disclosed herein are a method and system which combine gold nanorods , near infrared light and minor medical procedures to reduce and remove fatty tissue . by injecting a small volume of a solution of gold nanorods into the targeted area , the invention provides for the melting of fat ( lipolysis ) and the tightening of skin upon illumination using a low power , biologically benign near infrared ( nir ) laser . fig1 illustrates the process flow for the inventive method , with each process step linked by an arrow to a diagrammatic image of the step as performed on a target area of a patient . the flexibility in the laser diameter , shape and intensity allows precise control over the target area , which may vary from very small , on the order of a few millimeters , to relatively large , e . g ., several centimeters in diameter . in step 102 , the physician administers a subcutaneous injection into the target area of a solution of gold nanorods ( gnrs ) suspended in a sterile , inert liquid , e . g ., distilled water , using a fine syringe . in step 104 , the gnr solution diffuses through the adipose tissue to be targeted immediately after injection , or as soon as practically possible , nir laser light is focused onto the target area ( step 106 ) for a period that may range from a few seconds to several minutes , depending on the area and volume of the targeted fat , and at least for a sufficient period of time to induce surface plasmon resonance within the gnrs . the laser light has a wavelength within the range of 600 nm to 950 nm , preferably within the range of 700 nm to 900 nm , and more preferably about 800 nm . in step 108 , spr is induced , producing localized heating which , in step 110 , causes the solid fat to liquefy . finally , in step 112 , the physician inserts a syringe into the targeted area to aspirate the liquefied fat . a similar procedure may be used to heat and thus stimulate the surrounding skin to minimize sagging after adipose tissue removal . in such a procedure , the gnr solution may be applied directly to the skin or injected intradermally prior to irradiation by the nir laser light . fig2 is a representative schematic diagram of the components of the lipolysis system 10 of the present invention . the gnrs 8 ( in solution ) are injected into the target tissue 20 using syringe 24 . the gnrs are preferably suitable for in vivo use , for example , a polymer coating can be added for long circulation . the gnr &# 39 ; s should be sterilized and certified endotoxin - free . the nir laser energy 6 from the energy source 14 is directed into delivery device 16 via a delivery channel 18 , which may be a fiber optic , articulated arm , or other appropriate optical waveguide . in preferred embodiments , the nir laser is tunable to allow selection of a wavelength that is optimized for different size gnrs . the laser should preferably have adjustable power to modulate the degree of heating . control system 22 provides a user interface for use by the physician , or assisting nurse or technician , to select the appropriate laser wavelength , intensity , duration and other parameters that may affect the treatment . at the distal end of delivery device 16 is an energy directing means 28 for directing the pulsed energy toward the surface tissue 12 overlying the target tissue ( fat ) 20 . the directing means 28 may be one or more optical elements such as a lens or other focusing element , beam shaping optics , slits , apertures , gratings , an array of lenses and other optics or other focusing configuration , which focuses the beam within the targeted volume of fat containing the gnrs . in a preferred embodiment , the optical elements may include beam expanding lenses to allow adjustment of the beam spread to cover different size target areas . following irradiation of the gnrs in the fatty tissue to liquefy the fat 20 , the liquid is aspirated using syringe 26 that is inserted into the pocket of liquefied fat . the invention further includes a kit for performing lipolysis in conjunction with an existing nir laser unit . the kit includes the gnrs 8 in solution and syringes 24 and 26 . the syringe for extracting the liquefied fat may be replaced by a fine cannula connected to a vacuum source that is capable of generating suction at the distal end of the cannula sufficient to draw the liquefied fat from the target area and into a collection vessel . the inventive technique is possible because nir light of low power is minimally absorbed by endogenous components in the body , such as skin , water , hemoglobin . furthermore , low