A lithographic apparatus is a machine that applies a desired pattern onto a substrate, usually onto a target portion of the substrate. A lithographic apparatus can be used, for example, in the manufacture of integrated circuits (ICs). In that instance, a patterning device, which is alternatively referred to as a mask or a reticle, may be used to generate a circuit pattern to be formed on an individual layer of the IC. This pattern can be transferred onto a target portion (e.g., including part of, one, or several dies) on a substrate (e.g., a silicon wafer). Transfer of the pattern is typically via imaging onto a layer of radiation-sensitive material (resist) provided on the substrate. In general, a single substrate will contain a network of adjacent target portions that are successively patterned. In lithographic processes, it is desirable frequently to make measurements of the structures created, e.g., for process control and verification. Various tools for making such measurements are known, including scanning electron microscopes, which are often used to measure critical dimension (CD), and specialized tools to measure overlay, a measure of the accuracy of alignment of two layers in a device. Overlay may be described in terms of the degree of misalignment between the two layers, for example reference to a measured overlay of 1 nm may describe a situation where two layers are misaligned by 1 nm.
Various forms of metrology apparatuses, such as scatterometers, have been developed for use in the lithographic field. For example, certain metrology apparatuses direct a beam of radiation onto a target and measure one or more properties of the scattered radiation—e.g., intensity at a single angle of reflection, or over a range of angles of reflection, as a function of wavelength; intensity at one or more wavelengths as a function of reflected angle; or polarization as a function of reflected angle—to obtain a “spectrum” from which a property of interest of the target can be determined. Determination of the property of interest may be performed by various techniques: e.g., reconstruction of the target by iterative approaches implemented using rigorous coupled wave analysis or finite element methods; library searches; and/or principal component analysis.
Targets may be measured using dark field scatterometry in which the zeroth order of diffraction (corresponding to a specular reflection) is blocked, and only higher orders processed. Examples of dark field metrology can be found in PCT patent application publication nos. WO 2009/078708 and WO 2009/106279, which documents are hereby incorporated in their entireties by reference.
Intensity asymmetry between different diffraction orders (e.g. between −1st and the +1st diffraction orders) for a given overlay target provides a measurement of target asymmetry; that is, asymmetry in the target. This asymmetry in the overlay target can be used as an indicator of overlay (undesired misalignment of two layers).