BACKGROUND Second-generation supraglottic airway (SGA) devices are useful for airway management during positive pressure ventilation in general anesthesia and emergency medicine. In some clinical settings, such as the anesthetic management of awake craniotomy, SGAs are used in the head-rotated position, which is required for exposure of the surgical field, although this position sometimes worsens the efficiency of mechanical ventilation with SGAs. In this study, we investigated and compared the influence of head rotation on oropharyngeal leak pressures (OPLP) of the i-gel and LMA® Supreme™ which are second-generation SGA devices. METHODS Patients who underwent elective surgery under general anesthesia were enrolled in this study and randomly divided into i-gel or LMA Supreme groups. After induction of anesthesia with muscle relaxation, the i-gel or LMA Supreme was inserted according to computerized randomization. The primary outcome was the OPLP at 0°, 30°, and 60° head rotation. The secondary outcomes were the maximum airway pressure and expiratory tidal volume when patients were mechanically ventilated using a volume-controlled ventilation mode with a tidal volume of 10 mL/kg (ideal body weight), ventilation score, and fiber-optic views of vocal cords. RESULTS Thirty-four and 36 participants were included in the i-gel and LMA Supreme groups, respectively. The OPLPs of the i-gel and LMA Supreme significantly decreased as the head rotation angle increased (mean difference [95% confidence interval], P value: i-gel; 0° vs 30°: 3.5 [2.2-4.8], P < .001; 30° vs 60°: 2.0 [0.6-3.5], P = .002; 0° vs 60°: 5.5 [3.3-7.8], P < .001, LMA Supreme; 0° vs 30°: 4.1 [2.6-5.5], P < .001; 30° vs 60°: 2.4 [1.1-3.7], P < .001; 0° vs 60°: 6.5 [5.1-8.0], P < .001). There were statistically significant differences in expiratory tidal volume and ventilation score between 0° and 60° in the i-gel group and in ventilation score between 30° and 60° in the LMA Supreme group. There was no statistically significant difference between the 2 devices in all outcome measures. The incidences of adverse events, such as hoarseness or sore throat, were not significantly different between i-gel and LMA Supreme. CONCLUSIONS Head rotation to 30° and 60° reduces OPLP with both i-gel and LMA Supreme. There is no difference in OPLP between i-gel and LMA Supreme in the 3 head rotation positions.
Head Rotation Reduces Oropharyngeal Leak Pressure of the i-gel and LMA® Supreme™ in Paralyzed, Anesthetized Patients: A Randomized Trial.