a simplified perspective view of an exemplary cryosurgery system in which embodiments of the present invention may be implemented is illustrated in fig1 , 2 and 3 . cryosurgery system 100 comprises a pressurized cryogen storage tank 126 to store cryogen under pressure . in the following description , the cryogen stored in tank 126 is liquid nitrogen although cryogen may be other materials as described in detail below . the pressure for the liquefied gas in the tank may range from 5 psi to 90 psi . according to a more preferred embodiment , pressuring in the tank during storage is 40 psi or less , and pressure in the tank during operation is 35 psi or less . according to a more preferred embodiment , pressure in the tank during storage is 35 psi or less and pressuring during operation is 25 psi or less . according to a most preferred embodiment , pressure during operation at normal nitrogen flow is 22 ± 2 psi , and pressure during operation at low nitrogen flow is 14 ± 2 psi . when the pressure in the tank during operation is set to 22 psi , the flow rate / cooling capacity of the nitrogen is 25 w . when the pressure in the tank during operation is set to 14 psi , the flow rate / cooling capacity of the nitrogen is 12 . 5 w . in an alternate embodiment , the cryogen pressure may be controlled all the way to 45 psi and to deliver through smaller lumen catheters and additional feature sets . in such alternate embodiments the pressure in the tank during storage may be 55 psi or less . in the context of the output pressure of cryospray from the distal end of the catheter , the term low pressure means 2 psi to 20 psi . in the embodiment illustrated in fig1 , a conventional therapeutic endoscope 134 is used to deliver the nitrogen gas to target tissue within the patient . endoscope 134 may be of any size , although a smaller diagnostic endoscope is preferably used from the standpoint of patient comfort . in certain embodiments , a specially designed endoscope having a camera integrated therein may also be used . as is known , an image received at the lens on the distal end of the camera integrated into endoscope 134 may be transferred via fiber optics to a monitoring camera which sends video signals via a cable to the a conventional monitor or microscope , where the procedure can be visualized . by virtue of this visualization , the surgeon is able to perform the cryosurgery at treatment site 154 . as the liquid nitrogen travels from tank 126 to the proximal end of cryogen delivery catheter 128 , the liquid is warmed and starts to boil , resulting in cool gas emerging from the distal end or tip of catheter 128 . the amount of boiling in catheter 128 depends on the mass and thermal capacity of catheter 128 . since catheter 128 is of small diameter and mass , the amount of boiling is not great . ( the catheter would preferably be of size seven french .) when the liquid nitrogen undergoes phase change from liquid to gaseous nitrogen , additional pressure is created throughout the length of catheter 128 . this is especially true at the solenoid / catheter junction , where the diameter of the supply tube to the lumen of catheter 128 decreases from approximately 0 . 25 inches to approximately 0 . 070 inches , respectively . but the catheter range diameter of its lumen may be between 0 . 030 to 0 . 125 inches . in an alternate embodiment the gas boiling inside the catheter may be reduced even greater by the use of insulating materials such as ptfe , fep , pebax and others to help reduce its temperature coefficient . the addition of ptfe is especially desirable if done in the inner lumen because its lower coefficient of friction aids in laminar flow of the fluid and thus reducing turbulence and entropy . this reduces gas expansion and allows for good fluid velocity . when the liquid nitrogen reaches the distal end of catheter 128 it is sprayed out of cryogen delivery catheter 128 onto the target tissue . it should be appreciated that certain embodiments the cryosurgery system may be able to sufficiently freeze the target tissue without actual liquid nitrogen being sprayed from catheter 128 . in particular , a spray of liquid may not be needed if cold nitrogen gas is capable of freezing the target tissue . freezing of the target tissue is visually apparent to the physician by the acquisition of a white color , referred to as cryofrost , by the target tissue . the white color , resulting from surface frost , indicates the onset of mucosal or other tissue freezing sufficient to initiate destruction of the diseased or abnormal tissue . the operator may use the system timer to freeze for a specified duration once initial cryofrost is achieved in order to control the depth of injury . in one embodiment , the composition of catheter 128 or the degree of insulating capacity thereof will be selected so as to allow the freezing of the tissue to be slow enough to allow the physician to observe the degree of freezing and to stop the spray as soon as the surface achieves the desired whiteness of color . the operator may monitor the target tissue to determine when cryofrost has occurred via the camera integrated into endoscope 134 . the operator manipulates cryogen catheter 128 to freeze the target tissue . once the operation is complete , cryodecompression tube 132 , catheter 128 , and endoscope 134 are withdrawn . catheter length may be anywhere from 10 inches to 100 inches . inside diameter of the catheter may be anywhere from 0 . 