Determining optimal torsional ultrasound power for cataract surgery with automatic longitudinal pulses at maximum vacuum ex vivo.
To determine the optimal longitudinal power settings for Infiniti OZil Intelligent Phaco (IP) at varying torsional amplitude settings; and to test the hypothesis that increasing longitudinal power is more important at lower torsional amplitudes to achieve efficient phacoemulsification. Laboratory investigation. setting: John A. Moran Eye Center, University of Utah, Salt Lake City, Utah. procedure: Individual porcine nuclei were fixed in formalin, then cut into 2.0 mm cubes. Lens cube phacoemulsification was done using OZil IP at 60%, 80%, and 100% torsional amplitude with 0%, 10%, 20%, 30%, 50%, 75%, or 100% longitudinal power. All experiments were done using a 20 gauge 0.9 mm bent reverse bevel phaco tip at constant vacuum (550 mm Hg), aspiration rate (40 mL/min), and bottle height (50 cm). main outcome measure: Complete lens particle phacoemulsification (efficiency). Linear regression analysis showed a significant increase in efficiency with increasing longitudinal power at 60% torsional amplitude (R(2) = 0.7269, P = .01) and 80% torsional amplitude (R(2) = 0.6995, P = .02) but not at 100% amplitude (R(2) = 0.3053, P = .2). Baseline comparison of 60% or 80% vs 100% torsional amplitude without longitudinal power showed increased efficiency at 100% (P = .0004). Increasing longitudinal power to 20% abolished the efficiency difference between 80% vs 100% amplitudes. In contrast, 75% longitudinal power abolished the efficiency difference between 60% vs 100% torsional amplitudes. Results suggest that longitudinal power becomes more critical at increasing phacoemulsification efficiencies at torsional amplitudes less than 100%. Increasing longitudinal power does not further increase efficiency at maximal torsional amplitudes.