a clinical study was performed to assess the safety of transcomeal electrical stimulation ( tes ) and explore its efficacy in various subjective and objective parameters of visual function in patients with retinitis pigmentosa ( rp ); see schatz et al ., loc . cit . twenty - four patients in this prospective , randomized , partially - blinded , good - clinical - practice conform study received tes ( 5 ms biphasic pulses ; 20 hz ; dtl ( dawson - trick - litzkow ) electrodes ) 30 minutes per week for 6 consecutive weeks . patients were randomly assigned to one of three groups : sham , 66 %, or 150 % of individual electrical phosphene threshold ( ept ). visual acuity ( va ), visual field ( vf ; kinetic , static ), electroretinography ( ganzfeld , multifocal ), dark - adaptation , color discrimination , and epts were assessed at all visits or 4 times according to the study plan . after determination of the individual phosphene threshold in each case , the patients were randomized and divided into the 3 groups . the first ( sham ) group was treated without electric current , the second with 66 % of the individual phosphene threshold ( the base line stimulation current ), the third with 150 %. tes using dtl ( dawson - trick - lizkow ) electrodes was tolerated well ; all patients concluded the study . two adverse ( foreign body sensation ) but no serious adverse events were encountered . there was a tendency for most functional parameters to improve ( 8 / 18 ) or remain constant ( 8 / 18 ) in the 150 % group ; vf area and scotopic b - wave - amplitude reached statistical significance ( p & lt ; 0 . 027 and & lt ; 0 . 001 , respectively ). only desaturated colour discrimination and vf mean sensitivity decreased ( the latter due to the enlargement of vf area ). there was no obvious trend in the 66 % group . twenty - four eyes of said 24 patients were included ; all suffered from advanced rp ( rod - cone - dystrophy ). diagnosis was established by detailed history , funduscopy , electroretinography ( erg ), and visual field ( vf ) examination . inclusion criteria were age & gt ; 18 years , visual acuity ( va ) 0 . 02 - 0 . 9 ( snellen ) and recordable erg and vf results . exclusion criteria were other ocular diseases ( advanced diabetic retinopathy , retinal or choroidal neovascularization , exudative age - related macular degeneration ), silicon oil tamponade , and severe general disease . patients were recruited from the hospital &# 39 ; s special retinal degeneration clinic . after inclusion they were randomized to one of the three treatment arms : sham stimulation , stimulation with 66 % of their individual electrical phosphene threshold ( ept ) at 20 hz , or with 150 % of their individual ept . mean age was 55 . 63 ± 14 . 79 (± sd ), 51 . 63 ± 8 . 02 , and 56 . 38 ± 13 . 39 in the sham , 66 %, and 150 % group , respectively ( resp . ); there was no statistically significant difference between groups ( p = 0 . 72 ; anova ). patients and technicians who performed erg , multifocal erg ( mferg ), vf , darkadaptation ( da ) and color discrimination tests were blinded for their treatment group for the entire study period . the examining doctor who performed va , ept , intra - ocular pressure ( iop ), and slit - lamp examination was not blinded because he / she was responsible for setting of stimulation parameters . the study was approved by the local ethics committee and local agencies ; and was registered at clinicaltrials . gov ( nct00804102 ). all patients gave written informed consent after explanation of the nature and possible consequences of the study ; all procedures adhered to the declaration of helsinki . patients were seen at 9 visits : for one baseline visit ( visit 1 ), followed by 6 consecutive weekly visits ( visits 2 - 7 ) with tes ( dates varied max . ± 2 days ), and for 2 follow - up visits ( visit 8 and 9 ), one 2 weeks and one 11 weeks after the last stimulation visit ( the last dates varied max . ± 10 days ). visits 2 , 3 , 4 , 6 , 7 were ‘ short ’ visits which included : va , slit lamp biomicroscopy , and tes . visits 1 , 5 , 8 , 9 were ‘ long ’ visits and additionally included : vf , color test , dark - adaptation ( da ), erg and mferg , optical coherence tomography ( oct ). to adjust for learning effects in vf examinations this test was performed at visit 1 and visit 2 . for electrical stimulation a sterile single - use dtl electrode ( diagnosys uk ltd , cambridge , uk ) was used as active electrode as described in the original publication of dawson et al ., “ improved electrode for electroretinography ”, in invest ophthalmol vissci , 1979 ; 18 : 988 - 991 . on the patient &# 39 ; s request a local anesthetic was applied to the lower fornix ( e . g . oxybuprocaine hydrochloride ). a gold cup electrode ( lkc technologies , inc ., gaithersburg , md ., usa ) as counter electrode was attached to the ipsilateral temple after thorough cleaning of the skin and application of contact paste . for stimulation a commercially available neurostimulator was adapted by the manufacturer to deliver current pulses within the range of the study ( twister ®, dr . langer medical , waldkirch , germany ). rectangular biphasic current pulses ( 5 ms positive directly followed by 5 ms negative ) were applied for 30 minutes at a frequency of 20 hz . room light was switched on during the entire session . for sham stimulation all electrodes and cables were attached and epts were determined , but stimulation was not turned on . to assess epts an alternative forced choice method was used as described by gekeler et al ., loc . cit . the threshold was determined three times and the average was formed at each study visit for calculation of individual stimulation strength according to the treatment arm . ganzfeld ergs were registered according to the iscev standard ( international society for clinical electrophysiology of vision ) using a colordome ® controlled by an espion e 2 ® ( diagnosys llc , cambridge , uk ). after 30 minutes of dark adaptation and application of two drops of tropicamide 0 . 5 % ( mydriatikum stulln ®, stulln , germany ) self constructed dtl - electrodes and gold - cup electrodes ( viasys healthcare , warwick , uk ) were positioned on the forehead and the temples . the erg protocol consisted of 4 steps with intensities from 0 . 1 to 3 phot cd * s / m 2 and 4 ms duration ( white 6500 k ). two single flash responses were used as rod protocol and a single - flash response ( 3 phot cd * s / m 2 with a background illumination of 30 phot cd / m 2 = standard flash , sf ) and a 30 hz flicker were chosen as cone protocol . impedance level was & lt ; 10 kω . a band - pass filter was applied from 0 . 3 to 300 hz using the machine &# 39 ; s built - in software algorithm . erg potentials & lt ; 5 μv were excluded from the analysis . multifocal ergs were recorded with a veris ™ ( electro - diagnostic imaging inc ., redwood city , usa ; software version veris science ™ 4 . 9 . 1 ) system using a 21 inch screen ( resolution 1024 × 768 ; “ uhr21l ”, nortech imaging technologies , plymouth , usa ) positioned 32 cm in front of the subject . the visual field stimulated subtended an area of 60 °× 55 °; 61 hexagons were presented with alternating black ( 5 cd / m 2 ) and white ( 100 cd / m 2 ) fields . the built - in software algorithm allowed recordings between 10 and 100 hz ( band - pass filter ), amplified by a factor of 200 , 000 . multifocal ergs ( mfergs ) were analyzed by rings , according to the iscev guidelines . an octopus 900 ® perimeter ( haag - streit inc ., koeniz , switzerland ) was used to conduct vf examinations ; background luminance was 10 cd / m 2 . for semi - automatic kinetic perimetry up to 90 ° eccentricity white stimuli ( goldmann iii4e with a constant angular velocity of 3 °/ s ) were used . stimuli were presented every 15 °. isopter and scotoma areas ( in deg 2 ) were quantified using the built - in software algorithm . as quality criterion for the kinetic perimetry the blind spot was detected with at least 5 stimuli goldmann size 14e at 2 °/ s . white - on - white static perimetry was conducted using a fast threshold strategy gate ( german adaptive thresholding estimation ) up to 85 ° eccentricity . a test grid consisting of 86 stimulus locations which were condensed towards the centre was applied ( stimulus size : goldmann iii , presentation duration : 200 ms , response time : 1200 ms ). a stratus oct ® model 3000 ( carl zeiss meditec gmbh , oberkochen , germany ) scanned in “ radial lines ” mode ; foveal thickness was retrieved after manual verification of the detection borders of the software algorithm , and used for optical coherence tomography . for ophthalmological examination , snellen va was assessed using a projector at a viewing distance of 6 m ( scp - 660 ®, nidek inc ., fremont , calif ., usa ). intra - ocular pressure was tested using goldmann applanation tonometry ( haag streit , bern , switzerland ). a darkadaptometer ® ( roland consult stasche & amp ; finger gmbh , brandenburg , germany ) was used to determine final thresholds for cones ( red test light , 625 nm ) and rods ( green test light , 530 nm ) after 20 minutes of dark - adaptation with maximally dilated pupils . color discrimination was tested monocularly using farnsworth d - 15 ( saturated ) and lanthony desaturated d - 15 presented under a judge ii light testing box ® ( gretag macbeth , neu - isenburg , germany ) at 6500 k . the saturated test was scored using the color confusion index ( cci ); the desaturated test using the total error score ( tes ). scoring was performed by a web - based software ( http :// www . torok . info / colorvision / d15 . htm ). examinations at visits 1 and 9 included : general appearance , neck , extremities and joints , skin , lymph nodes , auscultation of lungs and heart , venous blood samples for : complete blood count , electrolytes , and basic renal and hepatic values .— pulse rate , blood pressure , and body temperature were taken at all visits . ten percent of all data extracted from the electronic case report forms ( ecrfs ) were randomly chosen to be double - checked manually using the monitoring process before statistical analysis was performed using jmp ® ( version 8 . 0 . 1 . 