BACKGROUND: A complication associated with the distal locking screw used in the proximal femoral nail (PFN) system is the formation of accidental additional holes. We hypothesized that an increase in stress around additional holes is a relevant factor contributing to fractures. This study aimed to evaluate stress changes in the cortical bone around additional screw holes using finite element analysis. METHODS: Proximal femoral nail antirotation (PFNA)-II (Synthes, Solothurn, Switzerland) was inserted into a femur model. An additional 4.9-mm transcortical hole was made either anteriorly (anterior hole model) or posteriorly (posterior hole model) to the distal locking screw. Finite element analysis was used to calculate compression, tension, and load limits to investigate stress around additional holes with respect to the direction of screw penetration and degree of osteoporosis. The results were then compared with those of mechanical testing. A 31A-21 type intertrochanteric fracture was applied. As a control group, a model without additional holes (no-hole model) was developed. Repeated load-loading tests were performed on 10 model bones per model group. RESULTS: Tensile stress was significantly greater in the no-hole model when additional screw holes were present, and the anterior hole showed a higher maximum stress value than the posterior hole, suggesting that the anterior hole was more susceptible to fracture. The change in tensile stress first appeared in the hole around the lateral cortical bone and proceeded to the medial side. Biomechanical testing showed that fractures around the distal locking screw occurred in 0 cases of the no-hole, 10 of the anterior hole, and 9 of the posterior hole models. CONCLUSIONS: During PFN surgery for intertrochanteric fracture, holes with distal locking screws fixed and removed at the anterior and posterior of the nail can be a risk factor for fractures in the surrounding area.
A finite element analysis and cyclic load experiment on an additional transcortical-type hole formed around the proximal femoral nail system’s distal locking screw