referring now to the drawing figures , the uvas is mounted on a rolling base 2 to provide portability . fig2 . an adjustable handle 3 is provided for transporting the device . the base includes a box 4 which could measure 30 × 20 cm , and in which is housed circuits , a power supply for the dc components , and the bulb ballasts . a central post rises 6 from the base to an overall height of , for example , 220 cm . around the central post are banks of uv - c emitting bulbs . in the embodiment as shown , six pairs of medium pressure mercury bulbs 8 are present , with each pair positioned equidistant from the pair on each side , so that they are present at 60 ° around the device . the bulbs may be 48 - inch long , 115 - watt germicidal lamps that produce 300 microwatts of ultraviolet radiation at 1 meter . each pair of bulbs is preferred to provide not less than 80 ° of coverage . a control box 10 on top of the unit contains wireless components , the uv - c sensor array , a bank of basic stamps , motion detectors 12 , and audible 20 and visible alarms 22 . a power switch 14 is provided on the exterior of the device . a series of plugs 16 for control functions are also provided . a power cord 18 is provided . referring now to fig4 three door contacts 24 are shown . these contacts are placed in one or more of the doors of the room in which the device is operating . the door contacts are switches which disable the device if nay one of the switches is opened , such as by opening the door . the motion detectors 12 are immediately activated upon activation of the device and prior to powering of the ballasts 26 and the bulbs , by means of the time delay 28 . if the motion detectors sense motion at any time during the operation of the device , power to the ballasts and the bulbs is immediately disabled . a preferred embodiment has three 120 ° passive ultrasonic motion detectors located on top of the device . the uvas is controlled by a series of programmable basic stamps which receive commands from a wireless remote control 30 . a switch activates the remote control . entry of a security code allows the user to begin sending commands to the bank of basic stamps . commands include activate , shutdown , enter sterilization mode , or enter sanitize mode . the remote is in two - way communication with the uvas and displays data from the sensor array , time left to sterilize or sanitize the room , and in case of bulb failure , the status of all numbered bank of bulbs . if two - way communication with the remote is lost , the unit shuts down . the basic stamps activate the motion detectors at least one minute prior to activation of the uv - c bulbs and continue to monitor the detectors . they perform all calculations regarding bactericidal doses , store cumulative dosing data , and system checks to alert the user of bulb failure . this is needed since no one can actually look at the unit to check for burned out bulbs or damaged banks . the stamps can be programmed by attaching them to a personal computer via a serial port connection , thus allowing alteration to the algorithms to accommodate special circumstances . an additional embodiment of the device provides uv - c bulbs or lamps that are placed strategically at various locations within , or directed towards , the space to be treated . the bulbs may be located at two or more locations in a room to be treated . the bulbs may be connected to the base unit , and positioned at remote locations from the base . the device in this embodiment may utilize a single bank of sensors to measure doses at locations within the area to be treated . alternatively , multiple remote sensors , or a combination of remote and central sensors , may be provided to measure and control the dosing of uv - c to the treated area . an example of a protocol for using the device is described . 1 . a user rolls the uvas into the room to be sterilized . after checking the room for bystanders , the user leaves with the remote control . 2 . after securing the room , enters into the remote control the security code , then is prompted to presses an “ on ” switch on the remote control , thus activating the uvas . 3 . the audible voice alarms and the motion detectors activate and stay on until the entire cycle has been complete . should the uvas detect motion , the unit automatically deactivates itself until the user re - enters the room and trips a breaker , thus preventing the user from re - activating the unit and harming a bystander left in the room . 4 . the motion detectors stay on for a preset time , such as one minute , prior to powering the mercury bulbs and then stay active until the cycle is complete and the bulbs are powered down . 5 . the bulbs are powered , and when sufficient time has elapsed to allow the bulbs to reach a steady state output ( one minute or less ), the basic stamp reads data from all the individual sensors located on the array . the array senses 360 degrees at a minimum with overlapping of their window of view . they are oriented away from the uvas , thus measuring the dose of uv - c reflected back to the unit . this data is fed into the microcontroller where it is integrated to compute cumulative exposure of uv - c reflected back from each sensor in the array . 