  We describe the calibration status and data products pertaining to the GR2and GR3 data releases of the Galaxy Evolution Explorer (GALEX). These releaseshave identical pipeline calibrations that are significantly improved over theGR1 data release. GALEX continues to survey the sky in the Far Ultraviolet(FUV, ~154 nm) and Near Ultraviolet (NUV, ~232 nm) bands, providingsimultaneous imaging with a pair of photon counting, microchannel plate, delayline readout detectors. These 1.25 degree field-of-view detectors arewell-suited to ultraviolet observations because of their excellent redrejection and negligible background. A dithered mode of observing and photonlist output pose complex requirements on the data processing pipeline,entangling detector calibrations and aspect reconstruction algorithms. Recentimprovements have achieved photometric repeatability of 0.05 and 0.03 mAB inthe FUV and NUV, respectively. We have detected a long term drift of order 1%FUV and 6% NUV over the mission. Astrometric precision is of order 0.5" RMS inboth bands. In this paper we provide the GALEX user with a broad overview ofthe calibration issues likely to be confronted in the current release.Improvements are likely as the GALEX mission continues into an extended phasewith a healthy instrument, no consumables, and increased opportunities forguest investigations.
A dithered mode of observing and photon list output pose complex requirements on the data processing pipeline, entangling detector calibrations and aspect reconstruction algorithms. Recent improvements have achieved photometric repeatability of 0.05 and 0.03 mAB in the FUV and NUV, respectively. We have detected a long term drift of order 1% FUV and 6% NUV over the mission.