fig1 represents a schematic illustration of a patient 10 undergoing a minimally invasive procedure to treat an aneurysm 12 using the methods and devices of this invention . the patient &# 39 ; s head located near a magnet system 14 . the magnet system 14 is controlled by the physician to generate and apply a magnetic field 16 at the treatment site . the externally applied magnetic fields are described and shown schematically herein but other incorporated references describe these magnet systems in great detail . for example , the incorporated reference describes a suitable magnet system , which could be used to apply the magnetic field and to also image the magnetic device in the body . however , for purposes of this disclosure , the primary purpose of the magnet system 14 is to exert a well - defined and controlled magnetic field and magnetic gradient at the treatment site . throughout the schematic illustrations , directional arrows are identified with labeling to indicate the preferred type of magnetic field and / or gradient for the magnetic device under control . these directional arrows are identified as a magnetic field 20 , general gradient 52 , axial gradient 79 and transverse gradient 81 . these reference numerals are used throughout to identify these magnetic properties . generally , the direction of the resultant attractive force generated by applied magnet is indicated in the figures by the direction of the arrow indicating the external magnetic field . the term general gradient 52 is used where the magnetic field direction with respect to the magnetic gradient direction is relatively unimportant for the process being described . when the magnet system 14 is positioned with axis 18 parallel to the tractive forces , as seen in fig7 the configuration may be referred to as an “ axial ” gradient 79 . when the position of magnet system 14 with the axis 18 is perpendicular to the tractive forces as seen in fig6 the configuration may be called a “ transverse ” gradient 81 . in general , the control of the forces can be achieved either mechanically by moving the magnet system 14 with respect to the patient 10 or electrically by adjusting the currents in an electromagnet . for example , an axial field can be converted to a transverse field by rotation and translation of the magnet with respect to the position of the patient as seen in fig1 . fig2 shows an enlarged view of a portion of the vasculature of a patient containing an aneurysm 22 . in practice , the physician will gain access to the vessel 24 and pass a catheter 26 through an introducer into the patient &# 39 ; s vasculature . the catheter 26 will then be navigated using conventional guidewire techniques or under magnetic control as set forth in the incorporated references , to the treatment site . the physician will position the catheter 26 in the “ neck ” 28 of the aneurysm 22 with or without the aid of the magnet system . with the distal tip 30 of a catheter device 26 located within the neck 28 of the aneurysm , the physician can inject a magnetic device 32 into the cavity . in general a syringe or push wire ( not shown ) is coupled to the lumen 32 of the catheter 26 , and the magnetic device 32 is extruded from the lumen of the catheter by the syringe or push wire . in fig2 the catheter 26 device does not have a distal magnetic tip . therefore , in this instance , the external gradient 52 along with syringe pressure serves to “ meter ” the magnetic device into the aneurysm by the forces resulting from the exterior magnet without the influence of any local magnetic field . in this particular drawing , the magnetic device 32 is a so - called “ ferro - fluid ”. the ferro - fluid type of magnetic device 32 may take any one of numerous formulations . in general , the material will be sufficiently viscous to be deliverable at reasonable flow rates trough a small catheter lumen under the pressure of a physician operated syringe . it has been found that the ferro - fluids used in experiments stiffen under the application of an external field . this effect must be taken into account during formulation . experimentation has been preformed with carbonyl iron compounds in a cyanoacrylate carrier . in this combination , the iron material is in particulate form and the cyanoacrylate acts as a glue to solidify the composite material in the aneurysm . it is likely that temperature activated and light activated glues may be combined with magnetic particles to improve the solidification of the ferro - fluid . it is also anticipated that other diluent materials may be added to the ferro - fluid to modify and enhance the delivery properties of the material . therefore , in this context , ferro - fluid should be understood to encompass microscopic particulate material , which is combined with a binder or glue agent which causes the ferrofluid to congeal into a solid mass after delivery . both magnetic and magnitisable materials are operative and both