the infusion pump ( 1 ) shown in fig1 has a septum ( 2 ), through which a medicament can be filled into the supply container ( 3 ) with a hypodermic needle ( not shown in the figure ) through the abdominal wall , when the infusion pump ( 1 ) has been implanted into the body of a patient . the supply container ( 3 ), which acts as the medicament chamber , is bounded by a bellows ( 4 ) which expands on filling . this bellows ( 4 ) consists of titanium . the nearly constant excess pressure of the propellant gas in the pressurized gas reservoir ( 5 ) exerts a pressure on this bellows ( 4 ), leading in the course of time to an emptying of the medicament from the supply container ( 3 ). the pressurized gas reservoir ( 5 ) is bounded laterally by a partition ( 9 ), and is dimensioned such that the gas exerts a nearly constant pressure on the supply container ( 3 ) for the whole time while this is being emptied . a sewing eyelet ( 10 ) is fitted laterally on the housing of the infusion pump ( 1 ); at it , the infusion pump ( 1 ) is sewn firmly to the surrounding tissue , so that during the duration of the implantation the infusion pump ( 1 ) does not change its position too much and thereby cause pain to the wearer . the throttle paths ( 7 : 7 . 11 ; 7 . 12 ; 7 . 21 ; 7 . 22 ) limit the quantity of medicament which can be delivered per unit time from the supply container . these throttle paths ( 7 ) begin behind the valve ( 6 : 6 . 1 ; 6 . 2 ), which opens or closes the throttle paths ( 7 ) for medicament flow . the throttle paths ( 7 ) all end in a bolus septum ( 11 ) and have an outlet there via a catheter ( 12 ) into the interior of the body . the physician can also inject directly into this bolus septum ( 11 ), in order to administer an additional dose of medicament to the wearer of the infusion pump ( 1 ). the septum ( 2 ) and the bolus septum ( 11 ) project out of the housing of the infusion pump ( 1 ), so that the physician can feel the two septa ( 2 , 11 ) when searching for them . in the septum ( 2 ), a needle stop ( 8 ) ensures that the injection needle ( not shown in the drawing ) is not damaged while the supply container ( 3 ) is being filled . the outlet of the medicament from the infusion pump ( 1 ) takes place through the throttle paths ( 7 ) which are connected following the valve ( 6 ), so that a constant flow rate is reached . an inlet from the supply container ( 3 ) goes into each of the two valves ( 6 . 1 and 6 . 2 ), and two throttle paths ( 7 . 11 and 7 . 12 ; 7 . 21 and 7 . 22 ) lead out from each valve ( 6 . 1 and 6 . 2 ). the flow rate then depends on the gas pressure and in addition primarily on the length of the throttle paths ( 7 ) and on their capillary diameter . if several throttle paths ( 7 ) are used , different flow rates can be achieved by the parallel and / or serial connection of various throttle paths ( 7 ). the changeover then takes place by means of a tri - stable electromechanical microvalve ( 6 ). this valve ( 6 ) is explained in more detail in the following fig2 and 4 . in fig2 there is shown a section through a electromechanical valve ( 15 ), having three stable operating positions and in fig3 a view of a section through the valve ( 15 ). the piston ( 16 ) consists of a permanent magnet material ( e . g ., vacodym ), and is encapsulated with a material ( e . g ., titanium ) of the encapsulation ( 23 , 24 ) which is compatible ( i . e ., not reacting ) with the medicaments used . the fastening of the piston ( 16 ) consists of a spring element ( 19 ) ( see fig4 and 5 ). the spring element ( 19 ) of fig2 centers , and likewise guides , the piston ( 16 ) within the valve chamber ( 17 ). the leaf springs of fig5 which run in two planes ( 18a , 18b ) exert a prestress which is respectively directed towards the center and which makes possible a stable middle position of the piston ( 16 ). the prestresses are produced by the deformation of the spring element ( 19 ) during mounting . the mounting ring ( 22 ) spreads a portion of the spring element ( 21a , b and c ), thus producing a force component on the piston ( 16 ) in the positive z - direction ( up direction according to cartesian coordinates ). the leaf springs ( 20a , b and c ) are prestressed inwards by the encapsulation ( 23 , 24 ) of the piston as in fig2 so that a force is produced in the negative z - direction ( down direction according to cartesian coordinates ). on motion of the piston in the negative or positive z - direction , dependent on direction , respective leaf springs of the one plane are relieved of stress and in the other plane are further stressed , or vice versa . the valve chamber ( 17 ) has a lateral inlet ( 25 ) and , arranged on the end faces , two opposed , central outlets ( 30a , 30b ), to each of which is fitted a respective capillary connection ( 26a , 26b ) to the throttle paths ( 7 ). a respective electromagnet ( 27a , 27b ), with a coil former ( 28a , 28b ) of low retentivity magnetic material , is seated on each end face of the valve chamber ( 15 ). by correct poling of the two coils ( 29a , 29b ), the permanent magnetic piston ( 16 ) can be moved out of its middle position which is defined by the leaf springs ( 20a , b , c ; 21a , b , c ) towards the + z direction or the - z direction . when the piston ( 16 ) has been completely pulled into a position at one end of its path , one of the two outlets ( 30a , 30b ) is closed . after the current is switched off , this position of the valve ( 15 ) is still maintained . the required holding forces are produced by the permanent magnet piston ( 16 ) and the magnetization which it produces of the respective coil formers ( 28a , 28b ) of the electromagnets ( 27a , 27b ). the dimensions are to be made such that the permanent magnetic holding force is greater than the restoring force of the leaf springs ( 20a , b , c ; 21a , b , c ). the position at any given time of the valve ( 15 ) is determined by sensors ( 31a , 31b ) which are sensitive to magnetic fields , and which are built in on the end face on the side of the electromagnets ( 27a , 27b ) facing the valve chamber ( 17 ). a maximum field strength is then set in the end positions for the sensor concerned ( 31a , 31b ). in the middle position ( m ) defined by the leaf springs ( 20a , b , c ; 21a , b , c ), the field strength is about equal for both end position sensors ( 31a , 31b ). this middle position can be achieved by a dynamically controlled adjustment of the valve ( 15 ). it is possible to bring the piston ( 16 ) into the middle position by means of the built - in sensors ( 31a , 31b ) by means of a suitable control electronics , ( e . g ., a pi controller , in which the flux through both magnetic field sensors is equal in the middle position : φ 1 = φ 2 → φ 1 - φ 2 = 0 ( reference value for the controller )). after the electromagnets ( 27a , 27b ) are switched off , this state is still maintained , since the spring force dominates the magnetic attractive forces , which then scarcely differ . the piston ( 16 ) can be directed into its stable positions by the orientation of the current through the coils ( 29a , 29b ) of the two electromagnets ( 27a , 27b ). here at any given time one of the two capillary connections ( 26a , 26b ) at the two outlets ( 30a , 30b ) is closed or opened , in an end position of the piston ( 16 ). in the middle position , both capillary connections ( 26a , 26b ) are opened . the extracorporeal service appliance ( 32 ) and the implanted infusion device ( 33 ) are shown schematically in fig6 . the infusion device ( 33 ) has an electronic control device ( 34 ) which controls all activities in the infusion device ( 33 ). this control device ( 34 ) consists essentially of an eeprom ( 35 ), a single chip computer ( 36 ), a serial interface ( 37 ) ( a parallel interface would also be conceivable , but without advantage ), and a port ( 38 ) to which various drivers and sensors can be connected . the serial interface ( 37 ) is connected to a modulator / demodulator ( 39 ) for the absorption telemetry ( 40 ). this absorption telemetry ( 40 ) consists of a lf / rf converter and serves for communication between the infusion device ( 33 ) and the extracorporeal service appliance ( 32 ). the same antenna serves both for energy transmission and also for bidirectional data transmission . there is contained in the service appliance ( 32 ) a coil ( 32a ) via which energy can be inductively transmitted into the infusion device ( 33 ). the transmitted energy is used to operate the single chip computer ( 36 ), the sensors , ( 43 , 44 ) the driver ( 45 ) and the valve ( 42 ). energy coupling takes place only in the presence of the extracorporeal service appliance ( 32 ). two sensors ( 43 , 44 ) are connected to the port ( 38 ) of the single chip computer ( 36 ) and ascertain the position of the valve ( 42 ). the mode of operation of these sensors ( 43 , 44 ) has already been explained in connection with fig2 and 4 . moreover a driver ( 45 ) for a valve changeover unit ( 46 ) is connected to the port ( 38 ). the position of the valve ( 42 ) can thus be changed by means of a corresponding instruction , received from the service appliance ( 32 ) by the receiver in the absorption telemetry ( 40 ) and transmitted via the serial interface ( 37 ) to the single chip computer ( 36 ). the corresponding control characteristic is produced by the single chip computer ( 36 ) and correspondingly acts on the changeover unit ( 46 ). furthermore a filling level sensor ( 47 ) is connected to the port ( 38 ) of the single chip computer ( 36 ), and transmits to the single chip computer ( 36 ) a message concerning the level , i . e ., concerning the quantity of medicament which is still present in the supply container ( 48 ). the single chip computer ( 36 ) calculates therefrom the infusion time still available and tests whether the amount of flow agrees with the values which have been set . additionally , the physician can add fluid into the supply container 48 via a needle 71 . a duct ( 49 ) leads from the supply container ( 48 ) to the valve ( 42 ). the supply container ( 48 ) is constructed as a bellows and is constantly under the pressure of a pressurized gas reservoir ( 50 ). the pressure exerted by the pressurized gas reservoir ( 50 ) on the supply container ( 48 ) drives the medicament contained in this in the direction towards the valve ( 42 ). two throttle paths ( 51 , 52 ) lead from the valve ( 42 ). the throttle paths ( 51 , 52 ) ensure that the supply container ( 48 ) empties only in a controlled manner . they consist of capillaries , and the quantity of liquid flowing through them per hour is dependent on the pressure difference over the throttle paths ( 51 , 52 ), on their capillary length , and on the capillary diameter of the individual throttle paths ( 51 , 52 ); this diameter can be different for each throttle path ( 51 , 52 ). the valve ( 42 ) opens both throttle paths ( 51 , 52 ) or only one . as long as the position of the valve ( 42 ) is not changed , the quantity of liquid delivered per hour remains constant . the two throttle paths ( 51 , 52 ) end in the bolus ( 53 ), at the outlet of which is located a catheter ( 54 ) into the interior of the body of the wearer of the implanted infusion device ( 33 ). the physician can also inject a medicament directly into the bolus ( 53 ) with an injection needle ( 55 ). the service appliance ( 32 ) has a transformer ( 41 ) as a component of the absorption telemetry ( 40 ) for the inductive transmission of energy to the infusion device ( 33 ). this transformer ( 41 ) is supplied by a battery ( 61 ). the battery ( 61 ) is also connected to a microcontroller 56 ( 36 ). an operator , who will normally be a physician , can by means of a keyboard ( 57 ) or by means of function keys ( 58 ) interrogate from the infusion device ( 33 ) the operating data determined by the single chip computer ( 36 ) in the infusion device ( 33 ), or can initiate an adjustment of the valve ( 42 ) in the infusion device ( 33 ). for this purpose , the service appliance ( 32 ) has a data communication device ( 59 ), which consists of a modulator and demodulator for the absorption telemetry ( 40 ). all the operating parameters transmitted from the infusion device ( 33 ), and also the instructions sent to it , can be displayed on a lcd display ( 60 ). the service appliance ( 32 ) thus performs two tasks . firstly , energy is inductively transmitted from the service appliance ( 32 ) into the implanted infusion device ( 33 ) when the service appliance ( 32 ) is brought into the vicinity of the infusion device ( 33 ) ( abdominal wall ). during the transmission of energy , the valve ( 42 ) can be controlled from the service appliance ( 32 ) by means of an absorption telemetry which is located both in the service appliance ( 32 ) and in the infusion device ( 33 ). a single chip computer ( 36 ) is mounted in the infusion device ( 33 ), effects the communication with the service appliance ( 32 ), and can change over as prescribed , and control the functioning of , the internal valve ( 42 ) having three stable operating positions located in the infusion device ( 33 ). no electrical energy is stored in the infusion device ( 33 ) in order to change over the valve ( 42 ). in order to change over the valve ( 42 ) contained in the infusion device ( 33 ), energy has to be transmitted inductively into the infusion device ( 33 ). if the service device ( 32 ) is moved away , the inductive coupling is broken and the implanted infusion device ( 33 ) becomes without current , so that an adjustment of the valve ( 42 ) cannot take place . the flow rate which has been set is however still maintained , since the valve or valves ( 42 ), connected in series or in parallel , remain in their stable operating position ( s ). the possibilities of variation of the flow rate can thus be greatly increased by the use of several valves ( 42 ). when the infusion device ( 33 ) is activated ( i . e ., energy is transmitted from the extracorporeal service device ( 32 )), the filling level can be measured by means of a sensor ( 47 ). from the level and the flow rate which has been set , the remaining time to the next filling can be indicated for the patient on the display ( 60 ) of the service appliance ( 32 ). the monitoring of the flow rate can then advantageously be determined by means of stored level values . eeproms which permanently store the level values are particularly suitable here . from the adjustability and the filling level measurement , and the possibilities resulting from these , a flexibility results which was otherwise reserved to the electronic infusion devices , but without having to accept the disadvantage of the greatly limited life due to battery operation . an absolute insensitivity of the final electrically controlled valve ( 42 ) to interference from external interfering effects is achieved by the realization of the energy coupling according to the invention . in contrast to this , in infusion devices in which the adjustment of the valve takes place by means of the current of an electrical storage device ( i . e ., with a permanent electrical supply , e . g ., a battery ), there is always the danger that these infusion devices are reprogrammed by interference effects from the exterior . in the infusion device ( 33 ) according to the invention , a doubled security is achieved . in the first place , the infusion device ( 33 ) is passive in the normal state . apart from this , data transmission is effected , for security reasons , by means of absorption telemetry . additional security against interfering pulses is realized by an additional coding of the data and instructions to be transmitted . moreover , the absorption telemetry is only effective at close range .
a process and an apparatus for adjustment of a switchable , flow - limiting device for limited flow of liquids or gases employs a remote service device . the flow - limiting device and the service device are spatially separated and there is no bodily connection between the two . energy necessary for adjustment of the flow - limiting device is transmitted from the service device into an infusion pump that contains the flow - limiting device . the flow - limiting device is a valve having three stable operating positional states . the valve can only be switched while energy is being transmitted from the service device .