8 mm to 5 mm , preferably from 1 mm to 4 mm . the tank size may be anywhere from 5 l to 100 l ; its diameter may range from 4 inches to 36 inches . the vent orifice of the manifold may be 0 . 01 inches to 0 . 1 inches . fig2 is a perspective view of a portion of a cryosurgery system 200 having a cryogen delivery apparatus 240 . cryosurgery system 200 comprises an endoscope 202 having lumens 210 , 212 and 216 therein . as shown , endoscope may be positioned in the esophagus 222 of patient 250 . lumen 212 , disposed in endoscope 202 , is configured to receive an endoscope camera 242 . lumen 210 may be configured to receive a light 244 for illumination of the treatment site . lumen 216 of scope 202 may be configured to receive cryogen delivery apparatus 240 . cryogen delivery apparatus 240 comprises a retroflex - capable cryogen delivery catheter 204 , catheter tip 206 , and one or more holes 214 . after insertion of the cryogen delivery apparatus into the patient , cryogen is provided to cryogen delivery catheter 204 from a cryogen source . tip 206 causes the cryogen to be sprayed on the target tissue via hole 214 . a dual lumen ( for both passive and active venting ) cryodecompression tube 208 may be provided to evacuate the treatment area of undesirable gases , particles , fluids etc . alternatively , the controlled pressure and pulsing , coupled with careful control of catheter diameter , length and material composition , helps further deliver controlled flow of volume over time that is consistent with the cryogenic property of the fluid being delivered . dual phase fluid flow is achieved out of the catheter distal tip and maintained constantly via the equilibrium that the system achieves after pre - cool and after the catheter achieves a cold temperature . the range of dual phase fluid cryogen delivery out of a cryogen catheter with this system can range from 5 lpm to 50 lpm ( once it all expands into gas ). fig3 is a perspective view of a portion of a cryosurgery system 41 having a cryogen delivery apparatus 42 . cryosurgery system 41 comprises a bronchoscope 40 and a catheter tip 42 exiting its working channel . as shown , bronchoscope 40 may be positioned in the trachea 44 , or bronchi — such as the principle bronchi 45 of patient . the catheter 48 is placed in the working channel lumen 46 of the scope 40 and exits the working channel at the distal tip of the scope . cryogen delivery apparatus 42 comprises a radial spray cryogen delivery catheter at distal end 42 , and one or more holes 47 . after insertion of the cryogen delivery apparatus into the patient , cryogen is provided to cryogen delivery catheter 48 from a cryogen source . catheter distal end with one or more holes 42 causes the cryogen to be sprayed on the target tissue via hole ( s ). a gas egress tube 43 that surrounds the scope may be utilized to provide additional means to evacuate the treatment area of the cryogenic gas out of the patient 49 . passive lumen egress 50 is also present via the management of the airway to ensure proper venting during the procedure . the catheter is designed to transport liquid nitrogen ( or other cryogen ) from the console to the patient treatment site . according to one embodiment , the catheter may contain ( 1 ) a bayonet and hub for attachment to the console at its proximal end , ( 2 ) a layered polyimide and stainless steel braided shaft to minimize kinking and breaking , ( 3 ) insulation to protect the user from cold , ( 4 ) a strain relief to help prevent kinking when torqued by users and ( 5 ) an atraumatic tip at its distal end to prevent damage to tissue . the laminated construction and braided material provides additional strength and flexibility , allowing the physician to retroflex the catheter during a treatment procedure , if needed . the catheter pouch may contain an rfid tag that the user scans prior to use to prevent reuse and track disposable information . the catheter pouch may also contain an introducer that provides reinforcement for the catheter and helps prevent kinking during use and when placing the catheter into the scope . an alternative construction locates the rfid tag on the connector area adjacent to the bayonet . according to a preferred embodiment , the delivery catheter may be constructed of three layers of flexible polyimide , surrounded by a stainless steel braid , which is in turn