fig1 is a perspective view of the surgical scalpel incinerator , showing a cross section of the container 1 , with a box at the front 2 in which scalpel scrap is deposited , and which is used to remove said scrap when it is full . the scrap is formed by the electromechanism that makes electrical contact with the scalpel in order to incinerate it . the electromechanism consists of a support 3 ( fig1 a , and 4 b ) and its component elements , including a ceramic plate 4 ( fig1 , 4 a , 4 b , 5 b , 5 c , 5 d , 5 e , 5 f , and 5 g ) on which four fixed metal contact plates 5 are mounted ( fig1 , 5 d , and 5 g ), which in turn make electrical contact with the scalpel ; the ceramic plate is mounted on four struts 11 fig1 , 5 b , 5 c , 5 e , and 5 g ) , which in turn support a rotating door 7 ( fig1 , 5 a , 5 b , and 5 c ) that turns on a pivot 21 ( fig1 , 5 a , 5 b , 5 c , 5 e , and 5 e ). when the door 7 opens , a scalpel can be placed inside the unit ; once inside , the scalpel is laid horizontally on a bed formed by the four fixed metal contact plates 5 and five supports 8 mounted on the rotating door 7 ( fig1 , 5 a , 5 b , and 5 c ). these supports 8 recede as the door 7 closes , leaving the scalpel supported only by the four fixed metal contact plates 5 . to ensure that the contact plates 5 make effective electrical contact with the scalpel , when the door 7 is fully closed the three metal pressure plates 6 mounted on the rotating door 7 push the scalpel against the four fixed metal contact plates 5 ( see fig1 , 5 a , 5 b , 5 c , and 6 a ). opening and closing the rotating door 7 activates several functions : when the door 7 closes , in addition to pushing the scalpel as mentioned above , it also activates the microswitch 9 ( fig1 , and 3 ) by moving its lever 23 ( fig2 and 3 ), so that it closes the power supply circuit to the transformer 10 ( fig1 , and 3 ), which in turn supplies dc power to the four fixed metal contact plates 5 , or so that they receive power from an electronic circuit 15 ( fig6 b ), which in turn is powered by the transformer 10 . the components in the electronic circuit are ac rectifying diodes mounted on a circuit card 13 ( fig1 , and 3 ). when the rotating door 7 opens , even when the unit &# 39 ; s power switch 12 ( fig1 , and 3 ) indicates ( by lighting up ) that the equipment is receiving power and is ready for operation , the microswitch 9 is inoperative , as it is open and no power is being supplied to the transformer 10 , so that the user can place the scalpel inside the unit without receiving an electric shock . with the door 7 open ; the user can also insert a metal object to remove any pieces of scalpel that may have stuck to the fixed metal contact plates 5 during the incineration process . fig2 and 3 are side and overhead views of the unit , showing the internal and external layout of the elements in the cabinet 1 . inside the unit , we can see the electromechanism support 7 , the electromechanism , consisting of a ceramic plate 4 , fixed metal contact plates 5 , support struts 11 for the rotating door 7 , the pivot 21 for the rotating door 7 , the rotating door 7 , metal pressure plates 6 mounted on the rotating door 7 , the support bed 8 fastened to the rotating door , a chute 19 to guide scrap into the scrap drawer 2 , the microswitch 9 , the operation of which is described in fig1 the microswitch lever 23 operated by the opening and closing of the door 7 , the microswitch support 18 , the transformer 10 , which supplies electrical current to the fixed metal contact plates 5 , or through the electronic circuit formed by rectifying diodes 15 mounted on an electronic circuit card 13 . the following components are mounted on the outside of the cabinet 1 : fuse holder 14 for the fuse that protects the transformer or electronic circuit , the unit &# 39 ; s power switch 12 , with an internal light that indicates when the equipment is ready to operate , an outlet for the power cable 17 , a chassis 16 that supports the electromechanism , the electronic card 13 , the transformer 10 , the cabinet 1 and the scrap receptacle 2 . fig4 a and 4b are front and side views of the electromechanism support 3 , the ceramic plate 4 , the struts 11 of the rotating door 7 , the pivot 21 the door turns on , and a ramp or chute 19 to guide scrap into the receptacle 2 . fig5 a - 5g are front , overhead and side views of the electromechanism , consisting of : a ceramic plate 4 , fixed metal contact plates 5 , mounted on the ceramic plate 4 , the support struts 11 of the door 7 , mounted on the ceramic plate 4 , the pivot 21 on which the door 7 turns , metal pressure plates 6 , and the support bed 8 , mounted on the rotating door 7 . fig6 a - 6c are wiring diagrams of the device , with connections for all the different elements , including : power supply cable 22 , fuse holder 14 , power switch 12 , microswitch 9 , microswitch lever 23 , transformer 10 , rotating door 7 , metal pressure plates 6 , and support bed 8 . points a and b show the two options for connecting the transformer 10 , either directly to the fixed metal contact plates 4 or to the electronic circuit 15 and from there to the fixed metal contact plates 5 mounted on the ceramic