A tool for reading, displaying and saving data from the NanoVNA
[![Latest Release](https://img.shields.io/github/v/release/NanoVNA-Saver/nanovna-saver.svg)](https://github.com/NanoVNA-Saver/nanovna-saver/releases/latest)
[![License](https://img.shields.io/github/license/NanoVNA-Saver/nanovna-saver.svg)](https://github.com/NanoVNA-Saver/nanovna-saver/blob/master/LICENSE)
[![Downloads](https://img.shields.io/github/downloads/NanoVNA-Saver/nanovna-saver/total.svg)](https://github.com/NanoVNA-Saver/nanovna-saver/releases/)
[![GitHub Releases](https://img.shields.io/github/downloads/NanoVNA-Saver/nanovna-saver/latest/total)](https://github.com/NanoVNA-Saver/nanovna-saver/releases/latest)
[![Donate](https://img.shields.io/badge/paypal-donate-yellow.svg)](https://www.paypal.com/cgi-bin/webscr?cmd=_donations&business=T8KTGVDQF5K6E&item_name=NanoVNASaver+Development&currency_code=EUR&source=url)

NanoVNASaver
============

A multiplatform tool to save Touchstone files from the NanoVNA,
sweep frequency spans in segments to gain more than 101 data
points, and generally display and analyze the resulting data.

- Copyright 2019, 2020 Rune B. Broberg
- Copyright 2020ff NanoVNA-Saver Authors

<a href="#built-with"></a>
It's written in __Python 3__ using __PyQt5__ and __scipy__.

<details open="open">
<summary>Table of Contents</summary>

- [About](#nanovnasaver)
  - [Built With](#built-with)
- [Introduction](#introduction)
  - [Current Features](#current-features)
  - [Screenshot](#screenshot)
  - [Binary Releases](#binary-releases)
  - [Installation](#installation)
    - [Detailed Installation Instructions](docs/INSTALLATION.md)
- [Usage](#using-the-software)
  - [Calibration](#calibration)
  - [TDR](#tdr)
- [Latest Changes](#latest-changes)
- [Contributing](#contributing)
  - [Contribution Guidlines](docs/CONTRIBUTING.md)
- [License](#license)
- [References](#references)
- [Acknowledgements](#acknowledgements)

</details>


Introduction
------------

This software connects to a NanoVNA and extracts the data for
display on a computer and allows saving the sweep data to Touchstone files.

<a href="#current-features"></a>
### Current features:

- Reading data from a NanoVNA -- Compatible devices: NanoVNA, NanoVNA-H,
  NanoVNA-H4, NanoVNA-F, AVNA via Teensy
- Splitting a frequency range into multiple segments to increase resolution
  (tried up to >10k points)
- Averaging data for better results particularly at higher frequencies
- Displaying data on multiple chart types, such as Smith, LogMag, Phase and
  VSWR-charts, for both S11 and S21
- Displaying markers, and the impedance, VSWR, Q, equivalent
  capacitance/inductance etc. at these locations
- Displaying customizable frequency bands as reference, for example amateur
  radio bands
- Exporting and importing 1-port and 2-port Touchstone files
- TDR function (measurement of cable length) - including impedance display
- Filter analysis functions for low-pass, high-pass, band-pass and band-stop
  filters
- Display of both an active and a reference trace
- Live updates of data from the NanoVNA, including for multi-segment sweeps
- In-application calibration, including compensation for non-ideal calibration
  standards
- Customizable display options, including "dark mode"
- Exporting images of plotted values

### Screenshot
![Screenshot of version 0.1.4](https://i.imgur.com/ZoFsV2V.png)

Running the application
-----------------------

The software was written in Python on Windows, using Pycharm, and the modules
PyQT5, numpy, scipy and pyserial.
Main development is currently done on Linux (Mint 21 "Vanessa" Cinnamon)

### Binary releases

You can find current binary releases for Windows, Linux and MacOS under
<https://github.com/NanoVNA-Saver/nanovna-saver/releases/latest>

The 32bit Windows binaries are somewhat smaller and seems to be a
little bit more stable.

Versions older than Windows 7 are not known to work.

#### Windows 7

It requires Service Pack 1 and [Microsoft VC++ Redistributable](
  https://support.microsoft.com/en-us/help/2977003/the-latest-supported-visual-c-downloads).
For most users, this would already be installed.

### Installation and Use with pip

Copy the link of the tgz from latest relaese and install it with pip install. e.g.:

    pip3 install https://github.com/NanoVNA-Saver/nanovna-saver/archive/refs/tags/v0.5.4.tar.gz

Once completed run with the following command

    NanoVNASaver

[Detailed installation instructions](docs/INSTALLATION.md)

Using the software
------------------

Connect your NanoVNA to a serial port, and enter this serial port in the serial
port box.  If the NanoVNA is connected before the application starts, it should
be automatically detected. Otherwise, click "Rescan". Click "Connect to device"
to connect.

The app can collect multiple segments to get more accurate measurements. Enter
the number of segments to be done in the "Segments" box. Each segment is 101
data points, and takes about 1.5 seconds to complete.

