&#34; subminiature connectors. The Signal and RF interfaces use 17-pin, coaxial &#34;D&#34; subminiature connectors. 
     e. Video Interface 
     The 360 provides two video outputs, as follows: 
     1. Composite Monochrome Output. This output has rear-panel screwdrive adjustments for relative mix levels of red, green, and blue. You can adjust the mix to display shades of gray on an external monitor. 2. Separate R-G-B, TTL Level Outputs. You control the level using the same adjustment as for 1, above. Additionally, the 360 has positive-true and negative-true horizontal and vertical drive signals available. All video signals are nominally either TTL levels or 1-volt, zero-to-peak-into-75-ohm levels. The signals appear on a dedicated 15-pin &#34;D&#34; subminiature connector mounted on the rear panel. The composite video is also available on the RCA-type phono connector. 
     MODEL 360 CONTROL PANEL CONTROLS 
     FIG. 52 illustrates the control panel layout. 
     1. LINE ON/OFF 
     Turns the 360 on and off. When pressed ON, the program runs a self test then recalls the parameters and functions in effect when powered down last. 
     2. GPIB INDICATORS 
     REMOTE 
     Lights when the 360 goes under GPIB control. IT remains lit until the unit returns to local control. 
     TALK 
     Lights when you address the 360 to talk and remains lit until unaddressed. 
     LISTEN 
     Lights when you address the 360 to listen and remains lit until unaddressed. 
     SRQ 
     Lights when the 360 requests service from the controller (sends out a SRQ). The LED remains lit until the 360 receives a serial poll or until the controller resets the SRQ function. 
     LOCAL LOCKOUT 
     Lights when a local lockout message is received. The LED remains lit until the message is rescinded. When lit, you cannot return the 360 to local control via the front panel. 
     3. SAVE/RECALL MENU 
     Displays the first of three-menus that let you save the current calibration or control panel setup or recall a previously saved calibration or setup. 
     4. BEGIN CAL 
     Calls up the first in a sequence of menus that guide you through a measurement calibration. See below for a detailed discussion of the calibration keys, indicators, and menus. 
     5. CALIBRATION Indicators 
     Shows the calibration state of the 360. 
     6. APPLY CAL 
     Turns on and off the applied error correction displayed by the calibration indicators. 
     7. MENU 
     The ↑ key moves the menu cursor up and the ↓ key moves it down to select between entries appearing in the menu area of the CRT. 
     8. ENTER 
     Implements the menu selection chosen using the MENU arrow keys. 
     9. ROTARY KNOB 
     Used to alter measurement values for the active parameter (Start Frequency, Stop Frequency, Offset, etc). 
     10. DISPLAY Keys 
     GRAPH TYPE: Displays either of two menus that let you choose the type of display and its S-parameter that appears for the active channel. 
     SET SCALE: Displays the appropriate scaling menu, based on the type of graph and its S-parameter being displayed for the active channel. 
     AUTO SCALE: Automatically scales the active channel for optimum viewing. 
     S-PARAMS: Displays Menu SP, which lets you choose between S 11 , S 12 , S 21 , or S 22 . You may display the same parameter on two or more channels. 
     REF DELAY: Displays the first of two menus that let you enter a reference-delay in time or distance. For a correct distance readout, you must set the dielectric constant to the correct value. Refer to the discussion in Menu RD2. 
     TRACE MEMORY: Displays a menu that lets you do one of the following. (1) Store the measured data in memory. (2) View the stored data. (3) Add, subtract, multiply, or divide the measured data from the stored data (normalize to the stored memory). (4) View both the measured and the stored data simultaneously on the active channel. Four memories exist--one for each channel. This lets you normalize the data in each channel independently. The LED on this button lights only when the active channel is displaying measurement data normalized to memory. 
     11. KEYPAD 
     Provides for entering values for the active parameter. The active parameter is the one to which the menu cursor is pointing. 
     12. ENHANCEMENT Keys and LED 
     IF BW: Cycles between NORMAL, REDUCED, and MINIMUM intermediate frequency (IF) bandwidths. The appropriate indicator lights to display the selected value. 
     AVG/SMOOTH MENU: Displays a menu that lets you enter values for AVERAGING and SMOOTHING. 
     OPTION MENU: Displays a menu showing the choice of options installed. (This key is not active unless you have options other than Time Domain installed). 
     TRACE SMOOTH: Turns the trace smoothing function on and off. 
     AVERAGE: Turns the averaging function on and off. 
     13. TERMINATOR Keys 
     Ghz/10 3  /μs/m: Terminates a value entered on the keypad in the units shown--that is; gigahertz for frequency, 1×10 -3  power for dimensionless or angle entries, microseconds for time, or meters for length. 
     Mhz/X1 /ns/cm: Terminates a value entered on the keypad in the units shown--that is; megahertz for frequency, unity for dimensionless or angle entries, nanoseconds for time, or centimeters for length. 
     Khz/10 -3  /ps/mm: Terminates a value entered on the keypad in the units shown--that is; kilohertz for frequency, 1×10 -3  power for dimensionless or angle entries, picoseconds for time, or millimeters for length. 
     CLEAR/RET LOC 
     a. Local (Non-GPIB) Mode: (1) The key clears entries not yet terminated by one of the terminator keys above, which allows the previously displayed values to redisplay. Or (2) the key turns off the displayed menu, if you have not made any keypad entries needing termination. 
     b. GPIB Mode: The key returns the instrument to local (control panel) control, unless the controller has sent a local lockout message (LLO) over the bus. 
     14. DISK DRIVE 
     Provides a drive for the 31/2 inch floppy disk used to store both the operating system and the selected front panel setups and calibration. Refer to paragraphs 3-6 for a detailed description. 
     15. OUTPUT Keys 
     MENU: Displays option menus that let you define what will happen each time you press the START 
     PRINT key. The displayed menu also selects disk I/O operations. 
     START PRINT: Tells the printer or plotter to start output based on the current selections or plotting. 
     STOP PRINT: Immediately stops printing the data, clears the print buffer, and sends a form-feed command to the printer. However, if the printer is not then printing data, the key only sends a form-feed command. 
     16. MEASUREMENT Keys and LED 
     SETUP MENU: Displays the first of three menus that let you enter start/stop frequencies, source power, and attenuation values. 
     DEVICE ID: Displays a menu asking you to name the test device. 
     DATA POINTS: Cycles between maximum, normal, and minimum resolution values. The appropriate MAXIMUM, NORMAL, or MINIMUM switch indicator lights to display the selected value. 
     DOMAIN: Displays the first of the menus that let you set the Time Domain display parameters. (This key is only active if you have the Time Domain option). 
     (1) If already in the DOMAIN menus, pressing this key will return to the first menu in the sequence. 
     (2) If in the DOMAIN menus and another (non-time domain) menu is displayed by pushing a menu key, the last displayed time domain menu redisplays when the DOMAIN key is next pressed. 
     HOLD: Toggles the instrument in and out of the hold mode or triggers a sweep, depending on the function selected in menu SU4. 
     17. SYSTEM STATE Keys 
     DEFAULT PROGRAM: Resets the control panel to the factory-preset state and displays Menu SU1. 
     CAUTION 
     Use of this key will destroy control panel and calibration setup data, unless they have been saved to disk. 
     UTILITY MENU: Displays the first in a series of menus that let you perform diskette and other utility-type functions and operations. 
     18. MARKERS/LIMITS Keys 
     MARKER MENU: Displays the first in a series of menus that let you set and manipulate marker frequencies, times, and distances. 
     READOUT MARKER: Displays a menu that lists all of the active markers. If no markers are active, the message &#34;NO ACTIVE MARKERS&#34; displays for four seconds in the message area of the screen. 
     LIMITS: Displays one of the menus that let you manipulate the Limit 1 and Limit 2 lines displayed on the CRT. 
     19. CHANNELS Keys 
     CHANNEL MENU: Displays a menu that lets you display format for the channels, 
     CH 1: Makes Channel 1 the active channel. The active channel is the one acted on by the keys in the DISPLAY section. Only one channel can be active at any one time. 
     CH 2: Makes Channel 2 the active channel. 
     CH 3: Makes Channel 3 the active channel. 
     CH 4: Makes Channel 4 the active channel. 
     20. CRT Display 
     Displays any or all of the four measurement channels. 
     CALIBRATION Keys and Indicators 
     FIG. 53 illustrates the calibration keys. 
     a. BEGIN CAL Key 
     This key displays a menu that lets you initiate the calibration sequence. That is, to begin a sequence of steps that corrects for errors inherent in a 2-port measurement setup. 
     b. APPLY CAL Key 
     This key turns on and off the error correction that you may apply to the displayed channel (s) using the currently valid error-correction indicator. 
     C. CALIBRATION Indicators 
     FULL 12 TERM: You have corrected for all twelve error terms associated with a two-port measurement. 
     NOTE 
     Choosing this calibration in Menu C5 corrects for all possible measurement error terms. 
     1 PATH, 2 PORT: You have corrected for the four forward-direction error terms (E DF , E SF , E RF , and E TF ). 
     FREQ RESPONSE: You have corrected for one or both of the forward-direction error terms associated with a measurement of S 11  and S 21 . This is a subset of the 12-term calibration. 
     REFLECTION ONLY: You have corrected for the three error terms associated with an S 11  measurement (E DF , E SF , and E RF ). This is a subset of the 12-term calibration. 
     NONE: No calibration data currently exists. 
     Calibration Menus 
     (The Calibration Sequence consists of several menus, similar in structure but different in content. The content differs only in the type of device (load, open, or short) the menus say to connect to ports 1 and 2. The menus automatically appear in a sequence based on choices made C7A through L. Each menu will specify a type of device to be connected to ports 1 and 2 (FIG. 55) and each will provide a procedural message (one of the three types shown in Table 93B).) 
     Measurement calibration is aided by the step-by-step procedures contained in the calibration menus. FIG. 54 provides a flowchart describing the calibration sequence. Apps. 1-18 and FIGS. 53-57 describe the calibration menus. 
     SAVE/RECALL MENU Key And Menus 
     Pressing the SAVE/RECALL MENU key shown in FIG. 52 displays the first of four menus (Apps. 19 and 22) that allow you to save or recall control panel setups and calibration data. 
     MEASUREMENT Keys and Menus 
     SETUP Key 
     Pressing the measurement keys shown in FIG. 52 calls Sweep Setup Menu SU1 (SU3). Depending upon which menu items you select, additional menus SU2 thru SU6 may also be called. Apps. 23-28 describe the six menus. 
     DEVICE ID Key 
     Pressing this key calls a menu that lets you name the test device. This key has the same effect as selecting &#34;Device ID&#34; in the PM2 menu. 
     c. HOLD Key 
     If the instrument is sweeping, pressing this key results in the immediate halt of the sweep at the current data point. The LED on the button lights, indicating that the Hold Mode is active. 
     The instrument may be taken out of the hold mode as follows: 
     1. By using any of the options described in Menu SU4, Select Function for Hold Button. 
     2. By pressing the Default Program button. This causes the 360 to revert to a predefined state. 
     3. By pressing the Begin Cal button. This causes the 360 to resume sweeping and begin the Calibration Menu sequence. 
     NOTE 
     See App. 26 (Menu SU4) for a description of the interaction between the Hold Mode and the selection of &#34;Single Sweep&#34; or &#34;Restart Sweep&#34;. 
     If you restart the sweep after performing any disk operations in the Hold Mode (sweep stopped at some data point), the sweep restarts from the beginning. 
     d. DATA POINTS Key 
     Pressing this key toggles between MAXIMUM, NORMAL, and MINIMUM resolution, lighting the appropriate LED. If MAXIMUM resolution is X data points, MINIMUM resolution will be approximately X/6 data points and NORMAL resolution will be approximately X/3 data points. The nominal values are X=501, X/3=167 and X/6=85. 
     e. DOMAIN Key 
     See FIG. 52 for a full description. Additionally, if the Time Domain option is installed, making a selection other than &#34;Frequency Domain&#34; lets you display measured data in the time domain. It also calls a further sequence of Time Domain Menus. Menu TD1 and all other time domain menus, along with a discussion of the time domain measurement, are provided below. 
     CHANNEL Keys and Menu 
     a. CH 1-4 Keys 
     The channel keys shown in FIG. 52 define the active channel. One (and only one) must always be active as indicated by the associated LED. Pressing a button makes the indicated channel active. If it is already the active channel, pressing the key has no effect. 
     The active channel will be the channel acted upon by the S PARAMS, GRAPH TYPE, REF DELAY, TRACE MEMORY, SET SCALE, AUTO SCALE and DOMAIN keys. When in the single channel display mode, the active channel will be the one displayed. 
     b. CHANNEL Menu 
     Pressing this key calls menu CM (App. 29). Here, you select the number of channels to be displayed. When in the single display mode, only the active channel will be displayed. 
     DISPLAY Keys and Menus (See FIG. 52) 
     a. S PARAMS Key 
     Pressing this key calls menu SP (App. 30). This menu allows you to select the S-Parameter to be displayed by the active channel for the selected S-parameter. 
     b. GRAPH TYPE Key 
     Pressing this key calls menu GT1 or GT2 (Apps. 31 or 32). These menus let you select the type of display to appear on the active channel for the selected S-parameter. 
     c. SET SCALE Key 
     Pressing this key calls the appropriate scaling menu (SSn) depending upon the graph type being displayed on the active channel for the selected S-parameter. 
     d. REF DELAY Key 
     Pressing this key calls menu RD1 (App. 36). This menu lets you input the reference delay in time or distance. You do this by selecting the appropriate menu item. For a correct distance readout, the dielectric constant must be set to the correct value. This is accomplished by selecting &#34;SET DIELECTRIC&#34;, which calls menu RD2 (App. 37). 
     On menu RD1, selecting &#34;AUTO&#34; and pressing ENTER automatically adjusts the reference delay to unwind the phase. The values for time and distance turn red for one second when you activate &#34;AUTO&#34;. 
     The 360 unwinds the phase as follows: 
     1. First, it sums the phase increments between each pair of measured data points, then it takes the average &#34;P delta  &#34; over the entire set of points. 
     2. Next, it corrects the phase data by applying the following formula: 
     P correct  =P measured  --Nx delta  (where P measured  is the measured phase of the Nth data point). 
     Assuming there are fewer than 360 degrees of phase rotation between each data point, the above-described operation removes any net phase offset. The endpoints of the phase display then fall at the same phase value. 
     e. AUTO SCALE Key 
     Pressing this key autoscales the trace or traces of the active channel. When in one of the scaling menus, the 360 indicates this is happening by turning the menu entries red for 1 second. The new scaling values are then displayed on the menu and graticule. The resolution will be selected from the normal sequence of values you have available using the knob. 
     f. TRACE MEMORY Key 
     Pressing this key brings up menu NO1 (App. 33). This menu--which relates to the active channel--allows you to store data to memory, view memory, perform operations with the stored memory, and view both data and memory simultaneously. Four memories exist, one for each channel. This allows each channel to be stored and normalized independent of the other channels. Data from the trace memory may be stored in the disk or recalled from it. The SELECT TRACE MATH choice brings up menu NO2 (App. 34). 
     9. DISPLAY Menus 
     Apps. 30-37 show the menus associated with the GRAPH TYPE, S PARAMS, and REF DELAY keys. Apps. 38 and 39 show two menus that are representative of the twelve associated with the SET SCALE key. 
     ENHANCEMENT Keys and Menus (See FIG. 52) 
     a. OPTION MENU Key 
     This key is reserved for future options. 
     b. AVG/SMOOTH MENU Key 
     Pressing this key brings up the EM Menu (App. 40). When pressed during calibration sequence, it brings up the EM Cal Menu (App. 41) instead. 
     c. TRACE SMOOTH and AVERAGE Keys 
     The AVERAGE and TRACE SMOOTH keys select their respective functions on and off with the appropriate LED indicating when the function is selected. 
     d. IF BW Key 
     Pressing this key cycles between three different IF bandwidths. The applicable NORMAL, REDUCED, or MINIMUM LED lights to indicate selection. 
     OUTPUT Keys and Menus (See FIG. 52) 
     a. MENU Key 
     Pressing this key brings up menu PM1 (App. 42). This menu allows you to define what will happen every time you press the START PRINT key. Apps. 42-45 describe the menu options. 
     b. START PRINT Key 
     Pressing this key starts outputting the measured data as defined by the setup defined by the selected MENU key. 
     c. STOP PRINT Key 
     Pressing this key can result in any of the following actions if the printer is selected: 
     If the 360 is not outputting data, the key sends a form feed command to the printer. 
     If the printer is active, the key aborts the printing and sends a form feed command to the printer. Aborting the printing clears the print buffer. 
     Pressing this key if the printer is not selected and another form of output is active, the key aborts it but does not send a form feed to the printer. 
     d. PLOTTING FUNCTIONS 
     The 360 can plot an image of either the entire screen or subsets of it. Plots can be either full sizes or they can be quarter size and located in any of the four quadrants. You can select different pens for plotting different parts of the screen. You cannot, however, plot tabular data. The menus for selecting plotter functions are shown in Apps. 46-48. 
     SYSTEM STATE Keys and Menus (See FIG. 52) 
     a. DEFAULT PROGRAM Key 
     Pressing this key brings up the default menu. If pressed again, it recalls the factory selected default values for the control panel controls. The values are defined in Table 3-1. 
     C A U T I O N 
     Use of this key will destroy control panel and calibration set up data, unless they have been saved to disk. 
     
                       TABLE 3-1                                                   
______________________________________                                    
Default Settings                                                          
Function  Default Setting                                                 
______________________________________                                    
Instrument                                                                
          Measurement Setup Menu Displayed                                
State                                                                     
Measurement                                                               
          Maximum sweep range of source and test set                      
          Source Power 0.0 dBm                                            
          Resolution: Normal                                              
Channel   (Quad (four-channel) display                                    
          Channel 1 active                                                
Display   Channel 1: S.sub.11, 1:1 Smith Chart                            
          Channel 2: S.sub.12, Log Magnitude and Phase                    
          Channel 3: S.sub.21, Log Magnitude and Phase                    
          Channel 4: S.sub.22, 1:1 Smith Chart                            
          Scale: 10 dB/Division or 90°/Division                    
          Offset: 0.000 dB or 0.00 degree                                 
          Reference Position: Midscale                                    
          Electrical Delay: 0.00 seconds                                  
          Dielectric: 1.00 (air)                                          
          Normalization: Off                                              
          Normalization Sets: Unchanged                                   
Enhancement                                                               
          Video IF Bandwidth: Reduced                                     
          Averaging: Off                                                  
          Smoothing: Off                                                  
Calibration                                                               
          Correction: Off                                                 
          Connector: SMA                                                  
          Load: Broadband                                                 
Markers Limits                                                            
          Markers On/Off: All off                                         
          Markers Enabled/Disabled: All enabled                           
          Marker Frequency: All set to the start-sweep                    
          frequency, or start-time distance                               
          ΔReference: Off                                           
System State                                                              
          Limits: All set to reference position value:                    
          all off, all enabled)                                           
          GPIB Addresses and Terminators: Unchanged                       
          Frequency Blanking: Disengaged                                  
          Error(s): All cleared                                           
          Measurement: Restarted                                          
Output    Output Type: Printer (full screen, clear headers)               
          Marker and Sweep Data: Enabled                                  
______________________________________                                    
 
    
     b. UTILITY MENU Key 
     Pressing this key calls menu U1. This menu accesses subordinate menus to perform system, disk, and service utilities, as described by the flowchart in FIG. 58. The only functions performed directly from the U1 Menu are &#34;Blank Frequency Information&#34; and &#34;Alternate Third Color Pen&#34; (blue or cyan). 
     c. UTILITY MENU Key Menus 
     Apps. 50-55 describe the UTILITY MENU Key menus. 
     Disk Storage Interface and 
     General Purpose Menus 
     a. The 360 has an integrally mounted disk drive of the 31/2 inch removable media type. The format, files and directory are compatible with PC(MS)-DOS, Version 3.2. 
     1. Disk Format. Disks are MS-DOS compatible. The 360 formats them to have 80 data tracks per side on 2 sides, with 9 sectors of 512 bytes per sector. This yields a total of 720 K Bytes per disk. Any of the following file types, in any combination may be stored on the disk. Table 3-2 shows these file extensions. 
     
                       TABLE 3-2                                                   
______________________________________                                    
File Extensions                                                           
File Type       File Extension                                            
______________________________________                                    
Calibration, Normal                                                       
                .CAL                                                      
Trace Memory    .NRM                                                      
Measured Data   .DAT                                                      
Program         (no extension)                                            
______________________________________                                    
 
    
     2. Disk Files. You may find any of the following file-types on the 360 disk. 
     a. Program Files. These are binary files used to load the operating program. Application-type programs cannot read them. 
     b. Calibration Data Files: These are binary files used to store and retrieve calibration and other data. Application-type programs cannot read them. File size depends on calibration type. For example, 58 KBytes for 123 term and 501 points. 
     c. Measured Data Files. These are ASCII files used to store actual measurement data. They can be read by application-type programs. File size depends on selected operations. For example, 25.6 KBytes for 501 points and 1 S-Parameter. 
     d. Trace Memory Files. These are data stored in a display, rather than in a floating-point format. Application-type programs cannot read them. You use them to perform trace math operations on data. File sizes is 4 KBytes. 
     b. Disk User Interface 
     A disk is capable of holding up to 720K bytes of data. Using the data size assumptions above, a disk would be capable of holding: 
     12-Plus Calibration and Front Panel Setups or 
     180 Sets of Normalization Data or 28-Plus Sets of Measurement Data 
     You can also store a downloaded operating program on the disk. However, this reduces the number of the above items that you could store. The disk format imposes a limitation of 112 on the total number of data items. This means the full 360 sets of normalization data could not be stored on the disk. 
     c. Disk and General Purpose Menus 
     1. Disk Menus. A set of DSK menus used to implement the disk functions are provided in Apps. 56-61. 
     2. General Purpose Menus. The menus shown in Apps. 62-64 appear as a result of choices made in certain of the Disk and Print menus. 
     Marker/Limits Keys and Menus (See FIG. 52) 
     a. Marker Menu Key 
     Pressing the MARKER MENU key calls Menu M1 (App. 65). This menu lets you toggle markers on and off and set marker frequencies, times or distances. 
     b. Marker Readout Key 
     1. Pressing this key calls menu M3 (App. 67) under the following conditions. 
     (a) If the Δ Reference mode is off and 
     (1) there is no selected marker, or 
     (2) the selected marker is not in the sweep range. 
     (b) If the Δ Reference mode is on and 
     (1) the Δ Reference marker is not in the sweep range, or 
     (2) no Delta ref marker has been selected. 
     2. Pressing this key calls menu M4 (App. 68) if the Δ Reference mode is off and the selected marker is in the current sweep rate (or time/distance). 
     3. Pressing this key calls menu M5 (App. 69) if the Δ Reference mode and marker are both on and the Δ Reference marker is in the selected sweep rate (or time/distance). 
     c. Markers Keys Menus 
     The menus associated with the two MARKERS keys are described in Apps. 65-69. 
     d. Limit Frequency Readout Function 
     The 360 is equipped with a Limit-Frequency Readout function. This function allows dB values to be read at a specified point (such as the 3 dB point) on the data trace. This function is only available for certain rectilinear graph-types. These graph-types are listed below. 
     1. Log Magnitude 
     2. Log Magnitude and Phase 
     3. Phase 
     4. Linear Magnitude And Phase 
     5. SWR 
     6. Group Delay 
     App. 70 shows an example of a Limit Frequency (LF) menu. 
     e. Limits Key 
     Pressing this key calls the appropriate Limit (Ln) menu. The Limit menus are described in Apps. 71-81. 
     ERROR AND STATUS MESSAGES 
     Error and status messages are described below. 
     Message Types 
     The basic types of messages are categorized by the first digit of their code number, as follows: 
     
         ______________________________________                                    
000-099        Power Up Self Diagnostic                                   
               000-091 Main #2                                            
               020-039 Main #1                                            
               040-049 IO Proc                                            
               050-099 Analog Hardware                                    
100-199        System Status                                              
               100-109 Program Load                                       
110-129        Program Initialization                                     
               130-149 Disk Related                                       
               150-169 Peripheral Related                                 
200-299        Front Panel Operations                                     
300-399        Measurement Related                                        
400-499        GPIB                                                       
______________________________________                                    
 
    
     Fatal Errors 
     Some errors are &#34;fatal&#34; in that they cause the instrument to terminate operations until you correct the condition causing the error. These errors are listed with an &#34;X&#34; in the column below marked &#34;FATAL&#34;. 
     Message Definitions 
     Table 3-3 provides a listing of error message definitions. 
     
                                           TABLE 3.3                               
__________________________________________________________________________
Error Codes and Status Messages                                           
Code                                                                      
   Message Text           Fatal                                           
                             Meaning                                      
__________________________________________________________________________
Self Test, Main Microprocessor #2                                         
000                                                                       
   FiFO RESET FAILURE     X  FiFO failed to reset, PCB A12                
002                                                                       
   PROM CHECKSUM FAILURE #2                                               
                          X  Prom failure, PCB A12                        
003                                                                       
   BATTERY BACKED RAM FAILURE                                             
                          X  Non volatile RAM failure, PCB A12            
004                                                                       
   EXTENDED MEMORY FAILURE                                                
                          X  Failure in the extended memory, PCB A12      
005                                                                       
   DYNAMIC RAM FAILURE #2 X  Dynamic RAM failure, PCB A12                 
006                                                                       
   TIMER FAILURE #2       X  Programmable timer failure, PCB A12          
007                                                                       
   INTERRUPT CONTROLLER FAILURE #2                                        
                          X  Interrupt controller failure PCB A12         
008                                                                       
   NUMERIC PROCESSOR FAILURE #2                                           
                          X  Numeric processor failure, PCB A12           
009                                                                       
   FRONT PANEL INTERFACE FAILURE                                          
                          X  Interface failure, front panel, PCB A12      
010                                                                       
   PRINTER INTERFACE FAILURE                                              
                          X  Printer interface failure PCB A12            
Self Test, Main Microprocessor #1                                         
020                                                                       
   FiFO TO #2 FAILED RESET                                                
                          X  Interface failure to FiFO, PCB A12           
022                                                                       
   FiFO TO I/O FAILED RESET  Interface failure with FiFO PCB A13          
023                                                                       
   PTOM VHRVKDUM GSILUTR 31                                               
                          X  Checksum error, PROM PCB A13                 
024                                                                       
   DYNAMIC RAM FAILURE #1 X  Dynamic RAM failure, PCB A13                 
025                                                                       
   TIMER FAILURE #1       X  Programmable timer failure PCB A13           
026                                                                       
   INTERRUPT CONTROLLER FAILURE 31                                        
                          X  Interrupt Controller failure PCB A13         
027                                                                       
   DISK DRIVE CONTROLLER FAILURE                                          
                             Disk drive controller failure, PCB A13       
028                                                                       
   DISK DRIVE FAILURE        Disk drive SEEK failure, PCB A13             
029                                                                       
   NUMERIC PROCESSOR FAILURE 31                                           
                          X  8087 math coprocessor failure                
030                                                                       
   PROM CARTRIDGE CHECKSUM   PROM cartridge failure PCB A13               
   ERROR                                                                  
031                                                                       
   DISK DRIVE NOT READY FOR TEST                                          
                             Diskette is not in disk drive                
040                                                                       
   PROM CHECKSUM FAILURE I/O                                              
                          X  PROM failure, PCB A11                        
041                                                                       
   RAM FAILURE I/O        X  RAM failure, PCB A11                         
042                                                                       
   TIMER/INTERRUPT LOOPBACK                                               
                          X  Programmable timer failure, PCB A11          
   FAILURE                                                                
043                                                                       
   GPIB INTERFACE FAILURE I/O                                             
                             GPIB failure, PCB A11                        
044                                                                       
   FiFO FAILURE I/O       X  FiFO failure, PCB A11                        
050                                                                       
   A1 COMMUNICATIONS FAILURE L01 Phase Lock PCB Error                     
051                                                                       
   A2 COMMUNICATIONS FAILURE L02 Phase Lock PCB Failure                   
052                                                                       
   A3 COMMUNICATIONS FAILURE Cal/Third Local Oscillator PCB Failure       
053                                                                       
   A4 COMMUNICATIONS FAILURE Analog to digital PCB failure                
054                                                                       
   A5 COMMUNICATIONS FAILURE 10 MHz Reference PDB failure                 
055                                                                       
   A6 COMMUNICATIONS FAILURE Source lock PCB failure                      
056                                                                       
   A10 COMMUNICATIONS FAILURE                                             
                             Bandswitch blank/sync PCB                    
057                                                                       
   8 BIT A/D CONVERTER FAILURE                                            
                             Failure A/D PCB A4                           
058                                                                       
   STEERING DAC FAILURE      Failure A/D PCB A4                           
059                                                                       
   12 BIT A/D OR STEERING DAC                                             
                             Failure A/D PCB A4                           
   FAILURE                                                                
060                                                                       
   TEST SET NOT CONNECTED OR NOT                                          
                             General failure of test set                  
   WORKING                                                                
061                                                                       
   TEST SET CHAN A CAL PHASING                                            
                             Test set CHAN A failure                      
   FAILURE                                                                
062                                                                       
   TEST SET CHAN A CAL LEVEL Test set CHAN A failure                      
   FAILURE                                                                
063                                                                       
   TEST SET CHAN A GAIN FAILURE                                           
                             Test set CHAN A failure                      
064                                                                       
   TEST SET CHAN A PHASE RANGING                                          
                             Test set CHAN a failure                      
   FAILURE                                                                
065                                                                       
   TEST SET CHAN B CAL PHASING                                            
                             Test set CHAN B failure                      
   FAILURE                                                                
067                                                                       
   TEST SET CHAN B GAIN FAILURE                                           
                             Test set CHAN B failure                      
068                                                                       
   TEST SET CHAN B PHASE RANGING                                          
                             Test set CHAN B failure                      
   FAILURE                                                                
069                                                                       
   TEST SET REF CHAN CAL PHASING                                          
                             Test set REF CHAN failure                    
   FAILURE                                                                
070                                                                       
   TEST SET REF CHAN CAL LEVEL                                            
                             Test set REF CHAN failure                    
   FAILURE                                                                
071                                                                       
   TEST SET REF CHAN GAIN FAILURE                                         
                             Test set REF CHAN failure                    
072                                                                       
   TEST SET REF CHAN PHASE   Test set REF CHAN failure                    
   RANGING                                                                
System Status, Program Load                                               
100                                                                       
   DISK DRIVE NOT READY   X  Program failed to load from disk, (disk      
                             installed)                                   
101                                                                       
   PROGRAM DATA ERROR     X  Program failed to load from disk             
102                                                                       
   PROGRAM FILE MISSING      Loader cold not find system files            
103                                                                       
   DISK ERROR                The 360 is unable to read the diskette       
104                                                                       
   UNKNOWN DISK ERROR        loader failed a consistency check            
105                                                                       
   PROGRAM DATA ERROR ON #2  Program for processor #2 failed to load      
Program initialization                                                    
110                                                                       
   SOURCE ID FAILURE         No sweeper ID on GPIB; sweeper may not       
                             be connected                                 
111                                                                       
   TEST SET NOT CONNECTED X  Initialization detects a discrepancy.        
112                                                                       
   TEST SET FREQ. RANGE DOES NOT                                          
                          X  Initialization detects a discrepancy.        
   MATCH SOURCE                                                           
113                                                                       
   CAL DATA NOT FOUND; CHANGE                                             
                             File not found on disk with name matching    
   DISK AND PRESS &lt;ENTER&gt;    that in battery pack RAM.                    
114                                                                       
   PROGRAM ERROR          X  Program corrupted.                           
115                                                                       
   PROCESSOR COM ERROR    X  FiFO Synch problem Disk Related.             
Program Initialization, Disk Related                                      
131                                                                       
   DISK READ ERROR           Hard error reading from disk                 
132                                                                       
   DISK WRITE ERROR          Hard error writing to disk                   
133                                                                       
   FILE DELETION ERROR       Write protect tab is in &#34;read                
                             only&#34; position.                              
134                                                                       
   DISK NOT READY            Disk not in unit or not formatted            
135                                                                       
   DISK WRITE PROTECTED      Write protect tab is in &#34;read                
                             only&#34; position.                              
136                                                                       
   OUT OF DISK SPACE         Disk file space full                         
137                                                                       
   FILE IS INCOMPATIBLE      File is not a 360 data or program file.      
138                                                                       
   NO SPACE FOR NEW DATA FILE                                             
                             Disk file space full                         
139                                                                       
   FILE MARKED READ ONLY     Read-only attribute set on file              
140                                                                       
   NO FILES REMAIN TO OVERWRITE                                           
                             All files of the type have been deleted.     
141                                                                       
   NO FILES REMAIN TO DELETE All files of the type have been deleted.     
Program initialization, Peripheral                                        
170                                                                       
   PRINTER NOT READY         Printor &#34;off line&#34; or not connected.         
171                                                                       
   PLOTTER NOT READY         Plotter &#34;off line&#34; or not connected.         
Control Panel                                                             
200                                                                       
   SELECTED FREQUENCY OUT OF CAL                                          
                             Cal. range does not include selected         
   RANGE                     frequency.                                   
201                                                                       
   MARKERS SELECTED FOR READOUT                                           
                             Sweep range doos not include selected        
   NOT ACTIVE IN SWEEP RANGE frequency.                                   
208                                                                       
   OUT OF RANGE              Attempted to enter an out-of-range           
                             parameter.                                   
209                                                                       
   START GREATER THAN STOP   Attempted to a set start frequency that      
                             was greater than the stop frequency.         
210                                                                       
   OUT OF RANGE .20 PERCENT MAX                                           
                             Attempted to enter a smoothing or group      
                             delay factor that was greater than 20%.      
213                                                                       
   OUT OF H/W RANGE          Attempted to enter a frequency that is       
                             outside of the system hardware range.        
216                                                                       
   TOO MANY POINTS.          Attempted to set too many discrete           
                             frequency points.                            
217                                                                       
   TOO FEW POINTS 2 MINIMUM  Attempted to set too few discrete            
                             frequency points.                            
219                                                                       
   DISCRETE FREQS LOST       Setup changed in N-discrete frequency        
                             mode.                                        
220                                                                       
   OUT OF SWEEP RANGE        Attempt to set marker outside sweep range.   
221                                                                       
   OPTION NOT INSTALLED      The selected option is not installed.        
222                                                                       
   MEAS. DATA NOT AVAILABLE FOR                                           
                             No measured data on channel to be stored.    
   STORAGE                                                                
223                                                                       
   NO STORED MEMORY DATA     No data available in memory for channel.     
224                                                                       
   SYSTEM BUS ADDRESSES MUST BE                                           
                             Attempt to set GPIB addresses to same        
   UNIQUE                    value.                                       
225                                                                       
   MEMORY LOCATION           no cal exists.                               
226                                                                       
   MEMORY LOCATION CORRUPTED Saved state data is invalid                  
227                                                                       
   SETUP INCONSISTENT RECALL Saved state not compatible with hardware     
   ABORTED                   or software versions                         
228                                                                       
   WINDOW TOO SMALL          Attempt to set start greater than or equal   
                             to stop.                                     
229                                                                       
   OUT OF WINDOW RANGE       Attempt to set marker outside start to stop  
                             range.                                       
230                                                                       
   ATTENUATOR UNAVAILABLE    Selected attenuators not available in test   
                             set                                          
231                                                                       
   START MUST BE LESS THAN STOP                                           
                             Attempt to set start greater than or equal   
                             to stop in marker sweep.                     
232                                                                       
   ILLEGAL IN C.W. MODE      Attempt to readout limit frequency.          
233                                                                       
   ILLEGAL IN TIME DOMAIN    Attempt to readout limit frequency.          
234                                                                       
   BOTH LIMITS MUST BE ON    Attempt to readout limit frequency.          
235                                                                       
   Stop is Over Range        Discrete fill parameters cause stop to go    
                             over hardware range.                         
238                                                                       
   Out of Range 10% Minimum  Attempt to set pen speed to below 10%        
270                                                                       
   UNCALIBRATED              Channel has S parameter for which            
                             calibration does not exist.                  
271                                                                       
   PRINTER NOT READY         Printer not connected or paper out.          
272                                                                       
   TOO MUCH PRINT DATA       Print buffer is full. Reduce number of       
                             channels or data points.                     
273                                                                       
   PLOTTER NOT READY         Plotter not connected.                       
280                                                                       
   CAL INVALID               Calibration is incorrect for S parameter     
                             displayed.                                   
281                                                                       
   TIME DOMAIN INVALID       time domain cannot be used in current        
                             setup                                        
282                                                                       
   GROUP DELAY INVALID       Group delay cannot be used in current        
                             setup                                        
283                                                                       
   GATE MUST BE ON           Time Gate with gate off.                     
284                                                                       
   SMOOTHING INVALID         Attempt to use smoothing while in C.W.       
                             mode                                         
285                                                                       
   MEMORY DATA INVALID       Setup has changed since data was stored.     
Measurement Related                                                       
300                                                                       
   LOW IF                    Insufficient signal level detected: Device   
                             under test may not be connected.             
301                                                                       
   LOCK FAILURE              RF source failed to lock to reference        
                             oscillator in 360 testset                    
302                                                                       
   A/D FAILURE               Analog to Digital convertor not functioning  
                             in 360 mainframe.                            
303                                                                       
   RF OVERLOAD               Test signal level is too high: reduce        
                             source                                       
                             level or add attenuation                     
310                                                                       
   SWPR ID FAILURE           Communications lost with RF source           
311                                                                       
   SWPR SELF TEST FAILURE    RF source failed power on self test program  
312                                                                       
   NO TEST SET               Test set not connected. Reconnect and        
                             cycle power to clear                         
GPIB Related                                                              
400                                                                       
   GPIB ERROR                Data transmission error on                   
__________________________________________________________________________
                             GPIB                                         
 
    
     Ill CONTROL PANEL OPERATION 
     DATA DISPLAYS 
     Display Modes and Examples 
     The 360 displays measurement data using a &#34;Channel Concept.&#34; This means that each channel can display a different S-Parameter and a different graph type for each S-Parameter. As you select each channel the graph type, scaling, reference delay, S-Parameter, etc. associated with that channel appear on the screen. You can display the same S-Parameter on two or more channels. FIG. 59 shows the possible displays. 
     Graph Types 
     Several graph-types are possible: polar, rectilinear, Smith chart. The rectilinear graph-type may be magnitude, phase, magnitude and phase, SWR, group delay, real, imaginary, and real and imaginary. The Smith chart graph-type is one specifically designed to plot complex impedances. 
     a. Graph Data Types 
     The data types (real, imaginary, magnitude, phase) used in the displayed graph-types reflect the possible ways in which S-Parameter data can be represented in polar, Smith, or rectilinear graphs. For example: Complex data--that is, data in which both phase and magnitude are graphed--may be represented and displayed in any of the ways shown below. 
     1. Complex Impedance; displayed on a Smith chart graph. 
     2. Real and imaginary; displayed on a real and imaginary graph. 
     3. Phase and magnitude components; displayed on a rectilinear (Cartesian) or polar graph. 
     In addition to the above, the 360 can display the data as a group delay plot. In this graph-type, the group-delay measurement units are time. Those of the associated aperture are frequency and SWR. The quantity group delay is displayed using a modified rectilinear-magnitude format. In this format the vertical scale is in linear units of time (ps-ns-μs). With one exception, the reference-value and -line functions operate the same as they do with a normal magnitude display. The exception is that they appear in units of time instead of magnitude. 
     Frequency Markers 
     a. Marker Annotation 
     FIG. 60 shows how the 360 annotates markers for the different graph-types. Each marker is identified with its own number. When a marker reaches the top of its graticule, it will flip over and its number will appear below the symbol. 
     When markers approach the same frequency, they will overlap. Their number will appear as close to the marker as possible without overlapping. 
     Details of the method whereby the system 100 of the present invention displays Markers is described in the commonly assigned co-pending Patent Application entitled, METHOD OF DISPLAYING GRAPHS WITH MARKERS, invented by David Peter Finch, having Ser. No. 176,097, filed on the same date as this Application and which is hereby expressly incorporated herein in its entirety. 
     b. Marker Designation 
     Depending on menu selection, you may designate a marker as the &#34;active&#34; or the &#34;delta reference&#34; marker. If you choose a marker to be active--indicated by its number being enclosed in a square box--you may change its frequency or time (distance) with the Data Entry keypad or knob. If you have chosen it to be the delta-reference marker, a delta symbol (Δ) appears one character space above the marker number (or one character space below a &#34;flipped&#34; marker). If the marker is both active and the delta reference marker, the number and the delta symbol appear above (below) the marker. The delta symbol appears above (below) the number. 
     Limits 
     Limit lines function as settable maximum and minimum indicators for the value of displayed data. These lines are settable in the basic units of the measurement on a channel-by-channel basis. If the display is rescaled the limit line(s) will move automatically and thereby maintain their correct value(s). 
     Each channel has two limit lines (four for dual displays), each of which may take on any value. Limit lines are either horizontal lines in rectilinear displays or concentric circles around the origin in Smith and polar displays. 
     Status Display 
     In addition to the graticules, data, markers, and marker annotation, the 360 displays certain instrument status information in the data display area. This information is described below. 
     a. Reference Position Marker 
     The Reference Position Marker indicates the location of the reference value. It is displayed at the left edge of each rectilinear graph-type. It consists of a triangular symbol identical to the cursor displayed in the menu area. You can center this symbol on one of the vertical graticule divisions and move it up or down using the &#34;Reference Position&#34; option. When you do this, the data trace will follow this marker. If you also select the value option; the marker will remain stationary and the trace will move with the maximum allowable resolution. When changing from a full-screen display or a half- or quarter-screen display, the marker will stay as close to the same position as possible. 
     b. Analog Instrument Status 
     The 360 displays analog-instrument-status messages (in red when appropriate) in the upper right corner of the data-display area (FIG. 61). They appear in the same vertical position as line 2 of the menu area. If more than one message appears, they stack up below that line. 
     c. Measurement Status 
     The 360 displays measurement-status messages (in red when appropriate) in the upper-right corner of the graticule (channel) to which they apply. 
     d. Sweep Indicator Marker 
     The sweep indicator marker (FIG. 61) indicates the progress of the current sweep. When measuring quiet data, that is, data having few or no perturbations, this indicator assures that the instrument is indeed sweeping. 
     The indicator, a blue horizontal line segment 15 pixels long by 1 pixel high, appears along the bottom edge of the data display area. Its position is proportional to the number of data points measured in the current sweep. When this sweep completes, the line segment will have traveled the full width of the data display area. If the sweep should stop for some reason, the position of the indicator will stop changing until the sweep resumes. 
     Data Display Control 
     a. Active Channel Selection 
     FIG. 62 shows the algorithm that the 360 uses to display the active channel. 
     b. S-Parameter Selection 
     If you select a new S-Parameter using Menu SP (App. 30), it appears on the then-active channel in the same graph-type it was displayed-in last. Table 3-4 shows the displayable S-Parameters based on the type of correction you have in place. If you attempt to display other S-Parameters, an error message displays. In cases when there is no last-displayed S-Parameter stored, the display will default as shown in the Table 3-4. 
     
