삼우 과학 - IFT 800 
 Lambert IFT 800 , Intensified CCD camera 
 LI - μ CAM camera head 
 LI - μ CAM mounted on a microscope 
 Lambert IFT 800 
 The IFT 800 is a sensitive low light level camera , based on an image intensifier fibre - optically coupled to a frame transfer CCD sensor . Programmable logic is used to implement several modes of operation . 
 Low light level performance : 
 In the standard video mode the camera acquires images at light levels between 3 microlux and 0 . 5 lux . This range can be further extended to lower light levels by applying the long time integration mode . At high light levels the dynamic range can be extended to daylight conditions by using an auto - iris lens . 
 For applications that require a high frame rate , a sub - frame corresponding to the region of interest , can be selected . The sub - frame rate is inverse proportional to the number of TV lines within this region . 
 The choice of different types of image intensifiers that can be applied in the IFT 800 camera results in flexibility to satisfy customer specifications , such as spectral sensitivity of the photocahtode , gain , phosphor characteristics . 
 The IFT 800 is a CCIR compatible intensified CCD camera . The camera can be operated stand - alone when connected to a 12 V dc power supply and a video monitor . Adjustments and connections are made on the camera backpanel . The camera can be remotely controlled via the DB 25 � REMOTE ? connector on the backpanel . 
 - general Low Light Level image acquisition 
 LI - μ CAM digital intensified camera 
 The LI - μ CAM is a versatile Imaging System for : 
 - Low Light Level Image Acquisition 
 - Triggered and Pulsed Experiments 
 - Calcium Ratio Measurements 
 - Fluorescent Lifetime Imaging 
 The technology of the LI - μ CAM : 
 The LI - μ CAM camera has its own 10 bits frame buffer , digital and standard analog video output and is fully computer controlled via an IEEE 488 interface to run it in a large number of modes . Together with its high sensitivity and fast shuttering possibilities , this makes the LI - μ CAM extremely useful for the most demanding scientific applications . 
 - Low Dark Current with optional Peltier cooling 
 - Low Read - out Noise and 10 bits dynamic range 
 - High speed ROI mode up to 1000 frames / s 
 - Ultrafast shuttering by gating intensifier 
 - Long Time Integration 
 - Acquisition Software for Windows 
 - Image Pro Plus driver 
 data sheet ( . pdf )