power near infrared light does not cause photodamage to tissue . nir light is currently used for imaging using indocyanine green ( icg ), an fda approved imaging agent able to absorb and emit in this region . while skin and adipose tissue do not absorb the nir wavelengths , gnrs do , enabling fine tuning of the spatiotemporal parameters of heating . because the fat is actually liquefied , the inventive method for lipolysis has the further advantage of being able to use needles or cannulas that are much smaller in diameter ( on the order of 16 or 18 gauge ) than those required for conventional liposuction , thus reducing patient discomfort , minimizing the risk of damage to surrounding tissue , reducing the risk of scarring and infection , and accelerating healing at the site of the procedure . another major improvement over the prior art methods is the duration of treatment . the highly selective and rapid heating produced by the excited gnrs is capable of producing the desired results within minutes , in contrast with the multiple hours required by typical liposuction procedures . to demonstrate the selective photothermal melting of fat , we performed experiments on a ˜ 2 mm layer of butter sandwiched between two slides separated by a silicone spacer small . gold nanorods ( gnrs ) were procured from nanopartz ™, specifically “ ntracker ™ for in vivo therapeutics ” gold nanorods coated in a proprietary dense layer of hydrophilic polymers , with 10 nm axial diameter and 42 nm length . according to information provided by nanopartz , at this aspect ratio , the plasmon absorption peaks are at 817 nm and 512 nm . laser heating was conducted on butter samples with and without gnrs using an unfocused (˜ 2 mm diameter ) 800 nm beam from a ti - sapphire ( 100 fs , 80 mhz ) laser . the gnr - butter samples were prepared from a mixture of 10 μl of 3 × 10 12 gnr / ml with ˜ 50 mg of butter . melting was monitored by visual inspection . the melting point of butter is 32 - 38 ° c . and its specific heat is ˜ 5 joules / g ° c . this means that with the ˜ 2 mm diameter beam at 800 nm at 0 . 45 w power ( 14 w / cm 2 ), the illuminated butter sample should heat at a rate of approximately 2 degrees every second . the input heat and resulting heating rate is likely less in actuality because of absorption of the microscope slide glass . the butter sample used in these experiments shows no absorption in the region of the laser illumination wavelength , 800 nm , as shown in fig3 a and 3b . the primary contribution to absorption is the fatty acids in the milk fat , which absorb in the visible range of the spectrum . the opacity of the sample limits the transmission of light through the butter so the optical density is high , as shown in the plot of fig3 a . if the contribution of the light scattering to the spectrum is removed , the absorption due to the butter can be better visualized , as shown in fig3 b . experiments on a plain butter sample showed that melting does not occur after 3 minutes , shown in the photos of fig4 , and up to 10 minutes , shown in fig5 a , of illumination with a 0 . 45 w laser beam . in the case of the gnr - butter sample under similar experimental conditions , melting of the butter was observed in the area irradiated by the nir laser beam after 2 . 5 minutes of illumination . fig5 b shows the butter before and after irradiation . testing was also performed on bacon samples to compare the heating behavior in fat versus meat . we added 10 μl of 3 × 10 12 gnr / ml in water onto the fatty sections of the bacon and illuminated the treated sections with a ˜ 2 mm diameter 800 nm beam at 2 . 5 w power . melting of the gnr - injected fat was observed after 45 sec in the volume traversed by the laser beam where gnrs were present . illumination was maintained for a total of 1 . 5 min to further melt the fat and determine whether charring can occur when high temperatures are attained . as shown in fig6 a , charring was observed . the melted fat ( grease ) became so hot that it splattered around the fat sample , indicated by the arrows in the figure . control experiments on similarly irradiated non - gnr fat showed no melting ( fig6 b ). after irradiation , the fat had the same appearance as non - irradiated samples . the irradiated meat sections without gnrs were similarly unaffected ( fig6 c ). these results demonstrate the highly selective nature of the heating in the gnr - injected areas of fat versus untreated areas . experiments indicate that a solution of approximately 3 × 10 12 gnr / ml in water would be an effective injectable