                       TABLE 3-4                                                   
______________________________________                                    
Display Types and Defaults by Correction Type                             
           Displayable                                                    
                     Defaults                                             
Correction Type                                                           
             S-Parameters                                                 
                         CH1    CH2  CH3  CH4                             
______________________________________                                    
None         All         S.sub.11                                         
                                S.sub.12                                  
                                     S.sub.21                             
                                          S.sub.22                        
Frequency Response                                                        
Transmission S21         S.sub.21                                         
                                S.sub.21                                  
                                     S.sub.21                             
                                          S.sub.21                        
Reflection   Sl1         S.sub.11                                         
                                S.sub.11                                  
                                     S.sub.11                             
                                          S.sub.11                        
Both         S11, S21    S.sub.11                                         
                                S.sub.11                                  
                                     S.sub.21                             
                                          S.sub.21                        
1-Path, 2-Port                                                            
             S11, S21    S.sub.11                                         
                                S.sub.11                                  
                                     S.sub.21                             
                                          S.sub.21                        
12-Term      All         S.sub.11                                         
                                S.sub.12                                  
                                     S.sub.21                             
                                          S.sub.22                        
Reflection Only                                                           
             S11         S.sub.11                                         
                                S.sub.11                                  
                                     S.sub.11                             
                                          S.sub.11                        
______________________________________                                    
 
    
     If you select an S-Parameter (or if the 360 defaults to one) for which there was no last-displayed graph-type, the display defaults to the following: 
     S 21  and S 12  : Log Magnitude and Phase 
     S 11  and S 22  : Smith. 
     c. Data Display Update 
     When you change a control panel parameter that affects the appearance of the display, the entire display changes immediately to reflect that change. For example, if you press AUTOSCALE, the entire display rescales immediately. You do not have to wait for the next sweep to see the results of the change. 
     The following parameters are supported for this feature: 
     Reference Delay, Offset, Scaling, Auto Scale, Auto 
     Reference Delay, Trace Math, IF BW, and Smoothing. 
     In the case of Averaging, the sweep restarts. 
     If the knob is used to vary any of the above parameters, the change occurs as the measurement progresses, that is, the continuing trace will reflect the new setting(s). 
     When you change a marker frequency or time (distance), the readout parameters will change. The changes reflect by the marker&#39;s new frequency, using data stored from the previous sweep. 
     d. Display of Markers 
     Once you have selected a marker display, it will always display. It does not matter what resolution you have selected. When you set a marker to another calibrated frequency the marker will continue to display. It will display even if its frequency is not consistent with the data points in the lower-resolution sweep. 
     e. Hard Copy and Disk Output 
     In addition to the CRT display, the Model 360 is capable of outputting measured data as a 
     1. tabular printout, 
     2. screen-image printout, 
     3. pen plot, 
     4. disk image of the tabular data values. 
     The selection and initiation of this output is controlled by the OUTPUT keys. 
     f. Tabular Printout 
     The tabular printout formats are used as follows: 
     1. Tabular Printout Format (FIG. 63): Used when printing three or four channels. 
     2. Alternate Data Format (FIG. 64): Used when printing one or two channels. 
     In tabular printouts, the 360 shifts the data columns to the left when an S-Parameter is omitted. Leading zeroes are always suppressed. The heading (Model, Device I/O, Date, Operator, Page) appears on each page. When using the 360SS sweeper, frequencies are in the format &#34;XX.XXXX.&#34; 
     g. Screen-Image Printout 
     In a ScreenImage Printout, the exact data displayed on the screen is dumped to the printer. The dump is in the graphics mode, on a pixel by pixel basis. A header (FIG. 65) prints before the screen data prints. 
     h. Plotter Output 
     The protocol used to control plotters is &#34;HP-GL&#34; (Hewlett-Packard Graphics Language). HP-GL contains a comprehensive set of &#34;vector graphics&#34; type commands. These commands are explained in the Interfacing and Programming Manual for any current model Hewlett-Packard plotter, such as the 7470A. 
     When the plotter is selected as the output device, it is capable of drawing the graph shown on the screen or of drawing only the data traces(s), so that multiple traces may be drawn on a single sheet of paper (in different colors, if needed). 
     i. Disk Output 
     The 360 can write-to or read-from the disk all measured data. This data is stored as an ASCII file in the exact same format as shown in FIG. 63. If read back from the disk, the data is output to the printer. There, it prints as tabular data. 
     TRANSMISSION AND REFLECTION MEASUREMENTS 
     This discussion provides information on general measurement considerations and transmission and reflection measurements using the 360. 
     Setup and Calibration Procedures and 
     Measurement Options 
     To get started, apply power to the system. Do this by first turning on the 360SS45 or 69 source power switch then the analyzer power--the order is important. If the analyzer is powered-up before the source, it will not be able to find the source and will probably fail the self test. 
     After turning on the power, allow the system to warm up for at least 30 minutes before operation. 
     In normal operation, the system comes on line in the state that it was in when last turned off. If you want to return the system to its default state, you can do so by pressing the DEFAULT PROGRAM key twice. 
     The default parameters provide a known starting point. For example, they reset the start and stop frequencies to their maximum values, the source power to 0 dBm, and the display resolution to 501 data points. 
     The Sweep Setup menu should now appear on the display (it also can be displayed using the SETUP MENU key). If you like, you can select a new start frequency, stop frequency, or source power. 
     The actual power level at Port 1 is about 10 dB less than the source power level due to internal losses in the test set. 
     If the 360 system has an Active Device Test Set, you can further reduce the power level at Ports 1 and 2 with the built-in attenuators. Using the Reduced Test Signals option in the Sweep Setup menu, you can change the settings of the Ports 1 and 2 source attenuators over a range of from 0 to 70 dB. The Port 2 test attenuator has a range of from 0 to 40 dB (in 10 dB steps). Selecting the BEGIN CAL key starts the calibration process. The Calibration menus step you through the calibration process, as follows: 
     1. Select the type of calibration desired. 
     2. Select the frequency range of calibration. You can choose the normal 501 points, CW (one point), or N-discrete frequencies (from 2 to 501 points). 
     3. Install the calibration kit devices to the test ports as instructed by the menu. Both the capacitance coefficients for the open and the offset lengths for the Open and Short can be modified or defined. 
     When the calibration is completed, you can store the calibration data in the internal memory or on a disk. You are now ready to install the test device and proceed with the measurement. At this point you have a number of measurement options to consider such as displays, markers, limits, outputs, sweeps, and enhancements. 
     You can select any of the available graph types and display them for any calibrated parameter on any of the four channels. 
     Up to six markers are available. Using the Marker Menu, you can set the frequency of each one, you can set each one in the delta marker mode, and you can set each marker&#39;s level to maximum or minimum. 
     In some cases--such as in a production environment --limit lines are desirable. Options within the menu called up using the LIMITS key, provide two limit lines for each channel. These limit lines function with all of the graph types, including Smith and admittance. The color of the limit lines differs from that of the measurement trace. This allows for easy analysis of results. 
     The Output Menu (App. 42) gives you a choice between a printer and a colored-pen plotter. It also lets you choose the data-output type, output head, and disk output functions. 
     To output the display, press the START PRINT key. The default setting provides for a full display printout from the associated printer. 
     To label the output, select Setup Output Headers in the Output Menu or press the DEVICE ID key. 
     On the output to the printer, plotter, or disk, a menu then appears that lets you specify the device name/serial number, the dater and the operator&#39;s name (App. 43). 
     Sweep frequencies can be changed with the calibration applied as long as the frequencies are between the calibration start and stop frequencies. 
     Additionally, a marker sweep can be selected from the Setup Menu. This allows you to sweep between any two active markers so long as the frequency of each falls between the calibrated start and stop frequencies. 
     Using the DATA POINTS key, you can select the number of data points for optimal resolution-vs-speed. 
     Finally, you can enhance the measurement data by reducing the IF bandwidth and using averaging and/or smoothing (see below). 
     Change the IF bandwidth by selecting the IF BW key. 
     Set the averaging and smoothing values by selecting the AVG/SMOOTH MENU key. 
     Turn on the averaging and smoothing using the TRACE SMOOTH and AVERAGE keys, which have LED&#39;s to let you know that the enhancement is being applied. 
     Transmission and Reflection Measurements 
     Before going any further, let us take a few moments to review some basic principles of network measurements. First, we apply incident energy to the input of a test device. If the device&#39;s input impedance differs from the measurement system&#39;s impedance, some of that energy is reflected. The remainder is transmitted through the device. We call the ratio of reflected-to-incident energy the reflection coefficient. The ratio of transmitted-to-incident energy we call the transmission coefficient. 
     These ratios are complex quantities that have magnitude and phase components. Using vector representation, the vector magnitude is the ratio of reflected-to-incident magnitude (or transmitted-to-incident magnitude), while the vector phase is the difference in phase between the incident energy and the reflected/transmitted energy. 
     The measurement reference for the incident energy is the point at which the device connects to the measurement system. We call this point the reference plane. The incident energy at the reference plane is defined as having a magnitude of 1 and a phase of 0 degrees. We establish this during the calibration. 
     The ratio of reflected and transmitted energy to the incident energy can be represented by a number of different measurements and units, as shown below. 
     The default display for reflection measurements is the Smith chart. The default display for transmission measurements is the Log Magnitude and Phase graph. 
     The Smith chart is a convenient way to display device impedance and is a useful aid for the graphical design and analysis of microwave circuits. 
     Let us assume both that our system is already calibrated and that we have equalized the system for the test port in use. We would then 
     1. Connect the Short. A Short always appears as a dot at the left-most edge of the Smith chart&#39;s horizontal axis. 
     2. Connect a Termination. Now you will see another dot located at the center (1+j0) of the chart (this assumes a 50-ohm load). 
     3. Connect the Open. An Open appears as an arc on the chart&#39;s right edge. This is due to the fringing capacitance of the Open standard. 
     Now let us perform a reflection measurement on a 20 dB attenuator over the 1-to-18 Ghz range. 
     We need to determine the setup, calibration, and measurement requirements. 
     A known good starting point is to reset with DEFAULT PARAMETERS. Since our measurement lies between 1 and 18 GHz, set the Start and Stop frequencies using the Sweep Setup menu that appears on the display following system reset. 
     Let us perform a simple calibration, REFLECTION ONLY, which uses an open, a shorts and a broadband load. To do this, press the BEGIN CAL key and follow the directions in the menu area. 
     When you complete the calibration, the &#34;CHANNEL 1 WITH S11&#34; Smith chart appears on the display. Now 
     1. Select the Log Magnitude display and install the attenuator. 
     2. Select AUTOSCALE to optimize the display data. 
     3. Use Markers 1 and 2 to find the maximum and minimum impedance. 
     Now let us perform transmission measurement on the same 20 dB attenuator over the same frequency range. We will follow the same steps as before, but this time we will use additional features. 
     Once again, reset the system using the DEFAULT PARAMETERS key. 
     In this calibration we will select the N-Discrete Frequencies menu option and step all frequencies in increments of 50 MHz. 
     When the calibration is complete, Channel 1 will display &#34;S12 FORWARD TRANSMISSION WITH LOG MAGNITUDE AND PHASE.&#34; You can use Markers 1 and 2 to find the maximum and minimum values of the attenuators insertion loss. 
     LOW LEVEL AND GAIN MEASUREMENTS 
     This discussion provides methods and techniques for making gain and low-signal-level measurements. It is divided into 360 system considerations and test device considerations. 
     360 System Considerations 
     The 360 system is limited in its ability to test low-signal levels by its dynamic range and low signal-to-noise-power ratio. First we will discuss dynamic range, which is the difference between the maximum and minimum acceptable signal levels (Dynamic Range=P max  -P min ). 
     a. Dynamic Range 
     The dynamic range of the 360 is limited by the 0.1 dB compression level of the samplers at high signal levels. It is further limited at low signal levels by leakage signals and noise. 
     The 0.1 dB compression level is on the order of -10 dBm. The 360 is designed such that all other conversions compress at a much greater level, which leaves the samplers as the main source of non-linearity. 
     The small signal response is limited by errors due to noise and leakage signals. The leakage signals are both from within the 360 and at the device-under-test (DUT) connectors. 
     The detected signal is the vector sum of the desired signals, the noise signals, and the leakage signals. These signals introduce an error or uncertainty. 
     Some of the possible leakage paths for the 360 are the transfer switch, the frequency conversion module, and leakage from the DUT. The system limits these leakages to greater than 100 dB. The 12-term error correction can reduce this leakage to better than 110 dB at 18 Ghz and 90 dB at 40 Ghz. 
     The DUT connectors should have internally captivated center pins. Those connectors which use external pins to captivate the center conductor should have silver loaded epoxy on the pins to reduce radiation to better than 90 dB. 
     b. Signal-to-Noise Power Ratio 
     The signal-to-noise power ratio for each of the test or reference channels is given by the formula 
     S/N Ratio=Signal Power/Noise Power 
     Where &#34;signal power&#34; is the power level of the 83.33 Khz IF signal at the internal synchronous detectors, and &#34;noise power&#34; is total power contained within the bandwidth of the bandpass filter at 83.33 Khz. 
     The uncertainty, or error, in a measurement is a function of the amplitude of leakage signals and of the noise level. The uncertainty in the measurement of magnitude and phase of the S-parameters are calculable. 
     The most difficult types of measurements 1 are those that exercise the full dynamic range of the 360, such as filters. Filter measurements are examples of where one must observe both low-insertion loss (in the passband) and high attenuation (in the stop band). 
     There are two techniques that you can use to optimize the signal-to-noise ratio. They are (1) maximizing the RF signal level and (2) using signal enhancement. 
     To maximize the RF signal level, use the default settings of the signal source. The 360SS45 or 69 defaults to 0 dBm--a power level that both maximizes dynamic range and optimizes linearity. 
     The 360 provides two enhancements for improving the signal-to-noise ratio: IF bandwidth reduction and averaging. 
     Reducing the IF bandwidth is a primary method for enhancing accuracy. The 360 has a choice of three bandwidths available from the front panel: Normal (10 Khz), Reduced (1 Khz), and Minimum (100 Hz). The noise level should decrease by a factor equal to the square root of the IF bandwidth. Using IF Bandwidth reduction makes for faster measurements than with the use of an equivalent amount of averaging. 
     Averaging is another way to improve accuracy. The improvement is proportional to the square root of the number of averages. The improvement from averaging, however, comes at the expense of increased sweep time. 
     EXAMPLE 
     Using 1 Khz BW reduction and 10 averages, you would increase the signal-to-noise ratio by 7.6 dB but would lengthen the time required for the measurement by a factor of 4.3. This example assumes a constant signal power. 
     Test Device (DUT) Considerations 
     In order to test a device, the required input RF level and the expected device output RF level must be determined. 
     The RF level at Port 1 must be set for the device input RF power level required. The power level at Port 1 is about 10 dB less than the RF source power level, or about -10 dBm. Attenuation can be added in steps of 10 dB up to 70 dB using the built-in source attenuator in the Models 3620 and 3621 Active Device Test Sets available for WILTRON Co. 
     The RF level into Port 2 should be kept to -10 dBm or less to ensure optimum linearity and to protect internal components from damage. The never-to-exceed RF level into either Port 1 or Port 2 is +20 dBm. You can add up to 40 dB of attenuation (in 10 dB steps) into Port 2 using the built-in test attenuator in the active device test sets. 
     If you are using a test set that does not have built-in attenuators, you should use external attenuators on Port 1 and Port 2 as needed. However, the use of external attenuators invalidates input and output match measurements; whereas, the built-in attenuators are compensated by the calibration and do not affect reflection measurements. 
     Before calibration, ensure that the test setup is correct by setting the power level and adding attenuation as needed. 
     The 360 uses enhancements in the calibration to ensure a wide dynamic range. It automatically selects the REDUCED IF bandwidth front panel setting and varies the number of averages with the calibration device. Terminations require the most averages. 
     If desired, the IF bandwidth and number of averages can be specified for the calibration measurements. Using 100 average (AVG=100) appears to be sufficient for most measurements. 
     To obtain the maximum performance from the 360 for measurements of attenuation, you can use the capability of the N discrete frequency calibration to spot check measurements in the frequency band of interest. 
     As described above, the measurement procedure is straightforward. 
     Wide Dynamic Range Device--Filter 
     Since you do both low-insertion-loss and high-attenuation measurements simultaneously, use the maximum RF signal level and no attenuation. Selecting the REDUCED IF BW setting and 100 averages will likely suffice for this kind of measurements. 
     High Gain Device--FET 
     This device has a typical 15 dB gain and requires an input level of about -30 dBm. Set the Port 1 Source Attenuator to 20 dB. Since the device RF output level is -15 dBm (-30 dBm 15 dB[gain]=-15 dBm) no attenuation is needed at Port 2. 
     Medium Power Device--Amplifier 
     Measure the small signal parameters of a 10 dB gain device that requires an input power level of -10 dBm. Here, Port 1 will have no attenuation. The device RF output level is -10 dBm. This level equals 0 dBm (-10 dBm+10 dB[gain]=0 dBm) into Port 2 and will cause compression in the measurement. At least 10 dB of test attenuation will be needed at Port 2, which will reduce the Port 2 RF level to -10 dB. 
     GROUP DELAY MEASUREMENTS 
     Group delay is the measure of transit time through a device at a particular frequency. Ideally, we want to measure a constant--or relatively constant--transit time over frequency. The top waveform shown in FIG. 66 is measured at one frequency. The bottom waveform is identical to the first, simply delayed in time. 
     Referring to FIG. 67, the first waveform shown is the original waveform. It is made up of many frequency components. After traveling through a device the signal is delayed in time. Some frequencies are delayed more than others and thus our waveform does not have exactly the same shape as before. 
     When delay is nonlinear, as shown above, distortion occurs. By measuring group delay with a network analyzer we can characterize the distortion that occurs from a signal traveling through our test device. 
     When designing components it is important to measure group delay so that you can compensate for any distortion caused by the component. 
     You may be able to tune the device so as to optimize the performance of group delay over the frequency range of interest. Outside of the specified frequency range, the group delay may or may not be linear. 
     So how is group delay measured? Signals travel too fast to enable measuring the input and output times of each frequency component. Consequently, we must use mathematical calculations to derive the group delay from the phase slope. 
     Group delay is mathematically represented by the following equations: ##EQU4## 
     What this equation shows is that group delay is a measure of the change in phase with relation to the change in frequency. 
     The change in frequency is referred to as an aperture. In other words, the aperture is equal to the frequency range divided by the number of data points. 
     To measure group delay the frequency aperture must be selected. Depending on the size of aperture, different levels of precision can result for the measurement of group delay. 
     A wide aperture results in a loss of fine-grain variations but gives more sensitivity in the measurement of time delay. A small aperture gives better frequency resolution, but at the cost of lost sensitivity. Thus, for any comparison of group delay data you must know the aperture used to make the measurement (FIG. 68). 
     Let us take a look at a group delay measurement made on the WILTRON 360 Vector Network Analyzer. Group delay, as a measurement option, can be found in the Graph Type menu. After selecting the option, the 360 displays the data in a time-vs-frequency graph, or to be more exact, a group-delay-vs-frequency graph. 
     The 360 automatically selects the frequency spacing between data points--that is, the aperture. This value is displayed on the screen with the measurement. 
     The aperture defaults to the smallest setting for the frequency range and number of data points selected. This value is displayed in the SET SCALING menu when measuring group delay. 
     Group delay applications are found throughout the microwave industry, although the majority of such measurements are made in the telecommunications area. 
     One occurrence of group delay that you may have experienced is with a long-distance telephone call. Occasionally a phone call can be disturbing because of the delay in time from when you speak and when the other person responds. If there is simply a delay, then time delay--or linear group delay--has occurred. But if the voices are also distorted, then non-linear group delay has occurred. It is this distortion that we must avoid. We can avoid linear group delay by measuring group delay both during the design and development stages and during recalibration in the field. 
     One final group-delay application is found in the development of components. In this application, group delay is measured for the transit time of a signal through the device. When time is of the essence in a fast switching system, as in a modern computer, the travel time through a device is critical. 
     ACTIVE DEVICE MEASUREMENTS 
     Active devices, for example FETs, amplifiers and MMICs, are key components in microwave systems. The microwave future is in smaller integrated microwave packages. 
     Actually, the measurements that are made are the same measurements made on passive devices. 
     Active devices come in many shapes and sizes. In most cases we are going to have to develop a fixture in which to mount the device, since it may have tabs, leads or pads instead of connectors. 
     Active devices require bias voltages, and in many cases they are easily damaged. High gain amplifiers may saturate with input signals of -50 dBm! With active devices, we have a new set of measurement requirements. 
     WILTRON has developed two models of active device test sets (Models 3620 and 3621) to help you make these types of measurements. These test sets include both step attenuators (70 dB) used to adjust operating power levels, and bias tees used to bias the device via the test port center conductor. This approach to bias is useful for testing transistors; however, MMIC&#39;s usually require bias injection at other points (FIG. 69). 
     Test fixtures are necessary for mounting the device so that it can be measured in a coaxial (or waveguide) measuring system. 
     Now we have an interesting situation. While we can measure the performance at the connector--which is the calibration plane--what we really want to know is how our device performs (FIG. 70). 
     You can consider the device embedded in the fixture and can measure the S-parameters of the fixture with the device installed. 
     The most elementary situation is a system in which the test fixture is electrically ideal or transparent. In this case the solution is simple--merely move the reference plane out to the device (FIG. 71). 
     In some cases--depending on the fixture or the device being measured--this is satisfactory. But when it is not, we need to employ other techniques. 
     One of the reasons that moving the reference plane out to the device does not always work, is that the test fixture includes a transition from coax to a structure such as microstrip, coplanar waveguide, or stripline (FIG. 72). 
     Engineers have come to grips with the general problem. However, there is no established standard approach. Two of the more common approaches are to calibrate the fixture as a part of the analyzer, and to characterize the fixture and compute the desired result. 
     In the discussion on calibration (above) we saw that the calibration components establish the reference plane and determine the quality of the measurement. If we have a good Open, Short and Z 0  load to place at the end of a microstrip line, we can calibrate the system at the point of measurement. 
     FIG. 73 shows some of the special test-fixture calibration standards that are available. 
     These special calibration kits are far from perfect. For instance opens are difficult due to radiation effects, and good terminations are hard to find, 20-30 db being often the best, terminations also determining &#34;effective directivity&#34;. Although the special calibration kits are far from perfect, they are superior to our perfect transmission line assumption. 
     You may have heard of the probe system built to permit on-wafer measurements by Cascade Microtech. It is a good example of a system conducive to this approach. 
     The Open, Short, termination approach provides three known standards that permit the analyzer to solve for three unknowns (FIG. 74). 
     CAUTION 
     You should turn off or disconnect the bias supplies during the calibration, since you are using a Short as the calibration standard. 
     It is also possible to use three known impedances. For instance, a varactor with three voltages applied (FIG. 75). 
     The second approach is to model the fixture. Modeling is elegant but of limited use due to the non-ideal characteristics of the fixture. Modeling can be accomplished in a CAD system like Touchstone or Compass. 
     In summary, there are quite a variety of approaches--all with their own characteristic pitfalls. Engineers try to choose the most appropriate technique for their application. 
     TIME DOMAIN MEASUREMENTS 
     Time Domain Measurements, Discussion 
     The Option 360-2 Time Domain feature for the WILTRON 360 analyzer is a useful measurement tool for determining the location of impedance discontinuities. Some typical applications are identifying and analyzing circuit elements, isolating and analyzing a desired response, locating faults in cables, and measuring antennas. 
     The relationship between the frequency-domain response and the time-domain response of a network is described mathematically by the Fourier transform. The 360 makes measurements in the frequency domain then calculates the inverse Fourier transform to give the time-domain response. The time-domain response is displayed as a function of time (or distance). This computational technique benefits from the wide dynamic range and the error correction of the frequency-domain data. Let us examine the time-domain capabilities of the Model 360 Vector Network Analyzer. Two measurement modes are available: lowpass and bandpass. 
     We use the lowpass mode with devices that have a dc or a low-frequency response. In the lowpass mode two responses to the device-undertest (DUT) are available: impulse or step stimulus. The frequencies used for the test must be harmonically related to the start frequency. 
     The lowpass impulse response displays the location of discontinuities as well as information useful in determining the impedance (R, L, or C) of each discontinuity. 
     The impulse response is a peak that goes positive for R&gt;Z 0  and negative for R&lt;Z O . The height of the response is equal to the reflection coefficient (rho=(R-Z 0 )/(R +Z 0 )). The impulse response for a shunt capacitance is a negative-then-positive peak and for a series inductance is a positive-then-negative peak (FIG. 76). 
     An example of using impulse response is circuit impedance analysis. With an impulse response, we can observe the circuit response of a passive device, such as a power splitter, and make final adjustments during the test (FIG. 77). The lowpass step response displays the location of discontinuities as well as information useful in determining the impedance (R, L, or C) of each discontinuity. If you are familiar with time-domain reflectometry (TDR) you may feel more comfortable with step response, as the displays are similar. 
     The lowpass step response for a resistive impedance is a positive level shift for R&gt;Z 0  and a negative level shift for R&lt;Z 0 . The height of the response is equal to the reflection coefficient (rho=(R-Z 0 )/(R+Z 0 )). The step response for a shunt capacitance is a negative peak, and for a series inductance it is a positive peak (FIG. 78). 
     An example of using the lowpass step response is cable fault location. In the frequency domain a cable with a fault exhibits much worse match than a good cable. Using lowpass step response, both the location of the discontinuity and information about its type are available (FIG. 79). 
     The 360 bandpass mode gives the response of the DUT to an RF-burst stimulus. Two types of response are available: impulse and phaser-impulse. An advantage of the bandpass mode is that any frequency range can be used. Use this mode with devices that do not have a dc or low-frequency path. 
     Use the bandpass-impulse response to show the location of a discontinuity in time or distance, as indicated by changes in its magnitude. Unlike the low-pass mode, no information as to the type of the discontinuity is available. A typical use for this mode is to measure devices--such as filters, waveguide, high-pass networks, bandpass networks--where a low-frequency response is not available. 
     The bandpass-impulse response for various impedance discontinuities is shown in FIG. 80. As we can see, no information about the type of discontinuity is available. 
     An example of using the bandpass-impulse response, is the pulse height, ringing, and pulse envelope of a bandpass filter. 
     Use the phaser-impulse response with bandpass response to determine the type of an isolated impedance discontinuity. 
     After the bandpass-impulse response has been isolated, the phaser-impulse response for a resistive-impedance-level change is a peak that goes positive (R&gt;Z 0 ) for the real part of S 11  and negative for R&lt;Z 0 . The imaginary part remains relatively constant. In each case the peak is proportional to the reflection coefficient. The phaser-impulse response for a shunt capacitance is a negative-going peak in the imaginary part of S 11 . For a series inductance, it is a positive going peak (FIG. 81). Next, let us look at a complex circuit comprising a resistor in series with an inductor shunted by a capacitor wherein R&lt;Z 0 . These impedance changes are shown in the time domain for the lowpass-impulse response, lowpass-step response, and bandpass-impulse response (FIG. 82). 
     The 360 processes bandpass-impulse-response data to obtain phasor-impulse response. This becomes most advantageous where both a reactive reflection and an impedance change occur at the same location. The real part of the time-domain response shows the location of impedance level changes, while the imaginary part shows the type of reactive discontinuity. Phasor-impulse response displays one discontinuity at a time (FIG. 83). 
     Details of the method and apparatus whereby the system 100 of the present invention produces Phasor-impulse response displays is described in the commonly assigned co-pending Patent Application entitled APPARATUS AND METHOD FOR LOW-PASS EQUIVALENT PROCESSING, S.N. 175,762, invented by Robert Huenemenn, filed on the same date as this Application and which is hereby expressly incorporated herein in its entirety. 
     a. Operating Time Domain 
     To operate in the time domain mode, press the DOMAIN key. A domain menu (App. 86) allows us to select the frequency- or time-domain modes by simple cursor selection. The 360 defaults to the frequency domain. 
     Select time or distance for the horizontal axis. The 360 defaults to time axis. 
     NOTE 
     If we select distance, be sure to set the dielectric constant in the reference Delay menu (App. 87). 
     Select SET RANGE and use the START/STOP or GATE/SPAN selections to set the range (App. 88). For the lowpass mode select either IMPULSE or STEP Response and set the DC term. The 360 defaults to the IMPULSE Response and the AUTO EXTRAPOLATE mode for the DC term (App. 89). 
     NOTE 
     The bandpass mode displays Bandpass Impulse Response unless we select Phasor Impulse Response. 
     The Marker Range menu allows us to zoom in and display the range between two selected markers (App. 90). 
     b. Windowing 
     Windowing is a frequency filter that we apply to the frequency-domain data when we convert it to time-domain data. This filtering rolls off the abrupt transition at F1 and F2. This effectively produces a time-domain response with lower sidelobes. Windowing allows a limited degree of control over the pulse shape, trading off ringing (sidelobes) for pulse width (FIG. 84). 
     We select windowing from the Time Domain Setup menu. Four different windows are available: RECTANGLE, NOMINAL, LOW SIDELOBE, and MINIMUM SIDELOBE. The RECTANGLE option provides the narrowest pulse width, while the MINIMUM SIDELOBE option provides the least ringing (fewest sidelobes). The 360 defaults to the NOMINAL option, which is acceptable for most measurements (App. 91). 
     c. Gating 
     Gating is a time filter that allows for removing unwanted time-domain responses by gating the desired response. We can view the isolated response in both the time domain--using the PHASOR IMPULSE RESPONSE option --and in the frequency domain--using the FREQUENCY WITH TIME GATE selection (FIG. 85). 
     There are four different gate shapes available: RECTANGLE, NOMINAL, LOW SIDELOBE, and MINIMUM SIDELOBE. The 360 defaults to the NOMINAL gate. To specify a different shape simply enter the Gate menu and select the desired gating shape. The RECTANGLE has the largest ripple, while MINIMUM SIDELOBE has the least (App. 92). 
     d. Gating Example 
     Let us look at a reflection measurement. A device at the end of a coax cable is measured in the frequency domain. We would like to measure the return loss of this device and characterize its impedance, but to do so we have to eliminate the response of the cable and connectors. Steps 1 thru 5 below describe a method for making this measurement. 
     1. Convert the frequency domain data into the time domain using TIME BANDPASS MODE. 
     2. Select SET GATE in the Domain menu then GATE DISP in the Gate menu. This allows us to put the gate around the discontinuity of interest using the START, STOP, or CENTER/SPAN selections. 
     3. Select GATE ON in the Gate menu and the unwanted responses are removed. 
     4. Select PHASER IMPULSE ON in the Bandpass menu. The real and imaginary responses of the Phaser Impulse Response are displayed. Based on the display, the device has a series inductance as well as resistance &gt;50 ohms. 
     5. Select FREQUENCY WITH TIME GATE in the Domain menu to display the frequency domain S 11  forward reflection of the gated time domain response. 
     An example of gating a transmission measurement is making an antenna measurement. Gating can remove unwanted ground or chamber reflections that interfere with characterizing an antenna&#39;s pattern (FIG. 86). 
     Finally, let&#39;s look at some measurement considerations and ways to optimize their time-domain results. 
     Small impedance changes cause small responses that can be lost in the noise floor. This is also true of long cable and waveguide runs with high insertion loss. 
     To optimize for small responses: 
     Use averaging and reduced IF bandwidth to lower the noise floor. 
     Use maximum power to provide maximum dynamic range. 
     Use the window with the lowest sidelobes to reduce ringing. 
     Elements that are physically close or have similar length transmission paths can have minimal or overlapping time domain responses. 
     To optimize for close-response measurements and attain the best resolution: 
     Use the widest sweep. 
     Use the window with the narrowest pulse shape. 
     To maximize the distance measurement capability without causing aliasing (false information), use the minimum-frequency-step size by selecting 501 points and the minimum-required-frequency range. 
     In summary, the 360 Time Domain capability is a powerful and versatile tool in performing network analyzer measurements. 
     During Time Domain Measurement, the system 100 of the present invention advantageously reduces the amount of processing necessary to make measurements by using the apparatus and method described in the commonly assigned co-pending Patent Application entitled CROSS-PATH OPTIMIZATION IN MULTI-TASK PROCESSING, invented by Douglas R. Thornton, Ser. No. 176,096, filed on the same date as this Application and which is expressly incorporated herein in its entirety by this reference. 
     Time Domain Menus 
     The menus associated with the Time Domain Option are described in Appso 93-104, 
     GPIB OPERATION--BASIC PROGRAMMING 
     This section provides a description of the GPIB and the network analyzer command codes. It also provides several examples of bus programming. 
     DESCRIPTION OF THE IEEE-488 
     (IEC-625) INTERFACE BUS 
     The IEEE-488 General Purpose Interface Bus (GPIB) is an instrumentation interface for integrating instruments, calculators, and computers into systems. The bus uses 16 signal lines to effect transfer of data and commands to as many as 15 instruments. 
     The instruments on the bus are connected in parallel, as shown in FIG. 87. Eight of the signal lines (DIO 1 thru DI08) are used to the transfer of data and other messages in a byte-serial, bit-parallel form. The remaining eight lines are used for communications timing (handshake), control, and status information. Data are transmitted on the eight GPIB data lines as a series of eight-bit characters, referred to as bytes. Data transferral is by means of an interlocked handshake technique. 
     This technique permits asynchronous communications over a wide range of data rates. The following paragraphs provide an overview of the data, and handshake buses, and describe how these buses interface with the network analyzer. 
     Data Bus Description 
     The data bus is the conduit for transmitting information and data between the controller and the network analyzer. It contains eight bi-directional, active-low signal lines. DIO 1 thru DIO 8. One byte of information (eight bits) is transferred over the bus at a time. DI01 represents the least-significant bit (LSB) in this byte and DIO 8 represents the most-significant bit (MSB). Each byte represents a peripheral address (either primary or secondary), a control word, or a data byte. 
     Management Bus Description 
     The management bus is a group of five lines used to control the operation of the bus system. Functional information regarding the individual control lines is provided below. 
     a. ATN (Attention): 
     When this line is TRUE, the network analyzer responds to appropriate interface messages such as, device clear and serial poll and to its own listen/talk address. 
     b. EOI (End Or Identify) 
     When this line is TRUE, the last byte of a multi-byte message has been placed on the line. The line is also used in conjunction with ATN to indicate a parallel poll. 
     c. (IFC Interface Clear) 
     When this line is TRUE, the network analyzer&#39;s interface functions are placed in a known state such as, unaddressed to talk, unaddressed to listen, and service request idle. 
     d. REN (Remote Enable) 
     When this line is TRUE the network analyzer is enabled upon receipt of its listen, address for entry into the remote state. This mode is exited either when the REN line goes FALSE (high) or when the network analyzer receives a go-to-local (GTL) message or an RTL (return to local) command. 
     e. SRO (Service Request) 
     This line is pulled LOW (true) by the network analyzer to indicate that certain preprogrammed conditions exist. 
     Data Byte Transfer Control (Handshake) 
     Bus Description 
     Information is transferred on the data lines by a technique called the three-wire handshake. The three handshake-bus signal lines (FIG. 88) are described below. 
     a. DAV (Data Valid) 
     This line goes TRUE (arrow 1) when the talker has (1) sensed that NRFD is FALSE, (2) placed a byte of data on the bus, and (3) waited an appropriate length of time for the data to settle. 
     b. NFRD (NOT READY FOR DATA) 
     This line goes TRUE (arrow 2) when a listener indicates that valid data has not yet been accepted. The time between the events shown by arrows 1 and 2 is variable and depends upon the speed with which a listener can accept the information. 
     c. NDAC (NOT DATA ACCEPTED). 
     This line goes FALSE to indicate that a listener has accepted the current data byte for internal processing. When the data byte has been accepted, the listener releases its hold on NDAC and allows the line to go FALSE. However, since the GPIB is constructed in a wired-OR configuration, NDAC will not go FALSE until all listeners participating in the interchange have also released the line. As shown by arrow 3, when NDAC goes FALSE, DAV follows suit a short time later. The FALSE state of DAV indicates that valid data has been removed; consequently, NDAC goes LOW in preparation for the next data interchange (arrow 4). 
     Arrow 5 shows the next action in time: NRFD going FALSE after NDAC has returned TRUE. The FALSE state of NRFD indicates that all listeners are ready for the next information interchange. The time between these last two events is variable and depends on how long it takes a listener to process the data byte. In summation, the wired-OR construction forces a talker to wait for the slowest instrument to accept the current data byte before placing a new data byte on the bus. 
     GPIB OPERATION 
     All front panel keys, except for LINE ON/OFF, are bus controllable. When used on the GPIB, the network analyzer functions as both listener and a talker. Table 4-1 provides a listing of the GPIB subset functions and gives the capability for each. 
     
                       TABLE 4-1                                                   
______________________________________                                    
360 IEEE-488 Bus Subset Capability                                        
GPIB SUBSET                                                               
          FUNCTION       DESCRIPTION                                      
______________________________________                                    
AH1       Acceptor Handshake                                              
                         Complete Capability                              
SH1       Source Handshake                                                
                         Complete Capability                              
T6        Talker         No Talk Only (TON)                               
TE0       Talker with    No Capability                                    
          Address Extension                                               
L4        Listener       No Listen Only (LON)                             
LE0       Listener with  No Capability                                    
          Address Extension                                               
SR1       Service Request                                                 
                         Complete Capability                              
RL1       Remote/Local   Complete Capability                              
PP1       Parallel Poll  Complete Capability                              
DC1       Device Clear   Complete Capability                              
DT1       Device Trigger Complete Capability                              
______________________________________                                    
 
    
     COMMAND CODES, DESCRIPTION 
     The following paragraphs and tables describe the various GPIB command codes used by the 360. 
     COMMAND CODES: Classifications 
     The GPIB interface for the 360 uses more than 300 commands to implement the various functions. For descriptive purposes, the commands are organized into functional classifications. 
     COMMAND CODES: Syntax And Programming Tips 
     a. Syntax 
     All mnemonics are three characters long and may be entered in either upper or lower case. Mnemonics which require data must have a valid terminator mnemonic after the data. Separators between mnemonics and either data or other mnemonics are optional. 
     b. Programming Tips 
     The 360 is a &#34;channel-based&#34; instrument, which means that most commands apply only to the current active channel. Therefore, to set up a desired state on multiple channels, a CH1-CH4 mnemonic should precede the setup. For example: 
     &#34;D14 CH1 S11 SM1 CH2 S12 MPH CH3 S21 MAG CH4 S22 ISM&#34; 
     This string sets up a quad display (D14) and then sets the s-parameter and graph type desired for each Channel (Channel 1: S 11 , Smith chart; Channel 2: S 12 , log magnitude and phase; Channel 3: S 21 , log magnitude; Channel 4: S 22 , inverted Smith chart). 
     Other command codes are &#34;global&#34; in their extent, meaning they apply to all channels. Examples of these mnemonics: start/stop frequency (SRT,STP), averaging (AVG,AOF), and source power (PWR). 
     COMMAND CODES: Response To Errors 
     The following describes how the 360 responds to error conditions. 
     a. SYNTAX ERROR 
     The 360 beeps and sends a Service Request (SRQ), if enabled. It also ignores any further commands until it is programmed to talk or be unlistened. 
     b. PARAMETER OUT OF RANGE ERROR 
     The 360 moves the cursor to be adjacent to the erroneous entry, beeps, displays the entry in red and sends an SRQ (if enabled). The error is cleared upon execution of the next instruction. 
     C. ACTION REQUESTED NOT POSSIBLE 
     The 360 sends an SRQ (if enabled) and ignores the command. 
     COMMAND CODES: Channel Control 
     The commands described in Table 4-4 set up the current display mode and active channel on the 360. The active channel specifies the channel to which channel-based changes apply. 
     
