Ultrasonic cleaner is then switch on according to the CSSD manual procedure.
At the end of the cycle the vials are visually checked for the color change from blue/green to yellow indicating positive result. NOTE: Strong ultrasonic energy can lead to a clear ULTRACHECK which is also a positive result.
The time needed for the color change will provide additional information regarding the ultrasonic energy.
The results are than recorded in the ULTRACHECK's documentation sticker.
In case of unsatisfactory results please refer to the trouble shooting guide.
Note: Do not use ULTRACHECK for washer-disinfectors because of high temperature during drying cycle.
STORAGE: Store ULTRACHECK box at +2°C – 25°C and do not freeze keep away from the light, heat and chemicals.