photothermal agent for melting adipose tissue upon irradiation with a nir laser as a prelude to in - vivo fat removal . for the removal of 50 ml of fat , less than 10 ml of the gnr may be required . at the price of $ 500 per liter of 3 × 10 12 gnr / ml , the method provides an affordable alternative to conventional liposuction approaches . the application of this technology has many secondary benefits in addition to the cosmetic effect of eliminating body fat . for example , illnesses such as diabetes mellitus are directly related to fat storage and obesity . insulin resistance can be eliminated by reducing body fat content . this scientific fact has significant implications on chronic illnesses such as diabetic nephropathy , diabetic retinopathy and coronary heart disease . to date , existing techniques have not exhibited the ability to remove an effective amount of fatty tissue without causing severe damage to adjacent tissue . in addition , during existing procedures , patients are exposed to the potentially dangerous effects of lidocaine toxicity , which is included in current tumescent solutions . the controlled thermal lipolysis protects all other vital structures , reducing post operative pain and , hence , reducing the amount of lidocaine needed in a tumescent solution and avoid life - threatening risks of lidocaine toxicity . the fact that no - to - minimal mechanical force is required to practice the inventive technique further eliminates the risk of penetrating deep tissues . penetration of tissues such as bowels , livers and lungs has been reported in the literature with use of excessive force to achieve adequate liposuction . 1 . r . weissleder , a clearer vision for in vivo imaging , nat . biotechnol . 19 ( 2001 ) 316 - 317 . 2 . s . j . oldenburg , j . b . jackson , s . l . westcott , n . j . halas , infrared extinction properties of gold nanoshells , appl . phys . lett . 75 ( 1999 ) 2897 - 2899 . 3 . l . r . hirsch , r . j . stafford , j . a . bankson , s . r . sershen , b . rivera , r . e . price , j . d . hazle , n . j . halas , j . l . west , nanoshell - mediated near - infrared thermal therapy of tumors under magnetic resonance guidance , proc . natl . acad . sci . u . s . a . 100 ( 2003 ) 13549 - 13554 . 4 . y . n . xia , p . d . yang , y . g . sun , y . y . wu , b . mayers , b . gates , y . d . yin , f . kim , y . q . yan , one - dimensional nanostructures : synthesis , characterization , and applications , adv . mater . 15 ( 2003 ) 353 - 389 . 5 . v . p . zharov , v . galitovsky , m . viegas , photothermal detection of local thermal effects during selective nanophotothermolysis , appl . phys . lett . 83 ( 2003 ) 4897 - 4899 . 6 . j . lee , a . o . govorov , n . a . kotov , nanoparticle assemblies with molecular springs : a nanoscale thermometer , angew . chem . int . ed . 44 ( 2005 ) 7439 - 7442 . 7 . w . huang , w . qian , m . a . el - sayed , gold nanoparticles propulsion from surface fueled by absorption of femtosecond laser pulse at their surface plasmon resonance , j . am . chem . soc . 128 ( 2006 ) 13330 - 13331 . 8 . h . takahashi , t . niidome , a . nariai , y . niidome , s . yamada , gold nanorodsensitized cell death : microscopic observation of single living cells irradiated by pulsed near - infrared laser light in the presence of gold nanorods , chem . lett . 35 ( 2006 ) 500 - 501 . 9 . j . y . chen , d . l . wang , j . f . xi , l . au , a . siekkinen , a . warsen , z . y . li , h . zhang , y . n . xia , x . d . li , immuno gold nanocages with tailored optical properties for targeted photothermal destruction of cancer cells , nano lett . 7 ( 2007 ) 1318 - 1322 . 10 . j . chen , f . saeki , b . j . wiley , h . cang , m . j . cobb , z . y . li , l . au , h . zhang , m . b . kimmey , x . d . li , y . xia , gold nanocages : bioconjugation and their potential use as optical imaging contrast agents , nano lett . 5 ( 2005 ) 473 - 477 . 11 . p . k . jain , k . s . lee , i . h . el - sayed , m . a . el - sayed , calculated absorption and scattering properties of gold nanoparticles of different size , shape , and composition : applications in biological imaging and biomedicine , j . phys . chem . b 110 ( 2006 ) 7238 - 7248 . 12 . s . link , m . a . el - sayed , shape and size dependence of radiative , non - radiative and photothermal properties of gold nanocrystals , int . rev . phys . chem . 19 ( 2000 ) 409 - 453 . 13 . b . g . prevo , s . a . esakoff , a . mikhailovsky , j . a . zasadzinski , scalable routes to gold nanoshells with tunable sizes and response to near - infrared pulsed - laser irradiation , small 4 ( 2008 ) 1183 - 1195 . 