                       TABLE 4-4                                                   
______________________________________                                    
Channel Control Command Codes                                             
Command Code                                                              
           Description                                                    
______________________________________                                    
DSP        Single Channel Display of Active Channel                       
D13        Dual Channel Display, Channels 1 and 3                         
D24        Dual Channel Display, Channels 2 and 4                         
D14        Quad Display, All Four Channels                                
CH1        Channel 1 Selected as Active Channel                           
CH2        Channel 2 Selected as Active Channel                           
CH3        Channel 3 Selected as Active Channel                           
CH4        Channel 4 Selected as Active Channel                           
______________________________________                                    
 
    
     COMMAND CODES: Data Entry 
     The command codes listed in Table 4-5 are used for mnemonics which must have numeric values. For mnemonics that take numeric values, an appropriate terminator is required, in addition to a numeric value. 
     
                       TABLE 4-5                                                   
______________________________________                                    
Data Entry Command Codes                                                  
Command Code  Description                                                 
______________________________________                                    
0,1,2,3,4,5,  Numerals for Numeric Entry                                  
6,7,8,9                                                                   
-             Minus Sign                                                  
.             Decimal Point                                               
GHZ           Gigahertz Data Terminator                                   
MHZ           Megahertz Data Terminator                                   
KHZ           Kilohertz Data Terminator                                   
PSC           Picoseconds Data Terminator                                 
NSC           Nanoseconds Data Terminator                                 
USC           Microseconds Data Terminator                                
DBL           dB Log Data Terminator                                      
DBM           dBm Data Terminator                                         
DEG           Degrees Data Terminator                                     
MMT           Millimeter Data Terminator                                  
CMT           Centimeter Data Terminator                                  
MTR           Meter Data Terminator                                       
XX1           Unitless Data Terminator, ×1                          
XX3           Unitless Data Terminator, ×10.sup.-3                  
XM3           Unitless Data Terminator, ×10.sup.-3                  
REU           Real Units Data Terminator                                  
IMU           Imaginary Units Data Terminator                             
______________________________________                                    
 
    
     Command Codes: Measurement Control 
     The command codes listed in Table 4-6 control the parameter being measured on the active channel (S 11 , S 21 , S 22  &amp; S 12 ) and the basic measurement setup. All command codes except S 11 , S 21 , S 22 , and S 12  are global, that is, they apply to the entire instrument. The SA1, SA2, and TA2 command codes can only be used with the Models 3630 and 3621 Test Sets with attenuators. Note that the two source attenuators have ranges of 0 to 70 dB while the test attenuator has a range of 0 to 40 dB. The HLD code holds the sweep at the current point and the CTN code continues sweeping from the current point. The TRS command code either restarts the sweep (continuous sweep mode) or triggers a single sweep (in hold mode). The WFS code causes the 360 to wait a full sweep so that any data on the display is valid. This is useful for scaling the display. It is required when outputting data from the 360, so as to ensure that the data being output is valid. The SWP code puts the 360 into continuous swept mode. 
     
                                           TABLE 4-6                               
__________________________________________________________________________
Measurement Control Command Codes                                         
COMMAND                                                                   
CODE   DESCRIPTION            VALUES   TERMINATORS                        
__________________________________________________________________________
S11    Selects S11 as S-Parameter On Active Channel                       
                              N/A      N/A                                
S21    Selects S21 as S-Parameter On Active Channel                       
                              N/A      N/A                                
S22    Selects S22 as S-Parameter On Active Channel                       
                              N/A      N/A                                
SRT    Sets Start Frequency   Depends on                                  
                                       GHZ, MHZ, KHZ                      
                              Frequency Range                             
                              of Instrument                               
STP    Sets Stop Frequency    Depends on                                  
                                       GHZ, MHZ, KHZ                      
                              Frequency Range                             
                              of Instrument                               
CWF    Sets CW Frequency      Depends on                                  
                                       GHZ, MHZ, KHZ                      
                              Frequency Range                             
                              of Instrument                               
PWR    Sets Source Power      Depends on                                  
                                       DBM, XX1, XX3,                     
                              Power Range of                              
                                       XM3                                
                              Source                                      
FHI    Sets Data Points to Maximum                                        
                              N/A      N/A                                
FME    Sets Data Points to Normal                                         
                              N/A      N/A                                
FL0    Sets Data Points to Minimum                                        
                              N/A      N/A                                
SA1    Sets Source Attenuator for Port 1                                  
                              0 dB to 70 dB                               
                                       DBL, DBM, XX1,                     
                                       XX3, XM3                           
SA2    Sets Source Attenuator for Port 2                                  
                              0 dB to 70 dB                               
                                       DBL, DBM, XX1,                     
                                       XX3, XM3                           
TA2    Sets Test Attenuator For Port 2                                    
                              0 dB to 40 dB                               
                                       DBL, DBM, XX1,                     
                                       XX3, XM3                           
HLD    Holds Sweep At Current Point                                       
                              N/A      N/A                                
CTN    Continue Sweep After Hold                                          
                              N/A      N/A                                
TRS    Triggers or Restarts a Sweep                                       
                              N/A      N/A                                
WFS    Wait full sweep        N/A      N/A                                
SWP    Selects Continuous Sweep Mode                                      
                              N/A      N/A                                
__________________________________________________________________________
 
    
     Command Codes: Display 
     The command codes listed in Table 4-7 are for setting up the graph type on the active channel. Most of the commands are straightforward with the exception of the SME, ISE, SMC and ISC codes. Both SME and ISE require values and only allow values of 10, 20 and 30. 
     
                                           TABLE 4-7                               
__________________________________________________________________________
Display Control Command Codes                                             
COMMAND                                                                   
       DESCRIPTION                 VALUES    TERMINATORS                  
__________________________________________________________________________
MAG    Selects Log Magnitude and Phase Display for Active                 
                                   N/A       N/A                          
       Channel                                                            
PHA    Selects Phase Display for Active Channel                           
                                   N/A       N/A                          
MPH    Selects Log Magnitude and Phase Display for Active                 
                                   N/A       N/A                          
       Channel                                                            
SMI    Selects Normal Smith Chart Display for Active Channel              
                                   N/A       N/A                          
SWR    Selects SWR Display for Active Channel                             
                                   N/A       N/A                          
ISM    Selects Inverted Normal Smith Chart Display for Active             
                                   N/A       N/A                          
       Channel                                                            
DLA    Selects Group Delay Display for Active Channel                     
                                   N/A       N/A                          
PLR    Selects Log Polar Display for Active Display                       
                                   N/A       N/A                          
LIN    Selects Linear Magnitude Display for Active Channel                
                                   N/A       N/A                          
LPH    Selects Linear Magnitude Display for Active Channel                
                                   N/A       N/A                          
REL    Selects Real Display for Active Channel                            
                                   N/A       N/A                          
IMG    Selects Imaginary Display for Active Channel                       
                                   N/A       N/A                          
RIM    Selects Real and Imaginary Display for Active Channel              
                                   N/A       N/A                          
SME    Selects Expanded Smith Chart Display for Active Channel            
                                   10, 20, 30                             
                                             DBL, XX1                     
ISE    Selects Inverted Expanded Smith Chart Display for                  
                                   10, 20, 30                             
                                             DBL, XX1                     
       Channel                                                            
SMC    Selects Compressed Smith Chart Display for Active                  
                                   3         DBL, XX1                     
       Channel                                                            
ISC    Selects Inverted Compressed Smith Chart Display Active             
                                   3         DBL, XX1                     
       Channel                                                            
SCL    Sets Scaling of Display On Active Channel                          
                                   Depends on Graph                       
                                             Depends on Graph             
                                   Type:     Type:                        
       Log Mag and Log Polar:      0.001 to 50                            
                                             DBL XX1, XX3,                
                                   db/div    XM3                          
       Phase:                      0.01 to 90                             
                                             DEG(,XX1, XX3,               
                                   degrees/div                            
                                             XM3 for PHA                  
                                             display)                     
       Group Delay:                1 femtosecond/div                      
                                             PSC, NSC, USC                
                                   to 999.999 s/div                       
       Linear Mag &amp; Linear Polar:  1 nanounit/div to                      
                                             XX1, XX3, SM3                
                                   999.999 units/div                      
       Imag:                       1 nanounit/div to                      
                                             IMU (,XX1, XX3,              
                                   999.999 units/div                      
                                             XM3 for IMG                  
                                             display)                     
       Smith/Inverted Smith        N/A       N/A                          
OFF    Set Offset of Display on Active Channel                            
                                   Depends on Graph                       
                                             Depends on Graph             
                                   Type:     Type:                        
       (This code moves the graph&#39;s reference position to the             
       offset value)                                                      
       Log Mag &amp; Log Polar:        -999.999 to                            
                                             DBL, XX1, XX3, XM            
                                   999.999 dB                             
                                             3                            
       PHASE:                      -180 to 180                            
                                             DEG(,XX1, XX3,               
                                   degrees   XM3 for PHA                  
                                             display)                     
       Group Delay:                -999.999 to                            
                                             PSC, NSC, USC                
                                   999.999 s                              
       Linear Mag &amp; Linear Polar:  0 to 999.999                           
                                             XX1, XX3, line 20,           
                                   units     XM3                          
       Imaginary                   -999.999 to                            
                                             IMU(,XX1, XX3,               
                                   99999.999 units                        
                                             XM3 for IMG                  
                                             display                      
       Smith/Inverted Smith:       N/A       N/A                          
REF    Set Reference Line of Display on Active Channel                    
                                   Depends on Graph                       
                                             Depends on Graph             
                                   Type:     Type:                        
       Log Magnitude, MAG Display: 0 to 8    DBL, XX1, XX3,               
                                             XM3                          
       Log Magnitude, MPH Display: 0 to 4    DBL, XX1, XX3,               
                                             XM3                          
       Phase, PHA Display:         0 to 8    DEG, XX1, XX3,               
                                             XM3                          
       Phase, MPH Display:         0 to 4    DEG                          
       Group Delay:                0 to 8    PSC, NSC, USC,               
                                             XX1, XX3, XM3                
       Linear Magnitude, LIN Display:                                     
                                   0 to 8    XX1, XX3, please             
                                             XM3                          
       Linear Magnitude, LPH Display                                      
                                   0 to 4    XX1, XX3,                    
                                             XM3                          
       Real, REL Display           0 to 8    REU, XX1, XX3,               
                                             XM3                          
       Real, RIM Display           0 to 4    REU, XX1, XX3,               
                                             XM3                          
       Imaginary, IMG Display:     0 to 8    IMU, XX1, XX3,               
                                             XM3                          
       Imaginary, RIM Display:     0 to 4    IMU                          
       Smith/Inverted Smith:       N/A       N/A                          
       Linear Polar/Log Polar:     N/A       N/A                          
ASC    Autoscale Display On Active Channel                                
                                   N/A       N/A                          
APR    Set Group Delay Aperture Percentage                                
                                   0 to 20   XX1, XX3, XM3                
__________________________________________________________________________
 
    
     EXAMPLE 
     &#34;SME 20 DBL&#34; This code selects a 20 dB expanded Smith chart on the active Channel. 
     Command Codes SMC and ISC also require values and only allow the value 3. 
     EXAMPLE 
     &#34;SMC 3 DBL&#34; This code selects a 3 dB compressed Smith chart on the active channel. 
     In addition to the brief description in Table 4-7, codes SCL and REF require additional description as provided in sub-paragraphs a and b, below. 
     a. SCL COMMAND CODE: 
     The SCL code sets the scaling per division of the graph on the active channel. Notice that for graph types with two types of information, the unitless terminators always apply to the first type of information. 
     EXAMPLE 
     &#34;MPH SCL 10 XXi&#34; This code will select a log magnitude and phase display on the active channel and set the magnitude scaling to 10 dB/div. The only way to scale the degrees part of the graph is by explicit use of the DEG terminator: 
     EXAMPLE 
     &#34;MPH SCL 45 DEG&#34; This code selects a log magnitude and phase display on the active channel and sets the phase scaling to 45 degrees/div. 
     NOTE: Smith charts and inverted Smith charts cannot be scaled using the SCL instruction. The different charts are selected using the SME, ISE, SMC and ISC mnemonics. 
     b. REF Command Code 
     The REF mnemonic selects which graticule line will be considered the &#34;reference&#34;. Notice that for graphs with one type of information such as MAG or PHA--the allowable reference line values are 0 to 8, while for graphs with two types of information the reference line value can only be 0 to 4. As described for the SCL code, for graphs having two types of information present, the unitless terminators apply to the first type of information. There is no reference line defined for Smith charts, inverted Smith charts, linear polar, or log polar displays. 
     Command Codes: Enhancement 
     The command codes listed in Table 4-8 control the data enhancement functions of IF bandwidth, averaging, and smoothing. Note that the maximum averaging number is 4095 and that the maximum smoothing number is 20%. 
     
                       TABLE 4-8                                                   
______________________________________                                    
Enhancement Command Codes                                                 
COMMAND                                                                   
CODE     DESCRIPTION   VALUE    TERMINATORS                               
______________________________________                                    
IFN      Selects Normal IF                                                
                       N/A      N/A                                       
         Bandwidth                                                        
IFR      Selects Reduced IF                                               
                       N/A      N/A                                       
         Bandwidth                                                        
IFM      Selects Minimum IF                                               
                       N/A      N/A                                       
         Bandwidth                                                        
AVG      Turns On      1 to 4095                                          
                                XX1, XX3, XM3                             
         Averaging and Sets                                               
         to Value                                                         
AOF      Turns Off     N/A      N/A                                       
         Averaging                                                        
SON      Turns On      0 to 20  XX1, XX3, XM3                             
         Smoothing and Sets                                               
         to Value                                                         
SOF      Turns Off     N/A      N/A                                       
         Smoothing                                                        
______________________________________                                    
 
    
     Command Codes: Reference Delay 
     The command codes listed in Table 4-9 are used to set up the reference delay applied to a channel and the relative dielectric constant of the system. Note that RDD, RDT and RDA change the active channel&#39;s reference delay while DIA, DIT, DIP, DIM, and DIE change the system&#39;s dielectric constant--a global change. The code RDA should only be used if a valid sweep is present. 
     
                                           TABLE 4-9                               
__________________________________________________________________________
Reference Delay Command Codes                                             
COMMAND                                                                   
CODE   DESCRIPTION                VALUES  TERMINATORS                     
__________________________________________________________________________
RDD    Sets Reference Delay As a Distance Value for the                   
                                  -999.999 to                             
                                          MMT, CMT,                       
       Channel                    999.999 MTR                             
RDT    Sets Reference Delay As a Time Value for the Active                
                                  -999.999 to                             
                                          PSC, NSC, USC                   
       Channel                    999.999,s                               
RDA    Selects Automatic Reference Delay for the Active                   
                                  N/A     N/A                             
       Channel                                                            
DIA    Selects Air Dielectric (1.00)                                      
                                  N/A     N/A                             
DIT    Selects Teflon Dielectric (2.10)                                   
                                  N/A     N/A                             
DIP    Selects Polyethylene Dielectric (2.26)                             
                                  N/A     N/A                             
DIM    Selects Microporous Teflon Dielectric (1.69)                       
                                  N/A     N/A                             
DIE    Sets Dielectric to Value   1 to 999.999999                         
                                          XX1, XX3, XM3                   
__________________________________________________________________________
 
    
     Command Codes: Trace Memory 
     The command codes listed in Table 4-10 control the trace memory function on the active channel and the trace math to be applied to it. These codes also do provide for storing and retrieving the active channel&#39;s trace memory to and from the disk. 
     
                       TABLE 4-10                                                  
______________________________________                                    
Trace Memory Command Codes                                                
COMMAND                                                                   
CODE     DESCRIPTION                                                      
______________________________________                                    
DAT      Displays Data Trace on Active Channel                            
MEM      Displays Memory Trace on Active Channel                          
DTM      Displays Data and Memory Traces on Active                        
         Channel                                                          
DNM      Displays Measured Data Normalized to Memory                      
         on Active Channel                                                
MIN      Selects Complex Subtraction As Trace Math on                     
         Active Channel                                                   
DIV      Selects Complex Division As Trace Math on                        
         Active Channel                                                   
ADD      Selects Complex Addition As Trace Math on                        
         Active Channel                                                   
MUL      Selects Complex Multiplication As Trace Math on                  
         Active Channel                                                   
STD      Stores Active Channel&#39;s Data Trace to Memory                     
SDK      Stores Active Channel&#39;s Trace Memory to Disk                     
         Under The Specified File Name                                    
RCK      Retrieves Active Channel&#39;s Trace Memory From                     
         Disk File Specified                                              
______________________________________                                    
 
    
     In order to view a display that involves trace memory (MEM, DTM and DNM) or to store trace memory to disk, the data for the channel must have been stored to memory first using the STD code. 
     EXAMPLE 
     &#34;WFS STD DIV DNM&#34; 
     This code causes the 360 to: 
     Wait a full sweep until data is valid (WFS). 
     Store that to memory (STD). 
     Select complex division as the trace math (DVI). 
     Display the data normalized to memory using this trace math(DNM). 
     Command Codes: Markers 
     The command codes listed in Table 4-11 control the location and display of the markers and the functions related to the markers. The MK1-MK6 codes are used to set a marker to a desired frequency, time or distance. The terminator mnemonics used must match the active channel&#39;s domain (frequency, time, or distance) Otherwise, an action-not-possible error will result. 
     
                                           TABLE 4-11                              
__________________________________________________________________________
Marker Command Codes                                                      
COMMAND                                                                   
CODE                                                                      
TERMINATORS                                                               
         DESCRIPTION                VALUES                                
__________________________________________________________________________
MK1-     Turns On Marker 1-6 and Sets Them to Value As                    
MK6      Shown Below:                                                     
         Frequency Markers          Limited to Current                    
                                              GHZ, MHZ, KHZ               
                                    Sweep Range                           
         Time Markers:              Limited to Current                    
                                              PSC, NSC, USC               
                                    Zoom Range                            
         Distance Markers:          Limited to Current                    
                                              MMT, CMT, MTR               
                                    Zoom Range                            
MOF      Disables Markers           N/A       N/A                         
MON      Enables Markers            N/A       N/A                         
M01-     Tums Off Marker 1-6        N/A       N/A                         
M06                                                                       
DR1-     Turns on Marker 1-6 As Delta Reference Marker                    
                                    N/A       N/A                         
DR6                                                                       
DRF      Turns On Delta Reference Marker Mode                             
                                    N/A       N/A                         
DR0      Turns Off Delta Reference Marker Mode                            
                                    N/A       N/A                         
MR1-     Selects Marker 1-6 As Readout Marker                             
                                    N/A       N/A                         
MR6                                                                       
MMX      Moves Active Marker to Maximum Trace Value                       
                                    N/A       N/A                         
M1S-     Marker Sweep With Marker 1-6 As Start Frequency                  
                                    N/A       N/A                         
M6S                                                                       
M1E-     Marker Sweep With Marker 1-6 As Stop Frequency                   
                                    N/A       N/A                         
M6E                                                                       
M1C-     CW Marker Sweep With Marker 1-6 As CW Frequency                  
                                    N/A       N/A                         
M6C                                                                       
__________________________________________________________________________
 
    
     EXAMPLE &#34;MK1 1.0000 NSC&#34; on a frequency domain channel generates an action-not-possible error 
     Markers can be individually turned off using the MO1-MO6 codes or markers can be disabled using the MOF code. A marker is turned on whenever any of the following conditions occur: 
     When the marker is set to a value, for example: &#34;MK2 4.5632 Ghz. 
     When the marker is selected for readout, for example: &#34;MR2&#34;. 
     When the marker is selected as the delta reference marker, for example: &#34;DR2&#34;. 
     The MMN and MMX codes move the active marker to the minimum and maximum trace values on the active channel, respectively. There must be an active marker selected for these mnemonics to execute. The M1-M6S, M1E-M6E and M1C-M6C mnemonics are used to define a marker sweep using the specified marker for either the start, stop, or CW frequency. 
     EXAMPLE &#34;WFS MR1 MMX M1S&#34; 
     This code sequence causes the 360 to 
     Wait for a full sweep of data to be present (WFS). 
     Turn on marker 1 and select it for readout (MR1). 
     Move marker 1 to the maximum value of the trace on the active channel (MMX). 
     Set the start frequency equal to the marker frequency (MlS). 
     Command Codes: Limits 
     The command codes listed in Table 4-12 are used to (1) set up the upper and lower limit values on the active channel and (2) set the limit delta for the limit frequency readout function. 
     
                                           TABLE 4-12                              
__________________________________________________________________________
Limits Command Codes                                                      
COMMAND                                                                   
CODE   DESCRIPTION                VALUES    TERMINATORS                   
__________________________________________________________________________
LUP    Turns On Limit 1 On the Active Channel and sets it                 
                                  Depends on Graph                        
                                            Depends on Graph              
       value, as Shown Below      Type:     Type:                         
LLO    Turn On Limit 2 On the Active channel And set it                   
                                  Depends on Graph                        
                                            Depends on Graph              
       Value, as Shown Below      Type:     Type:                         
LFD    Set Limit Delta On Active Channel for Limit Frequency              
                                  Depends On                              
                                            Depends On Graph              
       Readout, as Shown Below    Graph Type:                             
                                            Type:                         
       Log Mag &amp; Log Polar:       -999.999 to                             
                                            DBL, XX1, XX3,                
                                  999.999 dB                              
                                            XM3                           
       Phase:                     -180 to 180                             
                                            DEG(,XX1, XX3,                
                                  degrees   XM3 for PHA                   
                                            display)                      
       Group Delay:               -999.999 to                             
                                            PSC, NSC, USC                 
                                  999.999 s                               
       Linear Mag &amp; Linear Polar: 0 to 999.999 U                          
                                            XX1, XX3, XM3                 
       Real:                      -999.999 to                             
                                            REU, XX1, XX3,                
                                  999.999 U XM3                           
       Imaginary:                 -999.999 to                             
                                            IMU(,XX1, XX3,                
                                  999.999 U XM3 for IMG                   
                                            display)                      
       Smith/Inverted Smith:      0 to 1.413 units                        
                                            XX1, XX3, XM3                 
LOF    Disables Limits On Active Channel                                  
                                  N/A       N/A                           
LON    Enables Limits On Active Channel                                   
                                  N/A       N/A                           
LFR    Selects Limit Frequency Readout for Active                         
                                  N/A       N/A                           
       Channel                                                            
LFP    Selects Phase Limit Frequency Readout for Active                   
                                  N/A       N/A                           
       Channel for Log Magnitude/linear Magnitude and                     
       Phase Displays                                                     
__________________________________________________________________________
 
    
     The range of values and allowable terminator mnemonics are dependent on the graph type of the active channel much like the SCL, and REF codes described above. As described for these codes, for graphs with two types of information present, the unitless terminators apply to the first type of information. The second type of limit line value is accessed by explicit use of the appropriate data terminator mnemonic. 
     EXAMPLES 
     1. &#34;LUP 20 XXi&#34; on a log magnitude and phase display: sets the upper limit on the magnitude display to 20 dB. 
     2. &#34;LUP 45 DEG&#34; must be used to set the upper limit on the phase graph. 
     NOTE 
     The LFR, LFP, and LFD mnemonic codes, which deal with limit frequency readouts, are only available on the following graph types: log magnitude (M/G), log magnitude and phase (MPH), phase (PHA), linear magnitude (LIN), linear magnitude and phase (LPH), standing wave ratio (SWR), and group delay (DLA). The active channel must be a frequency domain channel. The LFP code can be used to select phase limit frequency readouts on log magnitude and phase and linear magnitude and phase graph types. If the LFR code is used on either of these graph types, the magnitude limit frequency readout menu for the channel is displayed. 
     Command Codes: Hard Copy 
     The command codes concerned with hard copy are listed in Table 4-13. 
     
                                           TABLE 4.13                              
__________________________________________________________________________
Hard Copy Command Codes                                                   
COMMAND                                                                   
CODE   DESCRIPTION                 VALUES   TERMINATORS                   
__________________________________________________________________________
PFS    Prints Full Screen          N/A      N/A                           
PGR    Prints Graph Area Only      N/A      N/A                           
PMK    Prints Tabular Marker Data  N/A      N/A                           
PTB    Prints Tabular Trace Data   N/A      N/A                           
PMT    Prints Tabular Marker and Trace Data                               
                                   N/A      N/A                           
PT0-PT9                                                                   
       Selects Tabular Printout Point Density                             
                                   N/A      N/A                           
PST    Stops Print/Plot            N/A      N/A                           
FFD    Printer Form Feed           N/A      N/A                           
PLS    Plots Entire Screen         N/A      N/A                           
PLD    Plots Graph Area            N/A      N/A                           
PLT    Plots Data Trace(s)         N/A      N/A                           
PMN    Plots Men                   N/A      N/A                           
PLM    Plots Markers and Limits    N/A      N/A                           
PLH    Plots Header                N/A      N/A                           
PGT    Plots Graticule             N/A                                    
PFL    Selects Full Size Plot      N/A      N/A                           
PTL    Selects 1/4 Size Plot, Top Left                                    
                                   N/A      N/A                           
PTR    Selects 1/4 Size Plot, Top Right                                   
                                   N/A      N/A                           
PBL    Selects 1/4 Size Plot, Bottom Left                                 
                                   N/A      N/A                           
PBR    Selects 1/4 Size Plot, Bottom Right                                
                                   N/A      N/A                           
DPN    Sets Pen Number for Plotting Data                                  
                                   1 to 8   XX1                           
GPN    Sets Pen Number for Plotting Graticule                             
                                   1 to 8   XX1                           
MPN    Sets Pen Number for Plotting Markers and Limits                    
                                   1 to 8   XX1                           
HPN    Sets Pen Number for Plotting Header                                
                                   1 to 8   XX1                           
LMS    Labels Model and Serial Number of Device                           
                                   String of                              
                                            N/A                           
                                   Characters Up To                       
                                   12 Characters                          
                                   Long                                   
LDT    Labels Date of Test         String of                              
                                            N/A                           
                                   Characters Up to                       
                                   12 Characters                          
                                   Long                                   
LID    Labels Device ID            String of                              
                                            N/A                           
                                   Characters Up to                       
                                   12 Characters                          
                                   Long                                   
LNM    Labels Operator&#39;s Name      String of                              
                                            N/A                           
                                   Characters Up to                       
                                   12 Characters                          
                                   Long                                   
TDD    Stores Tabular Data to Disk in File Specified (See                 
                                   N/Ae     N/A                           
       Naming Conventions in Paragraph 5-5.5)                             
RTB    Recalls Tabular Data From Disk File Specified to                   
                                   N/Anter  N/A                           
SPD    Sets Plotter Pen Speed Percentage                                  
                                   10 to 100                              
                                            XX1, XX3, XM3                 
__________________________________________________________________________
 
    
     These commands are straightforward with the exception of PT0-PT9. These ten codes set up (1) the density of tabular data points output to the printer for PTB and PMT and (2) the number of data points included in the output file for TDD (tabular data to disk). The number in the PT0-PT9 mnemonic codes specifies the number of points that are skipped during the print. Therefore, PT0 selects the densest printing mode while PT9 gives the fewest number of data points. 
     The command codes for plotting fall into two categories: setup and action. The setup codes are those that specify the desired size and location of the plot (PFL, PTL, PTR, PBL, PBR) and the pen numbers for each element of the plot (DPN, GPN, MPN, HPN). The action mnemonics actually initiate a plot for the subset of the display desired (PLS, PLD, PLT, PMN, PLM, PLH, PGT). 
     The LMS, LID, LDT and LNM codes require a string of characters to be sent over the GPIB along with the mnemonic. A string input to the 360 must have the quote characters (&#34;&#34;) surrounding the desired character for the string and cannot exceed the maximum number of characters specified for the mnemonics. An example of embedding quote characters in a string sent to the 360 is shown in App. 126. This example is in HP 85 BASIC. 
     The TDD code allows for outputting tubular data under the current print density mode (PT0-PT9) to an ASCII file. The RTB code causes the 360 to read an ASCII file assumed to be print data and output it to the printer. 
     NOTE 
     The maximum file size that can be handled with the RTB code is 58,000 bytes. Some miscellaneous commands are given in Table 4-14 as follows: 
     
                                           TABLE 4-14                              
__________________________________________________________________________
Miscellaneous Command Codes                                               
__________________________________________________________________________
RST Resets 360 to its Default                                             
                   Similar to pressing the &#34;DEFAULT PROGRAM&#34; key.         
    State                                                                 
FOF Implements Frequency                                                  
                   Instructs the 360 to blank any frequency information   
    Blanking       from the screen and any hard copy output.              
                   This code is useful for security reasons.              
FON Disables Frequency Blanking                                           
                   Frequency blanking can be turned off using this code.  
BCO Implements Display Blanking                                           
                   Allows for the ultimate in security -- a totally blank 
                   screen.                                                
                   In this mode, the 360 is fully operational over the    
                   GPIB                                                   
                   but nothing appears on the display.                    
BC1 Disables Display Blanking                                             
                   Screen blanking is turned off using the BC1 mnemonic.  
BLU Selects Blue As 3rd Color                                             
                   Allows selection of the third color used by the 360    
                   for                                                    
    Plane Color    markers, limits, and some menu annotation.             
CYN Selects Cyan As 3rd Color                                             
                   Allows selection of the third color used by the 360    
                   for                                                    
    Plane Color    markers, limits, and some menu annotations.            
TEST                                                                      
    Runs A Self Test                                                      
                   Instructs the 360 to perform a self test. An error in  
                   the                                                    
                   self test would be reported in the primary status      
                   byte,                                                  
                   bit 7 (see paragraph 5-5.7).                           
RTL Returns to Local Control                                              
                   Performs the same function as the control panel.       
                   RETURN                                                 
                   TO LOCAL key. This code has no effect if the           
                   360 is in local lockout.                               
DFP Displays Global Operating                                             
                   Displays Global Operating Parameters in the data       
    Parameters     area or the screen.                                    
DGS Displays GPIB System                                                  
                   Displays GPIB System Parameters in the data area of    
    Parameters     the screen.                                            
DCP Displays Calibration                                                  
                   Displays Calibration Pararmters in the data area of    
    Parameters     the screen.                                            
DC1 Displays Channel 1 and 2                                              
                   Displays Channels 1 and 2 Operating Parameters in the  
                   data                                                   
    Operating Parameters                                                  
                   area of the screen.                                    
DC3 Displays Channel 3 and 4                                              
                   Displays Channels 1 and 2 Operating Parameters in the  
                   data                                                   
    Operating Parameters                                                  
                   of the screen.                                         
__________________________________________________________________________
 
    
     Some more advanced command codes are described in Tables 5-1 through 5-4 as follows: 
     
                                           TABLE 5.1                               
__________________________________________________________________________
Calibration Command Codes                                                 
__________________________________________________________________________
RPC Repeat Previous Calibration                                           
                               N/A         N/A                            
LTC Specify Coaxial Lim        N/A         N/A                            
LTU Specify Microstrip Line Type                                          
                               N/A         N/A                            
LTW Specify Wavestrip Line Type                                           
                               N/A         N/A                            
SCM Specify Standard Calibration Method                                   
                               N/A         N/A                            
OCM Specify Offset Short Calibration Method                               
                               N/A         N/A                            
LCM Specify LRL Calibration Method                                        
                               N/A         N/A                            
C12 Begin 12-term Calibration  N/A         N/A                            
C8T Begin 8-term (I Port) Calibration                                     
                               N/A         N/A                            
CRF Begin Reflection Only (1 Port) Calibration                            
                               N/A         N/A                            
CFR Begin Frequency Response Calibration                                  
                               N/A         N/A                            
CFT Begin Reflection Only Frequency Response Calibration                  
                               N/A         N/A                            
CRL Begin Reflection Only Frequency Response Calibration                  
                               N/A         N/A                            
NOC Specify Normal 501-point Calibration                                  
                               N/A         N/A                            
DFC Specify Discrate Frequency Calibration                                
                               N/A         N/A                            
CWC Specify CW Calibration     N/A         N/A                            
TDC Specify Time Domain Harmonic Calibration                              
                               N/A         N/A                            
SRT Enter Start Frequency for Normal or Harmonic                          
                               Depends on Frequency                       
                                           GHZ, MHZ, KHZ                  
    Calibration                Range of Instrument                        
STP Enter Stop Frequency for Normal or Harmonic                           
                               Depends on Frequemy                        
                                           GHZ, MHZ, KHZ                  
    Calibration                Range of Instrument                        
DFQ Enter Discrate Frequency for Calibration                              
                               Depends an Frequency                       
                                           GHZ, MHZ, KHZ                  
    Range of Instrument                                                   
CWF Enter CE Frequency for Calibration                                    
                               Depends on Frequency                       
                                           GHZ, MHZ, KHZ-                 
                               Range of Instrument                        
__________________________________________________________________________
 
    
     App. 134 shows a sample program for controlling the calibration data collection, and FIG. 89 shows the binary data transfer message format. 
     Table 5--5 shows the output values versus various graph types, and App. 138 shows the response string from the OID comment as follows. 
     
                       TABLE 5.5                                                   
______________________________________                                    
Output Values Versus Various Graph Types                                  
Display Type   Output Values                                              
______________________________________                                    
Log Magnitude  dB, Degrees                                                
Phase          dB. Degrees                                                
Log Mag &amp; Phase                                                           
               dB, Degrees                                                
Linear Magnitude                                                          
               Lin Mag (Rho or Tau), Degrees                              
Linear Mag &amp; Phase                                                        
               Lin Mag (Rho or Tau), Degrees                              
Smith Chart    Ohms, Ohms (r + jx)                                        
Inverted Smith Siements, Siemens (G + o)                                  
Group Delay    Seconds, Degrees                                           
Log Polar      dB, Degrees                                                
Linear Polar   Lin.Mag (Rho or Tau), Degrees                              
Real           Real, imag                                                 
Imaginary      Real, imag                                                 
Real &amp; Imaginary                                                          
               Real, imag                                                 
SWR            SWR, Degrees                                               
______________________________________                                    
 
    
     Table 5-6 shows the calibration coefficient order for various calibration types, and tables 5-7 and 5-8 describe additional command codes. 
     
                       TABLE 5-6                                                   
______________________________________                                    
Calibration Coefficient Ordering Calbration Type                          
                                  Trans-                                  
                                  mssn  Refl                              
Co-                     Refl Freq Freq  Freq                              
efficient                                                                 
       12-term  8-term  Only Resp Resp  Resp                              
#      C12      C8T     CRF  CFR  CFT   CRL  None                         
______________________________________                                    
1      EDF      EDF     EDF  ERF  ETF   ERF  --                           
2      ESF      ESF     ESF  ETF  --    --   --                           
3      ERF      ERF     ERF  --   --    --   --                           
4      EXF      ETF     --   --   --    --   --                           
5      ELF      --      --   --   --    --   --                           
6      ETF      --      --   --   --    --   --                           
7      EDR      --      --   --   --    --   --                           
8      ESR      --      --   --   --    --   --                           
9      ERR      --      --   --   --    --   --                           
A      EXR      --      --   --   --    --   --                           
B      ELR      --      --   --   --    --   --                           
C      ETR      --      --   --   --    --   --                           
______________________________________                                    
 
    
     
                       TABLE 5-7                                                   
______________________________________                                    
Group Execute Trigger Command Codes                                       
Command                                                                   
Code    Description              Values                                   
______________________________________                                    
DEF     Begin Definition of Group Execute Trigger                         
                                 N/A                                      
        Response                                                          
END     End Definition of Group Execute Trigger                           
                                 N/A                                      
        Response                                                          
______________________________________                                    
 
    
     
                                           TABLE 5-8                               
__________________________________________________________________________
Disk Functions Command Codes                                              
Command                                                                   
Code  Description     Values                                              
__________________________________________________________________________
SDK   Store Active Channel&#39;s Trace                                        
                      String up to 8 Characters Long for File Name        
      Memory to Disk File                                                 
RCK   Recall Active Channel&#39;s Trace                                       
                      String up to 8 Characters Long for File Name        
      Memory From Disk File                                               
STO   Store Calibration Data and                                          
                      String up to 8 Characters Long for File Name        
      Front Panel Setup Information                                       
      to Disk File                                                        
RLD   Recall Calibration Data and                                         
                      String up to 8 Characters Long for File Name        
      Front Panel Setup Information                                       
      From Disk File                                                      
TDD   Store Tabular Printout Data to                                      
                      String up to 8 Characters Long for File Name        
      ASCII Disk File                                                     
RTB   Recall Tabular Data File From                                       
                      String up to 8 Characters Long for File Name        
      Disk for Output to Printer                                          
DEC   Delete Calibration and Front                                        
                      String up to 8 Characters Long for File Name        
      Panel Setup File From Disk                                          
DED   Delete Tabular Printout Data                                        
                      String up to 8 Characters Long for File Name        
      File From Disk                                                      
DEN   Delete Trace Memory File From                                       
                      String up to 8 Characters Long for File Name        
      Disk                                                                
INT   Initialize (Format) Disk in Drive                                   
                      N/A                                                 
      as a Data-Only Disk                                                 
LKT   Load Calibration Kit Information                                    
                      N/A                                                 
      From Disk                                                           
__________________________________________________________________________
 
    
     App. 141 is an example program illustrating the use of disk function command codes. Apps. 142 and 143 illustrate the two status bytes which describe the status of the 360, Table 5-9 describes the status byte command codes. 
     
                       TABLE 5-9.                                                  
______________________________________                                    
Status Byte Command Codes                                                 
Command                                                                   
Code    Description        Values                                         
______________________________________                                    
OPB     Output Primary Status Byte                                        
                           One Binary Byte                                
OEB     Output Extended (Secondary)                                       
                           One Binary Byte                                
        Status Byte                                                       
IPM     Input Primary Status Mask                                         
                           One Binary Byte                                
IEM     Input Extended (Secondary)                                        
                           One Binary Byte                                
        Status Mask                                                       
SQ0     Disable Service Requests                                          
                           N/A                                            
SQ1     Enable any Unmasked Service                                       
                           N/A                                            
        Requests                                                          
CSB     Clear Primary and Secondary                                       
                           N/A                                            
        Status Bytes                                                      
______________________________________                                    
 
    
     App. 145 shows a sample program for status byte enable mask setup and service request handling. Table 5-10 describes the Time Domain command codes. 
     
