pico+15/TF/U | ultrasonic level sensor | M22 | microsonic Back pico+15/TF/U Page content as online catalogue (download) download sensor as pdf download sensor line as pdf Download complete sensor range as PDF 1 x analogue 0-10 V 250 mm measuring range 20 - 250 mm design cylindrical M22 operating mode analogue distance measurements particularities hohe Chemiebeständigkeit PVDF-Gehäuse schlankes Schallfeld ultrasonic-specific means of measurement echo propagation time measurement transducer frequency 380 kHz blind zone 20 mm operating range 150 mm maximum range 250 mm resolution 0.069 mm reproducibility ± 0.15 % accuracy ± 1 % (temperature drift internally compensated) electrical data operating voltage UB 15 V bis 30 V DC, verpolfest voltage ripple ± 10 % no-load current consumption ≤ 40 mA type of connection 5-pin M12 initiator plug outputs output 1 analogue output voltage: 0-10 V, short-circuit-proof switchable rising/falling response time 32 ms delay prior to availability < 300 ms inputs input 1 com input synchronisation input teach-in input IO-Link housing material PVDF, PBT ultrasonic transducer coated with PTFE film, FFKM O-ring max. tightening torque of nuts 1 Nm class of protection to EN 60529 IP 67 operating temperature -25°C to +70°C storage temperature -40°C to +85°C weight 30 g technical features/characteristics temperature compensation yes controls com input scope for settings Teach-in via com input on pin 5 LCA-2 with LinkControl Synchronisation yes multiplex yes indicators 1 x LED green: working, 1 x LED yellow: switch status particularities hohe Chemiebeständigkeit PVDF-Gehäuse schlankes Schallfeld accessories deployable accessory KST5A-2/M12 KST5A-5/M12 KST5G-2/M12 KST5G-5/M12 LCA-2 LCA-2 Koffer documentation (download) user's manual BA_ENG_pico+xxxTF_IU.pdf STEP pico+15_TF_X_STP.zip IGS pico+YY_TF_X_IGS.zip IPT pico+YY_TF_X_IPT.zip PDF pico+xxx_TF_X.pdf SAT pico+YY_TF_X_SAT.zip EU-Declaration of conformity CE_CON_ENG_pico+xxxUI.pdf order no. pico+15/TF/U