  The construction of the ATLAS electromagnetic liquid argon calorimetermodules is completed and all the modules are assembled and inserted in thecryostats. During the production period four barrel and three endcap moduleswere exposed to test beams in order to assess their performance, ascertain theproduction quality and reproducibility, and to scrutinize the complete energyreconstruction chain from the readout and calibration electronics to the signaland energy reconstruction. It was also possible to check the full Monte Carlosimulation of the calorimeter. The analysis of the uniformity, resolution andextraction of constant term is presented. Typical non-uniformities of 0.5% andtypical global constant terms of 0.6% are measured for the barrel and end-capmodules.
During the production period four barrel and three endcap modules were exposed to test beams in order to assess their performance, ascertain the production quality and reproducibility, and to scrutinize the complete energy reconstruction chain from the readout and calibration electronics to the signal and energy reconstruction. It was also possible to check the full Monte Carlo simulation of the calorimeter.