easy measurement of Systolic and
Diastolic Blood Pressures
gives accurate and consistent blood pressure readings that
conform to values obtained by standard mercury sphygmomanometers.
Blood pressure measurement is extremely
simple using BIPITONE. No stethoscope is required and human errors
in auscultation is eliminated. The method used is an electronic adaption
of RIVA-ROCCI principle. Special cuff with multiple sensors cover
a wide pick-up area to determine the blood pressure easily.
BIPITONE is useful to doctors as well as for hypertensive patients
to keep a regular check of their blood pressure at home.
of Operation Wrap the velcro cuff on the arm; and inflate
bladder as usual. Press B.P. ON switch and gradually deflate cuff pressure
at rate of 2 to3 mm/Hg. When the Systolic pressure is reached , the electronic
beeper and the red LED flashes on the B.P.meter will start. As cuff pressure
is further reduced, the beeps/flashes will stop to indicate the Diastolic
Specifications Measurement Method
: Electronic adaption of
RIVA ROCCI principle Blood Pressure Indication : Systolic - Flashing LED
and beeper starts; Diastolic - Flashing LED and beeper stops. Measurement
Range: 20 to 300 mm/Hg; Sensor System
: Multiple HIR Electrosteth transducers in cuff. Electronic
Circuitry : Solid- State intergrated circuitry. Power
Source : One 9 - Volt Alkaline battery : Duracell MN1604 - 6LR61
; or equivalent. Alternatively, 220 or 110 volts/50 Hz mains supply
using optional battery eliminator ;
4 x 3 x 1 ½ inches / 500 gms.
of Biomedical/Medical Electronic Instruments & Scientific Pumps for
* Medicine * Pharmacology * Health * Biosciences * Research * Education