Input logic signals can be displayed as waveforms at 10 MS/s sampling resolution. Each port can measure 8 bits, and 2 ports of input are available on each module. Thus, a single module can observe 16-bit logic signals. Compatible with 700986, 700987, 702911 and 702912 logic probe.
Applicable to DL350 and DL850 series.
Input logic signals can be displayed as waveforms at a sampling resolution of up to 10 MS/s. Each port can measure 8 bits, and 2 ports of input are available on each module. Thus, a single module can observe 16-bit logic signals. You can turn display of individual bits ON/OFF, and give each bit its own unique label.
In all, you can input and display up to 128 bits of logic signals by installing up to 8 modules in a single DL850/DL850V. You can select from 4 different types of logic probe to best fit the input signal. For details on logic probes, see page 6 of this catalog. Also see the "Bulletin DL850-00E" catalog for logic probe adaptors and other accessories.)
The input format, maximum input voltage, threshold level, and other settings depend on the logic probe being used. Please refer to the specifications of your logic probe. The following logic probes can be used.
Logic Probe 702911/702912
High-Speed Logic Probe 700986
Isolation Logic Probe 700987
Isolated/Non-Isolated
Non-isolated Non-isolated Isolated
Max. Input Voltage
±35V 30Vrms 250Vrms
Threshold Level
Approx. 1.4 V Approx. 1.4 V 6VDC or 50VAC
3 µs or less 1 µs or less 1ms or less(DC), 20ms or less(AC)
Plug-in modules specifications ScopeCorder series DL850E/DL850EV/DL350/SL1000 (3.8 MB)