coated with an outer layer of pebax . it was discovered that that extrusion of pebax over the stainless steel braid allows the pebax to wick through the pitch of the steel braid , helping to prevent kinking , breaking , or delamination during retroflex of the catheter . the pebax also provides a desirable balance between hardness — important for smooth sliding of the catheter and general toughness , and softness , which is important for some degree of tackiness which allows the user to feel the movement of the catheter in the scope . the pitch of the stainless steel braid is configured to be fine enough to afford the required strength , not thick enough to allow the pebax to wick through . the distal end of the catheter is provided with an atraumatic tip comprised only of pebax , in the shape of a bullnose . this novel construction allows for retroflex of the catheter without kinking , breaking , or delamination of the catheter . for the purposes of this invention , retroflex is used to refer to the ability of a catheter to bend or turn approximately 180 ° about a radius of curvature of 1 inch or less . this is useful so that when the catheter is introduced into , for example , the stomach via the esophagus , the catheter can be turned approximately 180 ° in order to treat the roof of the stomach . fig4 shows the preferred embodiment catheter construction of the cryospray catheter 1 according to the invention . it includes a bayonet connection 2 , catheter connection housing 3 , insulation 4 , laser cut hypotube with fep or pebax heatshrink wrap 5 , nozzle connection of diminishing inner diameter 6 with wings for torqueing 7 , multilayer polymeric shaft 8 , radial spray pattern 9 , spray pattern indicator marking bands at tip 10 , spray pattern indicator marking band at other end of hole pattern 11 , s - curve shaped shaft area 12 . by adding very thin layers of metal to the catheter shaft or increasing the heat transfer coefficient in the shaft by adding a braided metal for example , the catheter may be constructed to provide optimal cryo delivery to the tip of the device in a very short cycle time . fig5 shows a close - up of a catheter tip 13 with alternate construction utilizing a self - expanding spherical polymeric frame 14 . this frame is preferably made out of a pebax extrusion and laser cut into multiple slits along its length . when the end shape is compressed , it forms the spherical frame shape 14 on the illustration . the frame is held in place by center shaft 15 containing a spring 16 which keeps the frame in compression until it is inserted into a small lumen which collapses the spherical frame 14 and stretch the center shaft spring 16 . fig5 also shows the radial spray pattern 9 bracketed by the markings that delineate the beginning and end of the radial spray . fig5 also shows a radial spray hole pattern that extends for 1 mm distance longitudinally and contains 8 rows of holes along the circumference to target a lumen at 360 degrees cryo spray coverage . the fenestrated hole pattern can vary with embodiments meant to target less or more longitudinal distance . the hole pattern can also be cut to target a quadrant or quadrants along a lumen ranging from 0 degrees to 360 depending on desired spray coverage . the size of the holes on the embodiment on the illustration are 0 . 015 inches . however , the hole sizes can vary from 0 . 004 inches to 0 . 030 inches in diameter . additionally , the holes can be any shape , e . g ., round , square , diamond , oval , rectangular , star - shaped , etc . continuing on fig5 , the spray is blocked by the center shaft and spring assembly at 17 . this ensures that the cryogen exits the fenestrated holes 9 and not the end of the shaft . the opposite end of the sphere 14 contains a machined piece 18 that is utilized to secure the spring 16 and form an atraumatic tip area at the end of the catheter . fig6 shows a typical hypotube 19 used for the construction of the proximal end of the catheter shaft 5 . it typically has a length of 50 inches but can vary from 24 to 96 inches in length . the internal diameter of the tube 19 is usually 0 . 104 inches but can vary between 0 . 045 to 0 . 150 inches . in the preferred embodiment , the hypotube 19 may be laser cut as a spiral , but other variable cuts can be present . the cuts provide flexibility to the metal tube . fig7 shows the s - curve 12 seen on fig4 when used with the scope for centering the spray with respect to the diameter of the scope 20 and the working channel exit 21 , which is off center on the scope . according to this embodiment , the catheter can be rotated along the axis of the working channel 21 , providing a level of alignment along the lumen that allows for centering of the spray pattern by the user as the tissue is targeted for cryo spray ablation . fig8 shows a catheter construction in which proximal metal tube with outer polymeric liner is constructed