0 ; sas institute , inc ., cary , n . c ., usa ). descriptive statistics were used to summarize data with illustrations of means and 95 % confidence intervals . to analyze the influence of treatment intra - individual differences were calculated for each subject between baseline and follow - up visits . comparison between treatment groups was performed using the method of restricted maximum likelihood ( reml ) to estimate the development of parameters under treatment during visits for each group . in contrast to maximum likelihood estimation , reml can produce unbiased estimates of variance parameters . to compare groups tukey - kramer post - hoc test analysis was applied with a global level of significance set at p & lt ; 0 . 05 . due to the exploratory character of the study no single endpoints were declared and no adjustment for multiple testing was performed . all 24 patients completed the entire follow - up period . electrical stimulation via dtl electrodes was tolerated well , as reported by gekeler et al ., loc . cit ., even when electrodes where not only used for threshold determination with short current pulses but continuously for 30 minutes with supra - threshold currents . base line stimulation current values of groups did not differ significantly , with the exception of cone threshold sensitivity ( p = 0 . 031 ). stimulation current was 0 . 23 ± 0 . 13 ma and 0 . 36 ± 0 . 18 ma in the 66 % and 150 % group , resp . ( overall mean ± sd ). one patient reported mild foreign body sensation and showed minor irritation of the conjunctiva after iop measurement , which we related to the anesthetizing drops . another patient experienced foreign body sensation once after stimulation , which resolved after prescription of artificial tears 2 hourly for one day . no other adverse events or serious adverse events ( saes ) were encountered . general examinations and blood tests were all within normal ranges at all visits . during study visits , the dark - adapted a - wave amplitude , elicited by an iscev sf , ( international society for clinical electrophysiology of vision standard flash ), changed by 6 . 72 μv , − 0 . 07 μv , and 0 . 64 μv in 150 %, 66 %, and sham group , resp . ( mean reml ; p = 0 . 49 ). the b - wave amplitude changed by 8 . 79 μv , − 8 . 82 , and − 8 . 43 μv in the 150 %, 66 %, and sham group , resp . ; the 150 % group differed significantly ( p = 0 . 027 ). no obvious change occurred for implicit times ( p = 0 . 90 and 0 . 96 for a - and b - wave , resp .). in the photopic erg the a - wave amplitude changed 1 . 11 μv , − 10 . 35 , and 2 . 77 μv in the 150 %, 66 %, and sham group resp . ; the incoherent behavior missed statistical significance ( p = 0 . 062 ). the b - wave amplitude changed by 6 . 66 μv , − 0 . 31 μv , and − 0 . 21 μv in the 150 %, 66 %, and sham group , resp . ( p = 0 . 082 ). no obvious change occurred for implicit times ( p = 0 . 80 and 0 . 25 for a - and b - wave , resp .). snellen va ( in logmar ) changed 0 . 017 , 0 . 039 , and 0 . 067 in the 150 %, 66 %, and sham group , resp . ; the changed missed statistical significance ( p = 0 . 22 ). visual field area changed by 17 %, − 9 %, and − 6 % in the 150 %, 66 %, and sham group , resp . ( p & lt ; 0 . 001 ). mean sensitivity in the vf changed by − 0 . 18 db , − 0 - 09 db , and 0 . 66 db in the 150 %, 66 %, and sham group , resp . ( p = 0 . 12 ). the final threshold of rod sensitivity decreased by 0 . 39 cd / m2 , 0 . 081 cd / m2 , and 0 . 043 cd / m 2 in the 150 %, 66 %, and the sham group , resp . ( p = 0 . 53 ). the final threshold of cone sensitivity decreased by 0 . 19 cd / m 2 , − 0 . 019 cd / m 2 , and − 0 . 084 cd / m2 in the 150 %, 66 %, and the sham group , resp . ( p = 0 . 46 ). sensitivity of electrically evoked phosphenes at 20 hz increased by 43 %, 25 %, and 27 % in the 150 %, 66 %, and sham group , resp . ( p = 0 . 73 ). the cci of the farnsworth d - 15 saturated changed by 0 . 058 , − 0 . 04 , and − 0 . 24 in the 150 %, 66 %, and sham group , resp . ; values changed significantly ( p = 0 . 034 ). the tes of the lanthony desaturated d - 15 changed by 27 . 99 , 2 . 98 , and 14 in the 150 %, 66 %, and sham group , resp . ; values did not change significantly ( p = 0 . 72 ). to summarize , the inventors found a tendency of improvement in the following 8 examinations after tes : scotopic a - and b - wave amplitudes of the erg , the photopic b - wave amplitude , snellen va , vf area , sensitivity for electrical phosphenes , and cone and rod threshold sensitivity . some of these changes were small or did not reach statistical significance . final evaluation showed an improvement in vision of about + 0 . 05 logmar in both the sham and 66 %- group , but an improvement of about + 0 . 1 logmar in the 150 % group . this means that about one line more can be read on a vision testing chart . study of the visual fields showed no change in the sham and 66 % groups , but a statistically highly significant improvement ( p & lt ; 0 . 