                                           TABLE 5-10                              
__________________________________________________________________________
Time Domain Command Codes                                                 
Command                                                                   
Code  Description            Values     Terminators                       
__________________________________________________________________________
FQD   Select Frequency Domain for Active Channel                          
                             N/A        N/A                               
TBP   Select Bandpass Mode With Time Readouts                             
                             N/A        N/A                               
      for active Channel                                                  
DBP   Select Bandpass Mode With Distance                                  
                             N/A        N/A                               
      Readouts for Active Channel                                         
TPI   Select Phasor Impulse Mado With Time                                
                             N/A        N/A                               
      Readouts for Active Channel                                         
DPI   Select Phasor Impulse Mode With Distance                            
                             N/A        N/A                               
      Readouts for Active Channel                                         
TLP   Select Lowpass Mode With Time Readouts                              
                             N/A        N/A                               
      for Active Channel                                                  
DLP   Select Lowpass Mode With Distance Readouts                          
                             N/A        N/A                               
      fot Active Channel                                                  
FGT   Select Frequency With Time Gate Made                                
                             N/A        N/A                               
LPI   Select Lowpass lmpulse Response for Active                          
                             N/A        N/A                               
      Channel                                                             
LPS   Select Lowpass Stop Response for Active                             
                             N/A        N/A                               
      Channel                                                             
GON   Turn Gate On on Active Channel                                      
                             N/A        N/A                               
GON   Turn Gate Off on Active Channel                                     
                             N/A        N/A                               
GDS   Display Gate Symbols on Active Channel With                         
                             N/A        N/A                               
      Gate Off                                                            
ZST   Sat Start of Time Domain Zoom Range for All                         
                             N/A        N/A                               
      Channels in Time Domain                                             
      Time Mode:             .999.999 to 999.999 us                       
                                        PSC, NSC, USC                     
      Distance Mode:         .999.999 to 999.999 m                        
                                        MMT, CMT, MTR                     
ZSP   Set Stop of Time Domain Zoom Range for All                          
      Channels in Time Domain                                             
      Time Mode:             .999.999 to 999.999 us                       
                                        PSC, NSC, USC                     
      Distance Mode:         .999.999 to 999.999 m                        
                                        MMT, CMT, MTR                     
ZCT   Set Center of Time domain Zoom Range for                            
      All Channels in Time Domain                                         
      Time Mode:             .999.999 to 999.999 us                       
                                        PSC, NSC, USC                     
      Distance Mode          .999.999 to 999.999 m                        
                                        MMT, CMT, MTR                     
ZSN   Sat Span of Time Domain Zoom Rangs for All                          
      Channels in Time Domain                                             
      Time Mode:             0 to 999.999 us                              
                                        PSC, NSC, USC                     
      Distance Mode          0 to 999.999 m                               
                                        MMT, CMT, MTR                     
GST   Set Gate Start Value for All Channels in Time                       
      Domain                                                              
      Time Mode:             -999.999 to 999.999 us                       
                                        PSC, NSC, USC                     
      Distance Made:         -999.999 to 999.999 m                        
                                        MMT, CMT, MTR                     
GSP   Set Gate Stop Value for all Channels in Time                        
      Domain                                                              
      Time Mode:             -999.999 to 999.999 us                       
                                        PSC, NSC, USC                     
      Distance Mode:         -999.999 to 999.999 m                        
                                        MMT, CMT, MTR                     
GCT   Set Gate Center Value for All Channels in                           
      Time Domain                                                         
      Time Mode:             0.0000 to 999.999 us                         
                                        PSC, NSC, USC                     
      Distance Mads:         0.0000 to 999.999 m                          
                                        MMT, CMT, MTR                     
GSN   Sat Gate Span Value for all Channels in Time                        
      Domain                                                              
      Time Mode:             0.0000 to 999.999 us                         
                                        PSC, NSC, USC                     
      Distance Mode          0.0000 to 999.999 m                          
                                        MMT, CMT, MTR                     
GRT   Select Rectangular Gate Shape for all Time                          
                             N/A        N/A                               
      Domain Channels                                                     
GNM   Select Nominal Gate Shape for all Time                              
                             N/A        N/A                               
      Domain Channels                                                     
GLS   Select Low Sidelobe Gate Shape for all Time                         
                             N/A        N/A                               
      Domain Channels                                                     
GMS   Select Minimum Sidelobe Gate Shape for all                          
                             N/A        N/A                               
      Time Domain Channels                                                
WRT   Select Rectangular Window Shape for all                             
                             N/A        N/A                               
      Time Domain Channels                                                
WNM   Select Nominal Window Shape for all Time                            
                             n/a        n/a                               
      Domain Channels                                                     
WLS   Select Low Sidelobe Windows Shape for all                           
                             N/A        N/A                               
      Time Domain Channels                                                
WMS   Select Minimum Sidelobe Window Shape for                            
                             N/A        N/A                               
      all Time Domain Channels                                            
MRR   Restore Original Range After a Marker Zoom                          
                             N/A        N/A                               
      Operation                                                           
DCA   Select Auto d.c. term for Low Pass                                  
                             N/A        N/A                               
DCZ   Select Line impedance d.c. term for Low pass                        
                             N/A        N/A                               
DCO   Select Open d.c. Term for Low Pass                                  
                             N/A        N/A                               
DCS   Select Short d.c. Term for Low Pass                                 
                             N/A        N/A                               
DCV*  Sect Low Pass d.c. Term to Value                                    
                             -100MΩ to 1000 MΩ                
                                        XX1, XX3, XM#                     
__________________________________________________________________________
 
    
     An extensive source code listing has been attached hereto as an Appendix and shall be construed as being incorporated herein in its entirety at the end of the description and prior to the claims. 
     The MODEL 360 VECTOR NETWORK ANALYZER OPERATION MANUAL, copyright 1987, printed December 1987, available from WILTRON Company, the assignor herein, is hereby incorporated herein in its entirety by this reference. 
     Thus, the scope of the invention is not intended to be limited except by the appended claims in which: 
     Appendix 
     -including- 
     MENU DISPLAYS FOR FUNCTIONS 
     AVAILABLE ON 
     THE MEASUREMENT SYSTEM 
     AND 
     ERROR CODES 
     PROVIDED FROM 
     THE MEASUREMENT SYSTEM 
     AND 
     COMMAND CODES 
     ACCEPTABLE TO 
     THE MEASUREMENT SYSTEM 
     
         ______________________________________                                    
MENU         DESCRIPTION                                                  
______________________________________                                    
SELECT                                                                    
CALIBRATION                                                               
DATA POINTS                                                               
NORMAL       Selects the standard calibration from a                      
(501 POINTS  start to a stop frequency that provides for                  
MAXIMUM)     up to 501 equally spaced (except the last)                   
             points of data for the defined frequency                     
             range. A flowchart of the calibration                        
             sequence is shown in FIG. 3-40.                              
C.W.         Selects the single frequency (C.W.)                          
(1 POINT)    calibration sequence that provides for 1                     
             data point at a selected frequency.                          
N-DISCRETE   Selects the discrete frequency calibration                   
FREQUENCIES  mode that lets you input a list of up to 501                 
(2 TO 501    individual data point frequencies.                           
POINTS)                                                                   
TIME DOMAIN  Selects the calibration mode for low-pass                    
(HARMONIC)   time-domain processing.                                      
PRESS &lt;ENTER&gt;                                                             
             Pressing the ENTER key implements                            
TO SELECT    your selection.                                              
______________________________________                                    
 
    
     
         ______________________________________                                    
MENU                                                                      
______________________________________                                    
FREQ RANGE OF                                                             
CALIBRATION                                                               
START        Enter the sweep-start frequency for                          
XX.XXXX GHz  calibration. If you desire, you can change                   
             this frequency for your meaurement when                      
             you reach Menu SU1, which follows the                        
             final calibration menu. The only                             
             restriction is that your start measurement                   
             frequency be greater than or equal to                        
             your start calibration frequency.                            
STOP         Enter the sweep-stop frequency for                           
XX.XXXX GHz  calibration. Like the start frequency,                       
             this too can be changed for your                             
             measurement. The stop frequency must be                      
             lower than or equal to your stop                             
             calibration frequency. In other words,                       
             your measurement frequency span must be                      
             equal to or smaller than your calibration                    
             frequency span.                                              
XXX DATA PTS The program automatically sets the step                      
USING ABOVE  size, based on the selected start and stop                   
START AND STOP                                                            
             frequencies. The step size will be the                       
XXX.X MHz    smallest possible (largest number of points                  
STEP SIZE    up to a maximum of 501), based on the                        
             chosen frequency span.                                       
NEXT CAL STEP                                                             
             Displays the next menu in the calibration                    
             sequence.                                                    
PRESS &lt;ENTER&gt;                                                             
             Pressing the ENTER key implements                            
TO SELECT    your menu selection.                                         
______________________________________                                    
 
    
     
         ______________________________________                                    
MENU         DESCRIPTION                                                  
______________________________________                                    
INSERT                                                                    
INDIVIDUAL                                                                
FREQUENCIES                                                               
INPUT A FREQ                                                              
PRESS &lt;ENTER&gt;                                                             
TO INSERT                                                                 
NEXT FREQ.   Move the cursor here and enter the next                      
XX.XXXXXX GHz                                                             
             frequency for which you wish calibration                     
             data taken. If the AUTO INCR option is                       
             ON, pressing ENTER automatically                             
             increments the calibration frequency by                      
             the interval in GHz that appears the                         
             option.                                                      
XXX FREQS.   Shows the number of frequencies that you                     
ENTERED,     have entered and reports the value of the                    
LAST FREQ WAS                                                             
             last frequency entered.                                      
XX.XXXXXX GHz                                                             
AUTO INCR ON Move the cursor here and press ENTER                         
(OFF)        to switch the Auto-Increment mode on or                      
XX.XXXXXX GHz                                                             
             off. If AUTO INCR is on, you may enter                       
             the frequency spacing.                                       
PREVIOUS MENU                                                             
             Displays C2D Menu.                                           
PRESS &lt;ENTER&gt;                                                             
             Pressing ENTER will cause actions as                         
TO SELECT    described above.                                             
OR TURN ON/OFF                                                            
______________________________________                                    
 
    
     
         ______________________________________                                    
MENU         DESCRIPTION                                                  
______________________________________                                    
SINGLE POINT                                                              
CALIBRATION                                                               
C.W. FREQ    Move cursor here and enter the frequency                     
XX.XXXXXX GHz                                                             
             for which calibration is to be done.                         
FINISHED     Move cursor here and press ENTER                             
ENTRY, NEXT  when finished.                                               
CAL STEP                                                                  
INPUT FREQ AND                                                            
             Input the frequency value and press the                      
PRESS &lt;ENTER&gt;                                                             
             ENTER key.                                                   
TO SELECT                                                                 
______________________________________                                    
 
    
     
         ______________________________________                                    
MENU         DESCRIPTION                                                  
______________________________________                                    
CALIBRATION                                                               
RANGE                                                                     
HARMONIC CAL                                                              
FOR                                                                       
TIME DOMAIN                                                               
START (STEP) Move cursor here to enter the desired                        
XX.XXXXXX GHz                                                             
             start frequency. This frequency also will                    
             be used as the frequency increment.                          
APPROX STOP  Move the cursor here to enter the                            
XX.XXXXXX GHz                                                             
             approximate desired stop frequency The                       
             frequency will be adjusted to the nearest                    
             harmonic multiple of the start frequency.                    
USING ABOVE  The program automatically indicates the                      
START AND STOP                                                            
             number of data points and the true                           
WILL RESULT IN                                                            
             (harmonic) stop frequency.                                   
XXX DATA                                                                  
POINTS AND                                                                
XX.XXXXXX GHz                                                             
TRUE STOP                                                                 
NEXT CAL STEP                                                             
             Move the cursor here and press ENTER                         
PRESS &lt;ENTER&gt;                                                             
             when finished.                                               
TO SELECT                                                                 
______________________________________                                    
 
    
     
         ______________________________________                                    
MENU         DESCRIPTION                                                  
______________________________________                                    
FILL FREQUENCY                                                            
RANGES                                                                    
INPUT START. This menu is used to create one or more                      
INCR. POINTS,                                                             
             ranges of discrete equally spaced                            
THEN SELECT  frequency points for calibration.                            
&#34;FILL RANGE&#34;                                                              
START FREQ   Enter the first frequency of the range.                      
0.5000 GHz                                                                
INCREMENT    Enter the increment (step size) between                      
0.1004 GHz   one frequency and the next.                                  
NUM OF PTS   Enter the number of frequency points in                      
82 POINTS    the range.                                                   
STOP FREQ                                                                 
XX-XXXX GHz                                                               
FILL RANGE   Moving the cursor here and pressing                          
(ENTERED     ENTER fills the range and shows the                          
82 FREQS)    number of frequencies selected (in NUM                       
             OF PTS above).                                               
INSERT       Calls Menu C2A, which allows you to set                      
INDIVIDUAL   the individual frequencies.                                  
FREQUENCIES                                                               
FINISHED     Calls Menu C3, the next menu in the                          
ENTRIES. NEXT                                                             
             calibration sequence.                                        
CAL STEP                                                                  
PRESS &lt;ENTER&gt;                                                             
             Pressing the ENTER key implements                            
TO SELECT    your menu selection.                                         
______________________________________                                    
 
    
     
         ______________________________________                                    
MENU         DESCRIPTION                                                  
______________________________________                                    
CONFIRM                                                                   
CALIBRATION                                                               
PARAMETERS                                                                
PORT 1 CONN: Displays the connector type for which the                    
SMA (M)      calibration is set to correct. This should                   
             agree with the connector type on both                        
             your calibration components and the test                     
             device.                                                      
PORT 2 CONN:                                                              
SMA (F)                                                                   
LOAD TYPE    Displays the type of load (termination)                      
(BROADBAND)  that you will use in the calibration. If you                 
             wish a different type, press the ENTER                       
             key to display Menu C4).                                     
CHANGE       Move cursor to appropriate line and press                    
PORT 1 CONN  ENTER to change connector type.                              
CHANGE       Move cursor to appropriate line and press                    
PORT 2 CONN  ENTER to change connector type.                              
CHANGE       Move cursor to appropriate line and press                    
LOAD TYPE    ENTER to change load type.                                   
START CAL    Starts the calibration sequence.                             
PRESS &lt;ENTER&gt;                                                             
             Pressing the ENTER key implements                            
TO SELECT    your menu selection.                                         
______________________________________                                    
 
    
     
         ______________________________________                                    
MENU          DESCRIPTION                                                 
______________________________________                                    
SELECT PORT 1 or 2                                                        
CONNECTOR TYPE                                                            
SMA (M)       Applies the four capacitance-coefficient                    
SMA (F)       values to the Open that are needed to                       
              correct for an SMA connector being                          
              installed on the test device. Refer to                      
              paragraph 3-11.4, &#34;Using Calibration                        
              Standards,&#34; for a discussion on this                        
              topic.                                                      
K-CONNECTOR (M)                                                           
              Same as above, except for K                                 
K-CONNECTOR (F)                                                           
              Connector.                                                  
TYPE N (M)    Same as above, except for Type-N                            
TYPE N (F)    connector.                                                  
GPC-7         Same as above, except for GPC-7                             
              connector.                                                  
OTHER         Calls Menu C12, which allow you to                          
              specify the connector coefficients.                         
PRESS &lt;ENTER&gt; Pressing the ENTER key implements                           
TO SELECT     your menu selection.                                        
______________________________________                                    
 
    
     
         ______________________________________                                    
MENU         DESCRIPTION                                                  
______________________________________                                    
SELECT                                                                    
CALIBRATION                                                               
TYPE                                                                      
KEEP EXISTING                                                             
             Keeps the existing calibration data, then                    
CAL DATA     exits the calibration sequence by calling                    
             up Menu C13.                                                 
FULL 12-TERM Corrects for error terms E.sub.TF, E.sub.TR, E.sub.RF,       
             E.sub.RR, E.sub.DF, E.sub.DR, E.sub.XF, E.sub.XR, E.sub.SF.  
             E.sub.SR, E.sub.LF,                                          
             and E.sub.LR (FIG. 3-39), which are all of the               
             error-terms associated with a two-port                       
             measurement. Refer to paragraph 3-11.1,                      
             &#34;Explaining Measurement Accuracy,&#34; for                       
             a detailed decussion of these error terms.                   
1 PATH       Corrects for forward-direction error                         
2 PORT       terms E.sub.TF, E.sub.RF, E.sub.DF, and E.sub.SF.            
FREQUENCY    Corrects for forward-direction error                         
RESPONSE ONLY                                                             
             terms E.sub.RF, and E.sub.DF. These provide a                
             frequency-response-only correction for                       
             Port 1.                                                      
REFLECTION   Corrects for forward-direction error                         
ONLY (PORT 1)                                                             
             terms E.sub.RF, E.sub.DF, and E.sub.SF. These provide        
             a reflection-only correction for Port 1.                     
PRESS &lt;ENTER&gt;                                                             
             Pressing the ENTER key implements                            
TO SELECT    your menu selection.                                         
______________________________________                                    
 
    
     
         ______________________________________                                    
MENU         DESCRIPTION                                                  
______________________________________                                    
SELECT                                                                    
FREQUENCY                                                                 
RESPONSE TYPE                                                             
TRANSMISSION For the calibration-correction of the                        
             transmission-only frequency-response                         
             error term.                                                  
REFLECTION   For the calibration-correction of the                        
             reflection-only frequency-response error                     
             term.                                                        
BOTH         For the calibration-correction of both                       
             transmission and reflection frequency-                       
             response error terms.                                        
PRESS &lt;ENTER&gt;                                                             
             Pressing the ENTER key implements                            
TO SELECT    your selection.                                              
______________________________________                                    
 
    
     
         ______________________________________                                    
MENU          DESCRIPTION                                                 
______________________________________                                    
SELECT                                                                    
TYPE OF LOAD                                                              
BROADBAND     Selects calibration based on the                            
FIXED LOAD    broadband load being used.                                  
SLIDING LOAD  Selects calibration based on the sliding                    
(FREQS BELOW  load being used. If your low-end                            
2 GHz ALSO    frequency is below 2 GHz, a fixed                           
REQUIRE FIXED broadband load is also required.                            
BROADBAND LOAD)                                                           
PRESS &lt;ENTER&gt; Pressing the ENTER key implements                           
TO SELECT     your menu selection.                                        
______________________________________                                    
 
    
     
         ______________________________________                                    
MENU        DESCRIPTION                                                   
______________________________________                                    
CALIBRATION                                                               
SEQUENCE                                                                  
CONNECT                                                                   
CALIBRATION                                                               
DEVICES:                                                                  
PORT 1:     Connect the required components to Port 1.                    
XXXXXXXXXXXX                                                              
PORT 2:     Connect the required components to Port 2.                    
XXXXXXXXXXXX                                                              
MESSAGE AREA                                                              
(See FIG. 3-47                                                            
______________________________________                                    
 
    
     
         ______________________________________                                    
MENU        DESCRIPTION                                                   
______________________________________                                    
CALIBRATION                                                               
SEQUENCE                                                                  
SLIDE LOAD  Slide the load to the next position,                          
TO POSITION X                                                             
            then press the ENTER key.                                     
            Moving the slide to six different positions                   
            provides sufficient data for the program to                   
            accurately calculate                                          
            the effective directory of the system                         
______________________________________                                    
 
    
     
         ______________________________________                                    
MENU        DESCRIPTION                                                   
______________________________________                                    
CALIBRATION                                                               
SEQUENCE                                                                  
CONNECT     Connect Ports 1 and 2 together using                          
THROUGHLINE the Throughline cable                                         
BETWEEN                                                                   
TEST PORTS                                                                
______________________________________                                    
 
    
     
         ______________________________________                                    
MENU         DESCRIPTION                                                  
______________________________________                                    
CALIBRATION                                                               
SEQUENCE                                                                  
COMPLETED                                                                 
PRESS        Pressing the SAVE/RECALL MENU                                
&lt;SAVE/RECALL&gt;                                                             
             Key displays Menu SR that lets you                           
TO STORE CAL save your calibration data onto a disk                       
DATA ON DISK or recall previously saved                                   
             calibration data from a disk.                                
             While this menu provides a                                   
             convenient point at                                          
             which to save the calibration data,                          
             it is not the only point allowed. You                        
             can use the SAVE/RECALL MENU                                 
             key at any point in                                          
OR           the measurment program.                                      
PRESS &lt;ENTER&gt;                                                             
             Pressing the ENTER key implements                            
TO PROCEED   your menu selection.                                         
______________________________________                                    
 
    
     
         ______________________________________                                    
MENU         DESCRIPTION                                                  
______________________________________                                    
PORT 1 OR 2  Enter the capacitance-coefficient values                     
OPEN DEVICE  needed to correct for your Open device.                      
ENTER THE                                                                 
CAPACITANCE                                                               
COEFFICIENTS                                                              
TERM 1-CO    Enter the term 1 coefficient value.                          
±XXX.XXe-15                                                            
TERM 2-C1    Enter the term 2 coefficient value.                          
±XXX.XXe-27                                                            
TERM 3-C2    Enter the term 3 coefficient value.                          
±XXX.XXe-36                                                            
TERM 3-C3    Enter the term 3 coefficient value.                          
±XXX.XXe-45                                                            
ENTER THE    Enter the offset length value needed to                      
OFFSET LENGTH                                                             
             correct for your Open device                                 
OFFSET LENGTH                                                             
±XXX.XX mm                                                             
PRESS &lt;ENTER&gt;                                                             
             Pressing the ENTER key implements                            
WHEN COMPLETE                                                             
             your menu selection                                          
______________________________________                                    
 
    
     
         ______________________________________                                    
MENU         DESCRIPTION                                                  
______________________________________                                    
PORT 2                                                                    
SHORT DEVICE                                                              
ENTER THE    Enter the length that the Short is offset                    
OFFSET LENGTH                                                             
             from the reference plane.                                    
OFFSET LENGTH                                                             
±XXX.XXX mm                                                            
PRESS &lt;ENTER&gt;                                                             
             Pressing the ENTER key implements                            
WHEN COMPLETE                                                             
             your menu selection.                                         
______________________________________                                    
 
    
     
         ______________________________________                                    
MENU       DESCRIPTION                                                    
______________________________________                                    
EXISTING   Retains the existing calibration data in                       
CALIBRATION                                                               
           memory. At the end of three seconds, the                       
KEPT       appropriate measurement setuop                                 
           menu appears.                                                  
______________________________________                                    
 
    
     
         ______________________________________                                    
MENU TEXT    DESCRIPTION                                                  
______________________________________                                    
SAVE/RECALL                                                               
FRONT PANEL                                                               
INFORMAITON                                                               
SAVE         Displays Menu SR2, which asks you to                         
RECALL       select a storage location-                                   
             internal memeory or disk.                                    
PRESS &lt;ENTER&gt;                                                             
             Pressing the ENTER key implements                            
TO SELECT    your menu selection. The menu remains                        
FUNCTION     on the screen until another menu is                          
             selected for display or until                                
             the CLEAR/RET LOC key is pressed.                            
______________________________________                                    
 
    
     
         ______________________________________                                    
MENU TEXT     DESCRIPTION                                                 
______________________________________                                    
RECALL (or SAVE)                                                          
FRONT PANEL   Calls Menu SR3, which lets                                  
SETUP IN      you save the control panel setup                            
INTERNAL      into or recalls it from                                     
MEMORY        internal memory.                                            
CAL DATA      Saves the calibration data and control                      
AND FRONT     panel setup onto the disk or recalls                        
PANEL SETUP   them from the disk. This selection                          
ON DISK       displays Menu DSK9(recall) or DSK11,                        
              and GP1-3(save) which asks you                              
              to select a disk file.                                      
PRESS &lt;ENTER&gt; Pressing the ENTER key implements                           
TO SELECT     your menu selection. The menu remains                       
              on the screen until another menu is                         
              selected-for display or until the                           
              CLEAR/RET LOC key is pressed.                               
______________________________________                                    
 
    
     
         ______________________________________                                    
MENU         DESCRIPTION                                                  
______________________________________                                    
SAVE TO                                                                   
INTERNAL                                                                  
MEMORY                                                                    
MEMORY 1     Causes the current control (front) panel                     
             setup to be saved to memory location 1.                      
MEMORY 2     Same as above, except the setup                              
             saves to memory location 2.                                  
MEMORY 3     Same as above, except the setup                              
             saves to memory location 3                                   
MEMORY 4     Same as above, except the setup                              
             saves to memory location 4                                   
PRESS &lt;ENTER&gt;                                                             
             You may press the ENTER key                                  
TO SELECT    or use the keypad to impelement                              
OR           your menu selection. The menu remains                        
USE KEYPAD   on the screen until another menu                             
             is selected for display or until the                         
             CLEAR/RET LOC key is pressed.                                
______________________________________                                    
 
    
     
         ______________________________________                                    
MENU         DESCRIPTION                                                  
______________________________________                                    
WARNING                                                                   
INTERNAL     Warns that the setup you are attempting                      
MEMORY       to recall is not compatible with the                         
DOES NOT     calibration data stored in memory.                           
MATCH                                                                     
CURRENT                                                                   
CAL SETUP                                                                 
CONTINUING   Recalling the setup in question will                         
RECALL WILL  destroy the calibration data stored                          
DESTROY      in memory.                                                   
CURRENT CAL                                                               
PRESS &lt;ENTER&gt;                                                             
             Pressing the ENTER key recalls the                           
TO RECALL    selected setup, while pressing the CLEAR                     
OR           key aborts the selection.                                    
PRESS &lt;CLEAR&gt;                                                             
TO ABORT                                                                  
______________________________________                                    
 
    
     
         ______________________________________                                    
MENU         DESCRIPTION                                                  
______________________________________                                    
SWEEP SETUP                                                               
START        Enter the sweep-start frequency in GHz.                      
XX.XXXXXX GHz                                                             
             The start frequency must be lower                            
             than the stop frequency.                                     
STOP         Enter the sweep-stop frequency in GHz.                       
XX.XXXXXX GHz                                                             
             The stop frequency must be higher                            
             than the start frequency.                                    
XXX DATA PTS Displays the number of frequency points                      
USING ABOVE  and the spacing between points for the                       
START AND STOP                                                            
             start and stop frequencies selected                          
XXX.X MHz    above. The number of points shown                            
STEP SIZE    provides the finest frequency resolution                     
             possible, based on your DATA POINTS                          
             key MAXIMUM-NORMAL- MINIMUM                                  
             selection.                                                   
C.W. MODE ON Move cursor here and press ENTER to                          
XX.XXXXXX GHz                                                             
             enable the CW mode. Enter CW                                 
             frequency for measurements.                                  
MARKER SWEEP Move cursor here and press ENTER to                          
             set the start and stop frequencies (SU5) or                  
             the CW frequency (SU6) to the values of                      
             any marker                                                   
HOLD BUTTON  Displays Menu SU4, which lets you                            
FUNCTION     set the action of the HOLD key.                              
REDUCED TEST Displays Menu SU2, which lets you                            
SIGNALS      set the source power and the values for                      
             the attenuators in the Model 3620 Series                     
             Test Set.                                                    
PRESS &lt;ENTER&gt;                                                             
             Pressing the ENTER key                                       
TO SELECT    implements your menu selection. The                          
OR TURN ON/OFF                                                            
             menu remains on the screen until another                     
             menu is selected for display or until the                    
             CLEAR/RET LOC key is pressed.                                
______________________________________                                    
 
    
     
         ______________________________________                                    
MENU         DESCRIPTION                                                  
______________________________________                                    
SWEEP SETUP                                                               
REDUCED                                                                   
TEST SIGNALS                                                              
SOURCE POWER Enter the output-power level for the                         
±XX.XdBm  sweep generator (frequency source)                           
             in dBm.                                                      
PORT 1 SOURCE                                                             
             Attenuates the microwave source                              
X0 dB (0-70) power from 0 to 70 dB, in 10                                 
             dB steps. The power is attenuated before                     
             being applied to Port 1 for a forward                        
             transmission or reflection test (S.sub.21 or                 
             S.sub.11, respectively).                                     
PORT 2 SOURCE                                                             
             Attenuates the microwave source                              
X0 dB (0-70) power from 0 to 70 dB, in 10 dB steps.                       
             The power is attenuated before being                         
             applied to Port 2 for a reverse                              
             transmission or reflection test                              
             (S.sub.12 or S.sub.22, respectively).                        
PORT 2 TEST  Attenuates from 0 to 40 dB (10 dB steps)                     
X0 dB (0-40) the microwave power being input                              
(THIS REDUCES                                                             
             to Port 2 from the                                           
SIGNAL FROM  device-under-test (DUT).                                     
AMPLIFIER                                                                 
UNDER TEST)                                                               
______________________________________                                    
 
    
     
         ______________________________________                                    
MENU         DESCRIPTION                                                  
______________________________________                                    
SINGLE POINT                                                              
MEASUREMENT                                                               
SETUP                                                                     
C.W. FREQ    Enter the measurement frequency in GHz                       
XX.XXXXXX GHz                                                             
             for continuous wave (CW) operation.                          
HOLD BUTTON  Displayes Menu SU4, which lets you                           
FUNCTION     set the action of the HOLD key.                              
REDUCED TEST Displays Menu SU2, which lets you                            
SIGNALS      set values for the source power and                          
             attenuators in the Model 3620                                
             Series Test Set.                                             
RETURN TO    Move cursor here and press ENTER to                          
SWEEP MODE   return to the F1-F2                                          
             sweep mode. (Menu SU1)                                       
PRESS &lt;ENTER&gt;                                                             
             Pressing the ENTER key implements                            
TO SELECT    your menu selection. The menu                                
             remains on the screen until another                          
             menu is selected for display or until the                    
             CLEAR/RET LOC key is pressed.                                
______________________________________                                    
 
    
     
         ______________________________________                                    
MENU         DESCRIPTION                                                  
______________________________________                                    
SELECT FUNC-                                                              
TION FOR                                                                  
HOLD BUTTON                                                               
HOLD/CONTINUE                                                             
             Causes the hold key (button) to                              
             stop and start the sweep.                                    
HOLD/RESTART Causes the hold key to stop and                              
             restart the sweep.                                           
SINGLE SWEEP Causes the hold key to trigger a single                      
AND HOLD     sweep and hold when finished.                                
             (Two sweeps, one from Port 1 to 2                            
             and another from Port 2 to 1, are                            
             accomplished for a 12-Term                                   
             measurement.)                                                
PRESS &lt;ENTER&gt;                                                             
             Pressing the ENTER key                                       
TO SELECT    implements your menu selection. The                          
             menu remains on the screen until another                     
             menu is selected for display or until the                    
             CLEAR/RET LOC key is pressed.                                
______________________________________                                    
 
    
     
         ______________________________________                                    
MENU         DESCRIPTION                                                  
______________________________________                                    
FREQUENCY                                                                 
MARKER SWEEP                                                              
START SWEEP  Pressing a number on the keypad causes                       
MARKER (n)   the associated marker to be the                              
XX.XXXX GHz  start frequency of the sweep.                                
STOP SWEEP   Pressing a number on the keypad causes                       
MARKER (n)   the associated marker to be the                              
XX.XXXX GHz  stop frequency of the sweep.                                 
USE KEYPAD   Use the keypad to select markers                             
TO SELECT    1, 2, 3, 4, 5, or 6.                                         
MARKER (1-6)                                                              
______________________________________                                    
 
    
     
         ______________________________________                                    
MENU         DESCRIPTION                                                  
______________________________________                                    
FREQUENCY                                                                 
MARKER C.W.                                                               
C.W. FREQ    Pressing a number on the keypad causes                       
MARKER (n)   the associated marker to be                                  
XX.XXXX GHz  the C.W. frequency.                                          
USE KEYPAD   Use the keypad to select markers                             
TO SELECT    1, 2, 3, 4, 5, or 6.                                         
MARKER (1-6)                                                              
______________________________________                                    
 
    
     
         ______________________________________                                    
MENU         DESCRIPTION                                                  
______________________________________                                    
SELECT                                                                    
DISPLAY MODE                                                              
SINGLE       Selects a single channel for display,                        
DISPLAY      which can be log magnitude, phase,                           
             log magnitude and phase, or Smith chart.                     
             You select the type of display in                            
             Menu GT.                                                     
DUAL         Selects Channels 1 and 3 for display.                        
DISPLAY      You select the type of display                               
CHANNEL 1-3  in Menu GT.                                                  
DUAL         Selects Channels 2 and 4 for display. You                    
DISPLAY      select the type of display in Menu GT.                       
CHANNEL 2-4                                                               
ALL FOUR     Selects all four channels for display. You                   
CHANNELS     select the type of display in Menu GT.                       
PRESS &lt;ENTER&gt;                                                             
             Pressing the ENTER key implements                            
TO SELECT    your menu selection. The menu remains                        
             on the screen until another menu is                          
             selected for display or until the                            
             CLEAR/RET LOC key is pressed.                                
______________________________________                                    
 
    
     
         ______________________________________                                    
MENU         DESCRIPTION                                                  
______________________________________                                    
SELECT                                                                    
S-PARAMETER                                                               
S21          Selects the S.sub.21 parameter to be                         
FWD TRANS    displayed on the active channel. The                         
             parameter can be displayed in any                            
             of the available formats.                                    
S11          Selects the S.sub.11 parameter to be                         
FWD REFL     displayed on the active channel. The                         
             parameter can be displayed in any                            
             of the available formats.                                    
S12          Selects the S.sub.12 parameter to be                         
REV TRANS    displayed on the active channel. The                         
             parameter can be displayed in any                            
             of the available formats.                                    
S22          Selects the S.sub.22 parameter to be                         
REV REFL     displayed on the active channel. The                         
             parameter can be displayed in any                            
             of the available formats.                                    
PRESS &lt;ENTER&gt;                                                             
             Pressing the ENTER key implements                            
TO SELECT    your menu selection. The menu remains                        
             on the screen until another menu is                          
             selected for display or until the                            
             CLEAR/LOC key is pressed.                                    
______________________________________                                    
 
    
     
         ______________________________________                                    
MENU         DESCRIPTION                                                  
______________________________________                                    
SELECT                                                                    
GRAPH TYPE                                                                
LOG          Selects a log magnitude graph for display                    
MAGNITUDE    on the active channel&#39;s selected                             
             s-parameter. The active channel is                           
             indicated by its key (CH1, CH2,                              
             CH3, CH4) being lit.                                         
PHASE        Selects a phase graph for display                            
             on the active channel.                                       
LOG MAGNITUDE                                                             
             Selects log magnitude and phase graphs                       
AND PHASE    for display on the active channel.                           
SMITH CHART  Selects a Smith chart for display                            
(IMPEDANCES) on the active channel.                                       
SWR          Selects an SWR display for                                   
             the active channel.                                          
GROUP DELAY  Selects a Group Delay display                                
             for the active channel.                                      
MORE         Takes you to additional graph type                           
             selections on Menu GT2.                                      
PRESS &lt;ENTER&gt;                                                             
             Pressing the ENTER key implements                            
TO SELECT AND                                                             
             your menu selection and resumes the                          
RESUME       calibration from where it left off, if in the                
CALIBRATION  calibration mode. The menu remains                           
             on the screen until another menu is                          
             selected for display or until the                            
             CLEAR/RET LOC key is pressed.                                
______________________________________                                    
 
    
     
         ______________________________________                                    
MENU         DESCRIPTION                                                  
______________________________________                                    
SELECT                                                                    
GRAPH TYPE                                                                
ADMITTANCE   Selects an Admittance Smith chart for                        
SMITH CHART  display on the active channel&#39;s                              
             s-parameter.                                                 
LINEAR POLAR Selects a Linear Polar graph for display                     
             on the active channel&#39;s s-parameter.                         
LOG POLAR    Selects a Log Polar graph for display                        
             of the active channel&#39;s s-parameter.                         
LINEAR MAG   Selects a Linear Magnitude graph for                         
             display on the active channel&#39;s                              
             s-parameter.                                                 
LINEAR MAG   Selectes Linear Magnitude and                                
AND PHASE    Phase graphs for display on the                              
             channel&#39;s s-parameter.                                       
REAL         Selects Real data for display on the active                  
             channel&#39;s s-parameter.                                       
IMAGINARY    Selects Imaginary data for display on the                    
             active channel&#39;s s-parameter.                                
REAL AND     Selects both Real and Imaginary data for                     
IMAGINARY    display on the active channel&#39;s                              
             s-parameter.                                                 
MORE         Takes you to additional graph type                           
             selections                                                   
PRESS &lt;ENTER&gt;                                                             
             Pressing the ENTER key implements                            
TO SELECT AND                                                             
             your menu selection and resumes the                          
RESUME       calibration from where it left off, if                       
CALIBRATION  in the calibration mode. The menu                            
             remains on the screen until another                          
             menu is selected for display or until                        
             the CLEAR/RET LOC key is pressed.                            
______________________________________                                    
 
    
     
         ______________________________________                                    
MENU         DESCRIPTION                                                  
______________________________________                                    
TRACE MEMORY                                                              
FUNCTIONS                                                                 
VIEW DATA    Displays measured data; that is, the data                    
             presently being taken.                                       
VIEW MEMORY  Displays stored data; that is, data that                     
             was previously taken and stored                              
             in memory.                                                   
VIEW DATA    Displays measured data superimposed                          
AND MEMORY   over stored data.                                            
VIEW DATA +  Displays measured data divided by                            
MEMORY       stored data.                                                 
SELECT       Takes you to Menu NO2 for selection of                       
TRACE MATH   the type of math operation to be                             
             performed.                                                   
STORE DATA   Stores the measured data to internal                         
TO MEMORY    memory.                                                      
DISK         Brings up Menu NO3, which allows data                        
FUNCTIONS    to be stored to or recalled from the disk.                   
MEMORY DATA  Indicates the reference delay applied to                     
REF, DELAY   the memory data being displayed.                             
XXX.XXX CM                                                                
PRESS &lt;ENTER&gt;                                                             
             Pressing the ENTER key implements                            
TO SELECT    your menu selection and remains on the                       
             screen until another menu is selected for                    
             display or until the CLEAR/RET LOC                           
             key is pressed.                                              
______________________________________                                    
 
    
     
         ______________________________________                                    
MENU         DESCRIPTION                                                  
______________________________________                                    
SELECT                                                                    
TRACE MATH                                                                
ADD (+)      Selects DATA + MEMORY                                        
             as the math function.                                        
SUBTRACT (-) Selects DATA - MEMORY                                        
             as the math function.                                        
MULTIPLY (×)                                                        
             Selects DATA × MEMORY                                  
             as the math function.                                        
DIVIDE (÷)                                                            
             Selects DATA ÷ MEMORY                                    
             as the math function.                                        
PRESS &lt;ENTER&gt;                                                             
             Pressing the ENTER key implements                            
TO SELECT    your menu selection. The menu returns                        
             to the NO1 menu.                                             
______________________________________                                    
 
    
     
         ______________________________________                                    
MENU         DESCRIPTION                                                  
______________________________________                                    
TRACE MEMORY                                                              
DISK FUNCTIONS                                                            
CHANNEL X    Indicates the channel to be                                  
             used (active channel).                                       
STORE TO DISK                                                             
             Displays GP1-3 or DSK11 menu to                              
             select file store data from                                  
             selected channel on disk                                     
RECALL FROM  Displays DSK9 menu to select                                 
DISK         file to recall from disk.                                    
PRESS &lt;ENTER&gt;                                                             
             Pressing the ENTER key implements                            
TO SELECT    your menu selection. The menu                                
             remains on the screen until another                          
             menu is selected for display or                              
             until the CLEAR/RET LOC                                      
             key is pressed.                                              
______________________________________                                    
 
    
     
         ______________________________________                                    
MENU         DESCRIPTION                                                  
______________________________________                                    
SET                                                                       
REFERENCE                                                                 
DELAY                                                                     
AUTO         Automatically sets the reference delay                       
             so that the cumulative phase shift is                        
             zero. This selection unwinds the phase                       
             in a Smith chart display or reduces the                      
             phase revolutions in a rectilinear                           
             display to less than one.                                    
DISTANCE     Electrically repositions the measurement                     
XXX.XXX mm   reference plane, as displayed on the                         
             active channel, by a distance value                          
             entered in millimeters. This selection                       
             lets you compensate for the phase                            
             reversals inherent in a length of                            
             transmission line connected between                          
             the test set&#39;s Port 1 connector                              
             and the device-under-test (DUT).                             
TIME         Electrically repositions the measurement                     
XXX.XXX ms   reference plane by a distance                                
             value that corresponds to the time in                        
             milliseconds.                                                
SET          Displays Menu RD2, which lets you                            
DIELECTRIC   enter a value for the dielectric                             
             constant of your transmission line.                          
PRESS &lt;ENTER&gt;                                                             
             Pressing the ENTER key implements                            
TO SELECT    your menu selection. The menu                                
             remains on the screen until another menu                     
             is selected for display or until the                         
             CLEAR/RET LOC key is pressed.                                
______________________________________                                    
 
    
     
         ______________________________________                                    
MENU         DESCRIPTION                                                  
______________________________________                                    
SET                                                                       
DIELECTRIC                                                                
CONSTANT                                                                  
AIR          Calculates reference delay based on                          
(1.000)      dielectric constant of air (1).                              
POLYETHYLENE Calculates reference delay based on                          
(2.260)      the dielectric constant of polyethylene                      
             (2.26).                                                      
TEFLON       Calculates reference delay based on the                      
(2.100)      dielectric constant of teflon (2.1).                         
MICROPOROUS  Calculates reference delay based on the                      
(1.69)       dielectric constant of microporous teflon                    
             (1.687).                                                     
OTHER        Calculates reference delay based on the                      
XXXX.XX      value you enter. Terminate your entry                        
             using any terminator and select                              
             with the ENTER key.                                          
PRESS &lt;ENTER&gt;                                                             
             Pressing the ENTER key implements                            
TO SELECT    your menu selection and returns                              
             you to the RD1 menu.                                         
______________________________________                                    
 
    
     
