throughout the following detailed description , the same reference numerals refer to the same elements in all figures . the microendoscopy apparatus 10 shown in fig1 is used for diagnosis and as a delivery conduit of the shock wave pulses generated by a laser or an electrohydraulic device to fracture salivary stones . these concrements or stones 12 can be in the submaxillary gland , seen in fig5 or the stones 14 can be in the parotid gland , shown in fig6 . the microendoscopy apparatus 10 used in this method employs a light source 16 which is linked to a 45 - 50 light guide fiber array 18 , as shown in fig4 . the fibers are encased in an extruded polytetrafluoroethylene or polyurethane jacket 20 . the jacket 20 is prevented from moving by an epoxy encasement 22 . the epoxy 22 spaces apart the various channels or bundles of fibers contained within the microendoscope 23 . the microendoscope 23 also has a polytetrafluoroethylene or polyurethane jacket 25 . a stainless steel tube 27 of 3 mm length at the distal end is designed to encase channels for video capacity . a magnification eyepiece 24 is connected to an image guide bundle 26 containing 4000 - 5000 pixels and the image guide grin lens 28 with a 70 ° degree angle of view shown in fig4 . a focus adjustment 60 is located on eyepiece 24 . an articulation wire 30 attached to a handle 32 is also mounted within the apparatus 10 . lastly , a flush channel 35 is encircled within the epoxy 22 and also acts as the working channel conduit 35 for the working apparatus . a polytetrafluoroethylene covering 34 encloses channel 35 . a branch coupler 38 unites the individual channels for the imagee bundle 26 and light bundle 18 . working channel 35 starts at opening 39 and articulation wire 30 is controlled in the body of the microendoscope 23 by handle 32 . optionally , as seen in fig2 the apparatus 10a includes a television monitor 40 attached to the eyepiece . this monitor 40 connected to a multi - system video recorder 42 and a video printer 44 as separate modules . still further , as seen in fig3 the apparatus 10b includes charge coupled device ( ccd ) 46 attached to the eyepiece 24 and this ccd can be directly connected to a television monitor 48 for viewing the movement of the working channel 35 , within the body of the patient . this whole system with components 48 , 46 and 16 can be enclosed in a housing 47 . the method is carried out by first introducing a dilation probe ( not shown ) into the salivary duct 50 . the papilla is dilated by dilators of increasing size . thereafter , the flexible microendoscope 23 having a diameter of about 1 . 4 - 1 . 6 mm is introduced into the duct 50 and advanced to the concrement 12 under visual guidance through image bundle 26 . the area is illuminated by the light source 16 through the fiber optic 18 light guides . a laser photo conductor of size 0 . 2 mm - 0 . 3 mm is then positioned on the stone 12 via the working channel 35 . the working channel 35 has an outer diameter of 0 . 5 mm . the concrement is fractured by a laser induced shock wave under continuous rinsing and flushing through the surrounding 0 . 3 - 0 . 77 mm area around the photoconductor . stone fragmented particles are either rinsed or drained off or leave the salivary duct by the re - channeled salivary flow . three short - pulse systems can be used . these are the excimer having a wave length of 308 nm and pulse length fwhm 60 ns ; a dye laser having a wave length of 504 nm and pulse length of fwhm 1200 ns ; and an alexandrite laser having a wave length of 755 nm and a pulse length fwhm of 300 to 500 ns . carrying out the method of this invention , the sialolithiasis is performed on out - patient basis while the patient is lying prone . the patient is unsedated . the head is slightly hyper extended to give the examiner a direct view of the oral cavity and the ostia of the salivary duct . after infiltration anaesthesia and administration of a local surface anaesthetic , the probe is introduced into the salivary duct . 1 to 2 ml of an anaesthetic is additionally instilled in the salivary duct via a sialography catheter to block sensory afferences of the area . during the advancement of the flexible microendoscope 23 , isotonic saline solution is flushed in continuously through the working channel 35 . use of the isotonic solution prevents soiling of the lens 28 attached to the silica image bundle at the distal end of the microendoscopic channel 26 in the stainless steel tip 27 . also , a better view of the lumen is gained by distension of the salivary duct . mucous , thickened or purulent secretion is drained via the working channel . secretion plugs can be immobilized , dissolved and drained by controlled forced rinsing . during introduction of the microendoscope 23 into the salivary duct 50 , the articulation wire 30 is controlled by handle 32 so that the tip 27 at the end of the microendoscope 23 can be deflected as necessary by the operating physician up to 90 degrees to position the microendoscope 23 adjacent the stones 12 to be fractured . the same procedure is followed , as shown in fig6 to penetrate the parotid gland duct 56 and fracture stone 14 . a point marker 54 is located at the distal end in tip 27 to provide an orientation marker for the operating physician .
a 1 . 4 - 1 . 6 mm diameter microendoscope is introduced into a duct leading to the submaxillary or parotid glands containing a concrement . the endoscope is adapted to contain a working channel , an image bundle , a light bundle and a wire , all structure fixedly spaced apart within the microendoscope . a laser fiber is inserted into the working channel to the concrement and the concrement is fragmented with a laser shock wave while flushing the fragments away through the working channel .