the processor or apparatus 1 for producing dental prostheses in heat curing acrylic , as shown in fig1 - 4 , comprises an internal cooking boiler or vessel 2 for boiling or heating water with compressed air injection via inlet 4 . caloric energy for the water is provided through an electric resistance heating means 6 . this boiler or vessel 2 is made of 2 mm . thick stainless steel . it comprises a cylinder 8 which is 232 mm . in diameter and 278 mm . high . the side 9 is cylindrical and both the top 10 and the bottom or floor 12 are concave . the top 10 has an oval opening 14 which is 132 mm . long and 105 mm . wide , in which is placed an oval lid 16 which is larger than the opening 14 . the lid 16 is 160 mm .× 130 mm . and its external border 18 has a 7 mm . thick ring 20 contouring it . this ring 20 is made of vulcanized , or heat molded rubber , and is used to prevent the compressed air from leaving when it is injected in the boiler 2 . the boiler 2 has an electric shielded resistance of 220 volts , and 800 watts of power , to heat the water inside the processor 1 . it is used as a means of cooking the heat curing acrylic of the different prostheses which are inside . moreover , as shown in fig4 inside there is a stainless steel table 30 having legs 31 and hinged folding portions 33 . on the table 30 there may be placed three or four curing containers 40 that fit inside the processor 1 for treatment of the plaster chamber 41 . the table 30 is 210 mm . in diameter and about 75 mm . high . the table 30 is perforated with perforations 35 , as shown in fig8 and 10 , to maintain a correct hot water circulation around the curing containers 40 . additionally , as shown in fig8 and 10 , the table 30 can be folded by means of hinges 32 to enable easy removal from the inside of the processor vessel 2 . the curing containers 40 are of stainless steel and , as shown in fig4 and 7 has a cylindrical body 42 , flat bottom 44 , a top edge 45 , and a top opening 46 . two curing containers are 102 mm . in diameter and 60 mm . high and another two are 85 mm . in diameter and 56 mm . high . to measure the air pressure that is injected via inlet 4 inside the vessel 2 , a pressure gauge or barometer 50 is provided in the upper portion of the side of the main cylinder 8 beside the top 10 . it is joined by a welded nipple 52 and a chromium bronze elbow 54 . in the processor two ball valves 56 and 58 connected to inlet 4 and outlet 5 , respectively , were placed for the injection of compressed air and the exhaust of air and steam in opposite positions on the cylinder 8 . a check valve 60 is connected to inlet ball valve 56 as shown in fig3 . a 0 ° c .- 150 ° c . thermometer 65 was also placed on the cylinder 8 to measure the temperature of the water 70 held by the vessel 2 . the water temperature is controlled by a thermostat 72 placed in a vertical box 74 welded to the external wall surface of the processor boiler 2 . this box 74 is 13 cm . long , 9 . 5 cm . wide and 7 . 5 cm . high . all the water 70 used by the processor boiler 2 is about 6 liters and it is used to cure the dental prostheses in heat curing acrylic , because this is the only way to harden the acrylic without burning , maintaining its homogeneous molecular structure and smooth texture . the curing containers 40 are used to place the plaster chamber with their prostheses into the vessel 2 , both for the cold process and for the heat - curing process when heat - curing acrylic is added . when a plaster chamber is placed in a curing container 40 , water has to be added inside up to 1 cm . below the upper edge 45 of the container 40 . in the upper part or top 10 beside the oval opening 14 , a safety valve 80 is placed in case the injected air exceeds the maximum pressure allowed for the process and prostheses . beside any damage to them being avoided , the security of the processor itself is ensured . the top 10 of the processor 1 has an oval lid 16 which is approximately 1 cm . larger than the oval opening 14 in all its perimeter , to adapt to the inside edge 85 of the opening 14 by forming an oval crown 87 that covers the opening 14 from the inside of the vessel 2 . when properly placed with the handle 90 of the lid 16 , it is adjusted on the fixed arm 92 of the processor vessel 2 with a ring 94 that fits in the narrow end part 96 of the arm 92 , fastening the handle 90 over bridge 98 like a spring . as the lid 16 a vulcanized rubber ring 20 , it must be placed obliquely and across the oval opening 14 and afterwards it must be rotated 90 ° until the longer part of the oval lid 16 coincides with the same part of the oval opening 14 . once a close fitting has been obtained , it must be sealed using the ring 94 . the vessel 2 can then be filled with compressed air until a pressure of 50 