         ______________________________________                                    
MENU         DESCRIPTION                                                  
______________________________________                                    
SET SCALING                                                               
OR PRESS                                                                  
&lt;AUTOSCALE&gt;                                                               
LOG MAG-                                                                  
RESOLUTION   Sets the resolution for the vertical axis                    
XX.XXX dB/DIV                                                             
             of the active channel&#39;s displayed log                        
             magnitude graph. Resolution can by set                       
             incrementally using the keypad or rotary                     
             knob. For the linear polar graph, the                        
             center is fixed at 0 units; therefore,                       
             changing the resolution also changes                         
             the reference value and vice versa                           
REF VALUE    Sets the value by which the active                           
XXX.XXX dB   channel&#39;s amplitude measurement                              
             is offset on the log-magnitude graph.                        
             The offset can be set in increments of                       
             0.001 dB using the keypad or rotary knob.                    
REFERENCE    Sets the reference line for the active                       
LINE         channel&#39;s amplitude measurement on the                       
             log-magnitude graph. This is the line                        
             about which the amplitude expands with                       
             different resolution values. The reference                   
             line can be set to any vertical division                     
             using the rotary knob.                                       
PHASE-                                                                    
RESOLUTION   Sets the resolution for the vertical                         
XX.XX DEG/DIV                                                             
             axis of the active channel&#39;s                                 
             displayed phase graph. Resolution can                        
             by set incrementally using the keypad or                     
             rotary knob.                                                 
REF VALUE    Sets the value by which the active                           
XXX.XX DEG   channel&#39;s phase measurement is offset                        
             on the phase graph. The offset can be                        
             set in increments of 0.01 degrees                            
             using the keypad or rotary knob.                             
REFERENCE    Sets the reference line for the active                       
LINE         channel&#39;s phase measurements on the                          
             phase graph. This is the line about which                    
             the phase expands with different                             
             resolution values. The reference line                        
             can be set to any vertical division using                    
             the rotary knob.                                             
______________________________________                                    
 
    
     
         ______________________________________                                    
MENU         DESCRIPTION                                                  
______________________________________                                    
SET SCALING                                                               
OR PRESS                                                                  
&lt;AUTOSCALE&gt;                                                               
IMPEDANCE    Scales an Impedance Smith chart for                          
(ADMITTANCE) display in the active channel.                               
SMITH CHART                                                               
NORMAL SMITH Selects a normal Smith chart for display                     
(REFL = 1.000                                                             
             in the active channel.                                       
FULL SCALE)                                                               
EXPAND 10 dB Selects a dB expansion of the Smith chart                    
(REFL = 0.316                                                             
             being displayed for the active channel.                      
FULL SCALE)                                                               
EXPAND 20 dB Selects a 20 dB expansion of the Smith                       
(REFL = 0.099                                                             
             chart being displayed for the                                
FULL SCALE)  active channel.                                              
EXPAND 30 dB Selects a 30 dB expansion of the Smith                       
(REFL = 0.031                                                             
             chart being displayed for the                                
FULL SCALE)  active channel.                                              
COMPRESS 3 dB                                                             
             Selects a 3 dB compression of the Smith                      
(REFL = 1.413                                                             
             Chart being displayed for the                                
FULL SCALE)  active channel.                                              
PRESS &lt;ENTER&gt;                                                             
             Pressing the ENTER key implements                            
TO SELECT AND                                                             
             your menu selection and resumes the                          
RESUME       calibration from where it left off,                          
CALIBRATION  if in the calibration mode.                                  
             The menu remains on the                                      
             screen until another menu is selected                        
             for display or until the                                     
             CLEAR/RET LOC key is pressed.                                
______________________________________                                    
 
    
     
         ______________________________________                                    
MENU         DESCRIPTION                                                  
______________________________________                                    
DATA                                                                      
ENHANCEMENT                                                               
AVERAGING    Averages the measured data over time,                        
XXXX MEAS.   as follows:                                                  
PER POINT    1. The sweep stops at the first frequency                    
             point and takes a number of readings,                        
             based on the selected numbvr of points.                      
             2. The program averages the readings and                     
             writes the average value for that                            
             frequency point in the displayed graph.                      
             3. The sweep then advances to the next                       
             sequential frequency point and repeats                       
             the process.                                                 
SMOOTHING    Smoothes the measured data over                              
XX.X PERCENT frequency, as follows:                                       
OF SPAN      1. The program divides the overall sweep                     
             into smaller segments, based on the                          
             selected percent-of-span. (Refer                             
             to paragraphs 3-5.3 and FIG.                                 
             3-29 for a description and example                           
             of smoothing.)                                               
             2. It takes a data reading at each                           
             frequency point within that percent-                         
             of-span segment.                                             
             3. It averages the readings with a raised                    
             Hamming window and writes that                               
             magnitude value at the mid-frequency                         
             point of the segment in the displayed                        
             graph or Smith chart.                                        
             4. It then advances the percent-of-span                      
             segment to encompass the next sequential                     
             group of frequency points and repeats                        
             the process.                                                 
______________________________________                                    
 
    
     
         ______________________________________                                    
MENU         DESCRIPTION                                                  
______________________________________                                    
DATA                                                                      
ENHANCEMENT                                                               
AVERAGING    Averages the measured data over time,                        
XXXX MEAS.   as follows:                                                  
PER POINT    1. The sweep stops at the first frequency                    
             point and takes a number of readings,                        
             based on the selected number of points.                      
             2. The program averages the readings and                     
             writes the average value for that                            
             frequency point in the displayed graph.                      
             3. The sweep then advances to the next                       
             sequential frequency point and repeats                       
             the process.                                                 
SMOOTHING    Smoothes the measured data over                              
XX.X PERCENT frequency, as follows:                                       
OF SPAN      1. The program divides the overall sweep                     
             into smaller segments, based on the                          
             selected percent-of-span. (Refer                             
             to paragraphs 3-5.3 and FIG.                                 
             3-29 for a description and example                           
             of smoothing.)                                               
             2. It takes a data reading at each                           
             frequency point within that percent-                         
             of-span segment.                                             
             3. It averages the readings with a raised                    
             Hamming window and writes that                               
             magnitude value at the mid-frequency                         
             point of the segment in the displayed                        
             graph or Smith chart.                                        
             4. It then advances the percent-of-span                      
             segment to encompass the next sequential                     
             group of frequency points and repeats                        
             the process.                                                 
______________________________________                                    
 
    
     
         ______________________________________                                    
MENU         DESCRIPTION                                                  
______________________________________                                    
SELECT OUTPUT                                                             
DEVICE                                                                    
PRINTER      Selects the printer as your output device.                   
PLOTTER      Selects the plotter as your output device.                   
SELECT PRINTER                                                            
OUTPUT TYPE                                                               
FULL SCREEN  Prints full-screen data, including                           
             the menu entries.                                            
GRAPH ONLY   Prints only the graph or Smith chart,                        
             including any and all data it contains.                      
TABULAR DATA Prints a tabulation of the measured data.                    
OUTPUT OPTIONS                                                            
SET UP OUTPUT                                                             
             Calls Menu PM2, which allows you to                          
HEADERS      enter the header information.                                
DISK         Calls Menu PM4, which allows you to                          
OPERATION    select disk operations.                                      
PLOT OPTIONS Calls Menu PL1, which allows you to                          
PRESS &lt;ENTER&gt;                                                             
             select between several plot options.                         
TO SELECT    Pressing the ENTER key implements                            
             your menu selection. The menu remains                        
             on the screen until another menu is                          
             selected for display or until the                            
             CLEAR/RET LOC key is pressed.                                
______________________________________                                    
 
    
     
         ______________________________________                                    
MENU         DESCRIPTION                                                  
______________________________________                                    
DATA OUTPUT                                                               
HEADERS                                                                   
MODEL ON (OFF) Selecting &lt;1&gt; displays Menu                                
XXXXXXXXXXXXXX GP5, which lets you select the letters                     
               and/or numbers in your model                               
               identifier.                                                
DEVICE ID      Selecting &lt;1&gt; displays Menu                                
ON (OFF)       GP5, which lets you select the letters                     
XXXXXXXXXXXX   and/or numbers in your Device                              
               I.D. identifier.                                           
DATE ON (OFF)  Selecting &lt;1&gt; displays Menu                                
XX/XX/XX       GP5, which lets you select the letters                     
               and/or numbers in the date.                                
OPERATOR       Selecting &lt;1&gt; displays Menu                                
ON (OFF)       GP5, which lets you select the letters                     
XXXXXXXXXXXXXXX                                                           
               identifying the operator.                                  
PRESS &lt;ENTER&gt;  Pressing the ENTER key selects                             
TO TURN ON/OFF between menu selections. Pressing                          
OR             the CLEAR/RET LOC key lets you                             
PRESS &lt;1&gt;      change the between ON and OFF                              
TO CHANGE      states                                                     
               Pressing &lt;1&gt; lets you enter the                            
               desired label in Menu GP5.                                 
______________________________________                                    
 
    
     
         ______________________________________                                    
MENU         DESCRIPTIONS                                                 
______________________________________                                    
TABULAR OUTPUT                                                            
FORMAT                                                                    
MARKER ON (OFF)                                                           
             Provides for the printing or                                 
DATA         not of markers.                                              
SWEEP ON (OFF)                                                            
             Provides for the printing or not                             
DATA         of frequency sweep data. If you elect to                     
             print the sweep data, you can choose how                     
             many frequency points to print out.                          
XXX POINTS   Outputs one point every X points                             
PRESS &lt;ENTER&gt;                                                             
             Pressing the ENTER key implements                            
TO TURN ON/OFF                                                            
             your menu selection. The menu remains                        
OR           on the screen until another menu is                          
TURN KNOB TO selected for display or until the                            
CHANGE NUMBER                                                             
             CLEAR/RET LOC key is pressed.                                
OF POINTS    Turning the knob on number of points                         
             changes the value of X to define                             
             the number of points printed.                                
______________________________________                                    
 
    
     
         ______________________________________                                    
MENU         DESCRIPTION                                                  
______________________________________                                    
DISK OUTPUT                                                               
OPERATIONS                                                                
TABULAR DATA Outputs tabular data to the disk and                         
TO DISK      takes you to GP1-3 or DSK 11 for                             
             selection of a file name.                                    
TABULAR DATA Brings up DSK9 for selection of a                            
FROM DISK    measurement data file to be output                           
TO PRINTER   to the printer.                                              
PRESS &lt;ENTER&gt;                                                             
             Pressing the ENTER key implements                            
TO SELECT    your menu selection. The menu remains                        
             on the screen until another menu is                          
             selected for display or until the                            
             CLEAR/RET LOC key is pressed.                                
______________________________________                                    
 
    
     
         ______________________________________                                    
MENU         DESCRIPTION                                                  
______________________________________                                    
PLOT OPTIONS                                                              
PLOT                                                                      
FULL PLOT    The plotter will plot everything displayed                   
             on the screen (data traces, graticule, menu                  
             text) when START PRINT is pressed.                           
OPTIONS                                                                   
HEADER ON (OFF)                                                           
             The plot will include                                        
             on information header if this                                
             option is on and START PRINT                                 
             is pressed.                                                  
MENU ON (OFF)                                                             
             The plot will include the menu text if                       
             this option is on and START PRINT                            
             is pressed.                                                  
MARKERS ON   The plot will include any marker or                          
(OFF) AND    limit lines if this option is on and START                   
LIMITS       PRINT is pressed.                                            
GRATICULE ON The plot will include the graticule and                      
(OFF)        annotation if this option is on and                          
             START PRINT is pressed. The plotter                          
             plots the graticule.                                         
DATA ON (OFF)                                                             
             The plot will include the data if this                       
TRACE(S)     option is on and START PRINT                                 
             is pressed. The plotter plots the                            
             graticule.                                                   
FORMAT                                                                    
PLOT SIZE    Calls Menu PL2, which lets you select the                    
             size and location of the plot.                               
PEN COLORS   Calls Menu PL3, which lets you select                        
             pen colors for the various elements of                       
             the plot: graticule, data traces, menu                       
             text and header. Also lets you select                        
             the relative pen speed.                                      
PRESS &lt;ENTER&gt;                                                             
             Pressing the ENTER key implements                            
TO SELECT    your menu selection. The menu remains                        
OR TURN ON/OFF                                                            
             on the screen until another menu                             
             is selected for display or until the                         
             CLEAR/RET LOC key is pressed.                                
______________________________________                                    
 
    
     
         ______________________________________                                    
MENU         DESCRIPTION                                                  
______________________________________                                    
SELECT                                                                    
PLOT SIZE                                                                 
FULL SIZE    Selects a full size (page) plot.                             
QUARTER                                                                   
SIZE PLOTS-                                                               
UPPER LEFT   Selects a quarter-size plot,                                 
             upper-left quadrant.                                         
UPPER RIGHT  Selects a quarter-size plot,                                 
             upper-right quadrant.                                        
LOWER LEFT   Selects a quarter-size plot,                                 
             lower-left quadrant.                                         
LOWER RIGHT  Selects a quarter-size plot,                                 
             lower-right quadrant.                                        
PRESS &lt;ENTER&gt;                                                             
             Pressing the ENTER key implements                            
TO SELECT    your menu selection. The menu remains                        
             on the screen until another menu is                          
             selected for display or until                                
             the CLEAR/RET LOC key is pressed.                            
______________________________________                                    
 
    
     
         ______________________________________                                    
MENU         DESCRIPTION                                                  
______________________________________                                    
SELECT                                                                    
PEN COLORS                                                                
DATA PEN     Selects the color in which the data will                     
n            be plotted. The number of the pen displays                   
             where the &#34;n&#34; is shown.                                      
GRATICLE PEN Selects the color in which the graticule                     
n            will be plotted. The number of the pen                       
             displays where the &#34;n&#34; is shown.                             
MARKERS AND  Selects the color in which the markers                       
LIMITS PEN   and limits will be plotted.                                  
n            The number of the pen displays where the                     
             &#34;n&#34; is shown.                                                
HEADER PEN   Selects the color in which the header                        
n            information will be plotted. The                             
             number of the pen displays where the                         
             &#34;n&#34; is shown.                                                
PEN SPEED    Selects the pen&#39;s speed as a percentage                      
100 PERCENT  of the plotter&#39;s maximum speed. (Used                        
OF MAXIMUM   to optimize plots on transparencies or                       
             with worn pens.)                                             
PREVIOUS MENU                                                             
             Recalls Menu PL1.                                            
PRESS &lt;ENTER&gt;                                                             
             Pressing the ENTER key implements                            
TO SELECT    your menu selection. The menu                                
             remains on the screen until another                          
             menu is selected for display or until the                    
             CLEAR/RET LOC key is pressed.                                
______________________________________                                    
 
    
     
         ______________________________________                                    
Function     Default Setting                                              
______________________________________                                    
INSTRUMENT   Measurement Setup Menu Displayed                             
STATE                                                                     
MEASUREMENT  Maximum sweep range of source and                            
             test set                                                     
             Source Power: 0.0 dBm                                        
             Resolution: Normal                                           
CHANNEL      Quad (four-channel) display                                  
             Channel 1 active                                             
DISPLAY      Channel 1: S.sub.11, 1:1 Smith Chart                         
             Channel 2: S.sub.12, Log Magnitude and Phase                 
             Channel 3: S.sub.21, Log Magnitude and Phase                 
             Channel 4: S.sub.22, 1:1 Smith Chart                         
             Scale: 10 dB/Division or 90°/Division                 
             Offset: 0.000dB or 0.00 degree                               
             Reference Position: Midscale                                 
             Electrical Delay: 0.00 seconds                               
             Dielectric: 1.00 (air)                                       
             Normalization: Off                                           
             Normalization Sets: Unchanged                                
ENHANCEMENT  Video IF Bandwidth: Reduced                                  
             Averaging: Off                                               
             Smoothing: OFF                                               
CALIBRATION  Correction: Off                                              
             Connector: SMA                                               
             Load: Broadband                                              
MARKERS LIMITS                                                            
             Markers On/Off: All off                                      
             Markers Enabled/Disabled: All enabled                        
             Marker Frequency: All set to the start-                      
             sweep frequency or start-time distance                       
             ΔReference: Off                                        
SYSTEM STATE Limits: All set to reference position                        
             value; all off (all enabled)                                 
             GP/B Addresses and Terminators:                              
             Unchanged                                                    
             Frequency Blanking: Disengaged,                              
             Error(s): All cleared                                        
             Measurement: Restarted                                       
OUTPUT       Output Type: Printer (full screen,                           
             clear headers)                                               
             Marker and Sweep Data: Enabled                               
______________________________________                                    
 
    
     
         ______________________________________                                    
MENU         DESCRIPTION                                                  
______________________________________                                    
SELECT UTILITY                                                            
FUNCTION TYPE                                                             
GENERAL DISK Calls Menu U2, which lets you select                         
UTILITIES    between several disk utilities.                              
CALIBRATION  Calls Menu U4, which lets you select                         
COMPONENT    between several calibration-component                        
UTILITIES    utilities.                                                   
GPIB         Displays the current GPIB addresses of                       
ADDRESSES    the various system instruments.                              
DISPLAY      Calls Menu U3, which lets you display                        
INSTRUMENT   the various instrument state                                 
STATE        parameters.                                                  
BLANK        Blanks all frequency-identifier information                  
FREQUENCY    from the 360 displays, if such information                   
INFORMATION  is presently being displayed.                                
ALTERNATE 3RD                                                             
             Switches between blue and cyan colors                        
COLOR PLANE                                                               
PRESS &lt;ENTER&gt;                                                             
             Switches between blue and cyan                               
TO SELECT    colors Pressing the ENTER key                                
             implements your menu selection. The                          
             menu remains on the screen until another                     
             menu is selected for display or until the                    
             CLEAR/RET LOC key is pressed.                                
______________________________________                                    
 
    
     
         ______________________________________                                    
MENU         DESCRIPTION                                                  
______________________________________                                    
DISK UTILITY                                                              
FUNCTIONS                                                                 
RESTORE                                                                   
DISPLAY                                                                   
DISPLAY DISK Displays disk directory in data area                         
DIRECTORY                                                                 
DELETE FILES Calls Menu DSK2, which lets you delete                       
FROM DISK    files from the disk.                                         
LOAD PROGRAM Reloads the operating program                                
FROM DISK    from the disk.                                               
TO 360                                                                    
              ##STR1##                                                    
INITIALIZE   Prepares (formats or initializes) the disk                   
DISK WITH    for use with the 360, including the                          
PROGRAM      operating program.                                           
INITIALIZE   Prepares the disk for use with the 360 but                   
DATA-ONLY    does not copy the operating program                          
DISK         to the disk                                                  
              ##STR2##                                                    
COPY DISK    Lets you copy one disk to another.                           
TO DISK                                                                   
______________________________________                                    
 
    
     
         ______________________________________                                    
MENU U3        DESCRIPTION                                                
______________________________________                                    
    DISPLAY                                                               
    INSTRUMENT                                                            
    STATE                                                                 
    RESTORE        Restores the normal data display.                      
    DISPLAY                                                               
A   SYSTEM         Displays all of the system parameters                  
    PARAMETERS     (Readout Text U4 a thru e. FIG.                        
                   3-101).                                                
B   CALIBRATION    Displays the calibration parameters.                   
    PARAMETERS                                                            
C   GLOBAL         Displays the global operating                          
    OPERATING      parameters.                                            
    PARAMETERS                                                            
D   CHANNEL 1-2    Displays the Channel 1-2                               
    OPERATING      operating parameters.                                  
    PARAMETERS                                                            
E   CHANNEL 3-4    Displays the Channel 3-4                               
    OPERATING      operating parameters.                                  
    PARAMETERS                                                            
    NEXT PAGE      Alternately displays Readout Text                      
                   U4 a thru e.                                           
    PRESS &lt;ENTER&gt;  Pressing the ENTER key implements                      
    TO SELECT      your menu selection. The menu                          
                   remains on the screen until another                    
                   menu is selected for display or until                  
                   the CLEAR/RET LOC key is                               
                   pressed.                                               
______________________________________                                    
 
    
     
         ______________________________________                                    
Readout Text U3e, Calibration Parameters                                  
Parameter  Display Format                                                 
______________________________________                                    
Number Of Points                                                          
           XXX                                                            
Source Power                                                              
           XXX dBm                                                        
Cal Type   xxxx                                                           
Start Frequency                                                           
           XX.XXX GHz                                                     
Stop Frequency                                                            
           XX.XXX GHz                                                     
Load Type  XXXXXXXXX                                                      
Connector                                                                 
PORT1-                                                                    
PORT2-                                                                    
           XXXXXXXXX      XXXXXXXXX                                       
Open Device                                                               
           *NOT INSTALLED *NOT INSTALLED                                  
 C0 (e-15)                                                                
XXX.XXX                                                                   
XXX.XXX                                                                   
 C1 (e-27)                                                                
XXX.XXX                                                                   
XXX.XXX                                                                   
 C2 (e-36)                                                                
XXX.XXX                                                                   
XXX.XXX                                                                   
 C3 (e-45)                                                                
XXX.XXX                                                                   
XXX.XXX                                                                   
 Offset Length                                                            
XXX.XXX mm                                                                
XXX.XXX mm                                                                
           •                                                        
 Serial Number                                                            
           XXXXXXXXX      XXXXXXXXX                                       
Short Device                                                              
           *NOT INSTALLED *NOT INSTALLED                                  
 Offset Length                                                            
XXX.XXX mm                                                                
XXX.XXX mm                                                                
 Serial Number                                                            
           XXXXXXXXX      XXXXXXXXX                                       
Atten Settings                                                            
 Source    XXdB           XX. dB                                          
 Test                     XX dB                                           
______________________________________                                    
 *If not installed, displays &#34;NOT INSTALLEED&#34;-                            
 
    
     
         ______________________________________                                    
MENU U4      DESCRIPTION                                                  
______________________________________                                    
CALIBRATION                                                               
COMPONENT                                                                 
UTILITIES                                                                 
INSTALL      Reads into memory the coefficient data                       
CALIBRATION  from the calibration-components disk                         
COMPONENT    supplied with the calibration kits.                          
INFORMATION                                                               
FROM DISK                                                                 
DISPLAY      Calls Menu U5, which lets you display the                    
INSTALLED    connector information for the                                
CALIBRATION  various connectors supported.                                
COMPONENT                                                                 
INFORMATION                                                               
PRESS &lt;ENTER&gt;                                                             
             Pressing the ENTER key implements                            
TO SELECT    your menu selection. The menu remains                        
             on the screen until another menu is                          
             selected for display or until the                            
             CLEAR/RET LOC key is pressed.                                
______________________________________                                    
 
    
     
         ______________________________________                                    
MENU U5      DESCRIPTION                                                  
______________________________________                                    
DISPLAY      This menu lets you view coefficient data                     
INSTALLED    on the components listed (FIG. 3-104).                       
CALIBRATION  The data appears in the display area of the                  
COMPONENT    screen.                                                      
INFORMATION                                                               
SMA (M)      Select coefficient data to display for the                   
             SMA male components.                                         
SMA (F)      Select coefficient data to display for the                   
             SMA female components.                                       
K - CONN (M) Select coefficient data to display for the                   
             K Connector ™ male components.                            
K - CONN (F) Select coefficient data to display for the                   
             K Connector female components.                               
TYPE N (M)   Select coefficient data to display for the                   
             Type N male components.                                      
TYPE N (F)   Select coefficient data to display for the                   
             Type N female components.                                    
GPC - 7      Select coefficient data to display for the                   
             sexless GPC-7 components.                                    
NEXT         Cycles through selections SMA (M)                            
COMPONENT    to GPG7.                                                     
PREVIOUS MENU                                                             
             Displays menu U4.                                            
PRESS &lt;ENTER&gt;                                                             
             Pressing the ENTER key implements                            
TO SELECT    your menu selection. The menu remains                        
             on the screen until another menu is                          
             selected for display or until the                            
             CLEAR/RET LOC key is pressed.                                
______________________________________                                    
 
    
     
         ______________________________________                                    
MENU DSK3    DESCRIPTION                                                  
______________________________________                                    
NO ROOM FOR                                                               
NEW DATA FILES                                                            
OVERWRITE    Allows you to overwrite an existing file                     
EXISTING     on the current data disk.                                    
FILES                                                                     
DELETE       Allows you to delete an existing file on                     
EXISTING     the current data disk.                                       
FILES                                                                     
PRESS &lt;ENTER&gt;                                                             
             Pressing ENTER implements your menu                          
TO SELECT    selection. You will be returned to the                       
             previous menu when your selection is                         
             made.                                                        
______________________________________                                    
 
    
     
         ______________________________________                                    
MENU DSK4    DESCRIPTION                                                  
______________________________________                                    
DISK UTILITY                                                              
FUNCTIONS                                                                 
DELETE CAL &amp; Allows you to delete specific calibration                    
FRONT PANEL  and control (front) panel setup files from                   
SETUPS FROM  the current data disk.                                       
DISK                                                                      
DELETE       Allows you to delete specific                                
NORMALIZATION                                                             
             normalization (trace memory) files from                      
FILES FROM   the current data disk.                                       
DISK                                                                      
DELETE       Allows you to delete specific measure-                       
MEASUREMENT  ment data files from the current data disk.                  
FILES FROM                                                                
DISK                                                                      
PREVIOUS MENU                                                             
             Returns you to the previous menu                             
PRESS &lt;ENTER&gt;                                                             
             Pressing the ENTER key implements                            
TO SELECT    your menu selection. The menu remains                        
             on the screen until another menu is                          
             selected for display or until the                            
             CLEARE/LOC key is pressed.                                   
______________________________________                                    
 
    
     
         ______________________________________                                    
MENU DSK5    DESCRIPTION                                                  
______________________________________                                    
SELECT CAL &amp;                                                              
FRONT PANEL                                                               
SETUP TO                                                                  
DELETE                                                                    
FILENAME 2   Selects the file named on this line to be                    
             deleted. (The actual name of the file,                       
             not &#34;FILENAME 2&#34; will appear.)                               
FILENAME 3   Same as for Filename 2 above.                                
FILENAME 4   Same as for Filename 2 above.                                
FILENAME 5   Same as for Filename 2 above.                                
FILENAME 6   Same as for Filename 2 above.                                
FILENAME 7   Same as for Filename 2 above.                                
FILENAME 8   Same as for Filename 2 above.                                
FILENAME 9   Same as for Filename 2 above.                                
MORE         The &#34;More&#34; option only displays if                           
             there are more than nine files.                              
PREVIOUS MENU                                                             
             Returns to the previous menu in this                         
             series.                                                      
PRESS &lt;ENTER&gt;                                                             
             Pressing the ENTER key implements                            
TO SELECT    your menu selection. You are                                 
             returned to the previous menu.                               
______________________________________                                    
 
    
     
         ______________________________________                                    
MENU DSK6    DESCRIPTION                                                  
______________________________________                                    
SELECT                                                                    
NORMALIZATION                                                             
FILE TO                                                                   
DELETE                                                                    
FILENAME 1   Selects the file named on this line to be                    
             deleted. (The actual name of the file,                       
             not &#34;FILENAME 1&#34; will appear.)                               
FILENAME 2   Same as for Filename 1 above.                                
FILENAME 3   Same as for Filename 1 above.                                
FILENAME 4   Same as for Filename 1 above.                                
FILENAME 5   Same as for Filename 1 above.                                
FILENAME 6   Same as for Filename 1 above.                                
FILENAME 7   Same as for Filename 1 above.                                
FILENAME 8   Same as for Filename 1 above.                                
MORE         The &#34;More&#34; option only displays if                           
             there are more than nine files.                              
PREVIOUS MENU                                                             
             Returns to the previous menu in this                         
             series.                                                      
PRESS &lt;ENTER&gt;                                                             
             Pressing the ENTER key implements                            
TO SELECT    your menu selection. You are                                 
             returned to the previous menu.                               
______________________________________                                    
 
    
     
         ______________________________________                                    
MENU DSK7    DESCRIPTION                                                  
______________________________________                                    
SELECT                                                                    
MEASUREMENT                                                               
FILE TO                                                                   
DELETE                                                                    
FILENAME 1   Selects the file named on this line to be                    
             deleted. (The actual name of the file,                       
             not &#34;FILENAME 1&#34; will appear.)                               
FILENAME 2   Same as for Filename 1 above.                                
FILENAME 3   Same as for Filename 1 above.                                
FILENAME 4   Same as for Filename 1 above.                                
FILENAME 5   Same as for Filename 1 above.                                
FILENAME 6   Same as for Filename 1 above.                                
FILENAME 7   Same as for Filename 1 above.                                
FILENAME 8   Same as for Filename 1 above.                                
MORE         The &#34;More&#34; option only displays if                           
             there are more than nine files.                              
PREVIOUS MENU                                                             
             Returns to the previous menu in this                         
             series.                                                      
PRESS &lt;ENTER&gt;                                                             
             Pressing the ENTER key implements                            
TO SELECT    your menu selection. You are                                 
             returned to the previous menu                                
______________________________________                                    
 
    
     
         ______________________________________                                    
MENU DSK9    DESCRIPTION                                                  
______________________________________                                    
SELECT FILE                                                               
TO READ                                                                   
FILENAME 1   Selects the file named on this line to be                    
             read. (The actual name of the file,                          
             not &#34;FILENAME 1&#34; will appear.)                               
FILENAME 2   Same as for Filename 1 above.                                
FILENAME 3   Same as for Filename 1 above.                                
FILENAME 4   Same as for Filename 1 above.                                
FILENAME 5   Same as for Filename 1 above.                                
FILENAME 6   Same as for Filename 1 above.                                
FILENAME 7   Same as for Filename 1 above.                                
FILENAME 8   Same as for Filename 1 above.                                
MORE         The &#34;More&#34; option only displays if                           
             there are more than nine files.                              
PREVIOUS MENU                                                             
             Returns to the previous menu in this                         
             series                                                       
PRESS &lt;ENTER&gt;                                                             
             Pressing the ENTER key implements                            
TO SELECT    your menu selection. You are                                 
             returned to the previous menu.                               
______________________________________                                    
 
    
     
         ______________________________________                                    
MENU GP1-3   DESCRIPTION                                                  
______________________________________                                    
SELECT FILE                                                               
TO OVERWRITE                                                              
OR                                                                        
CREATE NEW                                                                
FILE                                                                      
FILENAME 1   Selects disk File 1 for storing the                          
             calibration data or front (control) panel                    
             setup. Name the file using Menu GP5.                         
FILENAME 2   Selects disk File 2 for storage of data.                     
FILENAME 3   Selects disk File 3 for storage of data.                     
FILENAME 4   Selects disk File 4 for storage of data.                     
FILENAME 5   Selects disk File 5 for storage of data.                     
FILENAME 6   Selects disk File 6 for storage of data.                     
FILENAME 7   Selects disk File 7 for storage of data.                     
FILENAME 8   Selects disk File 8 for storage of data.                     
MORE         Displays additional menus.                                   
PREVIOUS MENU                                                             
             Displays the previous menu.                                  
PRESS &lt;ENTER&gt;                                                             
             Pressing the ENTER key implements                            
TO SELECT    your menu selection. The menu remains                        
             on the screen until another menu is                          
             selected for display or until the                            
             CLEAR/RET LOC key is pressed.                                
______________________________________                                    
 
    
     
         __________________________________________________________________________
MENU GP5     DESCRIPTION                                                  
__________________________________________________________________________
SELECT NAME  Name your file using the rotary knob to select letters,      
             numbers, or both.                                            
             A letter or number turns red to indicate that the            
             letter/number has been                                       
ABCDEFGHIJKLM                                                             
             chosen for selection. Pressing the ENTER key selects the     
             letter or num-                                               
             ber. The name you spell out displays in the area below       
             &#34;SELECT NAME.&#34;                                               
NOPQRSTUVWXYZ                                                             
             You are allowed up to eight characters for a file name and   
             twelve charac-                                               
             ters for a label.                                            
0123456789-/#                                                             
DEL CLEAR DONE                                                            
             Selecting &#34;DEL&#34; deletes the last letter in the name          
             displayed above.                                             
             Selecting &#34;CLEAR&#34; deletes the entire name.                   
             Selecting &#34;DONE&#34; signals that you have finished writing the  
             name.                                                        
TURN KNOB    Use the rotary knob to indicate the letter or number you     
             wish to select.                                              
TO INDICATE  You can use the up-arrow and down-arrow keys to move between 
             rows.                                                        
CHARACTER OR                                                              
FUNCTION                                                                  
PRESS &lt;ENTER&gt;                                                             
             Pressing the ENTER key implements your menu selection. The   
             menu                                                         
TO MAKE SELECTION                                                         
             remains on the screen until another menu is selected for     
             display or until                                             
             the CLEAR/RET LOC key is pressed.                            
NUMBERS MAY  You may also select numbers using the keypad.                
ALSO BE                                                                   
SELECTED                                                                  
USING KEYPAD                                                              
__________________________________________________________________________
 
    
     
         __________________________________________________________________________
MENU GP7   DESCRIPTION                                                    
__________________________________________________________________________
GPIB SETUP                                                                
360 GPIB-                                                                 
CR/CR-LF   Selects either the CR (carriage return) character or the CR-LF 
           (Carriage                                                      
           Return-Line Feed) characters as the data terminator for GPIB   
           transmis-                                                      
           sions.                                                         
ADDRESS:   Selects the GPIB address for the 360 analyzer. The 360 is set  
           to address                                                     
6          6 before leaving the factory.                                  
SYSTEM BUS-                                                               
CR/CR-LF   Selects either the CR (carriage return) character or the CR-LF 
           (Carriage                                                      
           Return-Line Feed) characters as the data terminator for System 
           Bus                                                            
           transmissions.                                                 
INSTR ADDR Selects the address for the System Bus controller. This is the 
           address the                                                    
0          360 uses to address the Source. The 360 is set to address 0    
           before leav-                                                   
           ing the factory.                                               
SOURCE ADDR                                                               
           Selects the address for the Source. The Source is set to       
           address 5 before                                               
5          leaving the factory.                                           
PLOTTER ADDR                                                              
           Selects the address for a compatible plotter.                  
PRESS &lt;ENTER&gt;                                                             
           Pressing the ENTER key implements your menu selection. The     
           menu                                                           
TOSELECT   remains on the screen until another menu is selected for       
           displayor until                                                
           the CLEAR/RET LOC key is pressed.                              
__________________________________________________________________________
 
    
     
         __________________________________________________________________________
MENU M1      DESCRIPTION                                                  
__________________________________________________________________________
SET MARKERS                                                               
MARKER 1 ON (OFF)                                                         
             Turns Marker 1 on or off (activates or deactivates). When on 
             (active), the                                                
XX.XXXXXX GHz                                                             
             frequency, time, or distance may be set using the keypad or  
             rotary knob.                                                 
             NOTE                                                         
             In this text, markers will be referred to as being active    
             and as                                                       
             being selected. Any marker that has been turned on and as-   
             signed a frequency is considered to be active. The marker    
             to                                                           
             which the cursor presently points is considered to be        
             selected.                                                    
             The selected marker is the only one for which you can        
             change                                                       
             the frequency.                                               
MARKER 2 ΔREF                                                       
XX.XXXXXX GHz                                                             
MARKER 3 ON (OFF)                                                         
             Turns Marker 3 on or off (activates or deactivates). When on 
             (active), the                                                
XX.XXXXXX GHz                                                             
             frequency, time, or distance may be set using the keypad or  
             rotary knob.                                                 
MARKER 4ON (OFF)                                                          
             Turns Marker 4 on or off (activates or deactivates). When on 
             (active), the                                                
             frequency, time, or distance may be set using the keypad or  
             rotary knob.                                                 
MARKER 5ON (OFF)                                                          
             Turns Marker 5 on or off (activates or deactivates). When on 
             (active), the                                                
             frequency, time, or distance may be set using the keypad or  
             rotary knob.                                                 
MARKER 6ON (OFF)                                                          
             Turns Marker 6 on or off (activates or deactivates). When on 
             (active), the                                                
XX.XXXXXX GHz                                                             
             frequency, time, or distance may be set using the keypad or  
             rotary knob.                                                 
MARKERS      Disables all markers.                                        
DISABLED                                                                  
ΔREF MODE ON (OFF)                                                  
             Selects and deselects the ΔReference Mode              
SELECT ΔREF                                                         
             Calls Menu M2, which lets you select the ΔREF Marker.  
MARKER                                                                    
PRESS &lt;ENTER&gt;                                                             
             Pressing the ENTER key implements your menu selection. The   
             menu                                                         
OR TURN ON/OFF                                                            
             remains on the screen until another menu is selected for     
             display or until                                             
TO SELECT    the CLEAR/RET LOC key is pressed.                            
__________________________________________________________________________
 
    
     
         __________________________________________________________________________
MENU M2    DESCRIPTION                                                    
__________________________________________________________________________
SELECT                                                                    
ΔREF MARKER                                                         
MARKER 1   Marker 1 only appears if it has been activated in Menu M1.     
           Placing the                                                    
XX.XXXXXX GHz                                                             
           cursor on Marker 1 and pressing the ENTER key here selects it  
           as the                                                         
           ΔREF marker. The ΔREF marker is the one from which 
           the other active                                               
           markers are compared and their difference frequency measured   
           and dis-                                                       
           played in Menu M3. The marker frequency may be set using the   
           keypad                                                         
           or rotary knob.                                                
MARKER 3   Same as above, but for Marker 3                                
XX.XXXXXX GHz                                                             
MARKER 4   Same as above, but for Marker 4                                
XX.XXXXXX GHz                                                             
PRESS &lt;ENTER&gt;                                                             
           Pressing the ENTER key implements your menu selection. The     
           menu                                                           
TO SELECT  remains on the screen until another menu is selected for       
           display or until                                               
           the CLEAR/RET LOC key is pressed.                              
__________________________________________________________________________
 
    
     
         __________________________________________________________________________
MENU M3     DESCRIPTION                                                   
__________________________________________________________________________
SELECT                                                                    
READOUT MARKER                                                            
MARKER 1    Displays the frequency and S-Parameter value(s) of Marker 1   
            on all CRT-                                                   
XX.XXXXXX GHz                                                             
            displayed graphs and Smith Charts. The frequency of Marker 1  
            also dis-                                                     
            plays here. If Marker 1 was activated in Menu M2 as the REF   
            marker,                                                       
            REF appears as shown for Marker M5 below.                     
MARKER 2    Same as for above, but for Marker 2.                          
XX.XXXXXX GHz                                                             
MARKER 5    Same as for above, but for Marker 5.                          
XX.XXXXXX GHz                                                             
ΔREF MODE                                                           
            Indicates the status of the ΔREF mode.                  
IS ON                                                                     
PRESS &lt;ENTER&gt;                                                             
            Pressing the ENTER key implements your menu selection. The    
            menu                                                          
TO SELECT   remains on the screen until another menu is selected for      
            display or until                                              
            the CLEAR/RET LOC key is pressed.                             
__________________________________________________________________________
 
    
     
         __________________________________________________________________________
MENU M4    DESCRIPTION                                                    
__________________________________________________________________________
C H 1-S 1 1                                                               
           Selects channel for readout                                    
XX.XXX PS DLY                                                             
MARKER 1   The selected marker-that is, the one to which the cursor       
           points in Menu                                                 
XX.XXXXXX GHz                                                             
           M1-and its frequency, time, or distance display here. This     
           could be any                                                   
XX.XXX dB  one of the six available markers: Marker 1 thru Marker 6.      
XXX.XXX DEG                                                               
MARKER TO MAX                                                             
           Causes the active marker to go to the frequency with the       
           greatest S-                                                    
           Parameter value on the active channel.                         
MARKER TO MIN                                                             
           Causes the selected marker to go to the frequency with the     
           smallest S-                                                    
           Parameter value on the active channel.                         
2 XX.XXXX GHz                                                             
           Displays the frequency, magnitude, and phase of the active S   
           Parameter                                                      
XX.XXX dB  at marker 2, if the marker is enabled.                         
XXX.XXX DEG                                                               
3 XX.XXXX GHz                                                             
           Displays the frequency, magnitude, and phase of the active S   
           Parameter                                                      
XX.XXX dB  at marker 3, if the marker is enabled.                         
XXX.XXX DEG                                                               
4 XX.XXXX GHz                                                             
           Displays the frequency, magnitude, and phase of the active S   
           Parameter                                                      
XX.XXX dB  at marker 4, if the marker is enabled.                         
XXX.XXX DEG                                                               
5 XX.XXXX GHz                                                             
           Displays the frequency, magnitude, and phase of the active S   
           Parameter                                                      
XX.XXX dB  at marker 5, if the marker is enabled.                         
XXX.XXX DEG                                                               
6 XX.XXXX GHz                                                             
           Displays the frequency, magnitude, and phase of the active S   
           Parameter                                                      
XX.XXX dB  at marker 6, if the marker is enabled.                         
XXX.XXX DEG                                                               
__________________________________________________________________________
 
    
     