14 . g . wu , a . mikhailovsky , h . a . khant , c . fu , w . chiu , j . a . zasadzinski , remotely triggered liposome release by near - infrared light absorption via hollow gold nanoshells , j . am . chem . soc . 130 ( 2008 ) 8175 - 8177 . 15 . d . v . volodkin , a . g . skirtach , h . moehwald , near - ir remote drug release from assemblies of liposomes and nanoparticles , angew . chem . int . ed . 48 ( 2009 ) 1807 - 1809 . 16 . x . h . huang , p . k . jain , i . h . el - sayed , m . a . el - sayed , plasmonic photothermal therapy ( pptt ) using gold nanoparticles , laser med . sci . 23 ( 2008 ) 217 - 228 . 17 . lynch , d . j ., iverson , r . e ., and the american society of plastic surgeons committee on patient safety . practice advisory on liposuction . plast . reconstr . surg . 113 : 1478 ; discussion 1491 ; discussion 1494 , 2004 . 18 . beran , s . j ., and rohrich , r . j . body contouring ( overview ). select . read . plast . surg . 8 : 1 , 1998 . 19 . weniger , f . g ., calvert , j . w ., and newton , e . d . liposuction of the legs and ankles : a review of the literature . plast . reconstr . surg . 113 : 1771 , 2004 . 20 . pitman , g . h . liposuction and body contouring . in s . j . aston ( ed . ), grabb and smith &# 39 ; s plastic surgery , 5th ed . philadelphia : lippincott - raven , 1997 . 21 . fodor , p . b ., and watson , j . p . wetting solutions in ultrasound - assisted lipoplasty . clin . plast . surg . 26 : 289 , 1999 . 22 . klein , j . a . tumescent technique for local anesthesia improves safety in large - volume liposuction . plast . reconstr . surg . 92 : 1085 , 1993 . 23 . pitman , g . h ., aker , j . s ., and tripp , z . d . tumescent liposuction . clin . plast . surg . 26 : 289 , 1999 . 24 . rubinstein , e . f . an anesthesiologist &# 39 ; s perspective of lipoplasty . clin . plast . surg . 26 : 423 , 1999 . 25 . brown , s . a ., lipschitz , a . h ., kenkel , j . m ., et al . pharmacokinetics and safety of epinephrine use in liposuction . plast . reconstr . surg . 114 : 756 , 2004 . 10 . friedberg , b . l . liposuction “ conscious sedation ” monitored anesthesia care and level of consciousness monitoring ( letter ). aesthetic plast . surg . 29 : 59 , 2005 . 26 . rohrich , r . j ., beran , s . j ., and fodor , p . b . the role of subcutaneous infiltration in suction - assisted lipoplasty : a review . plast . reconstr . surg . 99 : 514 , 1997 . 12 . commons , g . w ., halperin , b ., and chang , c . c . large - volume liposuction : a review of 631 consecutive cases over 12 years . plast . reconstr . surg . 108 : 1753 , 2001 . 27 . horton , j . b ., reece , e . m ., broughton , g ., janis , j . e ., thornton , j . f ., and rohrich , r . j . patient safety in the office - based setting . plast . reconstr . surg . 117 : 61e , 2006 . 28 . keyes , g . r ., singer , r ., iverson , r . e ., et al . analysis of outpatient surgery center safety using an internet - based quality improvement and peer review program . plast . reconstr . surg . 113 : 1760 , 2004 . 29 . trott , s . a ., beran , s . j ., rohrich , r . j ., kenkel , j . m ., adams , w . p ., jr ., and klein k . w . safety considerations and fluid resuscitation in liposuction : an analysis of 53 consecutive patients . plast . reconstr . surg . 102 : 2220 , 1998 . 30 . fodor , p . b . power - assisted lipoplasty versus traditional suction - assisted lipoplasty : comparative evaluation and analysis of output ( letter ). aesthetic plast . surg . 31 . gingrass , m . k . lipoplasty complications and their prevention . clin . plast . surg . 26 : 341 , 1999 . 32 . kim , j ., and stevenson , t . r . abdominoplasty , liposuction of the flanks , and obesity : analyzing risk factors for seroma formation . plast . reconstr . surg . 117 : 773 , 2006 . 33 . rohrich , r . j ., broughton , g ., horton , b ., lipschitz , a . h ., kenkel , j . m ., and brown , s . a . the key to long - term success in liposuction : a guide for plastic surgeons and patients . plast . reconstr . surg . 114 : 1945 , 2004 .
a system and method are provided for minimally - invasive lipolysis in a target area by injecting the area with a solution of photo - absorbing nanoparticles and irradiating the injected area with a beam of near infrared light . the nir emission wavelength is adapted to excite the nanoparticles to melt fat within the target area so that the liquefied fat can be aspirated from the target area . the nanoparticles may be gold nanorods having aspect ratios selected to produce surface plasmon resonance when irradiated with nir light around 800 nm .