out of a pre wound wire coil 22 . this coil provides the cold conduction as the cryo spray is applied ahead of the dual phase flow to help establish a low temperature gradient between the materials and the cryogenic fluid in the catheter . the coil shown in fig8 is then mated to a polymeric junction 6 that serves various functions , such as strain relief , fluid transition from a large diameter to a small diameter , and it provides a torque point for the distal catheter in catheters with the s - curve as the centering feature . to aid in the torqueing , wings 7 are provided on the junction . fig9 shows an s - curve feature 12 on the catheter that is pronounced or exaggerated beyond the center of the scope 20 . the pronounced or exaggerated s - curve provides improved positioning in the larger lumen and smaller lumen selection and navigation more navigation in larger lumen areas of the body . by torqueing the exaggerated curve 12 , along with the scope 20 manipulation and flexing , the navigation of the scope 20 may be enhanced . fig1 shows the junction 23 of the bayonet console connector 2 to the hypotube 19 or coil 22 ( not shown in fig1 ). the hypotube 19 may be welded to the bayonet 2 to create an all - around seal around the metal junction 23 . an fep heatshrink 24 may be applied to the entire length of the hypotube 19 or coil 22 . the heatshrink 24 can also be pebax or pet . fep heatshrink is preferred for cryogenic applications . fig1 is shows a transition 25 of a large diameter hypotube shaft 19 to a small diameter polymeric shaft 8 . the transition is so that a smaller diameter can be inserted into the working channel of a scope . in addition , the transition from large diameter to small diameter acts as a mixing point for the dual phase flow gas and liquid to interact along the catheter path and allow for the gas to once again attain the velocity of the liquid as they travel down the pipe . this transition is referred to as a “ nozzling ” transition . this transition can occur between two hypotubes , two polymeric shafts or between a coil and hypotube or coil and polymeric shaft . fig1 shows the insulator 4 and the connector housing 3 added to the catheter assembly 1 . fig1 shows a design in which the catheter radial spray pattern is supplemented by a straight end spray for such cases where lumen treatment of quadrants is desired at the same time that targeted spray is needed for lesions or specific tissue . this is accomplished by the addition pre - formed tip of a pebax extrusion or a molded polymer or cast metal that is adhered to the tip of the catheter . such pre - formed tip controls the size of the end spray radius . the diameter can vary from 0 . 010 inches to full inner diameter of the catheter shaft . the current typical diameter shaft of the preferred embodiment is 0 . 061 inches . in alternate embodiments the distal end of the catheter may be a preformed plastic tip ( typically pebax ) with a specific geometry that allows for specific spray patterns other than those coming out of the catheter shaft end ( aka . straight spray ). fig1 also illustrates an optional radial spray configuration according to which catheter 271 is fitted with a spray pattern tip 272 that includes holes 273 , 274 , 275 of different sizes at different distance positions that allow for gradual spray across a specific distance of the catheter shaft 271 . the hole patterns 273 , 274 , 275 may have dimensions that are between 0 . 005 ″ to 0 . 050 ″ in diameter . in this illustration , the hole at the distal end of the catheter 276 for straight spray may or may not be there and may have a diameter that is different from the rest . the diameter of hole 276 may have a range or 0 . 020 ″ to 0 . 085 inches . the construction of this tip may be achieved via drilling of the different hole sizes , fusing or adhering a preformed and predrilled tip or insert molded via micromolding techniques . fig1 shows an embodiment of a radial spray catheter where the self - centering mechanism is a cobalt chrome mesh basket 27 created out of a circular braid . the material can also be stainless steel wire or a polymeric molded process mesh . the main difference is that this centering basket is over the spray area 28 to create a well dispersed spray . in such case the mesh can be designed to be a visual aid for centering . the goal here is not necessarily to expand the mesh to touch all tissue , but instead to allow the basket to disperse the cryo spray more evenly in the spray treatment lumen . fig1 shows a fenestrated radial spray pattern illustration of different types of sprays attainable in the spray pattern area . the patterns from top to bottom demonstrate various hole patterns consisting of varying numbers of rows , varying hole sizes , number of holes per row , number of slits instead of rows , separation between holes , spiral hole patterns around the circumference , and variable