001 ) of about + 20 % in the 150 % group ; see fig1 . the visual field is to be regarded as particularly critical in importance for patient mobility , since it permits the patient at least to orient himself / herself independently both indoors and outdoors . the decision to use individual phosphene thresholds in this study instead of fixed predefined amplitudes in each group was based on several thoughts . first , the inventors believed that the stimulation strength which is necessary to produce an effect is dependent on the stage of the disease , i . e . more heavily degenerated retina should require higher current amplitudes . and since ept is a measure of the disease stage it can be taken as an indicator for stimulation strength . second , the inventors see a potential beneficial effect of stimulating supra - threshold since this ensures activation of remaining retinal cells , especially ganglion cells and also the whole visual system , an effect which would go beyond stimulation of e . g . müller cells to enhance production of neurotrophic factors . transcomeal electrical stimulation can have several potential advantages over other treatments . first , tes is little invasive and bears almost no danger of adverse or even serious adverse reactions , such as endophthalmitis after intraocular injections or very severe aes e . g . as gene therapy . second , est is cheap , costs of electrodes are low and a stimulator can be built using standard electronic components . furthermore , parameters of tes can be easily adjusted and there are possibilities to evaluate stimulation parameters after treatment periods through storage of impedance and other parameters in the stimulator unit . transcomeal electrical stimulation can also stimulate all ocular structures , including the retina , choroid , ciliary body , ganglion cells and optic nerve . on the hypothesis that tes exerts its effect — even if only partially — through activation of any these structures and their respective neurotrophic factors and pathways it could be beneficial to simultaneously affect the expression of a variety of these factors . the clinical study was carried out with standard commercial devices of types which have already been in use for years in electrophysiological eye examinations . determination of the individual phosphene threshold and the treatment parameters must be carried out by a physician . since the therapy may last for years , even blind persons should be able to apply the therapy themselves under the direction of a physician . the aim is that the patient should be able to administer the treatment in home surroundings . this makes local application of corneal anesthetics in particular impossible , since the risk of injury to visually handicapped patients would be increased by a suppression of the sense of pain . of all types of eye electrodes , dtl electrodes appear to be best tolerated by the patients ; see gekeler et la ., loc . cit . this means that a slender electrically conducting thread is positioned in the conjunctival sac or on the edge of the lid and fixed in place on the face with adhesive skin plasters ; see fig2 . however , fixation of dtl electrodes on the face involves problems . the slender fiber is difficult even for staff assistants to recognize , and its fixation on the skin is time - consuming , unreliable , and requires manual dexterity . prefabricated , disposable electrodes would be a solution for blind persons provided that reproducible positioning on the face would be feasible , with the proviso that only tactile sensation would be present . it must also be possible to check the correctness of electrode positioning on the eye by means of functions built into the stimulation device , since there is no way to ensure that the thread will always be felt . with regard to operation of the unit by persons with severe visual handicaps , the electrical current supply should require a minimum of supervision in operation and should also not require an ability to read display devices and the like ; rather , any required warning signals and operating instructions should be purely acoustical . only the attending physician should be allowed to determine treatment parameters . it is desirable that the physician should have access a posteriori to a treatment protocol showing the time and actual duration of treatment , since experience has shown that patient compliance is often difficult to assess . dtl electrodes consist of a plastic thread coated with an inert metal . depending on thickness , the thread is well tolerated . very thin filaments are not felt at all , but are also nearly invisible for staff assistants during the positioning process . thicker filaments which are easier to see are also a source of constant irritation . an empirical compromise must therefore be found . additional inquiries were made of several patients concerning their experiences , and electrodes of various thicknesses which had also been tolerance - tested by the investigators on themselves were presented to them . the threads which were selected ( monofilament , 0 . 