         __________________________________________________________________________
MENU M5    DESCRIPTION                                                    
__________________________________________________________________________
CH 1 - S11                                                                
XXX.XXX ns DLY                                                            
MARKER 1 ΔREF                                                       
           The REF marker, as activated in Menu M2, its frequency, its    
           reference                                                      
XX.XXXXXX GHz                                                             
           delay, and the channel on which it appears display here. The   
           REF marker                                                     
MARKER TO MAX                                                             
           could be any one of the six available markers: M1-M6. The      
           frequency of                                                   
MARKER TO MIN                                                             
           the REF marker can be changed using the keypad or rotary       
           knob.                                                          
Δ(1 - 2)                                                            
           The marker numbers of the REF marker and the next              
           lowest-numbered                                                
XX.XXXXX GHz                                                              
           active marker appear between the parentheses. This example     
           assumes                                                        
XX.XXX dB  Marker 1 as the Ref marker and Marker 2 as the next            
           lowest-numbered                                                
(XXX.XX DEG)                                                              
           active marker. The lines below display the difference          
           frequency, (or                                                 
           time/distance ) and trace value(s) between these two markers   
           on the ac-                                                     
           tive channel.                                                  
Δ(1 - 3)                                                            
           The marker numbers of the REF marker and the next              
           lowest-numbered                                                
XX.XXXXX GHz                                                              
           active marker appear between the parentheses. This example     
           assumes                                                        
XX.XXX dB  Marker 1 as the Ref marker and Marker 3 as the next            
           lowest-numbered                                                
(XXX.XX DEG)                                                              
           active marker. The lines below display the difference          
           frequency, (or                                                 
           time/distance ) and trace value(s) between these two markers   
           on the ac-                                                     
           tive channel.                                                  
Δ(1 - 4)                                                            
           The marker numbers of the REF marker and the next              
           lowest-numbered                                                
XX.XXXXX GHz                                                              
           active marker appear between the parentheses. This example     
           assumes                                                        
XX.XXX dB  Marker 1 as the Ref marker and Marker 4 as the next            
           lowest-numbered                                                
(XXX.XX DEG)                                                              
           active marker. The lines below display the difference          
           frequency, (or                                                 
           time/distance ) and trace value(s) between these two markers   
           on the ac-                                                     
           tive channel.                                                  
Δ(1 - 5)                                                            
           The marker numbers of the REF marker and the next              
           lowest-numbered                                                
XX.XXXXX GHz                                                              
           active marker appear between the parentheses. This example     
           assumes                                                        
XX.XXX dB  Marker 1 as the Ref marker and Marker5 as the next             
           lowest-numbered                                                
(XXX.XX DEG)                                                              
           active marker. The lines below display the difference          
           frequency, (or                                                 
           time/distance ) and trace value(s) between these two markers   
           on the ac-                                                     
           tive channel.                                                  
Δ(1 - 6)                                                            
           The marker numbers of the REF marker and the next              
           lowest-numbered                                                
XX.XXXXX GHz                                                              
           active marker appear between the parentheses. This example     
           assumes                                                        
XX.XXX dB  Marker 1 as the Ref marker and Marker 6 as the next            
           lowest-numbered                                                
(XXX.XX DEG)                                                              
           active marker. The lines below display the difference          
           frequency, (or                                                 
           time/distance ) and trace value(s) between these two markers   
           on the ac-                                                     
           tive channel.                                                  
__________________________________________________________________________
 
    
     
         ______________________________________                                    
MENU LFX    DESCRIPTION                                                   
______________________________________                                    
READOUT LIMIT                                                             
            This menu shows all of the frequencies                        
FREQUENCIES where the active S-Parameter value is                         
            equal to either Limit 1 or Limit 2.                           
LOG MAG-                                                                  
LIMIT 1 (REF)                                                             
            Displays the value of Limit 1.                                
x.xxx dB                                                                  
LIMIT 2     Displays the value of Limit 2.                                
x.xxx dB                                                                  
LIMIT Δ(1/2)                                                        
            Displays the difference in value between                      
x.xxx dB    Limit 1 and Limit 2.                                          
FREQUENCIES Displays all points where the S Parameter                     
AT LIMIT 2  is equal to Limit 2.                                          
2.9843 GHz                                                                
5.7210 GHz                                                                
7.4412 GHz                                                                
9.8764 GHz                                                                
10.3901 GHz                                                               
15.5648 GHz                                                               
______________________________________                                    
 
    
     
         __________________________________________________________________________
MENU L1    DESCRIPTION                                                    
__________________________________________________________________________
SET LIMITS                                                                
LOG MAG-                                                                  
LIMIT 1 ON (OFF)                                                          
           Turns the Limit 1 line on or off for the active channel. For   
           your con-                                                      
XXX.XXX dB venience, the arbitrarily set limit lines allow you to         
           delineate a go/no gl                                           
           line on your log-magnitude graph beyond which the measured     
           values are                                                     
           unacceptable.                                                  
LIMIT 2 ON (OFF)                                                          
           Turns the Limit 2 line on or off for the active channel on the 
           log-mag-                                                       
XXX.XXX dB nitude graph.                                                  
READOUT LIMIT                                                             
           Displays Menu LF1, which shows all points where the current    
FREQUENCIES                                                               
           s-parameter equals the limit values.                           
PHASE-                                                                    
LIMIT 1 ON (OFF)                                                          
           Turns the Limit 1 line on or off for the active channel on the 
           phase graph.                                                   
XXX.XX DEG                                                                
LIMIT 2 ON (OFF)                                                          
           Turns the Limit 2 line on or off for the active channel on the 
           phase graph.                                                   
XXX.XX DEG                                                                
READOUT LIMIT                                                             
           Displays Menu LF1, which shows all points where the current    
FREQUENCIES                                                               
           s-parameter equals the limit values.                           
LIMITS     Enables both limit lines for the active channel on both the    
           log-magnitude                                                  
ENABLED    and phase graphs.                                              
(DISABLED)                                                                
PRESS &lt;ENTER&gt;                                                             
           Pressing the ENTER key implements your menu selection. The     
           menu                                                           
TO SELECT  remains on the screen until another menu is selected for       
           display or until                                               
OR TURN ON/OFF                                                            
           the CLEAR/RET LOC key is pressed.                              
__________________________________________________________________________
 
    
     
         __________________________________________________________________________
MENU       DESCRIPTION                                                    
__________________________________________________________________________
SET LIMITS                                                                
LINEAR POLAR-                                                             
(-SMITH CHART-)                                                           
LIMIT 1 ON (OFF)                                                          
           Turns the Limit 1 line on or off for the active channel. For   
           your con-                                                      
(XXX.XXX Mv                                                               
           venience, the arbitrarily set limit lines allow you to         
           delineate a go/no go                                           
           line on your Smith chart or polar display beyond which the     
           measured values are unacceptable.                              
LIMIT 2 ON (OFF)                                                          
           Turns the Limit 2 line on or off for the active channel on     
           your Smith chart or polar display.                             
XXX.XXX mV                                                                
LIMITS     Enables both previously set limit lines to appear for the      
           active channel                                                 
ENABLED    on your Smith chart or polar display.                          
PRESS &lt;ENTER&gt;                                                             
           Pressing the ENTER key implements your menu selection. The     
           menu                                                           
TO SELECT  remains on the screen until another menu is selected for       
           display or until                                               
OR TURN ON/OFF                                                            
           the CLEAR/RET LOC key is pressed.                              
__________________________________________________________________________
 
    
     
         __________________________________________________________________________
MENU       DESCRIPTION                                                    
__________________________________________________________________________
SET LIMITS                                                                
LOG MAG-                                                                  
LIMIT 1 ON (OFF)                                                          
           Turns the Limit 1 line on or off for the active channel. For   
           your con-                                                      
XXX.XXX dB venience, the arbitrarily set limit lines allow you to         
           delineate a go/no go                                           
           line on your Log Mag display beyond which the measured values  
           are unacceptable.                                              
LIMIT 2 ON (OFF)                                                          
           Turns the Limit 2 line on or off for the active channel on     
           your Log Mag display.                                          
XXX.XXX dB                                                                
READOUT LIMIT                                                             
           Displays Menu LF1, which shows all points where the current    
FREQUENCIES                                                               
           s-parameter equals the limit values.                           
LIMITS     Enables both previously set limit lines to appear for the      
           active channel                                                 
ENABLED    on your Log Mag display.                                       
PRESS &lt;ENTER&gt;                                                             
           Pressing the ENTER key implements your menu selection. The     
           menu                                                           
TO SELECT  remains on the screen until another menu is selected for       
           display or until                                               
OR TURN ON/OFF                                                            
           the CLEAR/RET LOC key is pressed.                              
__________________________________________________________________________
 
    
     
         __________________________________________________________________________
MENU       DESCRIPTION                                                    
__________________________________________________________________________
SET LIMITS                                                                
PHASE-                                                                    
LIMIT I ON (OFF)                                                          
           Turns the Limit 1 line on or off for the active channel on the 
           phase graph.                                                   
XXX.XX DEG                                                                
LIMIT 2 ON (OFF)                                                          
           Turns the Limit 2 line on or off for the active channel on the 
           phase graph.                                                   
XXX.XX DEG                                                                
READOUT LIMIT                                                             
           Displays Menu LF1, which shows all points where the current    
FREQUENCIES                                                               
           s-parameter equals the limit values.                           
LIMITS     Enables both limit lines for the active channel on a phase     
           graph.                                                         
ENABLED                                                                   
PRESS &lt;ENTER&gt;                                                             
           Pressing the ENTER key implements your menu selection. The     
           menu                                                           
TO ELECT   remains on the screen until another menu is selected for       
           display or until                                               
OR TURN ON/OFF                                                            
           the CLEAR/RET LOC key is pressed.                              
__________________________________________________________________________
 
    
     
         __________________________________________________________________________
MENU       DESCRIPTION                                                    
__________________________________________________________________________
SET LIMITS                                                                
LOG POLAR-                                                                
LIMIT 1 ON (OFF)                                                          
           Turns the Limit 1 line on or off for the active channel. For   
           your con-                                                      
XXX.XXX dB venience, the arbitrarily set-limit lines allow you to         
           delineate a go/no go                                           
           line on your Log Polar display beyond which the measured       
           values are unacceptable.                                       
LIMIT 2 ON (OFF)                                                          
           Turns the Limit 2 line on or off for the active channel on     
           your Log Polar display.                                        
XXX.XXX dB                                                                
LIMITS     Enables both previously set limit lines to appear for the      
           active channel                                                 
ENABLED    on your Log Polar display.                                     
PRESS&lt;ENTER&gt;                                                              
           Pressing the ENTER key implements your menu selection. The     
           menu                                                           
TO SELECT  remains on the screen until another menu is selected for       
           display or until                                               
OR TURN ON/OFF                                                            
           the CLEAR/RET LOC key is pressed.                              
__________________________________________________________________________
 
    
     
         __________________________________________________________________________
MENU       DESCRIPTION                                                    
__________________________________________________________________________
SET LIMITS                                                                
GROUP DELAY-                                                              
LIMIT 1 ON (OFF)                                                          
           Turns the limit 1 line on or off for the active channel. For   
           your con-                                                      
XXX.XXX dB venience, the arbitrarily set limit lines allow you to         
           delineate a go/no go                                           
           line on your Group Delay display beyond which the measured     
           values are unacceptable.                                       
LIMIT 2 ON (OFF)                                                          
           Turns the Limit 2 line on or off for the active channel on     
           your Group Delay display.                                      
XXX.XXX dB                                                                
READOUT LIMIT                                                             
           Displays Menu LF1, which shows all points where the current    
FREQUENCIES                                                               
           s-parameter equals the limit values.                           
LIMITS     Enables both previously set limit lines to appear for the      
           active channel                                                 
ENABLED    on your Group Delay display.                                   
PRESS &lt;ENTER&gt;                                                             
           Pressing the ENTER key implements your menu selection. The     
           menu                                                           
TO SELECT  remains on the screen until another menu is selected for       
           display or until                                               
OR TURN ON/OFF                                                            
           the CLEAR/RET LOC key is pressed.                              
__________________________________________________________________________
 
    
     
         __________________________________________________________________________
MENU       DESCRIPTION                                                    
__________________________________________________________________________
SET LIMITS                                                                
LINEAR MAG-                                                               
LIMIT 1 ON (OFF)                                                          
           Turns the Limit 1 line on or off for the active channel. For   
           your con-                                                      
XXX.XXX dB venience, the arbitrarily set limit lines allow you to         
           delineate a go/no go                                           
           line on your Linear Mag display beyond which the measured      
           values are unacceptable.                                       
LIMIT 2 ON (OFF)                                                          
           Turns the Limit 2 line on or off for the active channel on     
           your Linear Mag display.                                       
XXX.XXX dB                                                                
READOUT LIMIT                                                             
           Displays Menu LF1, which shows all points where the current    
FREQUENCIES                                                               
           s-parameter equals the limit values.                           
LIMITS     Enables both previously set limit lines to appear for the      
           active channel                                                 
ENABLED    on your Linear Mag display.                                    
PRESS &lt;ENTER&gt;                                                             
           Pressing the ENTER key implements your menu selection. The     
           menu                                                           
TO SELECT  remains on the screen until another menu is selected for       
           display or until                                               
OR TURN ON/OFF                                                            
           the CLEAR/RET LOC key is pressed.                              
__________________________________________________________________________
 
    
     
         __________________________________________________________________________
MENU       DESCRIPTION                                                    
__________________________________________________________________________
SET LIMITS                                                                
LINEAR MAG-                                                               
LIMIT 1 ON (OFF)                                                          
           Turns the Limit 1 line on or off for the active channel. For   
           your con-                                                      
XXX.XXX dB venience, the arbitrarily set limit lines allow you to         
           delineate a go/no go                                           
           line on your Linear Mag display beyond which the measured      
           values are unacceptable.                                       
LIMIT 2 ON (OFF)                                                          
           Turns the Limit 2 line on or off for the active channel on     
           your Linear Mag display.                                       
XXX.XXX dB                                                                
READOUT LIMIT                                                             
           Displays Menu LF1, which shows all points where the current    
FREQUENCY  s-arameter equals the limit values.                            
PHASE-                                                                    
LIMIT ON (OFF)                                                            
           Turns the Limit 1 line on or off for the active channel. For   
           your con-                                                      
XXX.XXX dB venience, the arbitrarily set limit lines allow you to         
           delineate a go/no go                                           
           line on your Phase display beyond which the measured values    
           are unacceptable.                                              
LIMIT 2 ON (OFF)                                                          
           Turns the Limit 2 line on or off for the active channel on     
           your Phase display.                                            
XXX.XXX dB                                                                
READOUT LIMIT                                                             
           Displays Menu LF1, which shows all points where the current    
FREQUENCIES                                                               
           s-parameter equals the limit values.                           
LIMITS     Enables both previously set limit lines to appear for the      
           active channel                                                 
ENABLED    on your Phase display.                                         
PRESS &lt;ENTER&gt;                                                             
           Pressing the ENTER key implements your menu selection. The     
           menu                                                           
TO SELECT  remains on the screen until another menu is selected for       
           display or until                                               
OR TURN ON/OFF                                                            
           the CLEAR/RET LOC key is pressed.                              
__________________________________________________________________________
 
    
     
         __________________________________________________________________________
MENU       DESCRIPTION                                                    
__________________________________________________________________________
SET LIMITS                                                                
REAL-                                                                     
LIMIT 1 ON (OFF)                                                          
           Turns the Limit 1 line on or off for the active channel. For   
           your con-                                                      
XXX.XXX pU venience, the arbitrarily set limit lines allow you to         
           delineate a go/no go                                           
           line on your Real values display beyond which the measured     
           values are unacceptable.                                       
LIMIT 2 ON (OFF)                                                          
           Turns the Limit 2 line on or off for the active channel on     
           your Real values display.                                      
XXX.XXX pU                                                                
LIMITS     Enables both previously set limit lines to appear for the      
           active channel                                                 
ENABLED    on your Real values display.                                   
PRESS&lt;ENTER&gt;                                                              
           Pressing the ENTER key implements your menu selection. The     
           menu                                                           
TO SELECT  remains on the screen until another menu is selected for       
           display or until                                               
OR TURN ON/OFF                                                            
           the CLEAR/RET LOC key is pressed.                              
__________________________________________________________________________
 
    
     
         __________________________________________________________________________
MENU       DESCRIPTION                                                    
__________________________________________________________________________
SET LIMITS                                                                
IMAGINARY-                                                                
LIMIT 1 ON (OFF)                                                          
           Turns the Limit 1 line on or off for the active channel. For   
           your con-                                                      
XXX.XXX mU venience, the arbitrarily set limit lines allow you to         
           delineate a go/no go                                           
           line on your Imaginary values display beyond which the         
           measured values                                                
           are unacceptable.                                              
LIMIT 2 ON (OFF)                                                          
           Turns the Limit 2 line on or off for the active channel on     
           your Imaginary values display.                                 
XXX.XXX mU                                                                
LIMITS     Enables both previously set limit lines to appear for the      
           active channel                                                 
ENABLED    on your Imaginary values display.                              
PRESS &lt;ENTER&gt;                                                             
           Pressing the ENTER key implements your menu selection. The     
           menu                                                           
TO SELECT  remains on the screen until another menu is selected for       
           display or until                                               
OR TURN ON/OFF                                                            
           the CLEAR/RET LOC key is pressed.                              
__________________________________________________________________________
 
    
     
         __________________________________________________________________________
MENU       DESCRIPTION                                                    
__________________________________________________________________________
SET LIMITS                                                                
REAL-                                                                     
LIMIT 1 ON (OFF)                                                          
           Turns the Limit 1 line on or off for the active channel. For   
           your con-                                                      
XXX.XXX mU venience, the arbitrarily set limit lines allow you to         
           delineate a go/no go                                           
           line on your Real values display beyond which the measured     
           values are unacceptable.                                       
LIMIT 2 ON (OFF)                                                          
           Turns the Limit 2 line on or off for the active channel on     
           your Real values display.                                      
XXX.XXX mU                                                                
IMAGINARY-                                                                
LIMIT 1 ON (OFF)                                                          
           Turns the Limit 1 line on or off for the active channel. For   
           your con-                                                      
XXX.XXX pU venience, the arbitrarily set limit lines allow you to         
           delineate a go/no go                                           
           line on your Imaginiary values display beyond which the        
           measured values                                                
           are unacceptable.                                              
LIMIT 2 ON (OFF)                                                          
           Turns the Limit 2 line on or off for the active channel on     
           your Imaginiary values display.                                
XXX.XXX pU                                                                
LIMITS     Enables both previously set limit lines to appear for the      
           active channel                                                 
ENABLED    on your Imaginary values display.                              
PRESS      Pressing the ENTER key implements your menu selection. The     
           menu                                                           
TO SELECT  remains on the screen until another menu is selected for       
           display or until                                               
OR TURN ON/OFF                                                            
           the CLEAR/RET LOC key is pressed.                              
__________________________________________________________________________
 
    
     
                                           TABLE 3-3                               
__________________________________________________________________________
Error Codes and Status Messages                                           
CODE MESSAGE TEXT              FATAL                                      
                                    MEANING                               
__________________________________________________________________________
     Self Test, Main Microprocessor #2                                    
000  FIFO RESET FAILURE        X    FiFO failed to reset, PCB A12         
002  PROM CHECKSUM FAILURE #2  X    Prom failure, PCB A12                 
003  BATTERY BACKED RAM FAILURE                                           
                               X    Non volitile RAM failure, PCB A12     
004  EXTENDED MEMORY FAILURE   X    Failure in the extended memory, PCB   
                                    A12                                   
005  DYNAMIC RAM FAILURE #2    X    Dynamic RAM failure, PCB A12          
006  TIMER FAILURE #2          X    Programmable timer failure, PCB A12   
007  INTERRUPT CONTROLLER FAILURE #2                                      
                               X    Interrupt comtroller failure PCB,     
                                    A12                                   
008  NUMERIC PROCESSOR FAILURE #2                                         
                               X    Numeric processor failure, PCB A12    
009  FRONT PANEL INTERFACE FAILURE                                        
                               X    Interface failure, front panel, PCB   
                                    A12                                   
010  PRINTER INTERFACE FAILURE X    Printer or interface failure PCB A12  
     Self Test, Main Microprocessor #1                                    
020  FIFO TO # 2 FAILED RESET  X    Interface failure with FIFO, PCB A12  
022  FIFO TO I/O FAILED RESET       Interface failure with FIFO PCB A13   
023  PROM CHECKSUM FAILURE #1  X    Checksum error, PROM PCB A13          
024  DYNAMIC RAM FAILURE #1    X    Dynamic RAM failure, PCB A13          
025  TIMER FAILURE #1          X    Programmable timer failure PCB A13    
026  INTERRUPT CONTROLLOER FAILURE #1                                     
                               X    Interrupt controller failure PCB A13  
027  DISK DRIVE CONTROLLER FAILURE  Disk drive controller failure, PCB    
                                    A13                                   
028  DISK DRIVE FAILURE             Disk drive SEEK failure, PCB A13      
029  NUMERIC PROCESSOR FAILURE #1                                         
                               X    8087 math coprocessor failure         
030  PROM CARTRIDGEE CHECKSUM ERROR PROM cartridge failure PCB A13        
031  DISK DRIVE NOT READY FOR TEST  Diskette is not in disk drive         
     Self Test, I/O Processor                                             
040  PROM CHECKSUM FAILURE I/O X    PROM failure, PCB A11                 
041  RAM FAILURE I/O           X    RAM failure, PCB A11                  
042  TIMER/INTERRUPT LOOPBACK FAILURE                                     
                               X    Programmable timer failure, PCB A11   
043  GPIB INTERFACE FAILURE I/O     GPIB failure, PCB A11                 
044  FIFO FAILURE I/O          X    FIFO failure, PCB A11                 
050  A1 COMMUNICATIONS FAILURE      LO 1 Phase Lock PCB Error             
051  A2 COMMUNICATIONS FAILURE      LO 2 Phase Lock PCB Failure           
052  A3 COMMUNICATIONS FAILURE      Cal/Third Local O scillator PCB       
                                    Failure                               
053  A4 COMMUNICATIONS FAILURE      Analog to digital PCB failure         
054  A5 COMMUNICATIONS FAILURE      10 Mhz References PCB failure         
055  A6 COMMUNICATIONS FAILURE      Source lock PCB failure               
056  A10 COMMUNICATIONS FAILURE     Bandswitch blank/sync PCB             
057  8 BIT A/D CONVERTER FAILURE    Failure A/D PCB A4                    
058  STEERING DAC FAILURE           Failure A/D PCB A4                    
059  12 BIT A/D OR STEERING DAC FAILURE                                   
                                    Failure A/D PCB A4                    
060  TEST SET NOT CONNECTED OR NOT  General failure of test set           
     WORKING                                                              
061  TEST SET CHAN A CAL PHASING FAILURE                                  
                                    Test set CHAN A failure               
062  TEST SET CHAN A CAL LEVEL FAILURE                                    
                                    Test set CHAN A failure               
063  TEST SET CHAN A GAIN FAILURE   Test set CHAN A failure               
064  TEST SET CHAN A PHASE RANGING FAILURE                                
                                    Test set CHAN A failure               
065  TEST SET CHAN B CAL PHASING FAILURE                                  
                                    Test set CHAN B failure               
067  TEST SET CHAN B GAIN FAILURE   Test set CHAN B failure               
068  TEST SET CHAN B PHASE RANGING FAILURE                                
                                    Test set CHAN B failure               
069  TEST SET REF CHAN CAL PHASING FAILURE                                
                                    Test set REF CHAN failure             
070  TEST SET REF CHAN CAL LEVEL FAILURE                                  
                                    Test set REF CHAN failure             
071  TEST SET REF CHAN GAIN FAILURE Test set REF CHAN failure             
072  TEST SET REF CHAN PHASE RANGING                                      
                                    Test set REF CHAN failure             
     System Status, Program Load                                          
100  DISK DRIVE NOT READY      X    Program failed to load from disk,     
                                    (disk installed?)                     
101  PROGRAM DATA ERROR        X    Program failed to load from disk      
102  PROGRAM FILE MISSING           Loader could not find system files    
103  DISK ERROR                     The 360 is unable to read the         
                                    diskette                              
104  UNKNOWN DISK ERROR             Loader failed a consistency check     
105  PROGRAM DATA ERROR ON # 2      Program for processor #2 failed to    
                                    load                                  
     Program Initialization                                               
110  SOURCE ID FAILURE              No sweeper ID on GPIB; sweeper may    
                                    not be connected.                     
111  TEST SET NOT CONNECTED    X    Initialization detects a              
                                    descrepancy.                          
112  TEST SET FREQ. RANGE      X    Initialization detects a              
                                    descrepancy.                          
     DOES NOT MATCH SOURCE                                                
113  CAL DATA NOT FOUND: CHANGE     File not found on disk with name      
                                    matching                              
     DISK AND PRESS &lt;ENTER&gt;         that in battery backed RAM.           
114  PROGRAM ERROR             X    Program corrupted.                    
115  PROCESSOR COM ERROR       X    FIFO Synch problem Disk Releated.     
     Program Initialization, Disk Related                                 
131  DISK READ ERROR                Hard error reading from disk.         
132  DISK WRITE ERROR               Hard error writing to disk.           
133  FILE DELETION ERROR            Write protect tab is in &#34;read only&#34;   
                                    position.                             
134  DISK NOT READY                 Disk not in unit or not formatted     
135  DISK WRITE PROTECTED           Write protect tab is in &#34;read only&#34;   
                                    position.                             
136  OUT OF DISK SPACE              Disk file space full                  
137  FILE IS INCOMPATIBLE           File is not a 360 data or program     
                                    file.                                 
138  NO SPACE FOR NEW DATA FILE     Disk file space full.                 
139  FILE MARKED READ ONLY          Read-only attribute set on file.      
140  NO FILES REMAIN TO OVERWRITE   All files of the type have been       
                                    deleted.                              
141  NO FILES REMAIN TO DELETE      All files of the type have been       
                                    deleted.                              
     Program Initialization, Peripheral                                   
170  PRINTER NOT READY              Printer &#34;off line&#34; or not connected.  
171  PLOTTER NOT READY              Plotter &#34;off line&#34; or not connected.  
     Control Panel                                                        
200  SELECTED FREQUENCY OUT OF CAL RANGE                                  
                                    Cal. range does not include selected  
                                    frequency.                            
201  MARKERS SELECTED FOR READOUT NOT                                     
                                    Sweep range does not include selected 
                                    frequency                             
     ACTIVE IN SWEEP RANGE                                                
208  OUT OF RANGE                   Attempted to enter an out-of-range    
                                    parameter.                            
209  START GREATER THAT STOP        Attempted to a set start frequency    
                                    that was                              
                                    greater than the stop frequency.      
210  OUT OF RANGE.20 PERCENT MAX    Attempted to enter a smoothing or     
                                    group delay factor                    
                                    that was greater than 20%.            
213  OUT OF H/W RANGE               Attempted to enter a frequency that   
                                    is outside of                         
                                    the system hardware range.            
216  TOO MANY POINTS.               Attempted to set too many discrete    
     501 MAXIMUM                    frequency points.                     
217  TOO FEW POINTS.                Attempted to set too few discrete     
                                    frequency points.                     
     2 MINIMUM                                                            
219  DISCRETE FREQS LOST            Setup changed in N-discrete frequency 
                                    mode.                                 
220  OUT OF SWEEP RANGE             Attempt to set marker outside sweep   
                                    range.                                
221  OPTION NOT INSTALLED           The selected option is not installed  
222  MEAS. DATA NOT AVAILABLE       No measured data on channel to be     
                                    stored.                               
     FOR STORAGE                                                          
223  NO STORED MEMORY DATA          No data available in memory for       
                                    channel.                              
224  SYSTEM BUS ADDRESSES           Attempt to set GPIB addresses to same 
                                    value.                                
     MUST BE UNIQUE                                                       
225  SYSTEM UNCALIBRATED            no cal exists.                        
226  MEMORY LOCATION                Saved state data is invalid.          
     CORRUPTED                                                            
227  SETUP INCONSISTENT             Saved state not compatible with       
                                    hardware or                           
     RECALL ABORTED                 software version                      
228  WINDOW TOO SMALL               Attempt to set start greater than or  
                                    equal to stop.                        
229  OUT OF WINDOW RANGE            Attempt to set marker outside start   
                                    to stop range.                        
230  ATTENUATOR UNAVAILABLE         Selected attenuators not available in 
                                    test set                              
231  START MUST BE LESS THAN        Attempt to set start greater than or  
                                    equal to stop                         
     STOP                           in marker sweep                       
232  ILLEGAL IN C.W. MODE           Attempt to readout limit frequency.   
233  ILLEGAL IN TIME DOMAIN         Attempt to readout limit frequency.   
234  BOTH LIMITS MUST BE ON         Attempt to readout limit frequency.   
235  Stop is Over Range             Discrete fill parameters cause stop   
                                    to go over                            
                                    hardware range                        
238  Out of Ramge 10% Minimum       Attempt to set pen speed to below     
                                    10%                                   
270  UNCALIBRATED                   Channel has S parameter for which     
                                    calibration does not exist.           
271  PRINTER NOT READY              Printer not connected or paper out.   
272  TOO MUCH PRINT DATA            Print buffer is full. Reduce number   
                                    of channels                           
                                    or data points.                       
273  PLOTTER NOT READY              Plotter not connected.                
280  CAL INVALID                    Calibration is incorrect for S        
                                    parameter displayed.                  
281  TIME DOMAIN INVALID            Time domain cannot be used            
                                    in current setup                      
282  GROUP DELAY INVALID            Group delay cannot be used            
                                    in current setup                      
283  GATE MUST BE ON                Attempt to select Frequency With      
                                    Time Gate with gate off.              
284  SMOOTHING INVALID              Attempt to use smoothing while        
                                    in C.W. mode.                         
285  MEMORY DATA INVALID            Setup has changed since data was      
                                    stored.                               
     Measurement Related                                                  
300  LOW IF                         Insufficient signal level detected:   
                                    Device under test may not be          
                                    connected.                            
301  LOCK FAILURE                   RF source failed to lock to           
                                    reference                             
                                    oscillator in 360 testset.            
302  A/D FAILURE                    Analog to Digital convertor not       
                                    functioning in 360 mainframe.         
303  RF OVERLOAD                    Test signal level is too high:        
                                    reduce                                
                                    source level or add attenuation.      
310  SWPR ID FAILURE                Communications lost with RF source.   
311  SWPR SELF TEST FAILURE         RF source failed power on             
                                    self-test program.                    
312  NO TEST SET                    Test set not connected. Reconnect and 
                                    cycle                                 
                                    power to clear                        
     GPIB Related                                                         
400  GPIB ERROR                     Data transmission error on            
__________________________________________________________________________
                                    GPIB                                  
 ##STR3##
 
    
     
         ______________________________________                                    
MENU TD1     DESCRIPTION                                                  
______________________________________                                    
DOMAIN                                                                    
FREQUENCY    Displays the data in normal frequency                        
             domain format.                                               
FREQUENCY    Displays the data in the frequency domain                    
WITH TIME    after a specific time range has been                         
GATE         sampled by the gate function.                                
TIME         Displays the data in the time (distance)                     
LOWPASS MODE domain, using true lowpass processing.                       
             Data must be taken using a harmonic                          
             series calibration and sweep in order                        
             to use this mode.                                            
TIME         Displays the data in the time (distance)                     
BANDPASS MODE                                                             
             domain using bandpass processing. Any                        
             data sweep range using normal calibration                    
             can be used.                                                 
DISPLAY                                                                   
TIME/DISTANCE                                                             
             Switches the mode of display between                         
             time and distance. This does not affect the                  
             actual displayed data, but only the                          
             annotation.                                                  
SET RANGE    Takes you to Menu TD2, which lets you                        
             set range and other display parameters.                      
SET GATE     Takes you to menu TD4, which lets you                        
             set gate parameters.                                         
GATE ON/OFF  Switches the gate on or off each time                        
             ENTER is pressed.                                            
HELP         Displays an informational help menu.                         
PRESS &lt;ENTER&gt;                                                             
             Pressing the ENTER key implements                            
TO SELECT    your menu selection. The menu remains                        
OR SWITCH    on the screen until another menu is                          
             selected for display or until the                            
             CLEAR/RET LOC key is pressed.                                
______________________________________                                    
 
    
     
         ______________________________________                                    
MENU TD2tl   DESCRIPTION                                                  
______________________________________                                    
LOWPASS TIME                                                              
DOMAIN SETUP                                                              
START        Sets the start time of the display.                          
XXX.XXX ps                                                                
STOP         Sets the stop time of the display.                           
XXX.XXX ps                                                                
CENTER       Sets the center time of the display.                         
XXX.XXX ps                                                                
SPAN         Sets the span (Stop - Start) of the display.                 
XXX.XXX ps                                                                
MARKER RANGE Takes you to Menu TD7d/71, which lets                        
             you set the display to a range determined                    
             by two of the markers.                                       
RESPONSE     Switches between Impulse and Step                            
IMPULSE/STEP response each time ENTER is pressed.                         
MORE         Takes you to Menu TD3, which contains                        
             additional selections for display steup.                     
PRESS &lt;ENTER&gt;                                                             
             Pressing the ENTER key implements                            
TO SELECT    your menu selection. The menu remains                        
             on the screen until another menu is                          
             selected for display or until the                            
             CLEAR/RET LOC key is pressed.                                
______________________________________                                    
 
    
     
         ______________________________________                                    
MENU TD2dl   DESCRIPTION                                                  
______________________________________                                    
LOWPASS                                                                   
DISTANCE                                                                  
DISPLAY SETUP                                                             
START        Sets the start time of the display.                          
XXX.XXX mm                                                                
STOP         Sets the stop time of the display.                           
XXX.XXX mm                                                                
CENTER       Sets the center time of the display.                         
XXX.XXX mm                                                                
SPAN         Sets the span (Stop - Start) of the display.                 
XXX.XXX mm                                                                
MARKER RANGE Takes you to Menu TD7d/7l, which lets                        
             you set the display to a range determined                    
             by two of the markers.                                       
RESPONSE     Switches between Impulse and Step                            
IMPUSLE/STEP response each time ENTER is pressed.                         
MORE         Takes you to Menu TD3, which contains                        
             additional selections for display setup.                     
REL. VELOCITY                                                             
             Indicates the relative velocity of light,                    
X.X          as set by the dielectric constant in                         
             menu RD2.                                                    
PRESS &lt;ENTER&gt;                                                             
             Pressing the ENTER key implements                            
TO SELECT    your menu selection. The menu remains                        
             on the screen until another menu is                          
             selected for display or until the                            
             CLEAR/RET LOC key is pressed.                                
______________________________________                                    
 
    
     
         ______________________________________                                    
MENU TD2tb   DESCRIPTION                                                  
______________________________________                                    
BANDPASS TIME                                                             
DOMAIN SETUP                                                              
START        Sets the start time of the display.                          
XXX.XXX ps                                                                
STOP         Sets the stop time of the display.                           
XXX.XXX ps                                                                
CENTER       Sets the center time of the display.                         
XXX.XXX ps                                                                
SPAN         Sets the span (Stop - Start) of the display.                 
XXX.XXX ps                                                                
MARKER RANGE Takes you to Menu TD7d/7l, which lets                        
             you set the display to a range determined                    
             by two of the markers.                                       
PHASOR ON/OFF                                                             
             Switches Phasor Impulse processing on                        
IMPULSE      or off each time ENTER is pressed.                           
SET RANGE TO                                                              
ISOLATE ONE                                                               
DISCONTINUITY                                                             
MORE         Takes you to Meun TD3, which contains                        
             additional selections for display setup.                     
PRESS &lt;ENTER&gt;                                                             
             Pressing the ENTER key implements                            
TO SELECT    your menu selection. The menu remains                        
             on the screen until another menu is                          
             selected for display or until the                            
             CLEAR/RET LOC key is pressed.                                
______________________________________                                    
 
    
     
         ______________________________________                                    
MENU TD2db   DESCRIPTION                                                  
______________________________________                                    
BANDPASS                                                                  
DISTANCE                                                                  
DISPLAY SETUP                                                             
START        Sets the start time of the display.                          
XXX.XXX mm                                                                
STOP         Sets the stop time of the display.                           
XXX.XXX mm                                                                
CENTER       Sets the center time of the display.                         
XXX.XXX mm                                                                
SPAN         Sets the span (Stop - Start) of the display.                 
XXX.XXX mm                                                                
MARKER RANGE Takes you to Menu TD7d/7l, which lets                        
             you set the display to a range determined                    
             by two of the markers.                                       
PHASOR ON/OFF                                                             
             Switches Phasor Impulse processing on                        
IMPULSE      or off each time ENTER is pressed.                           
SET RANGE TO                                                              
ISOLATE ONE                                                               
DISCONTINUITY                                                             
MORE         Takes you to Menu TD3, which contains                        
             additional selections for display setup.                     
REL. VELOCITY                                                             
             Indicates the relative velocity of light,                    
X.X          as set by the dielectric constant                            
             in menu RD2.                                                 
PRESS &lt;ENTER&gt;                                                             
             Pressing the ENTER key implements                            
TO SELECT    your menu selection. The menu remains                        
             on the screen until another menu is                          
             selected for display or until the                            
             CLEAR/RET LOC key is pressed.                                
______________________________________                                    
 
    
     
         ______________________________________                                    
MENU         DESCRIPTION                                                  
______________________________________                                    
BANDPASS TIME                                                             
DOMAIN SETUP                                                              
WINDOW       Takes you to Menu TD5, which lets you                        
NOMINAL      change the window type.                                      
SET GATE     Takes you to menu TD4, which lets you                        
             set the gate.                                                
PREVIOUS MENU                                                             
             Returns you to Menu TD2.                                     
PRESS &lt;ENTER&gt;                                                             
             Pressing the ENTER key implements                            
TO SELECT    your menu selection. The menu remains                        
OR TURN ON/OFF                                                            
             on the screen until another menu is                          
             selected for display or until the                            
             CLEAR/RET LOC key is pressed.                                
______________________________________                                    
 
    
     
         ______________________________________                                    
MENU         DESCRIPTION                                                  
______________________________________                                    
LOWPASS TIME                                                              
DOMAIN SETUP                                                              
WINDOW       Takes you to Menu TD5, which lets you                        
NOMINAL      change the window type.                                      
SET GATE     Takes you to menu TD4, which lets you                        
             set the gate.                                                
SET D.C. TERM                                                             
             Takes you to Menu TD6, which lets you                        
XXX.XXX      set the D.C. term for lowpass processing.                    
PREVIOUS MENU                                                             
             Returns you to Menu TD2                                      
PRESS &lt;ENTER&gt;                                                             
             Pressing the ENTER key implements                            
TO SELECT    your menu selection. The menu remains                        
             on the screen until another menu is                          
             selected for display or until the                            
             CLEAR/RET LOC key is pressed.                                
______________________________________                                    
 
    
     
         ______________________________________                                    
MENU         DESCRIPTION                                                  
______________________________________                                    
GATE                                                                      
START        Sets the start time of the gate.                             
XXX.XXX mm                                                                
STOP         Sets the stop time of the gate.                              
XXX.XXX mm                                                                
CENTER       Sets the center time of the gate.                            
XXX.XXX mm                                                                
SPAN         Sets the span (Stop - Start) of the gate.                    
XXX.XXX mm                                                                
SHAPE        Takes you to Menu TD5, which lets you                        
NOMINAL      set the shape of the gate.                                   
GATE ON      Switches the gate on or off each time                        
             ENTER is pressed.                                            
PREVIOUS MENU                                                             
             Returns you to the previous menu.                            
PRESS &lt;ENTER&gt;                                                             
             Pressing the ENTER key implements                            
TO SELECT    your menu selection. The menu remains                        
             on the screen until another menu is                          
             selected for display or until the                            
             CLEAR/RET LOC key is pressed.                                
______________________________________                                    
 
    
     
         ______________________________________                                    
MENU         DESCRIPTION                                                  
______________________________________                                    
GATE                                                                      
START        Sets the start time of the gate.                             
XXX.XXX ns                                                                
STOP         Sets the stop time of the gate.                              
XXX.XXX ns                                                                
CENTER       Sets the center time of the gate.                            
XXX.XXX ns                                                                
SPAN         Sets the span (Stop - Start) of the gate.                    
XXX.XXX ns                                                                
SHAPE        Takes you to Menu TD5, which lets you                        
NOMINAL      set the shape of the gate.                                   
GATE ON      Switches the gate on or off each time                        
             ENTER is pressed.                                            
PREVIOUS MENU                                                             
             Returns you to the previous menu.                            
PRESS &lt;ENTER&gt;                                                             
             Pressing the ENTER key implements                            
TO SELECT    your menu selection. The menu remains                        
             on the screen until another menu is                          
             selected for display or until the                            
             CLEAR/RET LOC key is pressed.                                
______________________________________                                    
 
    
     