cyanoacrylate and albumen based glues are operable . it is believed that the number of candidate materials for ferro - fluid is large and may include for example , hydrogels or copolymers mixed with the magnetically active particles . throughout this specification , the term ferro - fluid is used to denote a material which is magnetically active and which is injected into the aneurysm or vascular defect . it should also be understood that other materials may be present in high concentrations to improve the delivery or reactive characteristics of the ferro - fluid . embolization times and or solidification times of solidifying or adhesive agents vary considerably . agents with shorter solidification times are most suitable . to improve the predictability of the solidification time , an agent could be mixed with the magnetic particles and activated with an external source . the external source could consist of ultra - violet light or radio frequency source . however , this method is not limited to a particular source used to activate the agent . the ferro - fluid embodiments illustrated in fig2 , 5 , 6 , 7 , have microscopic magnetic particulate . larger particles may be used as well but the geometry of the occlusion process differs from that associated with smaller particles as discussed in connection with fig3 . fig3 shows a catheter 26 device for delivering discrete magnetic objects or particles typified by particle 50 . the discrete objects may be as large as the catheter lumen 42 . these particles may be sequentially inserted into the aneurysm . an external magnetic field 52 attracts and groups the individual objects , causing a single solid object 54 to form . retention of the solid object 54 type of magnetic device , in the aneurysm reduces or prohibits blood flow through the aneurysm , causing it to clot . the shape of the magnetic objects can be spherical as shown or they may be cylindrical . however , many other shapes that are easily passed through a catheter device could be used as well . these magnetic devices may be coated with an adhesive forming a large particulate ferro - fluid . given the relatively large volume of each device , the distribution of magnetic material may be non - uniform in the particle . alternatively , the magnetic portion of the device may be concentrated and symmetrically placed . for example , the magnetic portion may be a small sphere of hiperco coated with a thick layer of hydrogel or the like . for use in treating avms as seen in fig1 the magnetic devices may have a coating such as polyvinyl alcohol ( pva ) enclosing a magnetically active material such as magnetite or hiperco . fig1 shows a catheter delivering pva coated magnetic particles into an avm 56 . in this particular procedure , the magnet gradient 52 attracts the magnetic devices typified by particle 58 along a path that lodges the particle in the branch 60 selected by the physician . in this instance , the external magnetic gradient is used along with blood flow in the vessel to direct the magnetic devices to the desired branch after they have left the distal tip of the catheter 26 . fig4 shows another instance of a ferro - fluid type of magnetic device 32 used to fill an aneurysm 22 , under the influence of an external magnetic gradient 52 . however , in this instance the catheter body 26 includes a magnetically active distal tip 29 . this tip 29 may be the most distal segment of the catheter body or it may be near the distal tip . in either event , the magnet or magnetic material 29 serves to constrict and control the flow of ferro - fluid out of the lumen 30 . in this procedure , the flow of blood around the interior of the aneurysm is insufficient to overcome the tractive force holding the magnetic device 32 in the cavity and there is little risk of the material escaping from the cavity . depending upon the glue or adhesive system used to prepare the ferrofluid , the external magnet remains on and in position long enough to allow the ferro - fluid to “ set up ” and become a solid mass . once the ferro - fluid material has set , the magnet may be removed . fig5 shows a catheter device 26 similar to that of fig4 but the relationship between the magnetic forces differs . in this embodiment the magnet structure 29 at the tip is relatively stronger than the applied force . in this situation the external magnetic field is “ weaker ” than the magnetic gradient 52 as in fig4 . in this instance , the ferro - fluid 32 is attracted to the distal magnetic tip structure 29 and forms a bloom 19 around the tip . the catheter would remain in place until the ferro - fluid 32 forms a solid or adheres to the aneurysm wall . using this method , it may be important to ensure that the ferro - fluid does not bond or adhere to the catheter tip . however , other methods could be incorporated to leave the distal magnet in the aneurysm following the injection of the ferro - fluid surrounding the magnet . in addition , the externally