Frequencies are entered in Hz, or suffixed with k or M.  Scientific notation
(6.5e6 for 6.5MHz) also works.

Markers can be manually entered, or controlled using the mouse. For mouse
control, select the active marker using the radio buttons, or hold "shift"
while clicking to drag the nearest marker. The marker readout boxes show the
actual frequency where values are measured.  Marker readouts can be hidden
using the "hide data" button when not needed.

Display settings are available under "Display setup". These allow changing the
chart colours, the application font size and which graphs are displayed.  The
settings are saved between program starts.

### Calibration

_Before using NanoVNA-Saver, please ensure that the device itself is in a
reasonable calibration state._

A calibration of both ports across the entire frequency span, saved to save
slot 0, is sufficient.  If the NanoVNA is completely uncalibrated, its readings
may be outside the range accepted by the application.

In-application calibration is available, either assuming ideal standards or
with relevant standard correction. To manually calibrate, sweep each standard
in turn and press the relevant button in the calibration window.
For assisted calibration, press the "Calibration Assistant" button.  If desired,
enter a note in the provided field describing the conditions under which the
calibration was performed.

Calibration results may be saved and loaded using the provided buttons at the
bottom of the window.  Notes are saved and loaded along with the calibration
data.

![Screenshot of Calibration Window](https://i.imgur.com/p94cxOX.png)

Users of known characterized calibration standard sets can enter the data for
these, and save the sets.

After pressing _Apply_, the calibration is immediately applied to the latest
sweep data.

\! _Currently, load capacitance is unsupported_ \!

### TDR

To get accurate TDR measurements, calibrate the device, and attach the cable to
be measured at the calibration plane - i.e. at the same position where the
calibration load would be attached.  Open the "Time Domain Reflectometry"
window, and select the correct cable type, or manually enter a propagation
factor.

Latest Changes
--------------

### Changes in 0.5.4

 - Bugfixes for Python3.11 compatability
 - Bugfix for Python3.8 compatability
 - use math instead of table for log step calculation
 - Support of NanoVNA V2 Plus5 on Windows
 - New SI prefixes added - Ronna, Quetta
 - addes a Makefile to build a packages
 - Simplyfied sweep worker
 - Fixed calibration data loading
 - Explicit import of scipy functions - #555
 - Refactoring of Analysis modules

### Changes in 0.5.3

 - Python 3.10 compatability fixes
 - Fix crash on open in use serial device
 - Use a Defaults module for all settings -
   ignores old .ini settings
 - Refactoring and unifying Chart classes
 - No more automatic update checks (more privacy)
 - Corrected error handling in NanaVNA\_V2 code 
 - Fixed man float related crashes with Qt and
   Python 3.10
 - Using more integer divisions to get right type for QPainter
   points
 - No more long lines in code (pycodestyle)

Contributing
------------

First off, thanks for taking the time to contribute! Contributions are what
make the open-source community such an amazing place to learn, inspire, and
create. Any contributions you make will benefit everybody else and are
**greatly appreciated**.


Please read [our contribution guidelines](docs/CONTRIBUTING.md), and thank you
for being involved!


License
-------

This software is licensed under version 3 of the GNU General Public License. It
comes with NO WARRANTY.

You can use it, commercially as well. You may make changes to the code, but I
(and the license) ask that you give these changes back to the community.

References
----------

- Ohan Smit wrote an introduction to using the application:
  [https://zs1sci.com/blog/nanovnasaver/]
- HexAndFlex wrote a 3-part (thus far) series on Getting Started with the NanoVNA:
  [https://hexandflex.com/2019/08/31/getting-started-with-the-nanovna-part-1/] - Part 3 is dedicated to NanoVNASaver:
  [https://hexandflex.com/2019/09/15/getting-started-with-the-nanovna-part-3-pc-software/]
- Gunthard Kraus did documentation in English and German:
  [http://www.gunthard-kraus.de/fertig_NanoVNA/English/]
  [http://www.gunthard-kraus.de/fertig_NanoVNA/Deutsch/]

Acknowledgements
----------------

Original application by Rune B. Broberg (5Q5R)

Contributions and changes by Holger Müller (DG5DBH), David Hunt and others.

TDR inspiration shamelessly stolen from the work of Salil (VU2CWA) at
<https://nuclearrambo.com/wordpress/accurately-measuring-cable-length-with-nanovna/>

TDR cable types by Larry Goga.

Bugfixes and Python installation work by Ohan Smit.

Thanks to everyone who have tested, commented and inspired.  Particular thanks
go to the alpha testing crew who suffer the early instability of new versions.

This software is available free of charge. If you read all this way, and you
*still* want to support it, you may donate to the developer using the button
below:

[![Paypal](https://www.paypalobjects.com/en_US/i/btn/btn_donate_LG.gif)](https://www.paypal.com/cgi-bin/webscr?cmd=_donations&business=T8KTGVDQF5K6E&item_name=NanoVNASaver+Development&currency_code=EUR&source=url)