         ______________________________________                                    
MENU        DESCRIPTION                                                   
______________________________________                                    
SHAPE                                                                     
RECTANGULAR Selects a Rectangular (one-term) shape.                       
NOMINAL     Selects a two-term Hamming shape.                             
LOW SIDELOBE                                                              
            Selects a three-term Blackman-Harris shape.                   
MIN SIDELOBE                                                              
            Selects a four-term Blackman-Harris shape.                    
HELP        Displays an informational help menu.                          
PRESS &lt;ENTER&gt;                                                             
            Pressing the ENTER key implements your                        
TO SELECT   menu selection. The menu remains on the                       
            screen until another menu is selected for                     
            display or until the CLEAR/RET                                
            LOC key is pressed.                                           
______________________________________                                    
 
    
     
         ______________________________________                                    
MENU         DESCRIPTION                                                  
______________________________________                                    
SET DC TERM  Since it is impossible to measure the true                   
FOR LOW PASS D.C. term required for lowpass processing,                   
PROCESSING   a value must be estimated. This menu allows                  
             a choice between five different selections                   
             for this value.                                              
AUTO         Sets the D.C. term to a value determined by                  
EXTRAPOLATE  extrapolating the data points near the zero                  
             frequency.                                                   
LINE         Sets the D.C. term to the characteristic                     
IMPEDANCE    impedance of the transmission medium (Z.sub.0).              
OPEN         Sets the D.C. term to correspond to an open                  
             circuit.                                                     
SHORT        Sets the D.C. term to correspond to a short                  
             circuit.                                                     
OTHER        Sets the D.C. term to the value entered.                     
XXX.XXX#                                                                  
ABOVE VALUE                                                               
REPRESENTS A                                                              
REFLECTION                                                                
COEFF. CF                                                                 
XX.XXX mU                                                                 
PRESS &lt;ENTER&gt;                                                             
             Pressing the ENTER key implements your                       
TO SELECT    menu selection. The menu remains on the                      
             screen until another menu is selected for                    
             display or until the CLEAR/RET                               
             LOC key is pressed.                                          
______________________________________                                    
 
    
     
         ______________________________________                                    
MENU         DESCRIPTION                                                  
______________________________________                                    
TIME                                                                      
MARKER RANGE                                                              
START TIME   Sets the start time to the value of the                      
MARKER ( )   selected marker.                                             
XXX.XXX nS                                                                
STOP TIME    Sets the stop time to the value of the                       
MARKER ( )   selected marker.                                             
XXX.XXX nS                                                                
RESTORE      Returns the display to the original time                     
ORIGINAL     range that was in effect before the marker                   
RANGE        range was selected.                                          
PREVIOUS MENU                                                             
             Returns you to the TD2 menu.                                 
USE KEYPAD   Pressing the ENTER key implements                            
TO CHOOSE    your menu selection. The menu remains                        
MARKER (1-6) on the screen until another menu is                          
OR           selected for display or until the                            
PRESS &lt;ENTER&gt;                                                             
             CLEAR/RET LOC key is pressed.                                
TO SELECT                                                                 
______________________________________                                    
 
    
     
                                           TABLE 4-2                               
__________________________________________________________________________
Command Code Classes                                                      
Command Code Description                    Para.                         
                                                Table                     
                                                    Page                  
__________________________________________________________________________
Channels     DSP, D13, D24, D14, CH1, CH2, CH3, CH4                       
                                            4-4.4                         
                                                4-4 4-16                  
Data entry   0 thru 9, -,.,GHZ, MHZ, KHZ, PSC, NSC, USC,                  
                                            4-4.5                         
                                                4-5 4-16                  
             DBM, DEG, MMT, CMT, MTR, XX1, XX3, XM3, REU, IMU             
Measurement Control                                                       
             S11, S21, S22, S12, SRT, STP, PWR, CWF, FHI,                 
                                            4-4.6                         
                                                4-6 4-17                  
             FLO, HLD, TRS, CTN, WFS, SA1, SA2, TA2, SWP                  
Display Control                                                           
             MAG, PHA, MPH, SWR, SMI, ISM, SME, ISE, SMC,                 
                                            4-4.7                         
                                                4-7 4-19                  
             DLA, PLR, PLG, LIN, LPH, REL, IMG, RIM, SCL, OFF,            
             REF, ASC, APR                                                
Enhancement  IFN, IFR, IFM, AVG, AOF, SON, SOF                            
                                            4-4.8                         
                                                4-8 4-22                  
Reference Delay                                                           
             RDD, RDT, RDA, DIA, DIT, DIP, DIM, DIE                       
                                            4-4.9                         
                                                4-9 4-22                  
Trace Memory DAT, MEM, DTM, DNM, MIN, DIV, ADD, MUL, STD,                 
                                             4-4.10                       
                                                 4-10                     
                                                    4-23                  
             SDK, RCK                                                     
Markers      MK1-MK6, MOF, MON, MO1-MO6, DR1-DR6, DRO,                    
                                             4-4.11                       
                                                 4-11                     
                                                    4-25                  
             DRF, MR1-MR6, MMN, MMX, M1S-M6S, M1E-M6E,                    
             M1C-M6C                                                      
Limits       LUP, LLO, LON, LOF, LFR, LFD, LFP                            
                                             4-4.12                       
                                                 4-12                     
                                                    4-26                  
Hard copy output                                                          
             TDD, PFS, PGR, PMK, PTB, PMT, PT0-PT9, PST,                  
                                             4-4.13                       
                                                 4-13                     
                                                    4-28                  
             PLS, PLD, PLT, PMN, PLH, PGT, PFL, PTL, PTR, PBL,            
             PBR, DPN, GPN, MPN, HPN, LDT, LNM, LMS, LID, TDD,            
             RTB, SPD                                                     
Miscellaneous                                                             
             RST, FON, FOF, BC0, BC1, BLU, CYN, TST, RTL,                 
                                             4-4.14                       
                                                 4-14                     
                                                    4-30                  
Functions    DGS, DCP, DC1, DC3                                           
Calibration  C12, C8T, CRF, CFR, CFT, CRL, NOC, SRT, STP,                 
                                            5-5.1                         
                                                *   *                     
             FRS, FRI, FRP, FIL, DFQ, DFD, CWC, CWF, TDC, P1C,            
             P2C, CMS, CFS, CMK, CFK, CMN, CFN, CM3, CF3, CNG,            
             CND, COO, COS, CC0, CC1, CC2, CC3, BBL, SLD, BEG,            
             TCD, NCS, KEC, CON, COF, A12, A8T, ARF, AFR, AFT,            
             ARL                                                          
Save/Recall  STO, RLD, SV1-SV4, RC1-RC4     5-5.2                         
                                                *   *                     
Data Transfer                                                             
             OM1-OM6, OAP, OKP, OID, ONP, FMA, FMB, FMC,                  
                                            5-5.3                         
                                                *   *                     
             MSB, OCl-OC9, OCA, OCB, OCC, OCL OCD, IC1-IC9                
             ICA, ICB, ICC, ICL, IFV, OS1-OS4, OFP, IS1-IS4,              
             IFP, ODR, ORD, OFD, ICD, IFD                                 
Group Execute Trigger                                                     
             DEF, END                       5-5.4                         
                                                *   *                     
Disk Functions                                                            
             SDK, RCK, STO, RLD, TDD, RTB, DEC, DEN, INT,                 
                                            5-5.5                         
                                                *   *                     
             LKT                                                          
Time Domain  FQD, DBP, TBP, DPI, TPI, DLP, TLP, DCA, DCZ,                 
                                            5-5.6                         
                                                *   *                     
             DCS, DCV, MRR, LPI, LPS, FGT, GON, GOF, GDS, SRT,            
             STP, GST, GSP, GCT, GSN, GRT, GNM, GLS, GMS, WRT,            
             WNM, WLS, WMS                                                
Status bytes/SRQ                                                          
             LPI, LPS, OPB, OEB, IPM, IEM, SQ0, SQ1, CSB                  
                                            5-5.7                         
                                                *   *                     
__________________________________________________________________________
 *Denotes material to be included in Section V of the followon packet     
 
    
     
                                           TABLE 4-3                               
__________________________________________________________________________
An Alphabetical List of Command Codes                                     
Command Code                                                              
          Description                  Para.                              
                                           Table                          
                                               Page                       
__________________________________________________________________________
ADD       Select Complex Addition as Trace Math                           
                                        4-4.10                            
                                            4-10                          
                                               4-23                       
AFR       Assume Frequency Response    *   *   *                          
AFT       Assumme Frequency Response - Transmission Only                  
                                       *   *   *                          
AOF       Averaging Off                4-4.8                              
                                           4-8 4-22                       
APR       Set Group Delay Aperature Percentage                            
                                       *   *   *                          
ARF       Assume Reflection Only - One Port                               
                                       *   *   *                          
ARL       Assume Frequency Response - Reflecton Only                      
                                       *   *   *                          
ASC       Autoscale Display            4-4.7                              
                                           4-7 4-19                       
AVG (value)                                                               
          Averaging On and Set to Value                                   
                                       4-4.8                              
                                           4-8 4-22                       
A8T       Assume 8 Term                *   *   *                          
A12       Assume 12 Term               *   *   *                          
BBL       Broadband Load               *   *   *                          
BC0       Blank CRT                     4-4.14                            
                                            4-14                          
                                               4-30                       
BC1       Unblank CRT                   4-4.14                            
                                            4-14                          
                                               4-30                       
BEG       Begin Calibration            *   *   *                          
BLU       Select Blue Color             4-4.14                            
                                            4-14                          
                                               4-30                       
C12       Calibrate Full 12 Term       *   *   *                          
C8T       Calibrate 8 Term - One Path Two Port)                           
                                       *   *   *                          
CC0-CC3 (value)                                                           
          Capacitance For Open Device  *   *   *                          
CFK       Female K Connector ™ for the Specified Port                  
                                       *   *   *                          
CFN       Female Type N Connector for the Specified Port                  
                                       *   *   *                          
CFS       Female SMA Connector for the Specified Port                     
                                       *   *   *                          
CFR       Calibrate Frequency Response *   *   *                          
CFT       Calibrate Frequency Response - Transmission Only                
                                       *   *   *                          
CH1-CH4   Select Active Channel 1 Through 4                               
                                       4-4.4                              
                                           4-4 4-16                       
CMK       Male K Connector For The Specified Port                         
                                       *   *   *                          
CMN       Male Type N Connector For The Specified Port                    
                                       *   *   *                          
CMS       Male SMA Connector For The Specified Port                       
                                       *   *   *                          
CMT       Centimeter                   4-4.5                              
                                           4-5 4-16                       
CM3       Male GPC-3.5 Connector for the Specified Port                   
                                       *   *   *                          
CF3       Female GPC-3.5 Connector for the Specified Port                 
                                       *   *   *                          
CND       Connector Type Other         *   *   *                          
CNG       Connector Type GPC-7         *   *   *                          
CNO (value)                                                               
          Connector Offset Value For &#34;OTHER&#34; connector                    
                                       *   *   *                          
CCF       Correction Off               *   *   *                          
CON       Apply Cal Correction         *   *   *                          
COO (value)                                                               
          Connector Offset for Open Device                                
                                       *   *   *                          
COS (value)                                                               
          Connector Offset for Short Device                               
                                       *   *   *                          
CRF       Calibrate Reflection Only - One Port                            
                                       *   *   *                          
CRL       Calibrate Frequency Response - Reflecion Only                   
                                       *   *   *                          
CSB       Clear Both Status Bytes      *   *   *                          
CTN       Continue Sweeping From Hold  4-4.6                              
                                           4-6 4-17                       
CWC       CW Cal                       *   *   *                          
CWF (value)                                                               
          CW Frequency                 4-4.6                              
                                           4-6 4-17                       
CYN       Select Cyan Color             4-4.14                            
                                            4-14                          
                                               4-30                       
D13       Dual Channel 1 and 3 Display 4-4.4                              
                                           4-4 4-16                       
D14       Four Channel Display 1 Through 4                                
                                       4-4.4                              
                                           4-4 4-16                       
D24       Dual Channel 2 and 4 Display 4-4.4                              
                                           4-4 4-16                       
DAT       Display Data                  4-4.10                            
                                            4-10                          
                                               4-23                       
DBL       dB                           4-4.5                              
                                           4-5 4-16                       
DBM       dBm                          4-4.5                              
                                           4-5 4-16                       
DBP       Distance Bandpass Mode       *   *   *                          
DC1       Display Channels 1 and 2 Operating Parameters                   
                                        4-4.14                            
                                            4-14                          
                                               4-30                       
DC3       Display Channels 3 and 4 Operating Parameters                   
                                        4-4.14                            
                                            4-14                          
                                               4-30                       
DCA       DC Term For Low Pass, Auto   *   *   *                          
DCO       DC Term For Low Pass, Open   *   *   *                          
DCP       Display Calibration Parameters                                  
                                       *   *   *                          
DCS       DC Term For Low Pass, Short  *   *   *                          
DCV (value)                                                               
          Set DC Term for Low Pass To Value Specified                     
                                       *   *   *                          
DCZ       DC Term for Lowpass, Line Impedance                             
                                       *   *   *                          
DEC (filename)                                                            
          Delete Cal File              *   *   *                          
DED (filename)                                                            
          Delete Data File             *   *   *                          
DEF       Begin Definition of GET Action                                  
                                       *   *   *                          
DEG       Degree                       4-4.5                              
                                           4-5 4-16                       
DEN (filename)                                                            
          Delete Norm File             *   *   *                          
DFC       Discrete Frequency Cal       *   *   *                          
DFD       Discrete Frequencies Done    *   *   *                          
DFP       Display Global Operating Parameters                             
                                        4-4.14                            
                                            4-14                          
                                               4-30                       
DFQ (value)                                                               
          Discrete Frequencies         *   *   *                          
DGS       Display GPIB Status           4-4.14                            
                                            4-14                          
                                               4-30                       
DIA       Set Dielectric Constant to 1.0 for Air                          
                                       4-4.9                              
                                           4-9 4-22                       
DIE (value)                                                               
          Enter Dielectric Constant    4-4.9                              
                                           4-9 4-22                       
DIM       Set Dielectric Constant to 1.69 for Microporous                 
                                       4-4.9n                             
                                           4-9 4-22                       
DIP       Set Dielectric Constant to 2.26 for Polyethylene                
                                       4-4.9                              
                                           4-9 4-22                       
DIT       Set Dielectric Constant to 2.1 for Teflon                       
                                       4-4.9                              
                                           4-9 4-22                       
DIV       Set Complex Division as Trace Math                              
                                        4- 4.10                           
                                            4-10                          
                                               4-23                       
DLA       Group Delay Display Format   4-4.7                              
                                           4-7 4-19                       
DLP       Distance Lowpass Mode        *   *   *                          
DNM       Display Data Normalized to Memory                               
                                        4-4.10                            
                                            4-10                          
                                               4-23                       
DPI       Distance Phasor Impulse Mode *   *   *                          
DPN (value)                                                               
          Data Pen Number               4-4.13                            
                                            4-13                          
                                               4-28                       
DR1-DR6   Select Delta Reference Marker                                   
                                        4-4.11                            
                                            4-11                          
                                               4-25                       
DRF       Delta Reference Mode On       4-4.11                            
                                            4-11                          
                                               4-25                       
DRO       Delta Reference Mode Off      4-4.11                            
                                            4-11                          
                                               4-25                       
DSP       Single Channel Display       4-4.4                              
                                           4-4 4-16                       
DTM       Display Data and Memory       4-4.10                            
                                            4-10                          
                                               4-23                       
END       End Definition of GET Action *   *   *                          
FFD       Form Feed Sent to Printer    *   *   *                          
FGT       Frequency With Time Gate     *   *   *                          
FHI       Frequency Resolution High    4-4.6                              
                                           4-6 4-17                       
FIL       Fill Range                   *   *   *                          
FLO       Frequency Resolution Low     4-4.6                              
                                           4-6 4-17                       
FMA       ASCII Format Data Transfer   *   *   *                          
FMB       Floating Point Format 64 Bit Data Transfer                      
                                       *   *   *                          
FMC       Floating Point Format 32 Bit Data Transfer                      
                                       *   *   *                          
FME       Frequency Resolution Medium  4-4.6                              
                                           4-6 4-17                       
FOF       Blank All Frequency Information                                 
                                        4-4.14                            
                                            4-14                          
                                               4-30                       
FON       Display All Frequency Information                               
                                        4-4.14                            
                                            4-14                          
                                               4-30                       
FQD       Frequency Domain             *   *   *                          
FRI (value)                                                               
          Fill Range Increment         *   *   *                          
FRP (value)                                                               
          Fill Range Points            *   *   *                          
FRS (value)                                                               
          Fill Range Start             *   *   *                          
GCT (value)                                                               
          Gate Center, Value in Time or Distance                          
                                       *   *   *                          
GDS       Gate Symbol Displayed        *   *   *                          
GHZ       Gigahertz                    4-4.5                              
                                           4-5 4-16                       
GLS       Gate Shape Low Sidelobe      *   *   *                          
GMS       Gate Shape Minimum Sidelobe  *   *   *                          
GNM       Gate Shape Nominal           *   *   *                          
GOF       Gate Off                     *   *   *                          
GON       Gate On                      *   *   *                          
GPN (value)                                                               
          Graticule Pen Number          4-4.13                            
                                            4-13                          
                                               4-28                       
GRT       Gate Shape Rectangular       *   *   *                          
GSN (value)                                                               
          Gate Span, Value in Time or Distance                            
                                       *   *   *                          
GSP (value)                                                               
          Gate Stop, Value in Time or Distance                            
                                       *   *   *                          
GST (value)                                                               
          Gate Start, Value in Time or Distance                           
                                       *   *   *                          
HLD       Hold Mode                    4-4.6                              
                                           4-6 4-17                       
HPN (value)                                                               
          Header Pen Number             4-4.13                            
                                            4-13                          
                                               4-28                       
IC1-IC9   Load Cal Coefficients 1 Through 9                               
                                       *   *   *                          
ICA-ICC   Load Cal Coefficients 10 Through 12                             
                                       *   *   *                          
ICD       Load Corrected Data          *   *   *                          
ICL       Input 12 Calibration Coefficients in String Format              
                                       *   *   *                          
IEM       Input Extended Status Byte Mask                                 
                                       *   *   *                          
IFN       Set IF Bandwith to Normal    4-4.8                              
                                           4-8 4-22                       
IFR       Set IF Bandwith to Reduced   4-4.8                              
                                           4-8 4-22                       
IFM       Set IF Bandwith to Minimum   4-4.8                              
                                           4-8 4-22                       
IFD       Load Final Data              *   *   *                          
IFV       Input List Of Frequencies    *   *   *                          
IMG       Imaginary Display Format     4-4.7                              
                                           4-7 4-16                       
IMU       Imaginary Units              4-4.5                              
                                           4-5 4-16                       
INT       Initalize Data Disk          *   *   *                          
IPM       Input Primary Status Byte Mask                                  
                                       *   *   *                          
ISC       Select Inverted Compressed Smith Chart for Active               
                                       4-4-7el                            
                                           4-7 4-16                       
ISE (value)                                                               
          Expand Inverted Smith Chart (10, 20, or 30 dB)                  
                                       4-4.7                              
                                           4-7 4-16                       
ISM       Inverted Smith Chart         4-4.7                              
                                           4-7 4-16                       
KEC       Keep Existing Cal            *   *   *                          
KHZ       Kilohertz                    4-4.5                              
                                           4- 5                           
                                               4-16                       
LDT (string)                                                              
          Label Date                    4-4.13                            
                                            4-13                          
                                               4-28                       
LFD (value)                                                               
          Set Limit Delta for Limit Frequency Readout on                  
                                        4-4.12                            
                                            4-12                          
                                               4-26                       
          Channel                                                         
LFP       Select Phase Limit Readout for Active Channel                   
                                        4-4.12                            
                                            4-12                          
                                               4-26                       
LFR       Select Limit Frequency Readout for Active Channel               
                                        4-4.12                            
                                            4-12                          
                                               4-26                       
LIN       Linear Magnitude Display Format                                 
                                       4-4.7                              
                                           4-7 4-19                       
LID (string)                                                              
          Label Device ID               4-4.13                            
                                            4-13                          
                                               4-28                       
LLO (value)                                                               
          Set Lower Limit               4-4.12                            
                                            4-12                          
                                               4-26                       
LMS (string)                                                              
          Label Model and Serial Number                                   
                                        4-4.12                            
                                            4-12                          
                                               4-26                       
LNM (string)                                                              
          Label Operator&#39;s Name         4-4.12                            
                                            4-12                          
                                               4-26                       
LOF       Disable Limits                4-4.12                            
                                            4-12                          
                                               4-28                       
LON       Enable Limits                 4-4.12                            
                                            4-12                          
                                               4-28                       
LPH       Linear Magnitude and Phase Display Format                       
                                       4-4.7                              
                                           4-7 4-19                       
LPI       Lowpass Impulse              *   *   *                          
LPS       Lowpass Step                 *   *   *                          
LSB       Least Significant Byte First Data Transfer Mode                 
                                       *   *   *                          
LUP (value)                                                               
          Set Upper Limit               4-4.12                            
                                            4-12                          
                                               4-28                       
MAG       Log Magnitude Display Format 4-4.7                              
                                           4-7 4-19                       
MEM       Display Memory                4-4.10                            
                                            4-10                          
                                               4-23                       
MHZ       Megahertz                    4-4.5                              
                                           4-5 4-16                       
MIN       Select Complex Subtraction as Trace Math                        
                                        4-4.10                            
                                            4-10                          
                                               4-23                       
MK1-MK6 (value)                                                           
          Set Marker To Value           4-4.11                            
                                            4-11                          
                                               4-25                       
MMN       Marker To Min                 4-4.11                            
                                            4-11                          
                                               4-25                       
MMT       Millimeter                    4-4.11                            
                                            4-11                          
                                               4-25                       
MMX       Marker To Max                 4-4.11                            
                                            4-11                          
                                               4-25                       
MO1-MO6   Markers Off                   4-4.11                            
                                            4-11                          
                                               4-25                       
M1C-M6C   Marker Sweep CW Frequency     4-4.11                            
                                            4-11                          
                                               4-25                       
M1E-M6E   Marker Sweep End Frequency    4-4.11                            
                                            4-11                          
                                               4-25                       
M1S-M6S   Marker Sweep Start Frequency  4-4.11                            
                                            4-11                          
                                               4-25                       
MON       Markers Enabled               4-4.11                            
                                            4-11                          
                                               4-25                       
MPH       Log Magnitude and Phase Display Format                          
                                       4-4.7                              
                                           4-7 4-19                       
MPN (value)                                                               
          Marker Pen Number             4-4.13                            
                                            4-13                          
                                               4-28                       
MR1-MR6   Marker Selected for Readout and Data Output                     
                                         4-4.11                           
                                            4-11                          
                                               4-25                       
MRR       Marker Range Restore         *   *   *                          
MSB       Most Signifigant Byte First Data Transfer Mode                  
                                       *   *   *                          
MTR       Meter                        4-4.5                              
                                           4-5 4-16                       
MUL       Select Complex Multiplication as Trace Math                     
                                        4-4.10                            
                                            4-10                          
                                               4-23                       
NCS       Next Cal Step                *   *   *                          
NOC       Normal Cal                   *   *   *                          
NSC       Nanosecond                   4-4.5                              
                                           4-5 4-16                       
OAP       Output Active Parameter      *   *   *                          
OC1-OC9   Output Cal Coefficients 1 Through 9                             
                                       *   *   *                          
OCA-OCC   Output Cal Coefficients 10 Through 12                           
                                       *   *   *                          
OCD       Output Corrected Data        *   *   *                          
OCL       Output All 12 Cal Coefficients in String Format                 
                                       *   *   *                          
ODR       Output Disk Directory        *   *   *                          
OEB       Output Extended Status Byte  *   *   *                          
OFD       Output Final Data            *   *   *                          
OFF (value)                                                               
          Set Offset Value             4-4.7                              
                                           4-7 4-19                       
OFP       Output Front Panel Setup in String Format                       
                                       *   *   *                          
OFV       Output Frequency Values      *   *   *                          
OID       Output Identify              *   *   *                          
OKP       Output Front Panel Key Pressed                                  
                                       *   *   *                          
OM1-OM6   Output Marker Value 1 Through 6                                 
                                       *   *   *                          
ONP       Output Number Of Data Points *   *   *                          
OPB       Output Primary Status Byte   *   *   *                          
ORD       Output Raw Data              *   *   *                          
OS1-OS4   Output Stored Setup In String Format                            
                                       *   *   *                          
P1C       Port 1 Connector Specification                                  
                                       *   *   *                          
P2C       Port 2 Connector Specification                                  
                                       *   *   *                          
PBL       Plot Bottom Left, 1/4 Screen  4-4.13                            
                                            4-13                          
                                               4-28                       
PBR       Plot Bottom Right, 1/4 Screen                                   
                                        4-4.13                            
                                            4-13                          
                                               4-28                       
PFS       Print Full Screen             4-4.13                            
                                            4-13                          
                                               4-28                       
PGR       Print Graph                   4-4.13                            
                                            4-13                          
                                               4-28                       
PGT       Plot Graticule                4-4.13                            
                                            4-13                          
                                               4-28                       
PHA       Phase Display Format         4-4.7                              
                                           4-7 4-19                       
PLD       Plot Data Area Only           4-4.13                            
                                            4-13                          
                                               4-28                       
PFL       Full Size Plot                4-4.13                            
                                            4-13                          
                                               4-26                       
PLG       Polar Log Display Format     4-4.7                              
                                           4-7 4-19                       
PLH       Plot Header                   4-4.13                            
                                            4-13                          
                                               4-28                       
PLM       Plot Markers and Limits       4-4.13                            
                                            4-13                          
                                               4-28                       
PLR       Polar Linear Display Format  4-4.7                              
                                           4-7 4-19                       
PLS       Plot Entire Screen            4-4.13                            
                                            4-13                          
                                               4-28                       
PLT       Plot Trace(s)                 4-4.13                            
                                            4-13                          
                                               4-28                       
PMK       Print Marker Data Only        4-4.13                            
                                            4-13                          
                                               4-28                       
PMN       Plot Menu                     4-4.13                            
                                            4-13                          
                                               4-28                       
PMT       Print Marker And Tabular Data                                   
                                        4-4.13                            
                                            4-13                          
                                               4-28                       
PSC       Picosecond                   4-4.5                              
                                           4-5 4-16                       
PST       Stop Print /Plot              4-4.13                            
                                            4-13                          
                                               4-28                       
PT0-PT9   Set Print Tabular Data Density                                  
                                        4-4.13                            
                                            4-13                          
                                               4-28                       
PTB       Print Tabular Data Only       4-4.13                            
                                            4-13                          
                                               4-28                       
PTL       Plot Top left, 1/4 Size       4-4.13                            
                                            4-13                          
                                               4-28                       
PTR       Plot Top Right, 1/4 Size      4-4.13                            
                                            4-13                          
                                               4-28                       
PWR (value)                                                               
          Set Power Level              4-4.6                              
                                           4-6 4-17                       
RC1-RC4   Recall Front Panel Setup From Internal Memory                   
                                        4-4.10                            
                                            4-10                          
                                               4-23                       
          Numbers 1 Through 4                                             
RCK (filename)                                                            
          Recall Normalization Data From Disk                             
                                       4-4.9                              
                                           4-9 4-22                       
RDA       Autoadjust Reference Delay   4-4.9                              
                                           4-9 4-22                       
RDD (value)                                                               
          Reference Delay (Value in Distance)                             
                                       4-4.9                              
                                           4-9 4-22                       
RDT (value)                                                               
          Reference Delay (Value in Time)                                 
                                       4-4.9                              
                                           4-9 4-22                       
REF       Set Reference Line of Display on Active Channel                 
                                       4-4.7                              
                                           4-7 4-19                       
REL       Real Display Format          4-4.7                              
                                           4-7 4-19                       
REU       Real Units                   4-4.5                              
                                           4-5 4-16                       
RIM       Real and Imaginary Display Format                               
                                       4-4.7                              
                                           4-7 4-19                       
RLD (filename)                                                            
          Reload Cal Data And Front Panel Setup From Disk                 
                                       *   *   *                          
RST       Reset 360 To Default Parameters                                 
                                        4-4.14                            
                                            4-14                          
                                               4-30                       
RTB (filename)                                                            
          Recall Tabular Data From Disk                                   
                                        4-4.13                            
                                            4-13                          
                                               4-28                       
RTL       Return to Local               4-4.14                            
                                            4-14                          
                                               4-30                       
S11       Measure S.sub.11 On Active Channel                              
                                       4-4.6                              
                                           4-6 4-17                       
S12       Measure S.sub.12 On Active Channel                              
                                       4-4.6                              
                                           4-6 4-17                       
S21       Measure S.sub.21 On Active Channel                              
                                       4-4.6                              
                                           4-6 4-17                       
S22       Measure S.sub.22 On Active Channel                              
                                       4-4.6                              
                                           4-6 4-17                       
SA1 (value)                                                               
          Set Source Attenuator, Port 1                                   
                                       4-4.6                              
                                           4-6 4-17                       
SA2 (value)                                                               
          Set Source Attenuator, Port 2                                   
                                       4-4.6                              
                                           4-6 4-17                       
SCL (value)                                                               
          Set Resolution (scale) of Display                               
                                       4-4.7                              
                                           4-7 4-19                       
SDK (filename)                                                            
          Store Data on Disk            4-4.10                            
                                            4-10                          
                                               4-23                       
SLD       Sliding Load                 *   *   *                          
SMC (value)                                                               
          Compressed Smith Chart (dB)  4-4.7                              
                                           4-7 4-19                       
SME (value)                                                               
          Expanded Smith Chart (10, 20, or 30 dB)                         
                                       4-4.7                              
                                           4-7 4-19                       
SMI       Smith Chart                  4-4.7                              
                                           4-7 4-19                       
SOF       Smoothing Off                4-4.8                              
                                           4-8 4-22                       
SON (value)                                                               
          Smoothing On and Set to Value                                   
                                       4-4.8                              
                                           4-8 4-22                       
SPD (value)                                                               
          Set Plotter Pen Speed Percentage                                
                                        4-4.13                            
                                            4-13                          
                                               4-28                       
SQ0       Disable SRQ                  *   *   *                          
SQ1       Enable SRQ                   *   *   *                          
SRT (value)                                                               
          Start Frequency, Distance, Or Time Value                        
                                       4-4.6                              
                                           4-6 4-17                       
STD       Store Active Channel Data Trace to Memory                       
                                        4-4.10                            
                                            4-10                          
                                               4-23                       
STO (filename)                                                            
          Store Cal Data And Front Panel Setup On Disk                    
                                       *   *   *                          
STP (value)                                                               
          Stop Frequency, Distance Or Time Value                          
                                       4-4.6                              
                                            4-10                          
                                               4-23                       
SV1-SV4   Save Front Panel Setup to Internal Memory                       
                                       *   *   *                          
          Numbers 1 Through 4                                             
SWP       Select Continuous Sweep Mode 4-4.6                              
                                            4-10                          
                                               4-23                       
SWR       Select SWR Display Format    4-4.7                              
                                           4-7 4-19                       
TA2 (value)                                                               
          Set Test Attenuator, Port 2  4-4.6                              
                                           4-6 4-17                       
TBP       Time Bandpass Mode           *   *   *                          
TCD       Take Cal Data                *   *   *                          
TDC       Time Domain (Harmonic) Cal   *   *   *                          
TDD (filename)                                                            
          Tabular Data to Disk          4-4.13                            
                                            4-13                          
                                               4-28                       
TLP       Time Lowpass Mode            *   *   *                          
TPI       Time Phasor Impulse Mode     *   *   *                          
TRS       Trigger/Restart Sweep        4-4.6                              
                                           4-6 4-17                       
TST       Self Test                     4-4.14                            
                                            4-14                          
                                               4-30                       
USC       Microsecond                  4-4.5                              
                                           4-5 4-16                       
WFS       Wait Full Sweep              4-4.6                              
                                           4-6 4-17                       
WLS       Window Shape Low Sidelobe    *   *   *                          
WMS       Window Shape Min Sidelobe    *   *   *                          
WNM       Window Shape Nominal         *   *   *                          
WRT       Window Shape Rectangular     *   *   *                          
XM3       Unitless Terminator × 10.sup.-3                           
                                       4-4.5                              
                                           4-5 4-16                       
XX1       Unitless Terminator          4-4.5                              
                                           4-5 4-16                       
XX3       Unitless Terminator × 10.sup.3                            
                                       4-4.5                              
                                           4-5 4-16                       
__________________________________________________________________________
 
    
     
                       TABLE 4-4                                                   
______________________________________                                    
Channel Control Command Codes                                             
COMMAND                                                                   
CODE        DESCRIPTION                                                   
______________________________________                                    
DSP         Single Channel Display of                                     
            Active Channel                                                
D13         Dual Channel Display,                                         
            Channels 1 and 3                                              
D24         Dual Channel Display,                                         
            Channels 2 and 4                                              
D14         Quad Display, All Four Channels                               
CH1         Channel 1 Selected As Active Channel                          
CH2         Channel 2 Selected As Active Channel                          
CH3         Channel 3 Selected As Active Channel                          
CH4         Channel 4 Selected As Active Channel                          
______________________________________                                    
 
    
     
                       TABLE 4-5                                                   
______________________________________                                    
Data Entry Command Codes                                                  
COMMAND                                                                   
CODE          DESCRIPTION                                                 
______________________________________                                    
0,1,2,3,4,5,  Numerals For Numeric Entry                                  
6,7,8,9                                                                   
-             Minus Sign                                                  
.             Decimal Point                                               
GHZ           Gigahertz Data Terminator                                   
MHZ           Megahertz Data Terminator                                   
KHZ           Kilohertz Data Terminator                                   
PSC           Picoseconds Data Terminator                                 
NSC           Nanoseconds Data Terminator                                 
USC           Microseconds Data Terminator                                
DBL           dB Log Data Terminator                                      
DBM           dBm Data Terminator                                         
DEG           Degrees Data Terminator                                     
MMT           Millimeter Data Terminator                                  
CMT           Centimeter Data Terminator                                  
MTR           Meter Data Terminator                                       
XX1           Unitless Data Terminator, ×1                          
XX3           Unitless Data Terminator, ×10.sup.-3                  
XM3           Unitless Data Terminator, ×10.sup.-3                  
REU           Real Units Data Terminator                                  
IMU           Imaginary Units Data Terminator                             
______________________________________                                    
 
    
     
                                           TABLE 4-6                               
__________________________________________________________________________
Measurement Control Command Codes                                         
COMMAND                                                                   
CODE   DESCRIPTION           VALUES      TERMINATORS                      
__________________________________________________________________________
S11    Selects S.sub.11 as S-Parameter On Active Channel                  
                             N/A         N/A                              
S21    Selects S.sub.21 as S-Parameter On Active Channel                  
                             N/A         N/A                              
S22    Selects S.sub.22 as S-Parameter On Active Channel                  
                             N/A         N/A                              
S12    Selects S.sub.12 as S-Parameter On Active Channel                  
                             N/A         N/A                              
SRT    Sets Start Frequency  Depends On Frequency                         
                                         GHZ, MHZ, KHZ                    
                             Range Of Instrument                          
STP    Sets Stop Frequency   Depends On Frequency                         
                                         GHZ, MHZ, KHZ                    
                             Range Of Instrument                          
CWF    Sets CW Frequency     Depends On Frequency                         
                                         GHZ, MHZ, KHZ                    
                             Range Of Instrument                          
PWR    Sets Source Power     Depends On Power                             
                                         DBM, XX1, XX3, XM3               
                             Range Of Source                              
FHI    Sets Data Points To Maximum                                        
                             N/A         N/A                              
FME    Sets Data Points To Normal                                         
                             N/A         N/A                              
FLO    Sets Data Points To Minimum                                        
                             N/A         N/A                              
SA1    Sets Source Attenuator For Port 1                                  
                             0 dB to 70 dB                                
                                         DBL, DBM, XX1, XX3, XM3          
SA2    Sets Source Attenuator For Port 2                                  
                             0 dB to 70 dB                                
                                         DBL, DBM, XX1, XX3, XM3          
TA2    Sets Test Attenuator For Port 2                                    
                             0 dB to 40 dB                                
                                         DBL, DBM, XX1, XX3, XM3          
HLD    Holds Sweep At Current Point                                       
                             N/A         N/A                              
CTN    Continue Sweep After Hold                                          
                             N/A         N/A                              
TRS    Triggers or Restarts a Sweep                                       
                             N/A         N/A                              
WFS    Wait full sweep       N/A         N/A                              
SWP    Selects Continuous Sweep Mode                                      
                             N/A         N/A                              
__________________________________________________________________________
 
    
     
                                           TABLE 4-7                               
__________________________________________________________________________
Display Control Command Codes                                             
COMMAND                                                                   
CODE   DESCRIPTION             VALUES       TERMINATORS                   
__________________________________________________________________________
MAG    Selects Log Magnitude Display for Active                           
                               N/A          N/A                           
       Channel                                                            
PHA    Selects Phase Display for Active Channel                           
                               N/A          N/A                           
MPH    Selects Log Magnitude and Phase Display for                        
                               N/A          N/A                           
       Active Channel                                                     
SMI    Selects Normal Smith Chart Display for Active                      
                               N/A          N/A                           
       Channel                                                            
SWR    Selects SWR Display for Active Channel                             
                               N/A          N/A                           
ISM    Selects Inverted Normal Smith Chart Display for                    
                               N/A          N/A                           
       Active Channel                                                     
DLA    Selects Group Delay Display for Active Channel                     
                               N/A          N/A                           
PLR    Selects Linear Polar Display for Active Channel                    
                               N/A          N/A                           
PLG    Selects Log Polar Display for Active Channel                       
                               N/A          N/A                           
LIN    Selects Linear Magnitude Display for Active                        
                               N/A          N/A                           
       Channel                                                            
LPH    Selects Linear Magnitude and Phase Display for                     
                               N/A          N/A                           
       Active Channel                                                     
REL    Selects Real Display for Active Channel                            
                               N/A          N/A                           
IMG    Selects Imaginary Display for Active Channel                       
                               N/A          N/A                           
RIM    Selects Real And Imaginary Display for Active                      
                               N/A          N/A                           
       Channel                                                            
SME    Selects Expanded Smith Chart Display for                           
                               10.20.30     DBL.XX1                       
       Active Channel                                                     
ISE    Selects Inverted Expanded Smith Chart Display                      
                               10.20.30     DBL.XX1                       
       for Active Channel                                                 
SMC    Selects Compressed Smith Chart Display for                         
                               3            DBL.XX1                       
       Active Channel                                                     
ISC    Selects Inverted Compressed Smith Chart                            
                               3            DBL.XX1                       
       Display for Active Channel                                         
SCL    Sets Scaling Of Display On Active Channel                          
                               Depends on Graph Type:                     
                                            Depends on Graph Type:        
       Log Mag and Log Polar:  0.001 to 50 dB/div                         
                                            DBL.XX1.XX3.XM3               
       Phase:                  0.01 to 90 degrees/div                     
                                            DEG (.XX1.XX3 XM3             
                                            for PHA display)              
       Group Delay:            1 femtosecond/div to                       
                                            PSC.NSC.USC                   
                               999.999 s/div                              
       Linear Mag &amp; Linear Polar:                                         
                               1 nanounit/div to                          
                                            XX1.XX3.XM3                   
                               999.999 units/div                          
       Real:                   1 nanounit/div to                          
                                            REU.XX1.XX3.XM3               
                               999.999 units/div                          
       Imag:                   1 nanounit/div to                          
                                            IMU (.XX1.XX3.XM3             
                               999.999 units/div                          
                                            for IMG display)              
       Smith/inverted Smith:   N/A          N/A                           
OFF    Set Offset of Display on Active Channel                            
                               Depends On Graph Type:                     
                                            Depends On Graph Type:        
       (This code moves the graph&#39;s reference                             
       position to the offset value)                                      
       Log Mag &amp; Log Polar:    -999.999 to 999.999 dB                     
                                            DBL.XX1.XX3.XM3               
       Phase:                  -180 to 180 degrees                        
                                            DEG (.XX1.XX3.XM3             
                                            for PHA display)              
       Group Delay:            -999.999 to 999.999 s                      
                                            PSC.NSC.USC                   
       Linear Mag &amp; Linear Polar:                                         
                               0 to 999.999 units                         
                                            XX1.XX3.XM3                   
       Real:                   -999.999 to 999.999 units                  
                                            REU.XX1.XX3.XM3               
       Imaginary               -999.999 to 999.999 units                  
                                            IMU (.XX1.XX3.XM3             
                                            for IMG display)              
       Smith/Inverted Smith:   N/A          N/A                           
REF    Set Reference Line of Display on Active                            
                               Depends on Graph Type:                     
                                            Depends on Graph Type:        
       Channel                                                            
       Log Magnitude, MAG Display:                                        
                               0 to 8       DBL.XX1.XX3.XM3               
       Log Magnitude, MPH Display:                                        
                               0 to 4       DBL.XX1.XX3.XM3               
       Phase, PHA Display:     0 to 8       DEG.XX1.XX3.XM3               
       Phase, MPH Display:     0 to 4       DEG                           
       Group Delay:            0 to 8       PSC.NSC.USC.XX1.              
                                            XX3.XM3                       
       Linear Magnitude, LIN Display:                                     
                               0 to 8       XX1.XX3.XM3                   
       Linear Magnitude, LPH Display:                                     
                               0 to 4       XX1.XX3.XM3                   
       Real, REL Display:      0 to 8       REU.XX1.XX3.XM3               
       Real, RIM Display:      0 to 4       REU.XX1.XX3.XM3               
       Imaginary, IMG Display: 0 to 8       IMU.XX1.XX3.XM3               
       Imaginary, RIM Display: 0 to 4       IMU                           
       Smith/Inverted Smith:   N/A          N/A                           
       Linear Polar/Log Polar: N/A          N/A                           
ASC    Autoscale Display On Active Channel                                
                               N/A          N/A                           
APR    Set Group Delay Aperture Percentage                                
                               0 to 20      XX1.XX3.XM3                   
__________________________________________________________________________
 
