<unk> ner with cover removed to show internal components. Legend: T: X-ray tube D: X-ray detectors X: X-ray beam R: Gantry rotation Left image is a sinogram which is a graphic representation of the raw data obtained from a<unk> . At right is an image sample derived from the raw data. Computed tomography operates by using an X-ray generator that rotates around the object; X-ray detectors are positioned on the opposite side of the circle from the X-ray source. As the X-rays pass through the patient, they are attenuated differently by various tissues according to the tissue density. A visual representation of the raw data obtained is called a sinogram, yet it is not sufficient for interpretation. Once the scan data has been acquired, the data must be processed using a form of tomographic reconstruction, which produces a series of cross-sectional images. These cross-sectional images are made up of small units of pixels or voxels. Pixels in an image obtained by<unk> ning are displayed in terms of relative radiodensity. The pixel itself is displayed according to the mean attenuation of the tissue(s) that it corresponds to on a scale from +3,071 (most attenuating) to −1,024 (least attenuating) on the Hounsfield scale. Pixel is a two dimensional unit based on the matrix size and the field of view. When the<unk>  slice thickness is also factored in, the unit is known as a voxel, which is a three-dimensional unit. Water has an attenuation of 0 Hounsfield units (HU), while air is −1,000 HU, cancellous bone is typically +400 HU, and cranial bone can reach 2,000 HU or more (os temporale) and can cause artifacts. The attenuation of metallic implants depends on the atomic number of the element used: Titanium usually has an amount of +1000 HU, iron steel can completely extinguish the X-ray and is, therefore, responsible for well-known line-artifacts in computed tomograms. Artifacts are caused by abrupt transitions between low- and high-density materials, which results in data values that exceed the dynamic range of the processing electronics. Two-dimensional<unk>  images are conventionally rendered so that the view is as though looking up at it from the patient's feet. Hence, the left side of the image is to the patient's right and vice versa, while anterior in the image also is the patient's anterior and vice versa. This left-right interchange corresponds to the view that physicians generally have in reality when positioned in front of patients. Initially, the images generated in<unk> s were in the transverse (axial) anatomical plane, perpendicular to the long axis of the body. Modern scanners allow the scan data to be reformatted as images in other planes. Digital geometry processing can generate a three-dimensional image of an object inside the body from a series of two-dimensional radiographic images taken by rotation around a fixed axis. These cross-sectional images are widely used for medical diagnosis and therapy. Contrast Contrast media used for X-ray<unk> , as well as for plain film X-ray, are called radiocontrasts. Radiocontrasts for<unk>  are, in general, iodine-based. This is useful to highlight structures such as blood vessels that otherwise would be difficult to delineate from their surroundings. Using contrast material can also help to obtain functional information about tissues. Often, images are taken both with and without radiocontrast.
CT scan
Fluoroscopy
Contrast CT
SAP scan
Imaging
Radiography
Exploratory surgery
Autoradiograph