applied magnetic gradient 52 could be eliminated with the distal magnet 29 supplying the entire attractive force to keep the ferro - fluid within the aneurysm . fig6 illustrates the application of a transverse gradient 81 to the ferro - fluid 32 as it exits the catheter 26 . in this instance , the magnetic field and the magnetic gradient are perpendicular resulting in a force , which tends to cause the ferro - fluid to form layers or lamella in the aneurysm 22 . with the application of a transverse gradient 81 at the aneurysm , the ferro - fluid 32 forms into structured layers or lamella as typified by layer 27 . in this instance , the layers , typified by layer 27 , adhere to one another and align with the field of the external magnet . it is preferred to align the layers perpendicular to the neck of the aneurysm to sufficiently “ pack ” the aneurysm and to prevent the material from forming in the nearby vessel branch . fig7 illustrates the application of an axial gradient 79 to the ferro - fluid 32 as it exits the catheter 26 . in this instance , the magnetic field and the magnetic gradient are parallel resulting in a force , which tends to cause the ferro - fluid to form peaks in the aneurysm 22 . these appear as columns of closely spaced structures in the aneurysm typified by structure 66 . utilizing this method , a smaller volume of ferro - fluid can effectively reduce the blood flow into the aneurysm , causing a clot to form . the choice between the use of a “ transverse ” or “ axial ” gradient or some intermediate field is essentially a medical choice driven in part by the volume and shape of the aneurysm as well as the relative size of the “ neck ”. fig1 illustrates the use of a ferro - fluid type magnetic device 61 to treat an avm 59 . in this instance , the catheter 26 ejects a stream of ferro - fluid that is magnetically solidified forming a “ toothpaste - like string ” 61 which is advanced into the branch 60 where occlusion occurs . once again the external field 52 can both direct and retain the ferro - fluid in a structured form while the adhesive material in the ferro - fluid composition sets up . using this approach , an adhesive or polymer agent can be administered to the avm without concern for the agent passing through the avm and appearing in the general circulation . the various ferro - fluid embodiments allow the physician to fill the vascular defect with a selectable amount of material that typically adheres to or conforms to the shape of the defect . that is , the granularity of the particulate is so fine that an excellent match between the shape of the defect and the shape of the treatment device can be achieved , along with a matching of volume as well . coil systems do not possess this feature . fig9 shows a problem present with prior - art , coil placement systems . it is common to have the coil 40 curl up and pass out of the neck 41 of the aneurysm 22 reentering the vasculature near the neck as seen by the position of the distal tip 44 of the coil 40 . in this prior - art system , the physician will retract the coil 40 back into the catheter 26 and try to twist and reposition the catheter to prevent the reoccurrence of the event . the physician may also retract and replace the coil with a smaller diameter coil to enable proper coil delivery . fig8 shows a coil type magnetic device that includes a magnetic or magnetically active tip structure 70 mounted on a coil 72 . the coil may be platinum or another non - magnetic material . the coil is temporarily attached to a push wire 74 . the push wire is used to advance the coil from the distal opening of a catheter to retrieve the coil back into the catheter . this form of magnetic device may be biased into a straight form as seen in the drawing or it may be biased into a preformed curve shape , which may be advantageously used to direct the device into aneurysms . typically coils are biased into curvilinear shapes to assist in placement . this particular construction is well suited for magnetically winding a coil using an external magnetic field in aneurysms . fig1 shows a magnetic device of fig8 being pushed out of a catheter 26 . the externally applied field 20 aligns the tip of the coil towards the center of the aneurysm , permitting the coil to be tightly wound into the aneurysm . this is shown in the drawing by the decreasing radius of the coil . this feature overcomes the “ first coil ” problem explained in connection with fig9 . fig1 shows a method for magnetically holding coils in an aneurysm . this magnetic coil device 78 , is formed from a magnetiseable material such as “ hiperco ” or 400 series stainless steel in a form similar to the coil 72 of fig8 . this magnetic device 78 is held in the dome of the aneurysm by the magnetic gradient 52 of the applied field . in this instance , the wire of the coil has an affinity for itself and the complex form taken by the wire is three - dimensional and essentially fills the entire