1 mm in diameter ) lie securely on the corneal surface or the edge of the lid and engender only the same kind of irritation as that created by an eyelash , for example . local anesthesia is unnecessary . for hygienic reasons , the electrodes were eo ( ethylene oxide ) sterilized and delivered in the form of a loop . a small button - shaped counterweight ensures a certain amount of mechanical tension while also giving the patient a means of grasping and holding them ; see fig3 . in order to position the thread electrode on the eye in spite of the patient &# 39 ; s visual handicap , it is necessary to establish reproducible points of reference at which the electrode can be attached to the head and remain in contact with the edge of the lid at the assigned position . two different approaches for creating these points of reference are disclosed in above mentioned wo 2011 / 086150 a2 . a mask is fitted to the patient &# 39 ; s facial eye area , much like a plaster cast for fractures . plastic sheets cut to size are heat - softened in warm water , then pressed gently into place around the facial eye area . the material remains pliable for about ½ hour and can be molded to the facial contours during that time . subsequently , smaller adjustments are possible by means of a jet of hot air for renewed , local warming . the facial area around the eyes is then cut away with scissors , and buttons which will later hold the electrodes are pressed into the mask at its corners . these buttons are pre - assembled with connection wires for the electric current supply or for connection to a standard commercial skin electrode which serves as a counter - electrode . the intention is for the patient simply to connect a thread electrode ( with matching buttons on its ends ) by pressing it into place ; after a bit of practice , this can easily be accomplished by searching and feeling with the fingers . after the facial mask has been correctly fitted , the thread electrode is allowed to fall onto the corneal surface of the open eye , where it is barely felt . it is recommended that the patients keep their eyes closed after emplacement of the electrodes in order to avoid excessive mechanical irritation of the cornea and prevent the thread electrode from slipping out of place . the use of plaster casts is standard practice in many orthopedic centers and hospitals , for example as a means of stabilizing bone fractures . the manual techniques for fitting masks as described above are thus already available . the technique of pinpointing positions on the face with the help of a lens frame has been known for centuries and has proven reliable . it is already known from the field of optometry that spectacles can be adjusted in this way with an accuracy of about 1 mm , which is extremely important for progressive lenses in particular . this alternative approach therefore uses the same type of lens frames as those used by competitive marksmen to support a retractable occluder or aperture . in the present case , two adjustable pins were attached to this frame for each eye ; these were horizontally and vertically adjustable in such a way that they nearly touched the corners of the eyelids ; see fig4 . this initially required a bit of manual dexterity at the beginning , but no special expertise . the pins are electrically conductive and have notches facing the eye ; in each case , the loop - shaped , thread electrodes are positioned in these notches . the button - shaped counterweight puts light tension on the thread , holding it against the corneal surface . the pins are connected by a wire to the electric power supply , and another wire conductor attached to the lens frame leads to a standard commercial skin electrode , also with a button , which serves as counter - electrode . this skin electrode is kept in position around the temple of the head by an electrically conducting gel . the electronic control unit must be capable of meeting demands which initially appear mutually exclusive . on the one hand , it should allow the attending physician to specify the individual phosphene threshold , thus offering a flexible array of settings . on the other , it should provide the patient with a treatment designed for him / her alone , without optional settings , and without requiring residual vision for its use . initial observations made it clear that the idea of two different devices , one for the physician and one for the patient , was too complicated , and that it would be simpler to use a single device which the physician could connect by usb to his pc ; he could then vary the stimulation current by means of a graphic user interface . as it turned out , the electronic control unit requires less than 1 minute in standard threshold mode to increase stimulation current in 60 logarithmic steps from 5 μa to 10 ma . when the patient reports seeing the first phosphene , which typically occurs at about 1 ma , the physician stops the process . he can then determine the desired individual treatment parameters ( pulse form , strength of current , duration of treatment ) and store them on a miniature usb flash memory stick . these data are stored in encrypted form so that the above - mentioned treatment parameters can be changed only with the physician &# 39 ; s own software . the electronic control unit also has a connection designed specifically for this memory stick format so that other standard commercial usb sticks cannot be inserted into it . typical pulse data ( 30 minutes a week ) are : biphasic , stimulus currents of 5 ms duration for each positive and negative pulse , and a repetition rate of 20 hz . the physician can now leave the device ( with the memory stick ) in the patient &# 39 ; s possession for use at home or , if the patient already has such a device , simply give the patient the memory stick . the treatment parameters actually used by the patient ( pulse data , current strength , time and duration of treatment ) are then stored on the memory stick ; the physician can check the patient &# 39 ; s compliance at their next meeting , and can also store and process the data himself . the device itself is battery - powered and transportable ( weight : 430 g ). a set of standard commercial batteries lasts for several weeks of stimulation in normal use . the patient &# 39 ; s foil - type keypad is designed with only three buttons for tactile recognition : on / off / stop , pause , and start ; see fig5 . constant pressure on on / off / stop turns the device on and off . pressing it briefly interrupts stimulation ; this is recorded in the protocol . the pause button , in turn , interrupts stimulation only temporarily . acoustic signals inform the patient of the status of the device at all times . a rhythmic tone with gradually increasing speed means that the device has just been turned on , a rhythmic , gradually slower tone means that it has just been turned off . correct stimulation is indicated by a tone which sounds like the ticking of a clock ( with a soothing effect ). if the thread - like electrode wire is not correctly positioned on the eye or if a short circuit occurs , this is indicated by corresponding , intensive warning tones ( or , if a pc is used as control device , only by an icon on the monitor screen ). for this purpose the device has a compliance detector which constantly compares current delivered with the actual electric potential in order to determine the impedance of the patient / electrode system which is connected . 200 ohm and 5 kohm were empirically determined as limit values . a value of less than 200 ohm can occur only if the electrodes are connected in such a way that a short circuit occurs . values of more than 5 kohm mean that the thread electrode wire is not well positioned on the eye but rather — for example — on the outer eyelid or that no thread electrode is present at all . the detector thus also gives a warning if the skin electrode is positioned — contrary to instructions — without prior removal of fat from the skin area . this detector has proven to be very helpful , since the correct position of the thread on the cornea cannot be reliably perceived by the sense of touch alone when the lens frame with its electrode is put on , and can be ascertained even by persons with normal vision only under optimum lighting . the actual treatment times are also recorded on the flash memory stick and provide information to the physician who reviews the protocol as to whether and to what extent the patient has successfully coped with the system . an lcd display shows patient impedance at every moment and notifies of any errors which may occur . however , such information is relevant only in case of problems with the device and is therefore intended only for the physician or persons caring for the patient . the system described here makes electrical stimulation available for treatment based on clinical data gathered by a universally respected german university hospital in accordance with good clinical practice . the treatment parameters will require further testing and optimization in further , more comprehensive clinical studies . with a bit of practice , the new stimulation system can be used under the direction of a physician by both medical staff assistants and by the patients themselves . the system also helps lay the groundwork for carrying out multicentered studies in this medical area . initial reactions from patients indicate major interest . as used herein , “ a ” or “ an ” means “ at lest one ” or “ one or more ”. unless defined otherwise , all technical and scientific terms used herein have the same meaning as is commonly understood by one of ordinary skill in the art to which this invention belongs . all patents , patent applications ( published or unpublished ), and other publications referred to herein are incorporated by reference in their entireties . if a definition set forth in the present application is contrary to or otherwise inconsistent with a definition set forth in the patents , applications , published applications and other publications that are herein incorporated by reference , the definition set forth in the present application prevails over the definition that is incorporated herein by reference .
in a method for treating at least one eye of a patient in need of such treatment with a pulsed electrical stimulation signal , first an individual parameter of the patient is determined , thereafter at least one stimulation parameter of the pulsed electrical stimulation signal is set depending on the at least one individual parameter , and then the pulsed electrical stimulation signal is applied to the at least one eye .