psi is obtained to carry out either moisturizing a plaster chamber in the cold process or heat - curing a prosthesis by adding heat curing acrylic . to control temperature from the outside and verify correct working of the resistance heating means 6 and the thermostat 72 the 0 ° c . to 150 ° c . thermometer 65 is placed to extend through the cylinder wall 8 . the thermometer 65 is located in a pipe 100 welded to the main wall 8 of the processor boiler 2 and is placed in the center of the resistance heating means 6 to get the best sensitivity to the temperature of the boiling water 70 inside the processor 1 . the thermostat 72 is connected to a thermostat filament 102 which is placed in a pipe 104 and welded to the processor i anywhere , passing through the center , but not reaching the other side , and passing over the resistance heating means 6 . both pipes 100 and 104 are made of stainless steel . as the use of the processor 1 requires a vulcanized rubber lid 16 and temperature is raised gradually to reach working temperature , steam ( water vapor ) is produced , which damages the rubber 20 if the lid 16 is placed on the opening 14 . for this reason , as shown in fig4 and 5 , another thinner stainless steel protective lid 110 with the same shape is made . also , a simple non - vulcanized rubber ring 112 which is fit to the edge 114 of the lid 110 is inserted into the slot 116 of the rubber ring 112 . the protection lid 110 has a small handle 120 and small legs 122 in the obvers in order to be placed correctly in the opening 14 . the function of this lid 110 is only to prevent water vapor from leaving the processor 1 . also , it helps to preserve the main lid 16 which is very expensive . the protective lid 110 is also used after the curing stage of any prosthesis to maintain water temperature , as long as possible until the next work is performed . in the external box 74 where the thermostat 72 and the light switch ( not shown ) are , an indicating neon light 125 has been placed . light 125 turns off when the water 70 is at the right temperature and helps to show that the thermostat 72 is working properly . a plastic pipe 127 for the water or steam had to be added to the exhaust valve ( ball valve ) 58 so that the air and steam can get out together and do not burn the hands of whoever is working at the time . the pipe 127 is approximately 25 cm . long . moreover , a bottle or container may be added to avoid spilling water on the work place . the electric energy that the resistance heating means 6 receives comes from a bipolar rubber protected power cord 129 , to avoid any damage . also , the wires of power cord 129 , which carry energy to the thermostat 72 , are shielded by glass fiber and are covered with a glass - fiber insulation to be isolated from the wall 8 of the processor 1 . for the cold process , or moistening step , the plaster chamber 41 , either complete , semichamber or special , is placed in the curing container 40 with the waxing , and submerged in running water for introduction into the processor 1 at ambient temperature ( 20 ° to 25 ° c .). the temperature is checked on the processor &# 39 ; s thermometer 65 . the quantity of water in the processor 1 should also be checked , and the process is as follows : take a curing container 40 filled with water , introduce it into the processor 1 , and set it on the inside table 30 checking that water 70 reaches up to 1 cm . below the upper 45 edge of the curing container 40 . then remove the corresponding plaster chamber 41 from the curing container 40 submerged in running water , and introduce it into the container 40 in the processor 92 . close the latter with the main lid 16 . then fit the ring 64 of the lid handle 90 with the fixed arm 92 of the processor . turn off the air outlet valve 58 and place the compressor hose ( not shown ) in the air intake valve 56 . inject compressed air up to a pressure of 3 . 5 bars , checking on the barometer 50 that said pressure has been reached . turn off the air intake valve 56 and remove the compressor hose , then turn the compressor ( not shown ) off . set a chronometer for a 3 minute moistening time of the plaster chamber 41 . if said time is surpassed , it will be too damp . after said 3 minutes , turn on the air outlet valve 58 and the barometer 50 will show that the pressure has fallen to zero . remove the ring 94 of the fixed arm 92 and raise the lid handle 90 with the lid 16 . remove the curing container 40 from the processor 1 with the help of long pliers . remove the plaster chamber 41 from the curing container 40 and continue with washings to eliminate wax . for heat curing , the plaster chamber 41 , either complete plaster chamber or special , with the heat curing acrylic already aggregated inside it , is permitted to rest