6 . based on the least reflective surface or direction ( of several thousand “ snapshots ”) the microcontroller calculates the time the unit will need to stay activated to allow a bactericidal dose of uv - c to be reflected back to the unit from all directions . 7 . once sufficient time for a lethal dose of uv - c to be reflected back to the unit has elapsed , the unit powers down the bulbs and sounds an “ all clear ” alert to the user . upon completion of the cycle , the unit has sterilized all the exposed surfaces within the room , including the primary shadows such as the back or wall side of all rails , cabinets which are not against the wall , and tables . trials of the uvas in actual operating rooms and endoscopy suites and exam rooms as well as simulated trials have been performed . at direct exposure from two meters , the unit is able to reduce colony counts of common hospital pathogens by a minimum of 99 . 9 % in one minute and achieved sterilization in 10 minutes . in one trial , the back of an exam chair was contaminated with pseudomonas a ., e . coli , and staphylococcus aureaus bacteria from slurries prepared by a hospital microbiology lab . the surface was the cultured for a control prior to using the unit . it was then cultured at ten minutes and twenty minutes after activation of the unit . the test surface was not in direct line of sight of the uvas and received only reflected doses of uv - c . cultures using convex agar plates designed for surface cultures were used and incubated by a hospital microbiology lab . the control cultures grew greater than 100 , 000 colonies of all three species . the ten minute and twenty minute cultures showed no growth , demonstrating the ability of the unit to sterilize surfaces using only reflected doses . the estimated reflection from the wall in the test room was only 3 %. through the use of paint that reflects 50 - 85 % of the uv - c , the efficiency of the device is increased , allowing for decreased exposure times . in most environments , there is a presence of what microbiology labs label as “ wild spore forms ” of bacteria . these bacteria are not known to cause human disease , and yet , are resistant to low doses of uv - c . the dual programming modes of the unit allow treatment as required . one mode ( sanitize ) kills all known pathogens and requires a lower exposure and thus shorter time . the other mode ( sterilize ) kills all species of bacteria and requires greater cumulative doses and therefore more time . the ultraviolet area sterilizer self monitors bactericidal levels . reflected doses of uv - c are measured , and the device remains activated until bactericidal levels are received . this ensures that areas in relative shadow and not in direct line of sight with the unit are sterilized . also , the unit can be set to sanitize ( kill common pathogens ) or sterilize ( kill all microbes ). without adequate safety features , daily use of intense uv - c is dangerous and impractical . the device has motion detectors that assure the room is vacant of personnel prior to activation . hard wired plugs on the unit are available for additional door , window , or other entry monitoring devices special situation may dictate . once activated , the unit shuts down instantly when motion occurs anywhere in the room being sterilized . if the uvas loses two - way communication with the remote control it also shuts down . in daily use , safety protocols commonly used in hospitals such as those in use for laser and x - ray devices may be implemented . the uvas is able to sanitize or sterilize all exposed surfaces in a room . it is able to do so safely , leave no residual toxins or radiation , and generates no adverse environmental side products . in addition , the uvas is able to notify the user of the time required to perform this task and automatically shuts down upon completion of sterilization . the inventor has performed tests to prove the efficacy of the uvas , all of which have been successful . the only limiting factor encountered to date is the reflectivity of some paints and other surfaces which absorb rather than reflect uv - c , requiring prolonged exposures of twenty minutes or greater . highly reflective paints are beneficial to the method of area sterilization disclosed herein .
an ultraviolet area sterilizer is mobile or stationary . the uvas is positioned in a room , such an operating room or intensive care unit . motion detectors sense movement , to assure that personnel have evacuated the space to be sterilized . subsequently , uv - c generators , such mercury bulbs , generate uv - c from multiple locations within the room or other enclosed space . multiple uv - c sensors scan the room , and determine the area reflecting the lowest level of uv - c back to the sensors . the device calculates the time required to obtain a bactericidal dose of uv - c reflected back to the sensors . once an effective bactericidal dose has been reflected to all the sensors , the unit notifies the operator and shuts down .