| ||Class 2 Recall|
AU5800 Clinical Chemistry Analyzer
||March 06, 2013
||Beckman Coulter AU5800 Clinical Chemistry Analyzer. An automated chemistry analyzer that measures analytes such as Glucose, Magnesium, and Potassium.
||All serial numbers and all software versions 4.05 and prior.
|Beckman Coulter Inc.
250 S Kraemer Blvd
Brea, California 92821-6232
|For Additional Information Contact
||Noreen Galvin, PhD
|Due to a software data storage limitation which only occurs under certain conditions, measurements of the Reagent Blank (RB) or Calibration (CAL) sample results are not saved and the software reverts to a previously stored, acceptable RB/CAL result.
||Beckman Coulter sent an "URGENT PRODUCT CORRECTION" letter dated November 28, 2012 to all affected customers. The letter identified the product, problem, and actions to be taken by the customers. Contact the Call Center Hotline at (800) 854-3633 for questions regarding the notice.
|Quantity in Commerce
||28 in U.S.; 4 Canada; 318 to other foreign countries.
||Worldwide Distribution-USA (nationwide) and the countries of Argentina, Australia, Austria, Brazil, China, Czech Republic, France, Germany, Hungary, India, Ireland, Israel, Italy, Korea Republic, Malaysia, Netherlands, Russian Federation, Spain, Switzerland, Taiwan, Turkey, United Kingdom, and Viet Nam.