for some minutes on a horizontal flat surface , in the open air and at ambient temperature . it is then varnished and is dried with a dryer . cover all inlets with paris plaster . meanwhile , switch on the processor 1 by turning the thermostat 72 , put the protective lid 110 on the open top 10 , and let the temperature rise until it is between 90 ° c . and 100 ° c . then , check remove the protective lid 110 and the processor conditions to see if the quantity of water inside is correct , because the quantity of water decreases by evaporation due to heat . then add any water to the processor 1 which is necessary to reach a height of 1 cm . from the upper edge 45 of the curing container 40 . to make the height adjustment the curing container 40 is placed , with water and without the plaster chamber 41 , inside the processor 1 onto the table 30 with the help of suitable pliers . then check if the water height 70 inside the processor 1 is one centimeter below the upper edge 45 of the curing container 40 . then , after adding any necessary water to the processor 1 , remove the curing container 40 from the processor 1 . keep the temperature between 90 ° c . and 100 ° c . sometimes , when water is added , the temperature can fall a little and it is necessary to replace the protective lid 110 and wait a few more minutes until the thermometer 65 shows it is at 90 ° c .- 100 ° c . at which time the protective lid 110 is removed for placement of the plaster chamber 41 in the processor 1 . place the plaster chamber 41 in the curing container 40 . add running water to cover the plaster chamber 41 up to its edge . submerge the curing container 40 in the processor 1 with the help of long pliers and put the lid 16 on the processor 1 . then place the ring 94 of the lid handle 90 with the fixed arm 92 of the processor 1 . turn off the air outlet valve 58 and place the compressor hose on the air intake valve 56 . inject compressed air up to a pressure of 3 . 5 bars checking on the barometer 50 that said pressure has been reached . turn off the air intake valve 56 and remove the compressor hose . turn off the compressor &# 39 ; s air valve . set the chronometer for a cooking time of the heat curing acrylic for about 20 to 25 minutes ( 25 minutes is the recommended time for dentures thicker than normal ). after that period of time , turn off the thermostat 72 , unplug the processor 1 and turn on the air outlet valve 58 and the pressure gage 50 will show that the pressure will fall to 0 . remove the ring 94 from the fixed arm 92 and raise the handle 90 with the lid 16 to remove it from the processor 1 . remove the curing container 40 from the processor 1 with the help of suitable long pliers , and leave it at rest for some 10 minutes to reduce the acrylic surface tension , without removing the plaster chamber 41 from the curing container 40 . cool it with running water for 5 minutes until the water is cold and leave it again at rest for some 10 more minutes . remove the plaster chamber 41 from the curing container 40 , and remove the plaster fronts with the help of a blunt knife . eliminate the acrylic entrances with the help of a bur , the two anterior ones and the two posterior ones . remove plaster from the counterpart with a well - sharpened chisel , and break it in pieces upwards . place the model with the cured denture once again in the occlusor to check if the vertical dimension has changed . once all registers have been checked , proceed to the corresponding filing and polishing .
in the present invention dental prostheses are cured without the use of dental flasks using an electrically heated pressure vessel . the dental prosthesis is cured while in a plaster chamber which is placed in a curing container with ambient water . the apparatus includes a perforated table for supporting the curing container during heat curing . the vessel has a lid for sealing the top , an inlet for injection of compressed air , an outlet for exhausting of compressed air , and an electrical resistance heating means for heating water held in the vessel . the curing container holds ambient temperature water at a level which covers the plaster chamber prior to submerging the curing container in the vessel containing water at a temperature of 90 ° c . to 100 ° c . the table supports the curing container in the vessel so that the top edge of the curing container is above the level of water held by the vessel . the present invention provides that the plaster chambers with the dental prosthesis inside of it being already aggregated heat curing acrylic , are those which maintain the vertical dimension . they are within the curing container while they are gradually heated by the hot water of the processor . pressure which could fracture the model and the prosthesis is avoided .