This study aimed to assess the anterior corneal wavefront aberrations, keratometry, astigmatism vectors and pupil size between Pentacam HR® (Oculus Optikgeraete GmbH, Wetzlar, Germany) and iTrace® (Tracey Technologies Corp., Houston, TX, USA). In this observational study, 100 eyes (50 healthy volunteers) were scanned in mesopic light condition with a Pentacam HR® and iTrace®. Anterior corneal aberrations (spherical aberration (Z40), vertical coma (Z3 - 1), horizontal coma (Z3 + 1)), keratometry in the flattest (K1) and steepest meridian (K2), mean astigmatism, astigmatic vectors (J0 and J45), and pupil size were measured. We found a significant difference in Z40 (Pentacam®: +0.30 ± 0.11 µm and iTrace®: -0.03 µm ± 0.05 µm; p < 0.01) with no correlation between the devices (r = -0.12, p = 0.22). The devices were in complete agreement for Z3 - 1 (p = 0.78) and Z3 + 1 (p = 0.39), with significant correlation between the machines (r = -0.38, p < 0.01 and r = -0.6, p < 0.01). There was no difference in K1, K2 and mean astigmatism. J0 was negative with both devices (against-the-rule astigmatism), but there was no correlation. J45 was negative with the Pentacam HR® (more myopic oblique astigmatism) but significantly correlated between the devices. Pupil size was smaller with Pentacam HR® (p < 0.01). In summary, these devices cannot be used interchangeably. Corneal Z40 was significantly different with more negative Z40 with iTrace® compared to Pentacam HR®. iTrace® operates with lower illumination, giving larger pupil size than Pentacam HR®, which uses intense blue light during measurement. No correlation was found for J0. Pentacam HR® had a trend to record more negative J45 (myopic oblique astigmatism).
Comparison of Anterior Corneal Aberrometry, Keratometry and Pupil Size with Scheimpflug Tomography and Ray Tracing Aberrometer