Strategies to return to dental practice in pandemic times is a new challenge due to the generation and spread of potentially contaminated dispersion particles (PCDP) that may contain the SARS-CoV-2, the etiological factor of the COVID-19 disease. Due to the significant dispersion of PCDP in the dental environment, the use of equipment such as ultrasonic tips have been inadvisable during the pandemic. Several clinical procedures, however, benefit from the use of such equipment. Thus, using a microbial dispersion model of PCDP, the aim of this study was to compare the dispersion caused by the dental drill (DD) an ultrasonic tip (UT) alone and the UT coupled with a Spray control (SC) device. The DD, UT (with or without the SC) were activated for one minute having had the water from the reservoir replaced with a suspension of Lactobacillus casei Shirota (1.5 x 108 CFU/mL). Petri dishes containing MRS agar were positioned at 50cm, 100cm and 150cm from the headrest of the dental chair at different angles (0 degree and 90 degrees). At 50 cm, the mean CFU (standard deviation) of L. casei Shirota was 13554.60 (4071.03) for the DD, 286.67 (73.99) for the US (97.89% reduction), and 4.5 (0.58) CFU for the UT-SC (p < 0.0001), establishing a further 98.43% reduction between UT and UT with SC. The UT with SC model proved effective in reducing dispersion from the UT, endorsing its use as an additional strategy to reduce PCDP in the dental environment in times of pandemic.
A novel dental biosafety device to control the spread of potentially contaminated dispersion particles from dental ultrasonic tips