hole patterns to compensate flow along the length of shaft . slits can either be vertical or horizontal with respect to the shaft length . individual hole sizes can vary from outer diameter to inner diameter . the holes can also be made at an angle within the wall thickness of the tube to direct spray in various directions . fig1 is an isometric view of the catheter with an s - curve centering feature built into its distal tip shape . it shows the bend 12 and the alignment line 29 that is the feature used to visually align the catheter with respect to the scope working channel offset . fig1 shows the s - curve 12 as seen through the scope 20 visualization system . the method of use is to target the area to be treated by locating the catheter section 11 between the marking bands , then rotating the catheter axially until the axial line 29 is visible and horizontal in the line of vision . at this point the catheter tip is relatively centered with the scope 20 centerline . this axial line is typically created via a pad printed or laser marking process . not shown is a thermocouple wire construction within the catheter assembly that may be integrated outside of the proximal coil or hypotube construction . in addition , the thermocouple wire may be integrated into the braiding of the polymeric distal shaft or run along the outer diameter of such shaft . the thermocouple may connect to the console via a set of contacts within the console bayonet housing . the distal tip of the catheter is located within 3 cm of the tip and is also laser welded . multiple thermocouple wires can be run along the shaft to create redundancy or report multiple catheter length locations . the typical wires used are copper and constantan . according to a further embodiment the catheter may be fitted with a temperature sensing probe attached to the distal end of the catheter . this is achieved by laying at least two wires longitudinally or in a coil pattern prior to the outer layer of polymer laminated onto the catheter outer layer . if the wires are thermocouple wires , then they can be terminated into a thermocouple . alternatively , a cryogenic thermistor can be attached to the distal end of the catheter . such thermistor can then be encapsulated via conductive epoxy and a polymeric sleeve . then the thermistor can be used to monitor both the temperature at the end of the catheter tip as well as the treatment area for both freezing and thawing temperature monitoring . according to yet a further embodiment , there is provided a dual lumen , lumen - within - lumen catheter construction , see , e . g ., fig2 . such construction provides a cryospray catheter that can be precooled via the recirculation of fluid all the way to its distal end . the precooling is either achieved by the console control or the user &# 39 ; s input command ( like via a foot pedal ). the cryospray catheter 281 contains a valve or shutter 282 that is then engaged either via the console control or the user . fig2 describes the trigger type mechanism 283 that is engaged by the user for the duration of the spray to the treatment site . the mechanism 283 can be spring loaded to allow it to retrieve to the close position after treatment time is done . the valve is mechanically connected remotely to the trigger mechanism 283 via an engagement wire 284 running along the length of the catheter shaft 285 . the wire 284 is connected to a sliding sleeve so that when the trigger is engaged the sleeve slides back and opens up the elastomeric diaphragm as shown retracted in dashed lines . a failsafe to the valve 282 opening and closing is the user can depress the console flow control that stops the recirculation along the catheter shaft 285 if the mechanical trigger fails to immediately retract due to freezing issues . the catheter shaft consists of dual lumens with an input and an output port for the path of recirculation . in fig2 , the recirculation path is shown via an inner lumen 288 that is surrounded by an outer lumen 289 which returns the dual phase fluid flow back to the console for recollection . holes on the inner lumen 288 allow for this to occur . in yet another alternate embodiment , the control of the cryospray is achieved through a nozzle flow created by shafts of a certain length and diameter size , previously referred to as “ nozzling .” fig2 demonstrates how the pressure of the console 277 may remain constant , but the combination of catheter shaft 278 and nozzle 279 are used to throttle the output flow at the distal end of the catheter 280 with a specific output flow . the nozzle 279 length can have a range of 0 . 050 inches to 48 inches in length and an inner diameter of 0 . 030 to 0 . 080 inches . likewise the catheter shaft 278 of this construction can have a range of 1 . 5 inches to 90 inches when coupled with the nozzle construction . the catheter shaft can have an inner diameter range of 0 . 30 inches to 0 . 