dome of the aneurysm 22 . this method for magnetically holding coils within an aneurysm is particularly useful for treating aneurysms with large necks . standard coils cannot be held within large neck aneurysms without applying a balloon to close off the neck . the balloon method presents a greater risk that can be obviated by the use of magnetically held coils . once a number of coils are inserted into the aneurysm , the interlocked coils are mechanically maintained in the aneurysm without the need for a sustained magnetic gradient . it is important to note that this approach allows for the reconstruction of a vascular defect that may not be treatable with more conventional coiling and clipping techniques . fig1 shows a number of coil type magnetic devices used to treat an aneurysm 22 . coil 84 and coil 82 are within the catheter 26 while coil 80 has been detached into the aneurysm 22 . winding multiple coils allows for the use of preloaded fixed length devices . the application of an external gradient creates sufficient force 52 that the coil 80 is detached from the remaining sequence of coils by overcoming the force of attraction between the magnet on coil 83 on coil 80 . using this approach , a number of coils can be inserted into an aneurysm and individually released without the need to load a new coil with push wire into the catheter between each coil release . this process speeds the treatment of a vascular defect . fig1 shows a use of a torque by an external magnetic field to detach a coil 80 from the complementary coil 82 . in this instance the catheter 26 retains the coil 82 and resists the motion induced in magnet 70 by the external field 20 . fig1 shows a use of a magnetic gradient 52 to detach a coil 80 from the push wire 74 . in this instance , the catheter 26 retains the coil 80 allowing the coupling force between magnet 70 and the push wire to be overcome by the gradient force 52 from the externally applied magnet . fig1 and fig1 should be considered together as they illustrate the same procedure . in fig1 a coil 80 is magnetically wound , crossing the neck 25 of the aneurysm several times . in fig1 , the view shows the cross - section of the neck 25 and aneurysm 22 as seen from inside the vessel 23 . by placing one large loop in the aneurysm crossing the neck several times , the neck 25 is almost completely blocked off without “ filling ” the aneurysm in the conventional sense . it is believed that the reduction in blood flow into and out of the aneurysm due to this blockage will be sufficient to encourage the formation of an embolism without the necessity of completely filling the aneurysm with coils . fig1 is a schematic view of a magnetically actuated detachment mechanism used to uncouple a magnetic device , such as coil 72 , from a push wire 73 . in this magnetic device , the push wire terminates in a cap 71 that is positioned near a first cantilevered latch arm 77 and a second latch arm 36 near the coil 72 . a magnet 75 is positioned for rotational motion near the cap 71 . fig2 shows an external magnetic field indicated by arrows 20 acting on the magnet 75 tilting it to force the latch arms to expand releasing the coil 72 . although this mechanism is shown on a push wire to coil connection , it can be used on a sequence of coils as seen in fig1 . fig2 shows an external energy source 47 delivering energy 48 to a ferro - fluid 32 . the gradient 52 retains the bolus of ferro - fluid in the aneurysm 22 while the energy 48 catalyses the reaction to solidify the ferro fluid 32 . the catheter 26 is retracted so that it does not adhere to the ferrro - fluid . it should be apparent that many modifications to the devices and delivery system are within the scope of this disclosure , and that these changes can be made without departing from the scope of the claims .
device and method for treating vascular defects by filling the defect with magnetically controlled devices . vascular defects include aneurysms and arteriovenous malformations , particularly in the head and neck of the patient . the magnetic devices fall into categories of ferro - fluids , magnetic objects and magnetic coils . ferro - fluids consist of a magnetic particulate that is combined with a glue or polymer agent . the ferro - fluid is magnetically held and solidified , while the adhesive agent cures . magnetic objects such as polyvinyl alcohol coated magnetic particles can be injected into a vascular defect such as an avm . the magnetic objects are attracted into pre - selected avm branches by the application of a magnetic gradient . magnetic coils are magnetically steered and held within an aneurysm to improve coil delivery . large neck aneurysms are particularly assisted by this method , due to the need to steer the coil towards the center of the aneurysm and to hold the coils within the aneurysm . various methods of magnetically controlling these devices are described to cover a broad range of new treatment approaches .