    
     
                                           TABLE 4-8                               
__________________________________________________________________________
Enhancement Command Codes                                                 
COMMAND                                                                   
CODE   DESCRIPTION        VALUES                                          
                                TERMINATORS                               
__________________________________________________________________________
IFN    Selects Normal IF Bandwidth                                        
                          N/A   N/A                                       
IFR    Selects Reduced IF Bandwidth                                       
                          N/A   N/A                                       
IFM    Selects Minimum IF Bandwidth                                       
                          N/A   N/A                                       
AVG    Turns ON Averaging and Sets to Value                               
                          1 to 4095                                       
                                XX1.XX3.XM3                               
AOF    Turns Off Averaging                                                
                          N/A   N/A                                       
SON    Turns ON Smoothing and Sets to Value                               
                          0 to 20                                         
                                XX1.XX3.XM3                               
SOF    Turns Off Smoothing                                                
                          N/A   N/A                                       
__________________________________________________________________________
 
    
     
                                           TABLE 4-9                               
__________________________________________________________________________
Reference Delay Command Codes                                             
COMMAND                                                                   
CODE   DESCRIPTION           VALUES     TERMINATORS                       
__________________________________________________________________________
RDD    Sets Reference Delay As a Distance Value for                       
                             -999.999 to 999.999 s                        
                                        MMT.CMT.MTR                       
       the Active Channel                                                 
RDT    Sets Reference Delay As a Time Value for the                       
                             -999.999 to 999.999 s                        
                                        PSC.NSC.USC                       
       Active Channel                                                     
RDA    Selects Automatic Reference Delay for the                          
                             N/A        N/A                               
       Active Channel                                                     
DIA    Selects Air Dielectric (1.00)                                      
                             N/A        N/A                               
DIT    Selects Teflon Dielectric (2.10)                                   
                             N/A        N/A                               
DIP    Selects Polyethylene Dielectric (2.26)                             
                             N/A        N/A                               
DIM    Selects Microporous Teflon Dielectric (1.69)                       
                             N/A        N/A                               
DIE    Sets Dielectric to Value                                           
                             1 to 999.999                                 
                                        XX1.XX3.XM3                       
__________________________________________________________________________
 
    
     
                       TABLE 4-10                                                  
______________________________________                                    
Trace Memory Command Codes                                                
COMMAND                                                                   
CODE     DESCRIPTION                                                      
______________________________________                                    
DAT      Displays Data Trace on Active Channel                            
MEM      Displays Memory Trace on Active Channel                          
DTM      Displays Data and Memory Traces on Active                        
         Channel                                                          
DNM      Displays Measured Data Normalized to Memory                      
         on Active Channel                                                
MIN      Selects Complex Subtraction As Trace Math on                     
         Active Channel                                                   
DIV      Selects Complex Division As Trace Math on                        
         Active Channel                                                   
ADD      Selects Complex Addition As Trace Math on                        
         Active Channel                                                   
MUL      Selects Complex Multiplication As Trace Math on                  
         Active Channel                                                   
STD      Stores Active Channel&#39;s Data Trace to Memory                     
SDK      Stores Active Channel&#39;s Trace Memory to Disk                     
         Under The Specified File Name                                    
RCK      Retrieves Active Channel&#39;s Trace Memory From                     
         Disk File Specified                                              
______________________________________                                    
 
    
     
                                           TABLE 4-11                              
__________________________________________________________________________
Marker Command Codes                                                      
COMMAND                                                                   
CODE   DESCRIPTION            VALUES    TERMINATORS                       
__________________________________________________________________________
MK1-MK6                                                                   
       Turns On Marker 1-6 and Sets Them to Value                         
       As Shown Below:                                                    
       Frequency Markers      Limited to Current                          
                                        GHZ.MHZ.KHZ                       
                              Sweep Range                                 
       Time Markers:          Limited to Current                          
                                        PSC.NSC.USC                       
                              Zoom Range                                  
       Distance Markers:      Limited to Current                          
                                        MMT.CMT.MTR                       
                              Zoom Range                                  
MOF    Disables Markers       N/A       N/A                               
MON    Enables Markers        N/A       N/A                               
MO1-MO6                                                                   
       Turns Off Marker 1-6   N/A       N/A                               
DR1-DR6                                                                   
       Turns On Marker 1-6 As Delta Reference                             
                              N/A       N/A                               
       Marker                                                             
DRF    Turns On Delta Reference Marker Mode                               
                              N/A       N/A                               
DRO    Turns Off Delta Reference Marker Mode                              
                              N/A       N/A                               
MR1-MR6                                                                   
       Selects Marker 1-6 As Readout Marker                               
                              N/A       N/A                               
MMX    Moves Active Marker to Maximum Trace Value                         
                              N/A       N/A                               
MMN    Moves Active Marker to Minimum Trace Value                         
                              N/A       N/A                               
M1S-M6S                                                                   
       Marker Sweep With Marker 1-6 As Start                              
                              N/A       N/A                               
       Frequency                                                          
M1E-M6E                                                                   
       Marker Sweep With Marker 1-6 As Stop                               
                              N/A       N/A                               
       Frequency                                                          
M1C-M6C                                                                   
       CW Marker Sweep With Marker 1-6 As CW                              
                              N/A       N/A                               
       Frequency                                                          
__________________________________________________________________________
 
    
     
                                           TABLE 4-12                              
__________________________________________________________________________
Limits Command Codes                                                      
COMMAND                                                                   
CODE   DESCRIPTION            VALUES       TERMINATORS                    
__________________________________________________________________________
LUP    Turns On Limit 1 On the Active Channel and                         
                              Depends On Graph Type:                      
                                           Depends On Graph Type:         
       sets It to value, as Shown Below                                   
LLO    Turns On Limit 2 On the Active Channel and                         
                              Depends On Graph Type:                      
                                           Depends On Graph Type:         
       sets It to Value, as Shown Below                                   
LFD    Set Limit Delta On Active Channel for Limit                        
                              Depends On Graph Type:                      
                                           Depends On Graph Type:         
       Frequency Readout, as Shown Below                                  
       Log Mag &amp; Log Polar:   -999.999 to 999.999 dB                      
                                           DBL.XX1.XX3.XM3                
       Phase:                 -180 to 180 degrees                         
                                           DEG (.XX1.XX3 XM3              
                                           for PHA display)               
       Group Delay:           -999.999 to 999.999 s                       
                                           PSC.NSC.USC                    
       Linear Mag &amp; Linear Polar:                                         
                              0 to 999.999 U                              
                                           XX1.XX3.XM3                    
       Real:                  -999.999 to 999.999 U                       
                                           REU.XX1.XX3.XM3                
       Imaginary:             -999.999 to 999.999 U                       
                                           IMU (.XX1.XX3.XM3              
                                           for IMG display)               
       Smith/Inverted Smith:  0 to 1.413 units                            
                                           XX1.XX3.XM3                    
LOF    Disables Limits On Active Channel                                  
                              N/A          N/A                            
LON    Enables Limits On Active Channel                                   
                              N/A          N/A                            
LFR    Selects Limit Frequency Readout for Active                         
                              N/A          N/A                            
       Channel                                                            
LFP    Selects Phase Limit Frequency Readout for Ac-                      
                              N/A          N/A                            
       tive Channel for Log Magnitude/linear                              
       Magnitude and Phase Displays                                       
__________________________________________________________________________
 
    
     
         ______________________________________                                    
10 ! EXAMPLE ON USE OF STRINGS                                            
20 Q$=CHR$ (34) ! QUOTE SYMBOL                                            
30 M$=&#34;4.sub.-- TO.sub.-- 8.sub.-- FILTR&#34; ! MODEL                         
40 I$=&#34;456789&#34; ! I.D.                                                     
50 D$=&#34;8/25/87&#34; ! DATE                                                    
60 O$=&#34;GPIB.sub.-- WHIZ&#34; ! OPERATOR                                       
70 OUTPUT 706 &#34;LMS&#34;&amp;Q$&amp;M$&amp;Q$                                              
80 OUTPUT 706 &#34;LID&#34;&amp;Q$&amp;I$&amp;Q$                                              
90 OUTPUT 706 &#34;LDT&#34;&amp;Q$&amp;D$&amp;Q$                                              
100 OUTPUT 706 &#34;LNM&#34;&amp;Q$&amp;O$&amp;Q$                                             
110 END                                                                   
______________________________________                                    
 
    
     
                                           TABLE 4-13                              
__________________________________________________________________________
Hard Copy Command Codes                                                   
COMMAND                                                                   
CODE   DESCRIPTION             VALUES      TERMINATORS                    
__________________________________________________________________________
PFS    Prints Full Screen      N/A         N/A                            
PGR    Prints Graph Area Only  N/A         N/A                            
PMK    Prints Tabular Marker Data                                         
                               N/A         N/A                            
PTB    Prints Tabular Trace Data                                          
                               N/A         N/A                            
PMT    Prints Tabular Marker and Trace Data                               
                               N/A         N/A                            
PT0-PT9                                                                   
       Selects Tabular Printout Point Density                             
                               N/A         N/A                            
PST    Stops Print/Plot        N/A         N/A                            
FFD    Printer Form Feed       N/A         N/A                            
PLS    Plots Entire Screen     N/A         N/A                            
PLD    Plots Graph Area        N/A         N/A                            
PLT    Plots Data Trace(s)     N/A         N/A                            
PMN    Plots Menu              N/A         N/A                            
PLM    Plots Markers and Limits                                           
                               N/A         N/A                            
PLH    Plots Header            N/A         N/A                            
PGT    Plots Graticule         N/A         N/A                            
PFL    Selects Full Size Plot  N/A         N/A                            
PTL    Selects 1/4 Size Plot, Top Left                                    
                               N/A         N/A                            
PTR    Selects 1/4 Size Plot, Top Right                                   
                               N/A         N/A                            
PBL    Selects 1/4 Size Plot, Bottom Left                                 
                               N/A         N/A                            
PBR    Selects 1/4 Size Plot, Bottom Right                                
                               N/A         N/A                            
DPN    Sets Pen Number for Plotting Data                                  
                               1 to 8      XX1                            
GPN    Sets Pen Number for Plotting Graticule                             
                               1 to 8      XX1                            
MPN    Sets Pen Number for Plotting Markers and Limits                    
                               1 to 8      XX1                            
HPN    Sets Pen Number for Plotting Header                                
                               1 to 8      XX1                            
LMS    Labels Model and Serial Number of Device                           
                               String of Characters Up                    
                                           N/A                            
                               To 12 Characters Long                      
LDT    Labels Date of Test     String of Characters Up                    
                                           N/A                            
                               to 12 Characters Long                      
LID    Labels Device ID        String of Characters Up                    
                                           N/A                            
                               to 12 Characters Long                      
LNM    Labels Operator&#39;s Name  String of Characters Up                    
                                           N/A                            
                               to 12 Characters Long                      
TDD    Stores Tabular Data to Disk in File Specified                      
                               N/A         N/A                            
       (See File Naming Conventions In Paragraph 5-5.5)                   
RTB    Recalls Tabular Data From Disk File Specified to                   
                               N/A         N/A                            
       Printer                                                            
SPD    Sets Plotter Pen Speed Percentage                                  
                               10 to 100   XX1,XX3,XM3                    
__________________________________________________________________________
 
    
     
                                           TABLE 4-14                              
__________________________________________________________________________
Miscellaneous Command Codes                                               
COMMAND                                                                   
CODE   DEFINITION             NOTES                                       
__________________________________________________________________________
RST    Resets 360 to its Default State                                    
                              Similar to pressing the &#34;DEFAULT PROGRAM&#34;   
                              key.                                        
FOF    Implements Frequency Blanking                                      
                              Instructs the 360 to blank any frequency    
                              information from the screen                 
                              and any hard copy output. This code is      
                              useful for security reasons.                
FON    Disables Frequency Blanking                                        
                              Frequency blanking can be turned off using  
                              this code.                                  
BC0    Implements Display Blanking                                        
                              Allows for the ultimate in secutriy -- a    
                              totally blank screen. In this               
                              mode, the 360 is fully operational over the 
                              GPIB but nothing appears                    
                              on the display.                             
BC1    Disables Display Blanking                                          
                              Screen blanking is turned off using the BC1 
                              mnemonic.                                   
BLU    Selects Blue As 3rd Color Plane Color                              
                              Allows selection of the third color used by 
                              the 360 for markers, limits,                
                              and some menu annotation.                   
CYN    Selects Cyan As 3rd Color Plane Color                              
                              Allows selection of the third color used by 
                              the 360 for markers, limits,                
                              and some menu annotation.                   
TST    Runs A Self Test       Instructs tho 360 to perform a self test.   
                              An error in the self test would             
                              be reported in the primary status byte, bit 
                              7 (see paragraph 5-5.7).                    
RTL    Returns to Local Control                                           
                              Performs the same function as the control   
                              panel RETURN TO                             
                              LOCAL key. This code has no effect if the   
                              360 is in local lockout.                    
DFP    Displays Global Operating Parameters                               
                              Displays Global Operating Parameters in the 
                              data area of the screen.                    
DGS    Displays GPIB System Parameters                                    
                              Displays GPIB System Parameters in the data 
                              area of the screen.                         
DCP    Displays Calibration Parameters                                    
                              Displays Calibration Parameters in the data 
                              area of the screen.                         
DC1    Displays Channel 1 and 2 Operating Parameters                      
                              Displays Channels 1 and 2 Operating         
                              Parameters in the data area of              
                              the screen.                                 
DC3    Displays Channel 3 and 4 Operating Parameters                      
                              Displays Channels 1 and 2 Operating         
                              Parameters in the data area of              
                              the screen.                                 
__________________________________________________________________________
 
    
     
                                           TABLE 5-1                               
__________________________________________________________________________
Calibration Command Codes                                                 
COMMAND                                                                   
CODE   DESCRIPTION               VALUES               TERMINATORS         
__________________________________________________________________________
RPC    Repeat Previous Calibration                                        
                                 N/A                  N/A                 
LTC    Specify Coaxial Line Type N/A                  N/A                 
LTU    Specify Microstrip Line Type                                       
                                 N/A                  N/A                 
LTW    Specify Waveguide Line Type                                        
                                 N/A                  N/A                 
SCM    Specify Standard Calibration Method                                
                                 N/A                  N/A                 
OCM    Specify Offset Short Calibration Method                            
                                 N/A                  N/A                 
LCM    Specify LRL Calibration Method                                     
                                 N/A                  N/A                 
C12    Begin 12-term Calibration N/A                  N/A                 
C8T    Begin 8-term (1 Port) Calibration                                  
                                 N/A                  N/A                 
CRF    Begin Reflection Only (1 Port) Calibration                         
                                 N/A                  N/A                 
CFR    Begin Frequency Response Calibration                               
                                 N/A                  N/A                 
CFT    Begin Transmission Only Frequency Response                         
                                 N/A                  N/A                 
       Calibration                                                        
CRL    Begin Reflection Only Frequency Response                           
                                 N/A                  N/A                 
       Calibration                                                        
NOC    Specify Normal 501-point Calibration                               
                                 N/A                  N/A                 
DFC    Specify Discrete Frequency Calibration                             
                                 N/A                  N/A                 
CWC    Specify CW Calibration    N/A                  N/A                 
TDC    Specify Time Domain Harmonic Calibration                           
                                 N/A                  N/A                 
SRT    Enter Start Frequency for Normal or Harmonic                       
                                 Depends on Frequency Range of            
                                                      GHZ,MHZ,KHZ         
       Calibration                                                        
STP    Enter Stop Frequency for Normal or Harmonic                        
                                 Depends on Frequency Range of            
                                                      GHZ,MHZ,KHZ         
       Calibration                                                        
DFQ    Enter Discrete Frequency for Calibration                           
                                 Depends on Frequency Range of            
                                                      GHZ,MHZ,KHZ         
CWF    Enter CW Frequency for Calibration                                 
                                 Depends on Frequency Range of            
                                                      GHZ,MHZ,KHZ         
FRS    Enter Fill Range Start Frequency                                   
                                 Depends on Frequency Range of            
                                                      GHZ,MHZ,KHZ         
FRI    Enter Fill Range Frequency Increment                               
                                 Depends on Frequency Range of            
                                                      GHZ,MHZ,KHZ         
FRP    Enter Fill Rangs Number of Points                                  
                                 1 to (501 -- Current Number of           
                                                      XX1,XX3,XM3         
IFV    Input Frequency List for Calibration                               
                                 List of Frequencies in Current Data      
                                 Format               N/A                 
FIL    Fill Frequency Range Defined by FRS, FRI, and FRP                  
                                 N/A                  N/A                 
DFD    Discrete Frequency Entry Done                                      
                                 N/A                  N/A                 
WCO    Enter Waveguide Cutoff Frequency                                   
                                 0 -- Current Start Freqency              
                                                      GHZ,MHZ,KHZ         
VSW    Enter Microstrip Width    0.001 mm-1.0 m       MMT,CMT,MTR         
SBT    Enter Microstrip Substrate Thickness                               
                                 0.001 mm-1.0 m       MMT,CMT,MTR         
SBD    Enter Microstrip Substrate Relative Dielectric                     
                                 1.0-9999.99          XX1,XX3,XM3         
       Constant                                                           
USE    Enter Microstrip Effective Relative Dielectric                     
                                 1.0-9999.99          XX1,XX3,XM3         
       Constant                                                           
USZ    Enter Microstrip Characteristic Impedance                          
                                 1.0-9999.99          XX1,XX3,XM3         
P1C    Set Up to Specify Port 1 Calibration Standards                     
                                 N/A                  N/A                 
P2C    Set Up to Specify Port 2 Calibration Standards                     
                                 N/A                  N/A                 
CMS    Male SMA Connector for Specified Port                              
                                 N/A                  N/A                 
CFS    Female SMA Connector for Specified Port                            
                                 N/A                  N/A                 
CMK    Male K ™ Connector for Specified Port                           
                                 N/A                  N/A                 
CFK    Female K Connector for Specified Port                              
                                 N/A                  N/A                 
CMN    Male Type N Connector for Specified Port                           
                                 N/A                  N/A                 
CFN    Female Type N Connector for Specified Port                         
                                 N/A                  N/A                 
CM3    Male GPC-3.5 Connector for Specified Port                          
                                 N/A                  N/A                 
CF3    Female GPC-3.5 Connector for Specified Port                        
                                 N/A                  N/A                 
CNG    GPC-7 Connector for Specified Port                                 
                                 N/A                  N/A                 
CND    Other Connector Specification                                      
                                 N/A                  N/A                 
COO    Connector Offset for Open Device                                   
                                 -999.999 m to 999.999 m                  
                                                      MMT,CMT,MTR         
COS    Connector Offset for Short Device                                  
                                 -999.999 m to 999.999 m                  
                                                      MMT,CMT,MTR         
CC0    Capacitance Coefficient for Open Device,                           
                                 -999.999 to 999.999  XX1                 
       Implied X 10 E-15                                                  
CC1    Capacitance Coefficient for Open Device,                           
                                 -999.999 to 999.999  XX1                 
       Implied X 10 E-27                                                  
CC2    Capacitance Coefficient for Open Device,                           
                                 -999.999 to 999.999  XX1                 
       Implied X 10 E-36                                                  
CC3    Capacitance Coefficient for Open Device,                           
                                 -999.999 to 999.999  XX1                 
       Implied X 10 E-45                                                  
BBL    Specify Broadband Load for Calibration                             
                                 N/A                  N/A                 
SLD    Specify Sliding Load for Calibration                               
                                 N/A                  N/A                 
BEG    Begin Calibration Data Collection Steps                            
                                 N/A                  N/A                 
TCD    Take Calibration Data for Current Standard                         
                                 N/A                  N/A                 
NCS    Go on to Next Calibration Step                                     
                                 N/A                  N/A                 
KEC    Keep Existing Calibration N/A                  N/A                 
COF    Turn Any Vector Error Correction Off                               
                                 N/A                  N/A                 
CON    Turn Vector Error Correcion On if Available                        
                                 N/A                  N/A                 
A12    Simulate 12-term Calibration                                       
                                 N/A                  N/A                 
A8T    Simulate 8-term (1 Port) Calibration                               
                                 N/A                  N/A                 
ARF    Simulate Reflection Only (1 Port) Calibration                      
                                 N/A                  N/A                 
AFR    Simulate Frequency Response Calibration                            
                                 N/A                  N/A                 
AFT    Simulate Transmission Only Frequency Response                      
                                 N/A                  N/A                 
       Calibration                                                        
ARL    Simulate Reflection Only Frequency Response                        
                                 N/A                  N/A                 
       Calibration                                                        
__________________________________________________________________________
 
    
     
                                           TABLE 5-2                               
__________________________________________________________________________
Calibration Code Ordering                                                 
                  Command Code       Required (R)                         
Order                                                                     
    Item          Examples           or Optional (O)                      
__________________________________________________________________________
1   Line Type     LTC, LTW, LTV      O                                    
2   Calibration Method                                                    
                  SCM, OCM, LCM      O                                    
3   Calibration Type                                                      
                  C12, C8T, CRF, CFR, CFT, CRL                            
                                     R                                    
4   Data Points   NOC, DFC, TDC, CWC O                                    
5   Frequency                                                             
    Range         SRT, STP           O                                    
    Discrete      DFQ, IFV, FRS, FRI, FRP, FIL                            
                                     R                                    
    CW            CWF                O                                    
6   Connector Type/Offset                                                 
                  P1C, P2C, CMS, CFS, CMK, CFK,                           
                                     O                                    
    Short Values  CMN, CFN, CM3, CF3, CNG, CND,                           
                  COO, COS, CC0, CC1, CC2, CC3,                           
                  SH1, SH2                                                
7   Load Type     SLD, BBL           O                                    
8   Begin Data Collection Steps                                           
                  BEG                R                                    
__________________________________________________________________________
 
    
     
         ______________________________________                                    
10   ! 1ST STEP - BROADBAND LOADS ON                                      
     BOTH PORTS                                                           
20   DISP &#34;CONNECT LOADS TO BOTH PORTS&#34;                                   
30   DISP &#34;HIT END LINE WHEN READY&#34;                                       
40   INPUT NS                                                             
50   ! NS JUST A &#34;DUMMY&#34;[- WAIT FOR USER                                  
60   ! TAKE LOAD MEASUREMENT USING                                        
     100 AVERAGES                                                         
70   OUTPUT 706; &#34;AVG 100 XX1 TCD NCS&#34;                                    
80   ! 2ND STEP - PORT 1: OPEN, PORT 2: SHORT                             
90   DISP &#34;CONNECT OPEN TO PORT 1&#34;                                        
100  DISP &#34;CONNECT SHORT TO PORT 2&#34;                                       
110  DISP &#34;HIT END LINE WHEN READY&#34;                                       
120  INPUT NS                                                             
130  ! TAKE OPEN/SHORT MEASUREMENT WITH                                   
     20 AVERAGES                                                          
140  OUTPUT 706; &#34;AVG 20 XX1 TCD NCS&#34;                                     
150  ! 3RD STEP - PORT 1: SHORT, PORT 2: OPEN                             
160  DISP &#34;CONNECT SHORT TO PORT 1&#34;                                       
170  DISP &#34;CONNECT OPEN TO PORT 2&#34;                                        
180  DISP &#34;HIT END LINE WHEN READY&#34;                                       
190  INPUT NS                                                             
200  OUTPUT 706; &#34;TCD NCS&#34;                                                
210  ! 4TH STEP - THROUGH LINE                                            
220  DISP &#34;CONNECT THROUGH LINE&#34;                                          
230  DISP &#34;BETWEEN PORTS&#34;                                                 
240  DISP &#34;HIT END LINE WHEN READY&#34;                                       
250  INPUT NS                                                             
260  OUTPUT 706; &#34;TCD NCS&#34;                                                
270  CALIBRATION COMPLETE - SAVE TO DISK                                  
______________________________________                                    
 
    
     
                                           TABLE 5-3                               
__________________________________________________________________________
Advanced Command Codes: Save/Recall                                       
COMMAND                                                                   
CODE   DESCRIPT1ON           VALUES     TERMINATORS                       
__________________________________________________________________________
STO    Store calibration data and front panel setup to                    
                             String for file name up                      
                                        N/A                               
       disk file.            to 8 characters long                         
RLD    Recall calibration data and front panel setup                      
                             String for file name up                      
                                        N/A                               
       from disk file        to 8 characters long                         
SV1-SV4                                                                   
       Save front panel setup to internal memory                          
                             N/A        N/A                               
       location 1-4                                                       
RC1-RC4                                                                   
       Recall front panel setup data from internal                        
                             N/A        N/A                               
       memory location 1-4                                                
__________________________________________________________________________
 
    
     
                                           TABLE 5-4                               
__________________________________________________________________________
Data Transfer Command Codes                                               
COMMAND                                                                   
CODE   DESCRIPTION                       DATA FORMATS                     
__________________________________________________________________________
OM1-OM6                                                                   
       Output Marker 1-6 Value           ASCII                            
OAP    Output Active Parameter Value     ASCII                            
OKP    Output Front Panel Key Pressed    ASCII                            
OID    Output Identify String            40 byte ASCII String             
ONP    Output Number of Points           ASCII                            
FMA    Select ASCII Data Format          N/A                              
FMB    Select 64-Bit IEEE 754 Floating Point Data Format                  
                                         N/A                              
FMC    Select 32-Bit IEEE 754 Floating Point Data Format                  
                                         N/A                              
LSB    Select Least Significant Byte First Data Transfer                  
                                         N/Ae                             
MSB    Select Most Significant Byte First Data Transfer                   
                                         N/Ae                             
OC1-OC9                                                                   
       Output Calibration Coefficient 1-9                                 
                                         FMA, FMB, FMC                    
OCA-OCC                                                                   
       Output Calibration Coefficient A, B or C                           
                                         FMA, FMB, FMC                    
OCL    Output All 12-Term Calibration Coefficients                        
                                         Binary String                    
IC1-IC9                                                                   
       Input Calibration Coefficient 1-9 FMA, FMB, FMC                    
ICA-ICC                                                                   
       lnput Calibration Coefficient A, B, or C                           
                                         FMA, FMB, FMC                    
ICL    Input All 12-term Calibration Coefficients                         
                                         Binary String                    
OFV    Output Frequency Values           FMA, FMB, FMC                    
IFV    Input List of Frequencies         FMA, FMB, FMC                    
OS1-OS4                                                                   
       Output Stored Setup 1-4           Binary String                    
OFP    Output Current Front Panel Setup  Binary String                    
IS1-IS4                                                                   
       Input Stored Setup 1-4            Binary String                    
IFP    Input Front Panel Setup           Binary String                    
ODR    Output Disk Directory             Binary String                    
ORD    Output Raw (Uncorrected) Data For S-Parameter on Acive             
                                         FMA, FMB, FMC                    
OCD    Output Corrected Data for S-Parameter on Active Channel            
                                         FMA, FMB, FMC                    
OFD    Output Final (display Format) Data For S-parameter on Active       
       Channel                           FMA, FMB, FMC                    
ICD    Input Corrected Data For S-Parameter on Active Channel             
                                         FMA, FMB, FMC                    
IFD    lnput Final (Display Format) Data For S-Parameter on Active        
       Channel                           FMA, FMB, FMC                    
__________________________________________________________________________
 
    
     
                       TABLE 5-5                                                   
______________________________________                                    
OutPut Values Versus Various Graph Types                                  
Display Type    Output Values                                             
______________________________________                                    
Log Magnitude   dB. Degrees                                               
Phase           dB. Degrees                                               
Log Mag &amp; Phase dB. Degrees                                               
Linear Magnitude                                                          
                Lin Mag(Rho or Tau), Degrees                              
Linear Mag &amp; Phase                                                        
                Lin Mag(Rho or Tau), Degrees                              
Smith Chart     Ohms, Ohms (r + jx)                                       
Inverted Smith  Siemens, Siemens (g + jb)                                 
Group Delay     Seconds, Degrees                                          
Log Polar       dB, Degrees                                               
Linear Polar    Lin Mag(Rho or Tau), Degrees                              
Real            Real, imag                                                
Imaginary       Real, imag                                                
Real &amp; Imaginary                                                          
                Real, imag                                                
SWR             SWR, Degrees                                              
______________________________________                                    
 
    
     
         ______________________________________                                    
Number of Bytes                                                           
4      9          9        6      6     6                                 
______________________________________                                    
xxxx   xx.xxxxxx  xx.xxxxxx                                               
                           Sxxx.x Sxxx.x                                  
                                        xxx.xx                            
Model #                                                                   
       Low        High     Low    High  S.W.                              
       Freq.      Freq.    Pwr    Pwr   Rev.                              
       GHZ        GHZ      dBM    dBM                                     
______________________________________                                    
 
    
     
                                           TABLE 5-6                               
__________________________________________________________________________
Calibration Coefficient Ordering                                          
Calibration Type                                                          
               Reflection                                                 
                     Frequency                                            
                           Transmission                                   
                                    Reflection                            
Coefficient                                                               
      12-term                                                             
           8-term                                                         
               Only  Response                                             
                           Freq. Response                                 
                                    Freq. Response                        
#     C12  C8T CRF   CFR   CFT      CRL      None                         
__________________________________________________________________________
1     EDF  EDF EDF   ERF   ETF      ERF      --                           
2     ESF  ESF ESF   ETF   --       --       --                           
3     ERF  ERF ERF   --    --       --       --                           
4     EXF  ETF --    --    --       --       --                           
5     ELF  --  --    --    --       --       --                           
6     ETF  --  --    --    --       --       --                           
7     EDR  --  --    --    --       --       --                           
8     ESR  --  --    --    --       --       --                           
9     ERR  --  --    --    --       --       --                           
A     EXR  --  --    --    --       --       --                           
B     ELR  --  --    --    --       --       --                           
C     ETR  --  --    --    --       --       --                           
__________________________________________________________________________
 
    
     
                                           TABLE 5-8                               
__________________________________________________________________________
Disk Functions Command Codes                                              
COMMAND                                                                   
CODE   DESCRIPTION               VALUES                                   
__________________________________________________________________________
SDK    Store Active Channel&#39;s Trace Memory to Disk File                   
                                 String Up to 8 Characters Long for File  
                                 Name                                     
RCK    Recall Active Channel&#39;s Trace Memory From                          
                                 String Up to 8 Characters Long for File  
                                 Name                                     
       Disk File                                                          
STO    Store Calibration Data and Front Panel Setup                       
                                 String Up to 8 Characters Long for File  
                                 Name                                     
       Information to Disk File                                           
RLD    Recall Calibration Data and Front Panel Setup                      
                                 String Up to 8 Characters Long for File  
                                 Name                                     
       Information From Disk File                                         
TDD    Store Tabular Printout Data to ASCII Disk File                     
                                 String Up to 8 Characters Long for File  
                                 Name                                     
RTB    Recall Tabular Data File From Disk to Output to                    
                                 String Up to 8 Characters Long for File  
                                 Name                                     
       Printer                                                            
DEC    Delete Calibration and Front Panel Setup File                      
                                 String Up to 8 Characters Long for File  
                                 Name                                     
       From Disk                                                          
DED    Delete Tabular Printout Data File From Disk                        
                                 String Up to 8 Characters Long for File  
                                 Name                                     
DEN    Delete Trace Memory File From Disk                                 
                                 String Up to 8 Characters Long for File  
                                 Name                                     
INT    lnitialize (Format) Disk in Drive as a Data-Only                   
                                 N/A                                      
       Disk                                                               
LKT    Load Calibration Kit Information From Disk                         
                                 N/A                                      
__________________________________________________________________________
 
    
     
         ______________________________________                                    
! EXAMPLE 1 - SAVE CAL AND FRONT                                          
! PANEL SETUP TO DISK                                                     
Q$ = CHR$(34)  ! DOUBLE QUOTE SYMBOL(&#34;)                                   
C$ = &#34;12.sub.-- TERM&#34;                                                     
               ! FILE NAME FOR CAL DATA                                   
! STORE TO DISK FILE &#34;12.sub.-- TERM. CAL&#34;                                
OUTPUT 706; &#34;STO&#34; &amp;Q$&amp;C$&amp;Q$                                               
! EXAMPLE 2 - SAVE TABULAR DATA                                           
! TO DISK FILE                                                            
Q$ = CHR$(34)  ! DOUBLE QUOTE SYMBOL(&#34;)                                   
T$ = &#34;S21.sub.-- THRU&#34;                                                    
               ! FILE NAME FOR TAB DATA                                   
! STORE TO DISK FILE &#34;S21.sub.-- THRU. DAT&#34;                               
OUTPUT 706; &#34;TDD&#34;&amp;Q$&amp;T$&amp;Q$                                                
! EXAMPLE 3 - SAVE TRACE MEMORY                                           
! TO DISK, RECALL IT ON A DIFFERENT                                       
! CHANNEL AND THEN DELETE FILE                                            
OUTPUT 706; &#34;CH1 D13 S11 CH3 S21 FHI WFS&#34;                                 
OUTPUT 706; &#34;CH1 STD&#34;                                                     
               ! STORE TRACE TO MEMORY                                    
Q$ = CHR(34)   ! DOUBLE QUOTE SYMBOL(&#34;)                                   
N$ = &#34;S11TRACE&#34;                                                           
               ! FILE NAME FOR TRACE DATA                                 
!  STORE TO DISK FILE &#34;S11TRACE.NRM&#34;                                      
OUTPUT 706; &#34;SDK&#34;&amp;Q$&amp;N$&amp;Q$                                                
! RECALL SAME DATA ON CHANNEL 3                                           
OUTPUT 706; &#34;CH3 RCK&#34; &amp;Q$&amp;N$&amp;Q$                                           
! DELETE THE TRACE MEMORY FILE                                            
OUTPUT 706; &#34;DKN&#34; &amp;Q$&amp;N$&amp;Q$                                               
______________________________________                                    
 
    
     
         ______________________________________                                    
Primary Status Byte:                                                      
Bit #                                                                     
7    6      5       4     3     2     1      0                            
______________________________________                                    
Self SRQ    2nd     Action                                                
                          Out   Syntax                                    
                                      Sweep  Cal                          
Test        Byte    Not   of    Error Complete                            
                                             Sweep                        
Fail        Has     Pos-  Range       in Hold                             
                                             Com-                         
            Status  sible                    plete                        
______________________________________                                    
 
    
     
         ______________________________________                                    
Secondary Status Byte:                                                    
Bit #                                                                     
7     6         5      4    3    2      1    0                            
______________________________________                                    
Power Key       X*     X    X    Hard-  X    Disk                         
On    Pressed                    ware        Error                        
                                 Error                                    
______________________________________                                    
 *X denotes not used.                                                     
 
    
     
                                           TABLE 5-10                              
__________________________________________________________________________
Time Domain Command Codes                                                 
COMMAND                                                                   
CODE   DESCRIPTION             VALUES      TERMINATORS                    
__________________________________________________________________________
FQD    Select Frequency Domain for Active Channel                         
                               N/A         N/A                            
TBP    Select Bandpass Mode With Time Readouts for                        
                               N/A         N/A                            
       active Channel                                                     
DBP    Select Bandpass Mode With Distance Readouts                        
                               N/A         N/A                            
       for Active Channel                                                 
TPI    Select Phasor Impulse Mode With Time                               
                               N/A         N/A                            
       Readouts for Active Channel                                        
DPI    Select Phasor lmpulse Mode With Distance                           
                               N/A         N/A                            
       Readouts for Active Channel                                        
TLP    Select Lowpass Mode With Time Readouts for                         
                               N/A         N/A                            
       Active Channel                                                     
DLP    Select Lowpass Mode With Distance Readouts                         
                               N/A         N/A                            
       for Active Channel                                                 
FGT    Select Frequency With Time Gate Mode                               
                               N/A         N/A                            
LPI    Select Lowpass Impulse Response for Active                         
                               N/A         N/A                            
       Channel                                                            
LPS    Select Lowpass Step Response for Active                            
                               N/A         N/A                            
       Channel                                                            
GCN    Turn Gate On on Active Channel                                     
                               N/A         N/A                            
GCF    Turn Gate Off on Active Channel                                    
                               N/A         N/A                            
GDS    Display Gate Symbols on Active Channel With                        
                               N/A         N/A                            
       Gate Off                                                           
ZST    Set Start of Time Domain Zoom Range for All                        
       Channels in Time Domain                                            
       Time Mode:              -999.999 to 999.999 us                     
                                           PSC, NSC, USC                  
       Distance Mode:          -999.999 to 999.999 m                      
                                           MMT, CMT, MTR                  
ZSP    Set Stop of Time Domain Zoom Range for All                         
       Channels in Time Domain                                            
       Time Mode:              -999.999 to 999.999 us                     
                                           PSC, NSC, USC                  
       Distance Mode:          -999.999 to 999.999 m                      
                                           MMT, CMT, MTR                  
ZCT    Set Center of Time Domain Zoom Range for All                       
       Channels in Time Domain                                            
       Time Mode:              -999.999 to 999.999 us                     
                                           PSC, NSC, USC                  
       Distance Mode:          -999.999 to 999.999 m                      
                                           MMT, CMT, MTR                  
ZSN    Set Span of Time Domain Zoom Range for All                         
       Channels in Time Domain                                            
       Time Mode:              0 to 999.999 us                            
                                           PSC, NSC, USC                  
       Distance Mode:          0 to 999.999 m                             
                                           MMT, CMT, MTR                  
GST    Set Gate Start Value for All Channels in Time                      
       Domain                                                             
       Time Mode:              -999.999 to 999.999 us                     
                                           PSC, NSC, USC                  
       Distance Mode:          -999.999 to 999.999 m                      
                                           MMT, CMT, MTR                  
GSP    Set Gate Stop Value for all Channels in Time                       
       Domain                                                             
       Time Mode:              -999.999 to 999.999 us                     
                                           PSC, NSC, USC                  
       Distance Mode:          -999.999 to 999.999 m                      
                                           MMT, CMT, MTR                  
GCT    Set Gate Center Value for All Channels in Time                     
       Domain                                                             
       Time Mode:              0.0000 to 999.999 us                       
                                           PSC, NSC, USC                  
       Distance Mode:          0.0000 to 999.999 m                        
                                           MMT, CMT, MTR                  
GSN    Set Gate Span Value for all Channels in Time                       
       Domain                                                             
       Time Mode:              0.0000 to 999.999 us                       
                                           PSC, NSC, USC                  
       Distance Mode:          0.0000 to 999.999 m                        
                                           MMT, CMT, MTR                  
GRT    Select Rectangular Gate Shape for all Time                         
                               N/A         N/A                            
       Domain Channels                                                    
GNM    Select Nominal Gate Shape for all Time Domain                      
                               N/A         N/A                            
       Channels                                                           
GLS    Select Low Sidelobe Gate Shape for all Time                        
                               N/A         N/A                            
       Domain Channels                                                    
GMS    Select Minimum Sidelobe Gate Shape for all                         
                               N/A         N/A                            
       Time Domain Channels                                               
WRT    Select Rectangular Window Shape for all Time                       
                               N/A         N/A                            
       Domain Channels                                                    
WNM    Select Nominal Window Shape for all Time                           
                               N/A         N/A                            
       Domain Channels                                                    
WLS    Select Low Sidelobe Window Shape for all Time                      
                               N/A         N/A                            
       Domain Channels                                                    
WMS    Select Minimum Sidelobe Window Shape for all                       
                               N/A         N/A                            
       Time Domain Channels                                               
MRR    Restore Original Range After a Marker Zoom                         
                               N/A         N/A                            
       Operation                                                          
DCA    Select Auto d.c. term for Low Pass                                 
                               N/A         N/A                            
DCZ    Select Line Impedance d.c. term for Low Pass                       
                               N/A         N/A                            
DCO    Select Open d.c. Term for Low Pass                                 
                               N/A         N/A                            
DCS    Select Short d.c. Term for Low Pass                                
                               N/A         N/A                            
DCV*   Sect Low Pass d.c Term to Value                                    
                               -100M Ω to 1000M Ω             
                                           XX1, XX3, XM3                  
__________________________________________________________________________