125 inches . more than one nozzle can be created along the catheter shaft length . the diameter of the area through which gas vents passively must be adequate to ensure organ or body cavity distention does not occur . passive venting may be used with a vent tube when spraying proximal to a resistor where the lumen is patent ( open ), or when the treatment area is open to atmospheric pressure ( e . g ., dermatological or open surgery ). a lumen sizing device ( e . g . stent sizer ) may be used to measure the lumen to aid in selection of vent tube size . the greater the vent area , the lower the pressure . the vent tube can be a separate tube used strictly for venting gas and creates a round vent area . the vent tube can also provide an annular vent area where the scope passes through the center of the tube . the distal end of the passive venting tube should be placed in an unobstructed cavity near the procedure area if area is not sufficiently open to atmospheric pressure . if used , the proximal end of the passive venting tube should be positioned outside the body where the pressure is atmospheric . in fig2 and 22 the vent tube 260 takes the shape of sleeve 262 with a lumen 261 . such sleeve 262 or grooved channel 262 can then be utilized to slip the scope 263 into it to allow for the scope insertion into the body cavity to be the placement mechanism . the vent tube is flexible enough that the functionality of the scope is not hindered . the tube ends with an open end 264 to vent to the atmosphere . fig1 through 20 show another version of the vent tube 266 with the sleeve 265 rolled up upon unpackaging , and a scope location opening 267 , and a vent orifice 268 . as shown in fig2 , it is unrolled over the scope shaft 269 and ready for use . fig2 also shows the cryospray catheter 270 located out of the scope working channel . the vent hole 268 may be of dual vent lumen or single vent lumen construction which in turn supports both passive and active ( suction ) venting . fig2 and 30 show an egress tube according to an embodiment of the invention . according to a preferred embodiment , the egress tube 51 may be manufactured from a flexible polymeric material that can be easily extruded . it can have varying durometer ( i . e ., more flexible at distal tip for maneuvering ). the exterior of the tube 51 maybe coated for lubricity for ease of insertion or made of a lubricious material such as ptfe . according to the embodiment shown in fig3 , the egress tube may be connected to a gasket 52 , preferably a large tuohy borst with a sideport , at the proximal end to lock the scope in position and to allow venting to a standard tube 53 for passive venting or active venting ( connected to a suction pump ). according to preferred embodiments of the invention , the exterior of the egress tube 51 may include finer measurement marks to provide guidance for placement of scope . according to further embodiments of the egress tube , it may be provided with a dedicated pressure lumen which can be constructed in a variety of ways ( dual lumen extrusion , reflowing or adhesive of a separate extrusion , etc ). according to the embodiment shown in fig2 , the interior surface of the egress tube 51 may be configured with ribs 54 ( or , alternatively , rows of teeth or studs ) for centering of the scope and creating the channels of egress . according to a preferred embodiment , the interior surface of the egress tube 51 has three ribs . according to preferred embodiments of the egress tube of the invention , the scope is additionally insulated , the following features and advantages obtain : completely encapsulates the scope for insulation ; affords a higher cross - sectional area for egress compared to prior art egress tube of the same size ; smallest outside diameter by utilizing ribs for preservation of egress area instead of material introduced between the scope and the egress areas ; allows for complete maneuverability of the scope by allowing for very small material wall thicknesses around the scope ; allows for treatment of tissue without the added management of a separate tube , as the egress is part of the scope assembly ; allows the scope to reach distal areas for treatment while maintaining vent egress up close ; ease of delivery into et tube due to lubricious outer coating ; monitors pressure through a dedicated lumen . the cryogen decompression tube 132 on fig1 aids evacuation of nitrogen gas from the treatment site . the cryogen decompression tube connects via supplied accessory connection tubing 167 to a disposable suction canister 169 on the front of the console . the dual lumens of the cryogen decompression tube are coupled to ports that provide both active ( to the suction pump ) and passive ( direct to ambient ) vent paths . the dual lumen cryodecompression tube may be of the form on fig2 , where each lumen is independently vented to either a suction pump tube connection