plate 4 . power switch 12 : 117 vac , at 50 amps ( with internal light ). fuse holder 14 ( american type ) with 15 , 30 or 45 amp internal fuse . transformer 10 : 117 - 250 vac , with output of 40 , 50 , 75 , or 100 amps . ac - dc rectifying diodes 15 at 6 , 12 ; 25 , 50 or 75 amps . the cabinet 1 is in two parts : the upper housing ( referred to herein as the cabinet 1 ), and the chassis 16 , with the following components mounted on the chassis 16 : electronic circuit card support 13 , electromechanism support 3 , and transformer 10 . the following components are mounted on the upper housing ( on the cabinet 1 ): fuse holder 14 , power switch 12 , and power cable with external plug ( not shown in figures ), all arranged as shown in fig1 and 3 . the assembly of the chassis 16 with the upper housing , and their respective elements and components , constitutes the surgical scalpel incinerator . the upper housing ( the cabinet 1 ) and chassis may be made from metal , plastic , or a combination of the two . the surgical scalpel incinerator uses an electromechnism , which , as shown in the diagrams , is a combination of mechanical elements whose function is to make contact with the scalpel in order to pass a specified electrical current through it , creating a short circuit that raises the temperature of the scalpel to over 1000 degrees centigrade in under 4 seconds , producing scrap resulting from the deformation or breakage of the scalpel , which in turn is rendered completely sterile . in order for the electromechanism to perform this function , it is supplied with electricity from the electronic circuit , which in turn is powered by the transformer , or connected directly to the transformer 10 . the receptacle 2 , chassis 16 , electrical components , etc . are complementary elements that give the unit its form and enable it to operate . i have proposed using mass production procedures to manufacture the surgical scalpel incinerator , defining which elements will be manufactured and which will be obtained from third parties for subsequent mounting and assembly , leading in turn to the elaboration of : documents for the following procedures : manufacture , assembly and mounting of elements , in addition to safety , training , environmental protection , etc . quality control catalogues for : components , raw materials , materials supplied by providers , packing materials , advertising , etc . logs for production , quality control , equipment and tool maintenance , cleaning , etc . chassis 16 : made with the necessary bends and perforations to mount other elements as shown in the diagrams . cabinet : manufactured in three parts with the necessary bends and perforations to chassis , and other elements , to ensure that all perforations are the same size and are made in the same place . bends in the cabinet will be made by heating if it is plastic , and by mechanical benders if it is metal . support and electromechanism : all parts for these elements will be made using templates with the necessary perforations , after which all parts will be assembled . transformer and components , including : diodes , switches , power cable with plug , fuse holder , fuse , internal wiring , hardware , etc . will be acquired from third parties under predefined specifications . electronic circuit card : will be manufactured with the necessary perforations to mount electronic components . manufacture and assembly of the device &# 39 ; s elements and components includes the following steps : g .— chassis - cabinet connection , sealing of the unit , and second round of operating tests . a variety of scalpels are used in medical practice , which standard nom - 087 - ecol - 1995 defines as infectious biological waste . this device has been invented to incinerate all types of scalpels , of any length or shape , provided they can be removed from their holder , rendering the resulting scrap completely sterile . we anticipate that the most frequent use of this device will be in hospital operating theaters . the surgical scalpel incinerator is intended to assist and facilitate mandatory sanitary practices indicated in the referenced standard with regard to incineration and sterilization of waste . due to its small size and simple operation , the device can be operated by any person ( not necessarily technicians or medical specialists ); these benefits , added to its low cost and portability , make it an excellent option for both large and small medical facilities , even those in remote locations , provided they have electricity . this invention will make a great contribution to improving public health care .
a surgical scalpel incinerator an electromechanism for thermally destroying and sterilizing scalpels : a cabinet having a door through which a scalpel blade is inserted and supported by supports above several metal plates . closing the door withdraws the supports , after which the blade contacts the plates . closing the door closes a switch , connecting the plates to a power supply , electrifying the blade . pressure plates on the door urge the blade against the plates . sterilized scrap of the blade falls between the plates and into a receptacle in a drawer beneath the plates .