or a passive open air connection 291 . the passive venting may serve the function of vent during cryospray , but also the function of working channel to supplement the absence of a working channel if the catheter is inserted into the working channel of the scope . such working channel can be used for tissue manipulation , forceps , biopsy , among other uses . when the catheter sprays out of the catheter distal tip it is described as straight spray . in the alternate embodiment illustrated in fig2 , the liquid nitrogen may be broken down into small droplets via a diffuser 295 or filter to allow for a very even spray pattern and avoid cold spots of spray pattern . the diffuser 295 may be constructed of filter paper , a grating patterned polymer , a metal or plastic mesh basket or laser cutting methods on the shaft itself to pattern it with very small holes . in such embodiment , the catheter ends in a cap 296 that contains small longitudinal cuts 297 that provide for controlled spray to exit as it initially hits a bounce plate 298 on fig2 . the bounce plate 298 is of a conical shape and helps distribute the spray evenly all around the diffuser 295 and cap 296 . fig3 is a side perspective and fig3 is an isometric view that shows a cyclone tube separator assembly 37 as it demonstrates the flow of liquid cryogen along the outer edges of the cyclone tube 30 and the cryogen gas is concentrated on the center part of the tube 30 . the concentrated the cooling power of the liquid is directed to the distal shaft 33 , while the gas portion of the cryogen is vented along the length of the proximal shaft 32 via a return jacket to create both a cooling effect on its surrounding air as well as an insulative layer of cryogenic gas flow . the cyclone tube 30 of the preferred embodiment is a cyclo - uniliner of quartz construction . other embodiments may utilize other types of phase separation device for the cyclone tube . the cyclone tube aids in the separation of the heavier liquid from the lighter gas as the fluid travels through the cyclone tube . spray is thus phase separated in the cyclone tube 30 . the cyclone tube 30 of an alternate embodiment is a specialty tube that is formed via a casting , molding or machining process . the preferred embodiment utilizes a cyclone tube of quartz construction and is encased in a manifold assembly 37 that contains the gas funneling and orifices needed for fluid pathways . the gas center is redirected into an outer lumen 31 of the proximal portion of the catheter . the gas then exits near the bayonet into the cryospray console body for safety . the proximal tube 32 is constructed of a hypotube or polymeric shaft for the cryogen path into the cyclone tube , the outer lumen return jacket 31 concentric to the hypotube is an polymeric tube . the return jacket polymeric tube runs into the connector housing which then vents into the console . the outer edge of the cyclone tube exits into a nozzle that receives the distal shaft 33 . the distal shaft then has the mostly liquid output with greater cryo cooling power . the return jacket and the hypotube form a coaxial double - pipe counter - flow heat exchanger 36 . fig3 shows the front end detail of the phase separator as equipped with a valve plug 38 that in the closed position would redirect the entire flow into the return jacket and heat exchanger section of the catheter and prevent any flow into the distal end of a catheter . this mode of operation would allow to precool section of the catheter between the console and phase separator and when the valve 38 is open , liquid enhanced spray can be injected into the distal portion 33 of the catheter . the separator 37 may work without the cyclone to precool the proximal end 32 of the catheter . in this embodiment , only the fluid manifold portion of the separator 37 is used and the spray is partially redirected into the return jacket to prevent the spray flow inside the hypotube from excessive heat losses . the valve plug 38 can again be employed to interrupt the flow through the distal end 33 of the catheter during the precooling stage .
a cryosurgery system for application of medical - grade liquid nitrogen to a treatment area via a small , low pressure , open tipped catheter . the system includes a console , including a touch panel computer , a cryogen module , a suction module and an electronics module , and a disposable spray kit . features include optional low cryogen flow setting to reduce the cryogen flow rate by 50 %, improved cryogen flow consistency reducing pressure pulses and peaks , an integrated suction pump for improved consistency and self - checks , specified vent tube areas and corresponding maximum expected pressures during cryospray procedure ; optional pressure sensing capability to monitor pressure during a treatment , and novel catheter designs of multilayer and flexible construction providing a variety of spray patterns .